# Message Decoder for Solo2, Alto and Deep Solo Floats; v260727
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11330    -5  276 000000000000009  3265     0   2 2026/08/24, 13:43:04,  41.6168  -67.8238 243      0 14
SBD 11330    -5  102 000000000000009  3266     0   1 2026/08/24, 13:43:13,  41.5413  -70.6851   6      0 13
SBD 11330    -5  339 000000000000009  3267     0   0 2026/08/24, 13:43:23,  41.5413  -70.6851   6      0 13
# Delayed Transmission - (3/3) Messages transmitted 9732.6 days after surfacing on 2000/01/01 00:00:00 (refer to Surface Pressure Start Time)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       3          681           36         717
# Sensor_ID bytes     % packets Description
         13   417 61.23       2  Parameters
         14   264 38.77       1 Engineering
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -5  2.4    1   150   150   180   360     0     0     11     1    2160    0  0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS           25      10      1000       5      1000       1 
# Engineering Data
#       Alto BIT Test
#    cycle software   BIT         Total
#          version    result      time
E011   -5  0x0a0d0013 0x00000000  1006
 # Skyview           BIT_Start             BIT_End
         0 1970/01/01 00:00:35 1970/01/01 00:17:21
# Successful Bit Test
# CTD Status temp    salinity
 0x64697361  23.350   0.011
# Pressure offsets 
# before   now     after [dBar]
    0.100    0.000    0.099
# Pcase vacuum [inHg]
   5.140   5.980
# Battery Voltages [V]
  15.270  15.140
# Air Pump test
# Air Valve Positions [V]
   1.140   0.062   0.665
# Air Valve Calibrations [V]
   1.086   0.116   0.601
# duration count max P max currents      max voltage
     360     2    7.66    0.09    0.08   15.28
# Oil Pump test
# Oil Valve Positions [V]
   1.805   0.779
# Oil Valve Calibrations [V]
   1.753   0.831
# duration max currents [mAmp]      max voltage [V]
     600      0.20    0.16        15.24
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   -5         0    2026/08/24 13:43:23
