# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   944  340 000000000432020  5273     0   3 2022/07/26, 00:02:19,  67.4029  -54.8093   3      0 13
SBD 11146   944  335 000000000432020  5274     0   2 2022/07/26, 00:02:29,  67.4029  -54.8093   3      0 15,13,6,14,14
SBD 11146   944  303 000000000432020  5275     0   1 2022/07/26, 00:02:42,  67.4299  -54.7555   6      0 5,14,0,1,2,3,8
SBD 11146   944  336 000000000432020  5276     0   0 2022/07/26, 00:02:53,  67.4029  -54.8093   4      0 0,4,14,14,9,10,11,7
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1266           48        1314
# Sensor_ID bytes     % packets Description
          0    48  3.79       2         GPS
          1    42  3.32       1    Pressure
          2    58  4.58       1 Temperature
          3    42  3.32       1    Salinity
          4   138 10.90       1        Fall
          5    98  7.74       1        Rise
          6    81  6.40       1        Pump
          7    28  2.21       1 Drift_times
          8     8  0.63       1     Surface
          9    34  2.69       1  HR/Drift_P
         10    28  2.21       1  HR/Drift_T
         11    28  2.21       1  HR/Drift_S
         13   404 31.91       2  Parameters
         14   163 12.88       5 Engineering
         15    66  5.21       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  944 1 2022/07/25 18:24 +67.36524  -54.77734    0  5 29 36 45  1.5     0
G  944 2 2022/07/25 23:56 +67.40845  -54.79564   30  5  0 30 44  1.4     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   944   36    1     3         0
#      Pres      Temp    Salinity
      1.040     3.8690    33.3350
      2.000     3.8520    33.3420
      2.960     3.8530    33.3430
      3.960     3.8320    33.3450
      5.000     3.8290    33.3470
      6.040     3.8100    33.3460
      7.000     3.8160    33.3450
      7.960     3.8040    33.3480
      9.000     3.7530    33.3460
     10.000     3.7360    33.3480
     11.040     3.7340    33.3460
     12.040     3.7110    33.3470
     13.000     3.6940    33.3480
     13.960     3.6900    33.3470
     14.960     3.6910    33.3470
     16.000     3.7010    33.3480
     17.040     3.6960    33.3480
     18.040     3.6970    33.3470
     19.080     3.7060    33.3480
     20.080     3.7050    33.3480
     21.000     3.7070    33.3470
     22.040     3.6990    33.3480
     23.040     3.6970    33.3480
     24.000     3.6930    33.3470
     25.040     3.6970    33.3480
     26.000     3.6930    33.3470
     27.040     3.6900    33.3470
     28.000     3.6860    33.3470
     29.000     3.6830    33.3470
     30.040     3.6800    33.3480
     30.960     3.6830    33.3470
     32.000     3.6790    33.3470
     33.040     3.6760    33.3470
     34.040     3.6760    33.3470
     35.080     3.6760    33.3470
     36.040     3.6760    33.3470
# Drift Data
#      C Npts comp n_var
I0   944    8    1     4
#      Pres       Temp   Salinity     Time
     41.440     3.6300    33.3430        6
     41.800     3.6360    33.3430     1812
     39.880     3.6360    33.3440     3617
     39.320     3.6400    33.3440     5423
     33.960     3.6410    33.3450     7228
     34.320     3.6470    33.3450     9034
     40.400     3.6520    33.3450    10840
     37.520     3.6620    33.3460    12645
# Fall Time Series
#  Cycle npts     Start                Date
F00  944   26 712088679 2022/07/25 18:24:39
# time     Pres phase                Date
     0    0.120     1 2022/07/25 18:24:39
   120    3.080    10 2022/07/25 18:26:39
   143    6.280    10 2022/07/25 18:27:02
   163    9.240    10 2022/07/25 18:27:22
   182   12.200    10 2022/07/25 18:27:41
   202   15.200    10 2022/07/25 18:28:01
   221   18.040    10 2022/07/25 18:28:20
   241   21.080    10 2022/07/25 18:28:40
   260   24.320    10 2022/07/25 18:28:59
   280   27.440    10 2022/07/25 18:29:19
   285   28.160     2 2022/07/25 18:29:24
   297   30.000    10 2022/07/25 18:29:36
   320   33.400    10 2022/07/25 18:29:59
   339   36.240    10 2022/07/25 18:30:18
   359   39.120    10 2022/07/25 18:30:38
   660   41.600    10 2022/07/25 18:35:39
   960   41.680    10 2022/07/25 18:40:39
  1261   41.600    10 2022/07/25 18:45:40
  1561   41.600    10 2022/07/25 18:50:40
  1862   41.560    10 2022/07/25 18:55:41
  2162   41.840    10 2022/07/25 19:00:41
  2449   41.840     3 2022/07/25 19:05:28
  2463   41.920    10 2022/07/25 19:05:42
  2763   42.040    10 2022/07/25 19:10:42
  3064   40.800    10 2022/07/25 19:15:43
  3088   41.200    10 2022/07/25 19:16:07
# Rise Time Series
#  Cycle npts     Start                Date
R00  944   18 712107977 2022/07/25 23:46:17
# time     Pres phase                Date
     0   35.800     7 2022/07/25 23:46:17
    33   36.160    13 2022/07/25 23:46:50
    64   35.840    13 2022/07/25 23:47:21
    96   34.520    13 2022/07/25 23:47:53
   127   32.320    13 2022/07/25 23:48:24
   159   30.520    13 2022/07/25 23:48:56
   190   28.360    13 2022/07/25 23:49:27
   222   26.240    13 2022/07/25 23:49:59
   253   23.640    13 2022/07/25 23:50:30
   285   20.800    13 2022/07/25 23:51:02
   316   17.160    13 2022/07/25 23:51:33
   337   14.880    13 2022/07/25 23:51:54
   369   12.160    13 2022/07/25 23:52:26
   400    9.280    13 2022/07/25 23:52:57
   432    6.280    13 2022/07/25 23:53:29
   463    2.760    13 2022/07/25 23:54:00
   555    0.160    13 2022/07/25 23:55:32
   566    0.120     8 2022/07/25 23:55:43
# Pump Time Series
#   Cycle npts
B02   944    7
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    28.160  361   15.12       0    624    -9     2
    35.800  127   15.21       0    627    -9     7
    29.960   21   15.21       0    629    -9    13
    25.440   21   15.21       0    631    -9    13
    12.160   21   15.21       0    632    -9    13
  9999.000   51   15.19       0    635    -9    13
     0.160   60   15.18       0    640    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   944  2.4   18    90    60   600    51    10   270      8     1    4020    5  0.2929
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  944 2022/07/25 18:24:37
# Pcase vacuums
   6.370   6.180
# Oil valve positions
   1.684   0.509   0.669
# Seek attempts and pump time
      1      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  944 2022/07/25 19:16:07 9999/99/99 99:99:99
# Pcase vacuums
   6.170   6.180
# Average 
#   P      T      S
   38.560   3.643  33.344
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  944 2022/07/25 23:46:17 9999/99/99 99:99:99
# Pcase vacuums
   6.170   6.350
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  944 2022/07/25 23:55:52
# number of packets previous packets
      4               5
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.350   6.430
# Air Pump max current time  max pressure
     119       360     0.000
# Air Valve Positions
   1.280   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.534
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   944      2
#               Date   Code Description
 2022/07/25 23:54:41 0x0114 AscendCtd
 2022/07/25 23:55:11 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  944  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.040    279   5.960
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  944       197    2022/07/26 00:02:29
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      944	(cycle number) |
     MPS =        0	(phase) |
     MPC =      335	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      941	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      900	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        3	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      941	(cycle) |
  PTO[0] =    16200	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      941	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        3	(dBar) |
     ADT =       30	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =      941	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

