# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   997  246 000000000432020  5507     0   3 2022/08/07, 09:35:03,  67.5769  -54.7463   5      0 13,6,14,14
SBD 11146   997  340 000000000432020  5508     0   2 2022/08/07, 09:35:16,  67.5769  -54.7463   4      0 13
SBD 11146   997  277 000000000432020  5509     0   1 2022/08/07, 09:35:33,  67.5960  -54.8231   2      0 1,2,3,0,8,15
SBD 11146   997  324 000000000432020  5510     0   0 2022/08/07, 09:36:38,  67.5769  -54.7463   6      0 0,4,14,14,9,10,11,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1139           48        1187
# Sensor_ID bytes     % packets Description
          0    48  4.21       2         GPS
          1    45  3.95       1    Pressure
          2    43  3.78       1 Temperature
          3    43  3.78       1    Salinity
          4    68  5.97       1        Fall
          5    53  4.65       1        Rise
          6    70  6.15       1        Pump
          7    26  2.28       1 Drift_times
          8    44  3.86       1     Surface
          9    26  2.28       1  HR/Drift_P
         10    26  2.28       1  HR/Drift_T
         11    26  2.28       1  HR/Drift_S
         13   404 35.47       2  Parameters
         14   151 13.26       5 Engineering
         15    66  5.79       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  997 1 2022/08/07 04:01 +67.55560  -54.75854    0  8 24 39 47  0.9     0
G  997 2 2022/08/07 09:29 +67.55321  -54.76278   50  7  0 34 44  1.0     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   997   38    1     3         0
#      Pres      Temp    Salinity
      1.080     4.1730    33.2280
      2.000     4.1730    33.2270
      3.000     4.1730    33.2270
      3.960     4.1720    33.2280
      4.960     4.1710    33.2280
      6.040     4.1700    33.2280
      7.000     4.1700    33.2280
      7.960     4.1710    33.2280
      9.040     4.1700    33.2280
     10.000     4.1700    33.2280
     11.000     4.1710    33.2280
     12.040     4.1700    33.2280
     13.000     4.1710    33.2280
     14.000     4.1710    33.2280
     15.000     4.1700    33.2280
     16.000     4.1700    33.2280
     17.000     4.1700    33.2280
     18.040     4.1700    33.2280
     18.960     4.1700    33.2280
     20.000     4.1700    33.2290
     21.000     4.1700    33.2280
     22.000     4.1700    33.2280
     23.040     4.1700    33.2290
     24.000     4.1700    33.2280
     25.000     4.1700    33.2280
     26.000     4.1700    33.2290
     27.040     4.1700    33.2290
     28.000     4.1700    33.2290
     29.040     4.1700    33.2290
     29.960     4.1710    33.2290
     31.000     4.1700    33.2290
     31.960     4.1700    33.2290
     32.960     4.1700    33.2290
     34.040     4.1700    33.2300
     34.960     4.1700    33.2300
     36.040     4.1700    33.2300
     37.400     4.1690    33.2300
     37.520     4.1690    33.2300
# Drift Data
#      C Npts comp n_var
I0   997    4    1     4
#      Pres       Temp   Salinity     Time
     38.720     4.1890    33.2380        6
     38.720     4.1850    33.2380     3612
     38.680     4.1780    33.2380     7217
     38.400     4.1770    33.2370    10823
# Surface Pressure Time Series
#    C npts n_var               Start
S  997     4     6 2022/08/07 09:28:05
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    84     -0.020      4.169   -999.000      14 2022/08/07 09:29:29
    97      0.020      4.171   -999.000      14 2022/08/07 09:29:42
   108      0.010      4.172   -999.000      14 2022/08/07 09:29:53
   119     -0.030      4.172   -999.000      14 2022/08/07 09:30:04
# Fall Time Series
#  Cycle npts     Start                Date
F00  997   12 713160152 2022/08/07 04:02:32
# time     Pres phase                Date
     0    0.040     1 2022/08/07 04:02:32
   147    5.120    10 2022/08/07 04:04:59
   179   10.280    10 2022/08/07 04:05:31
   212   15.280    10 2022/08/07 04:06:04
   245   20.240    10 2022/08/07 04:06:37
   277   25.120    10 2022/08/07 04:07:09
   295   27.840     2 2022/08/07 04:07:27
   310   30.200    10 2022/08/07 04:07:42
   343   35.320    10 2022/08/07 04:08:15
   646   38.800    10 2022/08/07 04:13:18
   948   38.640    10 2022/08/07 04:18:20
   964   38.640    10 2022/08/07 04:18:36
# Rise Time Series
#  Cycle npts     Start                Date
R00  997    9 713179126 2022/08/07 09:18:46
# time     Pres phase                Date
     0   37.440     7 2022/08/07 09:18:46
   232   32.520    13 2022/08/07 09:22:38
   285   27.960    13 2022/08/07 09:23:31
   337   22.960    13 2022/08/07 09:24:23
   390   17.920    13 2022/08/07 09:25:16
   442   12.680    13 2022/08/07 09:26:08
   495    7.440    13 2022/08/07 09:27:01
   547    1.840    13 2022/08/07 09:27:53
   558    0.760     8 2022/08/07 09:28:04
# Pump Time Series
#   Cycle npts
B02   997    6
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    27.840  361   15.11       0    621    -9     2
    37.440  127   15.21       0    623    -9     7
    36.960   21   15.20       0    626    -9    13
    32.520   21   15.20       0    628    -9    13
    11.800   21   15.19       0    629    -9    13
     0.040   81   15.18       0    635    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   997  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  997 2022/08/07 04:02:30
# Pcase vacuums
   6.400   6.180
# Oil valve positions
   1.688   0.505   0.672
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  997 2022/08/07 04:18:36 9999/99/99 99:99:99
# Pcase vacuums
   6.180   6.140
# Average 
#   P      T      S
   38.640   4.182  33.238
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  997 2022/08/07 09:18:46 9999/99/99 99:99:99
# Pcase vacuums
   6.140   6.290
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  997 2022/08/07 09:28:04
# number of packets previous packets
      4               4
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.290   6.380
# Air Pump max current time  max pressure
     119       386     0.000
# Air Valve Positions
   1.280   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.532
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   997      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  997  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.000    289   5.930
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  997       197    2022/08/07 09:35:16
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      997	(cycle number) |
     MPS =        0	(phase) |
     MPC =      388	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      994	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      994	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      994	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =      994	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

