# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   979  204 000000000432020  5434     0   3 2022/08/03, 04:51:44,  67.5987  -55.1621  33      0 6,14,14
SBD 11146   979  340 000000000432020  5435     0   2 2022/08/03, 04:52:03,  67.5689  -54.8772  27      0 13
SBD 11146   979  329 000000000432020  5436     0   1 2022/08/03, 04:52:15,  67.5552  -54.8772  24      0 0,1,2,3,8,15,13
SBD 11146   979  330 000000000432020  5437     0   0 2022/08/03, 04:52:33,  67.5552  -54.8772  21      0 0,4,14,14,9,10,11,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1155           48        1203
# Sensor_ID bytes     % packets Description
          0    48  4.16       2         GPS
          1    51  4.42       1    Pressure
          2    46  3.98       1 Temperature
          3    46  3.98       1    Salinity
          4    68  5.89       1        Fall
          5    58  5.02       1        Rise
          6    92  7.97       1        Pump
          7    26  2.25       1 Drift_times
          8     8  0.69       1     Surface
          9    27  2.34       1  HR/Drift_P
         10    26  2.25       1  HR/Drift_T
         11    26  2.25       1  HR/Drift_S
         13   404 34.98       2  Parameters
         14   163 14.11       5 Engineering
         15    66  5.71       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  979 1 2022/08/02 23:16 +67.51865  -54.82206   20  1  0  0  0  2.4     0
G  979 2 2022/08/03 04:45 +67.53493  -54.81574   30  5  0 26 35  7.3     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   979   44    1     3         0
#      Pres      Temp    Salinity
      0.960     3.8740    33.2060
      2.040     3.8740    33.2060
      3.000     3.8750    33.2060
      4.040     3.8740    33.2060
      5.040     3.8740    33.2060
      5.960     3.8740    33.2060
      7.040     3.8740    33.2060
      8.040     3.8750    33.2060
      9.080     3.8750    33.2060
     10.000     3.8750    33.2060
     11.000     3.8740    33.2050
     12.000     3.8740    33.2050
     13.040     3.8760    33.2060
     14.000     3.8760    33.2060
     15.040     3.8760    33.2060
     16.000     3.8740    33.2050
     17.000     3.8750    33.2050
     18.080     3.8750    33.2050
     18.960     3.8750    33.2050
     20.040     3.8750    33.2040
     21.080     3.8750    33.2050
     22.040     3.8750    33.2040
     23.000     3.8750    33.2040
     24.040     3.8750    33.2040
     25.040     3.8750    33.2040
     26.000     3.8750    33.2050
     27.000     3.8750    33.2050
     28.000     3.8760    33.2050
     29.040     3.8760    33.2050
     30.000     3.8760    33.2050
     30.920     3.8760    33.2050
     32.000     3.8760    33.2050
     33.000     3.8760    33.2060
     34.000     3.8760    33.2050
     35.000     3.8760    33.2050
     35.960     3.8760    33.2050
     37.000     3.8760    33.2050
     38.000     3.8760    33.2050
     39.000     3.8760    33.2050
     40.040     3.8760    33.2050
     41.080     3.8760    33.2050
     42.000     3.8770    33.2050
     43.160     3.8770    33.2050
     43.600     3.8760    33.2050
# Drift Data
#      C Npts comp n_var
I0   979    4    1     4
#      Pres       Temp   Salinity     Time
     36.280     3.8960    33.1870        5
     36.280     3.8900    33.1890     3611
     36.560     3.8910    33.1910     7217
     37.600     3.8860    33.1930    10822
# Fall Time Series
#  Cycle npts     Start                Date
F00  979   12 712797418 2022/08/02 23:16:58
# time     Pres phase                Date
     0    0.160     1 2022/08/02 23:16:58
   140    5.360    10 2022/08/02 23:19:18
   169   10.520    10 2022/08/02 23:19:47
   198   15.120    10 2022/08/02 23:20:16
   231   20.080    10 2022/08/02 23:20:49
   267   25.360    10 2022/08/02 23:21:25
   284   27.960     2 2022/08/02 23:21:42
   300   30.160    10 2022/08/02 23:21:58
   332   35.120    10 2022/08/02 23:22:30
   632   36.440    10 2022/08/02 23:27:30
   934   36.360    10 2022/08/02 23:32:32
   950   36.200    10 2022/08/02 23:32:48
# Rise Time Series
#  Cycle npts     Start                Date
R00  979   10 712816378 2022/08/03 04:32:58
# time     Pres phase                Date
     0   43.760     7 2022/08/03 04:32:58
   159   38.800    13 2022/08/03 04:35:37
   222   33.720    13 2022/08/03 04:36:40
   295   28.360    13 2022/08/03 04:37:53
   348   23.680    13 2022/08/03 04:38:46
   400   18.680    13 2022/08/03 04:39:38
   453   14.000    13 2022/08/03 04:40:31
   516    8.360    13 2022/08/03 04:41:34
   568    3.480    13 2022/08/03 04:42:26
   681    0.160     8 2022/08/03 04:44:19
# Pump Time Series
#   Cycle npts
B02   979    8
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    27.960  362   15.11       0    623    -9     2
    43.760  127   15.22       0    624    -9     7
    38.000   21   15.22       0    627    -9    13
    32.040   21   15.22       0    629    -9    13
    27.480   21   15.21       0    630    -9    13
    10.240   21   15.21       0    631    -9    13
  9999.000   52   15.20       0    635    -9    13
     0.120   71   15.19       0    641    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   979  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  979 2022/08/02 23:16:55
# Pcase vacuums
   6.370   6.190
# Oil valve positions
   1.686   0.508   0.669
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  979 2022/08/02 23:32:49 9999/99/99 99:99:99
# Pcase vacuums
   6.190   6.150
# Average 
#   P      T      S
   36.680   3.891  33.190
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  979 2022/08/03 04:32:58 9999/99/99 99:99:99
# Pcase vacuums
   6.150   6.350
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  979 2022/08/03 04:44:20
# number of packets previous packets
      4               8
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.350   6.440
# Air Pump max current time  max pressure
     119       396     0.000
# Air Valve Positions
   1.280   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.532
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   979      2
#               Date   Code Description
 2022/08/03 04:43:17 0x0114 AscendCtd
 2022/08/03 04:43:47 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  979  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.160    279   5.950
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  979       197    2022/08/03 04:52:15
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      979	(cycle number) |
     MPS =        0	(phase) |
     MPC =      370	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      976	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      976	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      976	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =      976	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

