# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   998  246 000000000432020  5511     0   3 2022/08/07, 15:09:53,  67.5389  -55.1387 680      2 13,6,14,14
SBD 11146   998  340 000000000432020  5512     0   2 2022/08/07, 15:10:03,  67.5306  -55.2692 680      2 13
SBD 11146   998  279 000000000432020  5513     0   1 2022/08/07, 15:10:16,  67.5222  -55.3997 680      2 1,2,3,0,8,15
SBD 11146   998  325 000000000432020  5514     0   0 2022/08/07, 15:10:29,  67.5222  -55.3997 202      2 0,4,14,14,9,10,11,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1142           48        1190
# Sensor_ID bytes     % packets Description
          0    48  4.20       2         GPS
          1    46  4.03       1    Pressure
          2    52  4.55       1 Temperature
          3    44  3.85       1    Salinity
          4    68  5.95       1        Fall
          5    53  4.64       1        Rise
          6    70  6.13       1        Pump
          7    26  2.28       1 Drift_times
          8    35  3.06       1     Surface
          9    27  2.36       1  HR/Drift_P
         10    26  2.28       1  HR/Drift_T
         11    26  2.28       1  HR/Drift_S
         13   404 35.38       2  Parameters
         14   151 13.22       5 Engineering
         15    66  5.78       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  998 1 2022/08/07 09:36 +67.55332  -54.76337    0  6 30 38 44  0.9     0
G  998 2 2022/08/07 15:04 +67.54987  -54.77325   80  7  0 31 44  1.6     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   998   39    1     3         0
#      Pres      Temp    Salinity
      1.120     4.1930    33.2370
      2.000     4.1930    33.2370
      3.040     4.1830    33.2380
      4.040     4.1810    33.2380
      5.000     4.1850    33.2370
      6.040     4.1790    33.2380
      7.040     4.1790    33.2380
      8.000     4.1790    33.2370
      9.000     4.1770    33.2380
     10.000     4.1750    33.2380
     11.000     4.1740    33.2380
     12.040     4.1730    33.2380
     13.040     4.1700    33.2380
     14.000     4.1690    33.2370
     14.960     4.1720    33.2380
     16.000     4.1690    33.2370
     17.000     4.1680    33.2380
     17.960     4.1650    33.2380
     18.960     4.1640    33.2370
     20.000     4.1610    33.2380
     21.040     4.1610    33.2380
     22.040     4.1610    33.2380
     23.040     4.1640    33.2370
     24.000     4.1620    33.2380
     25.000     4.1630    33.2380
     26.040     4.1600    33.2390
     27.000     4.1580    33.2400
     27.960     4.1610    33.2410
     29.000     4.1600    33.2410
     30.000     4.1610    33.2410
     31.000     4.1600    33.2410
     32.040     4.1600    33.2410
     33.040     4.1620    33.2420
     34.000     4.1620    33.2410
     35.000     4.1620    33.2410
     36.040     4.1620    33.2420
     37.040     4.1620    33.2420
     38.120     4.1620    33.2410
     38.640     4.1620    33.2410
# Drift Data
#      C Npts comp n_var
I0   998    4    1     4
#      Pres       Temp   Salinity     Time
     36.800     4.1680    33.2310        5
     38.480     4.1650    33.2300     3611
     39.000     4.1620    33.2310     7217
     38.040     4.1630    33.2360    10822
# Surface Pressure Time Series
#    C npts n_var               Start
S  998     3     6 2022/08/07 15:02:55
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    84     -0.040      4.209   -999.000      14 2022/08/07 15:04:19
    97      0.040      4.206   -999.000      14 2022/08/07 15:04:32
   108     -0.010      4.205   -999.000      14 2022/08/07 15:04:43
# Fall Time Series
#  Cycle npts     Start                Date
F00  998   12 713180251 2022/08/07 09:37:31
# time     Pres phase                Date
     0    0.040     1 2022/08/07 09:37:31
   140    5.000    10 2022/08/07 09:39:51
   176   10.440    10 2022/08/07 09:40:27
   206   15.080    10 2022/08/07 09:40:57
   238   20.240    10 2022/08/07 09:41:29
   271   25.400    10 2022/08/07 09:42:02
   288   28.080     2 2022/08/07 09:42:19
   301   30.040    10 2022/08/07 09:42:32
   333   35.080    10 2022/08/07 09:43:04
   633   36.800    10 2022/08/07 09:48:04
   935   36.760    10 2022/08/07 09:53:06
   954   36.760    10 2022/08/07 09:53:25
# Rise Time Series
#  Cycle npts     Start                Date
R00  998    9 713199215 2022/08/07 14:53:35
# time     Pres phase                Date
     0   38.680     7 2022/08/07 14:53:35
   169   33.840    13 2022/08/07 14:56:24
   232   28.520    13 2022/08/07 14:57:27
   285   23.840    13 2022/08/07 14:58:20
   348   18.200    13 2022/08/07 14:59:23
   400   13.160    13 2022/08/07 15:00:15
   453    8.040    13 2022/08/07 15:01:08
   505    3.160    13 2022/08/07 15:02:00
   558    0.000     8 2022/08/07 15:02:53
# Pump Time Series
#   Cycle npts
B02   998    6
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    28.080  362   15.11       0    621    -9     2
    38.680  127   15.21       0    623    -9     7
    33.840   21   15.21       0    626    -9    13
    22.040   21   15.20       0    628    -9    13
     4.160   21   15.20       0    629    -9    13
    -0.040   81   15.19       0    634    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   998  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  998 2022/08/07 09:37:29
# Pcase vacuums
   6.380   6.180
# Oil valve positions
   1.689   0.505   0.671
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  998 2022/08/07 09:53:26 9999/99/99 99:99:99
# Pcase vacuums
   6.180   6.140
# Average 
#   P      T      S
   38.080   4.164  33.232
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  998 2022/08/07 14:53:35 9999/99/99 99:99:99
# Pcase vacuums
   6.140   6.280
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  998 2022/08/07 15:02:54
# number of packets previous packets
      4               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.290   6.380
# Air Pump max current time  max pressure
     119       386     0.000
# Air Valve Positions
   1.281   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.532
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   998      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  998  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.000    283   5.940
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  998       197    2022/08/07 15:10:03
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      998	(cycle number) |
     MPS =        0	(phase) |
     MPC =      389	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      995	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      995	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      995	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =      995	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

