# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   875  244 000000000432020  4942     0   3 2022/07/10, 17:22:58,  67.3158  -54.0410  23      0 13,6,14,14
SBD 11146   875  340 000000000432020  4943     0   2 2022/07/10, 17:24:00,  67.2601  -54.3545  16      0 13
SBD 11146   875  308 000000000432020  4944     0   1 2022/07/10, 17:24:12,  67.2483  -54.1498  15      0 1,2,3,8,15
SBD 11146   875  324 000000000432020  4945     0   0 2022/07/10, 17:24:24,  67.2408  -54.2792  13      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      17         1168           48        1216
# Sensor_ID bytes     % packets Description
          0    48  4.11       2         GPS
          1    49  4.20       1    Pressure
          2    80  6.85       1 Temperature
          3    57  4.88       1    Salinity
          4   178 15.24       1        Fall
          5    23  1.97       1        Rise
          6    37  3.17       1        Pump
          8    44  3.77       1     Surface
         13   404 34.59       2  Parameters
         14   182 15.58       5 Engineering
         15    66  5.65       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  875 1 2022/07/10 16:16 +67.22978  -54.32635    0  8 35 40 44  1.1     0
G  875 2 2022/07/10 17:17 +67.23309  -54.33367   30  6  0 32 45  1.1     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   875   49    1     3         0
#      Pres      Temp    Salinity
      1.040     3.7650    32.9510
      2.000     3.7110    32.9800
      2.960     3.6370    33.0090
      4.000     3.5660    33.0350
      5.080     3.4610    33.0770
      6.040     3.4250    33.0900
      7.000     3.3970    33.1040
      8.040     3.2220    33.2150
      9.040     3.1340    33.2650
     10.040     3.0920    33.2850
     11.040     3.0660    33.2980
     12.040     3.0470    33.3070
     13.000     3.0360    33.3130
     13.960     3.0120    33.3280
     15.000     3.0020    33.3350
     15.960     2.9870    33.3450
     16.960     2.9710    33.3510
     18.000     2.9500    33.3590
     19.000     2.9340    33.3640
     20.000     2.9140    33.3700
     20.960     2.8890    33.3790
     22.000     2.8560    33.3890
     23.040     2.8310    33.3960
     24.040     2.8200    33.4010
     25.000     2.8100    33.4050
     26.040     2.7910    33.4120
     26.960     2.7660    33.4160
     28.000     2.7540    33.4160
     29.000     2.7530    33.4170
     30.000     2.7490    33.4160
     30.960     2.7430    33.4170
     31.920     2.7400    33.4170
     33.000     2.7370    33.4180
     34.040     2.7350    33.4180
     35.000     2.7330    33.4180
     35.960     2.7250    33.4190
     37.000     2.7220    33.4180
     38.000     2.7030    33.4190
     38.960     2.6500    33.4150
     40.000     2.6490    33.4140
     41.000     2.6520    33.4150
     41.960     2.6520    33.4150
     43.000     2.6430    33.4140
     44.000     2.6370    33.4140
     44.960     2.6410    33.4150
     46.000     2.6580    33.4120
     47.000     2.6560    33.4130
     48.040     2.6380    33.4130
     48.920     2.6120    33.4100
# Surface Pressure Time Series
#    C npts n_var               Start
S  875     4     6 2022/07/10 17:16:28
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    70     -0.030      4.186   -999.000      14 2022/07/10 17:17:38
    93     -0.040      4.199   -999.000      14 2022/07/10 17:18:01
   114     -0.030      4.225   -999.000      14 2022/07/10 17:18:22
   135     -0.020      4.406   -999.000      14 2022/07/10 17:18:43
# Fall Time Series
#  Cycle npts     Start                Date
F00  875   34 710784988 2022/07/10 16:16:28
# time     Pres phase                Date
     0    0.000     1 2022/07/10 16:16:28
   199    5.200    10 2022/07/10 16:19:47
   245   10.160    10 2022/07/10 16:20:33
   290   15.280    10 2022/07/10 16:21:18
   329   20.040    10 2022/07/10 16:21:57
   372   25.320    10 2022/07/10 16:22:40
   393   28.000     2 2022/07/10 16:23:01
   411   30.400    10 2022/07/10 16:23:19
   447   35.040    10 2022/07/10 16:23:55
   493   40.320    10 2022/07/10 16:24:41
   538   45.320    10 2022/07/10 16:25:26
   584   50.280    10 2022/07/10 16:26:12
   639   55.200    10 2022/07/10 16:27:07
   731   60.040    10 2022/07/10 16:28:39
   848   55.880    10 2022/07/10 16:30:36
   933   50.840    10 2022/07/10 16:32:01
  1013   45.920    10 2022/07/10 16:33:21
  1088   40.920    10 2022/07/10 16:34:36
  1157   35.960    10 2022/07/10 16:35:45
  1225   30.840    10 2022/07/10 16:36:53
  1289   26.000    10 2022/07/10 16:37:57
  1355   20.840    10 2022/07/10 16:39:03
  1416   15.840    10 2022/07/10 16:40:04
  1473   10.960    10 2022/07/10 16:41:01
  1522    5.960    10 2022/07/10 16:41:50
  1565    0.800    10 2022/07/10 16:42:33
  1866    0.000    10 2022/07/10 16:47:34
  2166    0.000    10 2022/07/10 16:52:34
  2467    0.000    10 2022/07/10 16:57:35
  2564    0.040     3 2022/07/10 16:59:12
  2916    0.000    10 2022/07/10 17:05:04
  3216    0.000    10 2022/07/10 17:10:04
  3517    0.000    10 2022/07/10 17:15:05
  3520    0.000    10 2022/07/10 17:15:08
# Rise Time Series
#  Cycle npts     Start                Date
R00  875    3 710788522 2022/07/10 17:15:22
# time     Pres phase                Date
     0    0.000     7 2022/07/10 17:15:22
    63   -0.040    13 2022/07/10 17:16:25
    65   -0.040     8 2022/07/10 17:16:27
# Pump Time Series
#   Cycle npts
B02   875    3
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    28.000  368   15.09       0    628    -9     2
     0.000   63   15.12       0    622    -9     7
     0.000   60   15.12       0    628    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   875  2.4   18    90    60   600    51    10   270      8     1    4020    5  0.2929
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  875 2022/07/10 16:16:26
# Pcase vacuums
   6.280   6.180
# Oil valve positions
   1.683   0.512   0.673
# Seek attempts and pump time
      1     -3
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  875 2022/07/10 17:15:08 9999/99/99 99:99:99
# Pcase vacuums
   6.180   6.180
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  875 2022/07/10 17:15:22 9999/99/99 99:99:99
# Pcase vacuums
   6.180   6.220
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  875 2022/07/10 17:16:28
# number of packets previous packets
      4               4
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.220   6.310
# Air Pump max current time  max pressure
     119       372     0.000
# Air Valve Positions
   1.282   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.549
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   875      4
#               Date   Code Description
 2022/07/10 17:15:14 0x0615 ParkDepthMin
 2022/07/10 17:15:20 0x0615 ParkDepthMin
 2022/07/10 17:15:42 0x0114 AscendCtd
 2022/07/10 17:16:12 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  875  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.000    386   5.980
   3     0.040      7   6.180
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  875       197    2022/07/10 17:24:00
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      875	(cycle number) |
     MPS =        0	(phase) |
     MPC =      266	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      872	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      900	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      367	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65533	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      872	(cycle) |
  PTO[0] =    16200	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   300000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      300	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      872	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      100	(sec) |
     APN =       20	(sec) |
     APF =       20	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =       60	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =      872	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    20000	(ms) |
     SDS =       20	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

