# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   882  231 000000000432020  4972     0   4 2022/07/11, 05:52:40,  67.3201  -54.3759   2      0 6,14,14
SBD 11146   882  340 000000000432020  4973     0   3 2022/07/11, 05:52:52,  67.3277  -54.2461   4      0 13
SBD 11146   882  262 000000000432020  4974     0   2 2022/07/11, 05:53:02,  67.3201  -54.3759   2      0 3,0,8,15,13
SBD 11146   882  319 000000000432020  4975     0   1 2022/07/11, 05:53:13,  67.3201  -54.3759   2      0 11,7,5,14,1,2
SBD 11146   882  324 000000000432020  4976     0   0 2022/07/11, 05:53:25,  67.3201  -54.3759   2      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1416           60        1476
# Sensor_ID bytes     % packets Description
          0    48  3.39       2         GPS
          1    46  3.25       1    Pressure
          2    62  4.38       1 Temperature
          3    58  4.10       1    Salinity
          4   108  7.63       1        Fall
          5    78  5.51       1        Rise
          6   125  8.83       1        Pump
          7    51  3.60       1 Drift_times
          8    26  1.84       1     Surface
          9    69  4.87       1  HR/Drift_P
         10    67  4.73       1  HR/Drift_T
         11    51  3.60       1  HR/Drift_S
         13   404 28.53       2  Parameters
         14   157 11.09       5 Engineering
         15    66  4.66       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  882 1 2022/07/11 00:12 +67.34144  -54.33051    0  5 25 37 42  3.4     0
G  882 2 2022/07/11 05:47 +67.32093  -54.31193   50  6  0 34 45  1.0     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   882   43    1     3         0
#      Pres      Temp    Salinity
      1.000     2.8910    33.2710
      2.040     2.8920    33.2710
      2.960     2.8930    33.2710
      4.040     2.8930    33.2710
      5.040     2.8930    33.2710
      6.000     2.8920    33.2710
      7.000     2.8920    33.2710
      7.960     2.8950    33.2720
      8.920     2.9040    33.2730
     10.040     2.9040    33.2730
     11.000     2.9040    33.2730
     12.000     2.9040    33.2730
     13.040     2.9050    33.2730
     13.920     2.9040    33.2730
     15.040     2.9050    33.2730
     16.000     2.9100    33.2740
     16.920     2.9100    33.2760
     18.000     2.9100    33.2750
     19.000     2.9120    33.2770
     20.040     2.9110    33.2770
     21.080     2.9130    33.2790
     22.040     2.9300    33.2910
     22.960     2.9330    33.2940
     24.000     2.9410    33.3030
     24.960     2.9440    33.3070
     26.000     2.9410    33.3030
     27.040     2.9410    33.3030
     28.000     2.9480    33.3090
     29.040     2.9430    33.3060
     30.040     2.9480    33.3080
     31.000     2.9570    33.3170
     32.000     2.9580    33.3170
     33.040     2.9590    33.3180
     34.040     2.9610    33.3200
     34.920     2.9610    33.3210
     36.000     2.9660    33.3290
     37.000     2.9720    33.3460
     38.000     2.9720    33.3460
     38.960     2.9730    33.3480
     40.000     2.9730    33.3480
     41.080     2.9730    33.3490
     42.080     2.9730    33.3510
     42.880     2.9720    33.3530
# Drift Data
#      C Npts comp n_var
I0   882   54    1     4
#      Pres       Temp   Salinity     Time
     43.200     2.8290    33.3920        6
     43.120     2.8270    33.3930      311
     43.120     2.8290    33.3920      617
     43.160     2.8320    33.3920      922
     43.120     2.8350    33.3910     1228
     43.040     2.8380    33.3910     1533
     43.040     2.8460    33.3900     1838
     42.960     2.8480    33.3930     2144
     42.680     2.8490    33.3910     2449
     42.600     2.8640    33.3950     2755
     42.600     2.8680    33.4040     3060
     42.680     2.8680    33.4060     3366
     42.720     2.8680    33.4060     3671
     42.840     2.8700    33.4060     3976
     42.960     2.8710    33.4060     4282
     43.040     2.8700    33.4070     4587
     43.120     2.8710    33.4060     4893
     43.160     2.8730    33.4060     5198
     42.920     2.8750    33.4060     5504
     42.760     2.8780    33.4040     5809
     41.040     2.8830    33.4030     6114
     41.320     2.8830    33.4030     6420
     41.920     2.8830    33.4030     6725
     42.160     2.8900    33.4020     7031
     42.400     2.8940    33.4010     7336
     42.600     2.8960    33.4000     7641
     43.000     2.8980    33.3990     7947
     43.480     2.9020    33.3970     8252
     43.960     2.9040    33.3960     8558
     44.640     2.9070    33.3950     8863
     45.440     2.9100    33.3950     9169
     46.280     2.9150    33.3930     9474
     47.680     2.9180    33.3920     9779
     48.240     2.9210    33.3910    10085
     49.320     2.9280    33.3880    10390
     49.960     2.9340    33.3850    10696
     50.560     2.9330    33.3860    11001
     50.960     2.9370    33.3840    11306
     51.280     2.9400    33.3820    11612
     51.360     2.9460    33.3810    11917
     51.160     2.9480    33.3790    12223
     50.880     2.9510    33.3770    12528
     50.400     2.9550    33.3760    12834
     49.440     2.9630    33.3720    13139
     48.360     2.9570    33.3750    13444
     47.400     2.9580    33.3740    13750
     46.560     2.9620    33.3710    14055
     45.800     2.9630    33.3690    14361
     45.160     2.9670    33.3640    14666
     44.120     2.9690    33.3610    14971
     43.440     2.9690    33.3610    15277
     43.120     2.9700    33.3580    15582
     43.240     2.9720    33.3560    15888
     43.040     2.9720    33.3540    16193
# Surface Pressure Time Series
#    C npts n_var               Start
S  882     2     6 2022/07/11 05:46:09
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    69     -0.050      2.544   -999.000      14 2022/07/11 05:47:18
    91     -0.060      2.427   -999.000      14 2022/07/11 05:47:40
# Fall Time Series
#  Cycle npts     Start                Date
F00  882   20 710813580 2022/07/11 00:13:00
# time     Pres phase                Date
     0   -0.040     1 2022/07/11 00:13:00
   186    5.040    10 2022/07/11 00:16:06
   235   10.280    10 2022/07/11 00:16:55
   277   15.360    10 2022/07/11 00:17:37
   316   20.240    10 2022/07/11 00:18:16
   352   25.160    10 2022/07/11 00:18:52
   373   27.960     2 2022/07/11 00:19:13
   391   30.240    10 2022/07/11 00:19:31
   427   35.080    10 2022/07/11 00:20:07
   470   40.160    10 2022/07/11 00:20:50
   772   43.480    10 2022/07/11 00:25:52
  1074   43.440    10 2022/07/11 00:30:54
  1375   43.360    10 2022/07/11 00:35:55
  1675   43.320    10 2022/07/11 00:40:55
  1976   43.280    10 2022/07/11 00:45:56
  2276   43.240    10 2022/07/11 00:50:56
  2417   43.240     3 2022/07/11 00:53:17
  2577   43.240    10 2022/07/11 00:55:57
  2879   43.160    10 2022/07/11 01:00:59
  3056   43.160    10 2022/07/11 01:03:56
# Rise Time Series
#  Cycle npts     Start                Date
R00  882   14 710832852 2022/07/11 05:34:12
# time     Pres phase                Date
     0   42.920     7 2022/07/11 05:34:12
    65   42.920    13 2022/07/11 05:35:17
   128   42.400    13 2022/07/11 05:36:20
   191   41.120    13 2022/07/11 05:37:23
   254   38.760    13 2022/07/11 05:38:26
   317   35.160    13 2022/07/11 05:39:29
   380   31.240    13 2022/07/11 05:40:32
   432   27.000    13 2022/07/11 05:41:24
   485   21.720    13 2022/07/11 05:42:17
   537   15.880    13 2022/07/11 05:43:09
   579   10.840    13 2022/07/11 05:43:51
   621    5.600    13 2022/07/11 05:44:33
   704   -0.040    13 2022/07/11 05:45:56
   715   -0.040     8 2022/07/11 05:46:07
# Pump Time Series
#   Cycle npts
B02   882   11
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    27.960  241   15.06       0    617    -9     2
    42.920  107   15.25       0    619    -9     7
    42.400   21   15.26       0    622    -9    13
    41.760   21   15.25       0    623    -9    13
    40.480   21   15.25       0    626    -9    13
    38.760   21   15.24       0    627    -9    13
    36.880   21   15.24       0    629    -9    13
    34.200   21   15.24       0    631    -9    13
    31.240   21   15.23       0    632    -9    13
  9999.000   21   15.23       0    633    -9    13
    -0.040   66   15.21       0    638    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   882  2.4   19    90    60   600    49    10   270      8     1    4020    5  0.2915
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  882 2022/07/11 00:12:58
# Pcase vacuums
   6.290   6.130
# Oil valve positions
   1.683   0.512   0.668
# Seek attempts and pump time
      1      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  882 2022/07/11 01:03:56 9999/99/99 99:99:99
# Pcase vacuums
   6.130   6.130
# Average 
#   P      T      S
   44.960   2.904  33.389
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  882 2022/07/11 05:34:12 9999/99/99 99:99:99
# Pcase vacuums
   6.120   6.330
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  882 2022/07/11 05:46:07
# number of packets previous packets
      5               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.330   6.410
# Air Pump max current time  max pressure
     119       371     0.000
# Air Valve Positions
   1.282   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.534
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   882      1
#               Date   Code Description
 2022/07/11 05:45:35 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  882  1      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.040    367   5.980
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  882       197    2022/07/11 05:53:02
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      882	(cycle number) |
     MPS =        0	(phase) |
     MPC =      273	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      879	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      900	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      240	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65534	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      879	(cycle) |
  PTO[0] =    16200	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   300000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      300	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      879	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      100	(sec) |
     APN =       20	(sec) |
     APF =       20	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =       60	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =      879	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    20000	(ms) |
     SDS =       20	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

