# Message Decoder for Solo2, Alto and Deep Solo Floats; v210819
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11154    38  306 000000000000009   327     0   2 2020/10/07, 18:37:40, -42.5918  112.8336   3      0 13,6,14,14
SBD 11154    38  340 000000000000009   328     0   1 2020/10/07, 18:38:01, -42.5653  112.8336   3      0 13
SBD 11154    38  278 000000000000009   329     0   0 2020/10/07, 18:38:14, -42.5918  112.8336   3      0 0,4,14,14,7,5,14,0,8,15
# Delayed Transmission - (3/3) Messages transmitted   9.5 days after surfacing on 2020/09/28 06:15:46 (refer to 1st Rise scan < 3dbar)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      15          888           36         924
# Sensor_ID bytes     % packets Description
          0    48  5.41       2         GPS
          4    38  4.28       1        Fall
          5    18  2.03       1        Rise
          6    70  7.88       1        Pump
          7    25  2.82       1 Drift_times
          8     8  0.90       1     Surface
         13   404 45.50       2  Parameters
         14   211 23.76       5 Engineering
         15    66  7.43       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   38 1 2020/10/07 10:45 -42.59718 +112.78819  140  6  0 32 42  4.4     0
G   38 2 2020/10/07 18:36 -42.57936 +112.84260  150  3 35 38 42  2.4     0
# Drift Data
#      C Npts comp n_var
I0    38    1    1     4
#      Pres       Temp   Salinity     Time
   -999.000  -999.0000  -999.0000 15943390
# Fall Time Series
#  Cycle npts     Start                Date
F00   38    1 654576901 2020/09/28 02:55:01
# time     Pres phase                Date
     0 1130.080     1 2020/09/28 02:55:01
# Rise Time Series
#  Cycle npts     Start                Date
R00   38    1 654588946 2020/09/28 06:15:46
# time     Pres phase                Date
     0    1.480     7 2020/09/28 06:15:46
# Pump Time Series
#   Cycle npts
B02    38    6
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
  1130.080  770   15.04     143    749    -9     2
  1130.080    6   15.19     233    743    -9     3
  1130.080    6   15.18     231    744    -9     3
  1130.080    6   15.21     251    744    -9     3
     1.480   76   15.19     186    744    -9     7
     0.000   80   15.18     163    825    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    38  2.4   13  2000  1000   600   188   120    60     15     3    3860   60  0.5796
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   38 2020/10/07 10:45:19
# Pcase vacuums
   9.520   7.450
# Oil valve positions
   1.662   0.588   0.537
# Seek attempts and pump time
      3     18
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   38 2020/10/07 18:19:13 9999/99/99 99:99:99
# Pcase vacuums
   7.450   7.450
# Average 
#   P      T      S
 9999.000  99.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   38 2020/10/07 18:31:45 9999/99/99 99:99:99
# Pcase vacuums
   7.450   7.460
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   38 2020/10/07 18:34:19
# number of packets previous packets
      3               4
# CTD Status pres offset
 0x00000000   0.000
# Pcase vacuums
   7.460   8.820
# Air Pump max current time  max pressure
      52       160     5.304
# Air Valve Positions
   1.121   0.000
# Air Valve Calibrations
   1.059   0.123   0.591
# Oil Valve Positions
   0.000   0.598
# Oil Valve Calibrations
   1.606   0.656
# Encountered Error Codes
#    Cycle Errors
EE20    38     10
#               Date   Code Description
 2020/10/07 10:45:44 0x000C Ctd
 2020/10/07 10:45:44 0x080C CtdRetry
 2020/10/07 10:46:09 0x000C Ctd
 2020/10/07 10:46:09 0x080C CtdRetry
 2020/10/07 10:46:35 0x000C Ctd
 2020/10/07 10:46:35 0x080C CtdRetry
 2020/10/07 10:46:35 0x0113 DescendCtd
 2020/10/07 10:46:35 0x0013 Descend
 2020/10/07 10:47:01 0x000C Ctd
 2020/10/07 10:47:01 0x080C CtdRetry
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   38  1      0
# MC    Pres      Time  Pcase Vacuum
   1  1130.080      2   7.330
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   38       197    2020/10/07 18:38:01
Parameter Table Param_Table-V17.txt |
     IFV = 8.0.2.0	(IFV) |
     VER = 17	(version) |
     STY =        0	(Sensor 0=RBR,1=SBE) |
     MSN =       38	(cycle number) |
     MPS =        0	(phase) |
     MPC =        1	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =        0	(count) |
  MNM[2] =        0	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        0	(count) |
  MNS[2] =        0	(count) |
  DTD[0] =     1000	(dBar) |
  DTD[1] =      400	(dBar) |
  DTD[2] =      400	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        2	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        1	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =      300	(dBar) |
  CPB[3] =      100	(dBar) |
  CPN[1] =       20	(dBar) |
  CPN[2] =       10	(dBar) |
  CPN[3] =       10	(dBar) |
  CPP[1] =     2000	(ms) |
  CPP[2] =     1000	(ms) |
  CPP[3] =     1000	(ms) |
     CRD =       50	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    60000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       37	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       20	(dBar) |
  DSD[1] =      400	(dBar) |
  DSD[2] =      400	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     7200	(sec) |
  DST[2] =     7200	(sec) |
     DDF =      200	(sec) |
     DDC =      600	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      726	(sec) |
  DFT[1] =      130	(sec) |
  DFT[2] =      130	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =    10400	(sec) |
  DCT[1] =    10800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =       18	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        5	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     3600	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =       25	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        2	(dBar) |
     DDS =      180	(sec) |
     PNS =       37	(cycle) |
  PTO[0] =     3600	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =    32768	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        1	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =     7200	(sec) |
  PFT[1] =     3600	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =     2000	(dBar) |
  PFD[1] =        0	(dBar) |
  PFD[2] =        0	(dBar) |
  PFV[0] =      400	(sec) |
  PFV[1] =        0	(sec) |
  PFV[2] =        0	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =     3600	(sec) |
     PSV =       30	(sec) |
     PSW =     1300	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       37	(cycle number) |
     ARR =       15	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       20	(sec) |
     APN =       30	(sec) |
     APF =       10	(sec) |
     ASI =     1000	(ms) |
     ASD =        2	(dBar) |
     ADD =       10	(dBar) |
     ADT =      600	(sec) |
     ARC =     3000	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       37	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =     7000	(ms) |
     SDS =       60	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =        0	(sec) |
     SSP =       80	(sec) |

