# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   994  159 000000000432020  5495     0   3 2022/08/06, 16:49:16,  67.5361  -54.8004   3      0 6,14,14
SBD 11146   994  340 000000000432020  5496     0   2 2022/08/06, 16:49:42,  67.5361  -54.8004   3      0 13
SBD 11146   994  329 000000000432020  5497     0   1 2022/08/06, 16:50:05,  67.5361  -54.8004   2      0 1,2,3,0,8,15,13
SBD 11146   994  332 000000000432020  5498     0   0 2022/08/06, 16:50:22,  67.5632  -54.7463   2      0 0,4,14,14,9,10,11,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1112           48        1160
# Sensor_ID bytes     % packets Description
          0    48  4.32       2         GPS
          1    45  4.05       1    Pressure
          2    53  4.77       1 Temperature
          3    45  4.05       1    Salinity
          4    68  6.12       1        Fall
          5    58  5.22       1        Rise
          6    59  5.31       1        Pump
          7    26  2.34       1 Drift_times
          8     8  0.72       1     Surface
          9    28  2.52       1  HR/Drift_P
         10    27  2.43       1  HR/Drift_T
         11    26  2.34       1  HR/Drift_S
         13   404 36.33       2  Parameters
         14   151 13.58       5 Engineering
         15    66  5.94       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  994 1 2022/08/06 11:16 +67.53509  -54.74243    0  4 32 39 44  4.3     0
G  994 2 2022/08/06 16:43 +67.52776  -54.77167   80  8  0 34 43  1.0     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   994   41    1     3         0
#      Pres      Temp    Salinity
      1.080     4.3040    33.2350
      2.000     4.2780    33.2330
      3.040     4.2640    33.2360
      3.960     4.2320    33.2360
      5.000     4.1530    33.2360
      6.000     4.1410    33.2360
      7.000     4.1370    33.2370
      7.960     4.1330    33.2370
      9.000     4.1280    33.2380
     10.000     4.1270    33.2380
     10.960     4.1230    33.2390
     12.000     4.1200    33.2410
     13.000     4.1190    33.2420
     13.960     4.1170    33.2420
     14.960     4.1170    33.2420
     15.960     4.1160    33.2430
     17.040     4.1140    33.2430
     18.040     4.1140    33.2430
     19.000     4.1160    33.2440
     20.000     4.1170    33.2440
     21.000     4.1160    33.2440
     22.000     4.1160    33.2450
     22.960     4.1160    33.2460
     24.000     4.1150    33.2460
     24.960     4.1150    33.2460
     26.040     4.1140    33.2460
     26.960     4.1150    33.2460
     27.960     4.1150    33.2460
     29.040     4.1150    33.2470
     30.040     4.1150    33.2470
     31.000     4.1160    33.2470
     32.000     4.1160    33.2470
     33.000     4.1160    33.2470
     34.040     4.1160    33.2470
     35.000     4.1170    33.2470
     36.080     4.1180    33.2480
     37.000     4.1190    33.2500
     38.040     4.1190    33.2500
     39.040     4.1190    33.2490
     40.080     4.1190    33.2500
     40.920     4.1200    33.2500
# Drift Data
#      C Npts comp n_var
I0   994    4    1     4
#      Pres       Temp   Salinity     Time
     35.280     4.1150    33.2340        6
     38.080     4.0990    33.2400     3612
     34.000     4.1080    33.2470     7217
     37.120     4.1230    33.2520    10823
# Fall Time Series
#  Cycle npts     Start                Date
F00  994   12 713099833 2022/08/06 11:17:13
# time     Pres phase                Date
     0    0.080     1 2022/08/06 11:17:13
   143    5.040    10 2022/08/06 11:19:36
   176   10.200    10 2022/08/06 11:20:09
   208   15.400    10 2022/08/06 11:20:41
   241   20.280    10 2022/08/06 11:21:14
   273   25.560    10 2022/08/06 11:21:46
   288   28.080     2 2022/08/06 11:22:01
   300   30.520    10 2022/08/06 11:22:13
   326   35.160    10 2022/08/06 11:22:39
   626   35.880    10 2022/08/06 11:27:39
   928   35.240    10 2022/08/06 11:32:41
   957   35.400    10 2022/08/06 11:33:10
# Rise Time Series
#  Cycle npts     Start                Date
R00  994   10 713118800 2022/08/06 16:33:20
# time     Pres phase                Date
     0   41.400     7 2022/08/06 16:33:20
   127   36.400    13 2022/08/06 16:35:27
   190   31.200    13 2022/08/06 16:36:30
   243   26.480    13 2022/08/06 16:37:23
   295   21.440    13 2022/08/06 16:38:15
   348   16.440    13 2022/08/06 16:39:08
   400   11.520    13 2022/08/06 16:40:00
   453    6.440    13 2022/08/06 16:40:53
   505    1.280    13 2022/08/06 16:41:45
   537    0.040     8 2022/08/06 16:42:17
# Pump Time Series
#   Cycle npts
B02   994    5
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    28.080  362   15.13       0    621    -9     2
    41.400  127   15.22       0    623    -9     7
    32.080   21   15.21       0    626    -9    13
    12.640   21   15.22       0    627    -9    13
     0.040   81   15.20       0    632    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   994  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  994 2022/08/06 11:17:10
# Pcase vacuums
   6.370   6.180
# Oil valve positions
   1.688   0.506   0.670
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  994 2022/08/06 11:33:10 9999/99/99 99:99:99
# Pcase vacuums
   6.180   6.140
# Average 
#   P      T      S
   36.120   4.111  33.243
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  994 2022/08/06 16:33:20 9999/99/99 99:99:99
# Pcase vacuums
   6.140   6.270
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  994 2022/08/06 16:42:17
# number of packets previous packets
      4               5
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.270   6.360
# Air Pump max current time  max pressure
     119       385     0.000
# Air Valve Positions
   1.280   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.533
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   994      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  994  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.040    282   5.930
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  994       197    2022/08/06 16:50:05
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      994	(cycle number) |
     MPS =        0	(phase) |
     MPC =      385	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      991	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      991	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      991	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =      991	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

