# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   982  170 000000000432020  5447     0   3 2022/08/03, 21:40:01,  67.5361  -54.8004   2      0 6,14,14
SBD 11146   982  340 000000000432020  5448     0   2 2022/08/03, 21:40:30,  67.5361  -54.8004   2      0 13
SBD 11146   982  328 000000000432020  5449     0   1 2022/08/03, 21:41:00,  67.5361  -54.8004   3      0 1,2,3,0,8,15,13
SBD 11146   982  326 000000000432020  5450     0   0 2022/08/03, 21:41:21,  67.5090  -54.8545   3      0 0,4,14,14,9,10,11,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1116           48        1164
# Sensor_ID bytes     % packets Description
          0    48  4.30       2         GPS
          1    46  4.12       1    Pressure
          2    52  4.66       1 Temperature
          3    44  3.94       1    Salinity
          4    68  6.09       1        Fall
          5    53  4.75       1        Rise
          6    70  6.27       1        Pump
          7    26  2.33       1 Drift_times
          8     8  0.72       1     Surface
          9    27  2.42       1  HR/Drift_P
         10    27  2.42       1  HR/Drift_T
         11    26  2.33       1  HR/Drift_S
         13   404 36.20       2  Parameters
         14   151 13.53       5 Engineering
         15    66  5.91       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  982 1 2022/08/03 16:07 +67.51335  -54.81713    0  5 26 38 44 12.9     0
G  982 2 2022/08/03 21:35 +67.52601  -54.82930  140  5  0 33 44  8.4     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   982   39    1     3         0
#      Pres      Temp    Salinity
      1.120     3.9660    33.1890
      1.960     3.9660    33.1890
      2.920     3.9650    33.1890
      4.000     3.9650    33.1890
      4.960     3.9660    33.1890
      6.040     3.9660    33.1890
      7.000     3.9660    33.1890
      8.080     3.9660    33.1890
      8.960     3.9660    33.1890
     10.000     3.9660    33.1890
     11.000     3.9660    33.1890
     12.000     3.9640    33.1890
     13.000     3.9650    33.1890
     13.920     3.9650    33.1890
     15.000     3.9650    33.1890
     15.920     3.9640    33.1890
     16.960     3.9630    33.1900
     18.000     3.9620    33.1890
     19.080     3.9610    33.1890
     20.040     3.9590    33.1900
     21.000     3.9570    33.1900
     22.000     3.9590    33.1890
     23.000     3.9610    33.1890
     24.040     3.9600    33.1890
     25.000     3.9550    33.1900
     26.040     3.9480    33.1910
     27.000     3.9480    33.1900
     28.000     3.9440    33.1910
     29.000     3.9420    33.1910
     30.000     3.9530    33.1900
     31.000     3.9470    33.1910
     32.040     3.9400    33.1920
     33.040     3.9400    33.1920
     34.040     3.9400    33.1910
     35.000     3.9400    33.1910
     36.000     3.9400    33.1920
     37.000     3.9380    33.1920
     38.080     3.9380    33.1920
     38.760     3.9380    33.1920
# Drift Data
#      C Npts comp n_var
I0   982    4    1     4
#      Pres       Temp   Salinity     Time
     34.880     3.9160    33.1910        5
     37.000     3.9290    33.1880     3611
     39.880     3.9400    33.1880     7217
     41.320     3.9400    33.1910    10822
# Fall Time Series
#  Cycle npts     Start                Date
F00  982   12 712858064 2022/08/03 16:07:44
# time     Pres phase                Date
     0    0.000     1 2022/08/03 16:07:44
   192    5.200    10 2022/08/03 16:10:56
   225   10.080    10 2022/08/03 16:11:29
   261   15.200    10 2022/08/03 16:12:05
   293   20.120    10 2022/08/03 16:12:37
   326   25.160    10 2022/08/03 16:13:10
   343   27.840     2 2022/08/03 16:13:27
   359   30.120    10 2022/08/03 16:13:43
   395   35.040    10 2022/08/03 16:14:19
   697   34.960    10 2022/08/03 16:19:21
   999   34.680    10 2022/08/03 16:24:23
  1009   34.800    10 2022/08/03 16:24:33
# Rise Time Series
#  Cycle npts     Start                Date
R00  982    9 712877083 2022/08/03 21:24:43
# time     Pres phase                Date
     0   38.880     7 2022/08/03 21:24:43
   159   33.880    13 2022/08/03 21:27:22
   222   28.680    13 2022/08/03 21:28:25
   285   23.280    13 2022/08/03 21:29:28
   337   18.480    13 2022/08/03 21:30:20
   390   13.120    13 2022/08/03 21:31:13
   432    8.560    13 2022/08/03 21:31:55
   484    2.880    13 2022/08/03 21:32:47
   506    0.920     8 2022/08/03 21:33:09
# Pump Time Series
#   Cycle npts
B02   982    6
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    27.840  362   15.12       0    622    -9     2
    38.880  127   15.22       0    624    -9     7
    33.080   21   15.22       0    627    -9    13
    26.880   21   15.22       0    629    -9    13
    21.520   21   15.21       0    630    -9    13
     0.040   79   15.20       0    635    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   982  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  982 2022/08/03 16:07:42
# Pcase vacuums
   6.440   6.190
# Oil valve positions
   1.687   0.507   0.669
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  982 2022/08/03 16:24:34 9999/99/99 99:99:99
# Pcase vacuums
   6.190   6.150
# Average 
#   P      T      S
   38.280   3.931  33.189
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  982 2022/08/03 21:24:43 9999/99/99 99:99:99
# Pcase vacuums
   6.150   6.300
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  982 2022/08/03 21:33:10
# number of packets previous packets
      4               5
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.300   6.390
# Air Pump max current time  max pressure
     119       384     0.000
# Air Valve Positions
   1.280   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.533
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   982      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  982  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.000    338   5.940
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  982       197    2022/08/03 21:41:00
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      982	(cycle number) |
     MPS =        0	(phase) |
     MPC =      373	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      979	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      979	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      979	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =      979	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

