# Message Decoder for Solo2, Alto and Deep Solo Floats; v220516
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   845   88 000000000432020  4791     0   4 2022/07/03, 23:30:43,  67.2483  -54.1498  10      0 14
SBD 11146   845  340 000000000432020  4792     0   3 2022/07/03, 23:30:55,  67.2483  -54.1498   9      0 13
SBD 11146   845  291 000000000432020  4793     0   2 2022/07/03, 23:31:33,  67.2483  -54.1498   6      0 8,15,13,6,14
SBD 11146   845  328 000000000432020  4794     0   1 2022/07/03, 23:31:52,  67.2483  -54.1498   5      0 5,14,1,2,3,0
SBD 11146   845  336 000000000432020  4795     0   0 2022/07/03, 23:33:46,  67.2483  -54.1498   3      0 0,4,14,14,9,10,11,7
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1323           60        1383
# Sensor_ID bytes     % packets Description
          0    48  3.63       2         GPS
          1    54  4.08       1    Pressure
          2    83  6.27       1 Temperature
          3    78  5.90       1    Salinity
          4    88  6.65       1        Fall
          5    58  4.38       1        Rise
          6    92  6.95       1        Pump
          7    37  2.80       1 Drift_times
          8    26  1.97       1     Surface
          9    41  3.10       1  HR/Drift_P
         10    49  3.70       1  HR/Drift_T
         11    41  3.10       1  HR/Drift_S
         13   404 30.54       2  Parameters
         14   158 11.94       5 Engineering
         15    66  4.99       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  845 1 2022/07/03 17:40 +67.22662  -54.14270    0  5 28 37 43  1.1     0
G  845 2 9999/99/99 99:99 +99.99000 +999.99000    0  0  0  0  0  0.0     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   845   44    1     3         0
#      Pres      Temp    Salinity
      2.320     5.5210    29.0730
      2.640     5.4050    29.2990
      4.000     2.8020    33.0900
      5.040     2.3860    33.3440
      5.960     2.2970    33.3650
      7.000     2.1390    33.3910
      8.040     1.9620    33.4070
      9.000     1.8860    33.4150
     10.040     1.8480    33.4160
     11.040     1.8410    33.4160
     12.000     1.8310    33.4170
     13.040     1.8240    33.4170
     14.040     1.8200    33.4180
     14.960     1.8170    33.4180
     16.000     1.8150    33.4190
     16.960     1.8100    33.4190
     18.000     1.8080    33.4190
     19.040     1.8060    33.4200
     20.040     1.8060    33.4190
     21.000     1.8070    33.4200
     21.960     1.8050    33.4220
     23.000     1.8030    33.4220
     24.040     1.8010    33.4220
     25.080     1.7990    33.4230
     26.040     1.7990    33.4240
     27.000     1.8180    33.4270
     27.960     1.8230    33.4310
     29.000     1.8150    33.4260
     30.000     1.8230    33.4280
     30.960     1.8270    33.4300
     31.960     1.8330    33.4330
     33.000     1.8410    33.4380
     33.960     1.8400    33.4390
     34.960     1.8530    33.4420
     35.960     1.8530    33.4440
     37.000     1.8590    33.4450
     38.000     1.8740    33.4500
     39.000     1.8770    33.4510
     40.000     1.9010    33.4590
     41.000     1.9100    33.4600
     42.040     1.9130    33.4620
     43.040     1.9120    33.4620
     44.000     1.9170    33.4630
     45.200     1.9170    33.4630
# Drift Data
#      C Npts comp n_var
I0   845   26    1     4
#      Pres       Temp   Salinity     Time
     43.280     1.8330    33.4270        6
     43.200     1.8200    33.4260      611
     43.080     1.8170    33.4260     1217
     43.120     1.8030    33.4300     1822
     43.160     1.8050    33.4300     2427
     43.120     1.8040    33.4310     3033
     43.080     1.8070    33.4310     3638
     43.080     1.8060    33.4320     4244
     43.120     1.8040    33.4320     4849
     43.240     1.8050    33.4320     5454
     43.440     1.8070    33.4340     6060
     43.640     1.8070    33.4330     6665
     43.680     1.8070    33.4340     7271
     43.600     1.8090    33.4340     7876
     43.600     1.8120    33.4350     8482
     43.560     1.8120    33.4350     9087
     43.520     1.8070    33.4340     9692
     43.440     1.8050    33.4350    10298
     43.520     1.8080    33.4350    10903
     43.600     1.8150    33.4380    11509
     43.560     1.8150    33.4380    12114
     43.600     1.8180    33.4390    12720
     43.800     1.8350    33.4420    13325
     43.600     1.9290    33.4650    13930
     44.360     1.9290    33.4690    14536
     42.560     1.9160    33.4620    15141
# Surface Pressure Time Series
#    C npts n_var               Start
S  845     2     6 2022/07/03 23:19:45
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    63      2.000      5.555   -999.000      14 2022/07/03 23:20:48
   428     -0.030      5.563   -999.000      14 2022/07/03 23:26:53
# Fall Time Series
#  Cycle npts     Start                Date
F00  845   16 710185246 2022/07/03 17:40:46
# time     Pres phase                Date
     0    0.440     1 2022/07/03 17:40:46
   153    5.080    10 2022/07/03 17:43:19
   189   10.280    10 2022/07/03 17:43:55
   222   15.280    10 2022/07/03 17:44:28
   254   20.480    10 2022/07/03 17:45:00
   275   23.960     2 2022/07/03 17:45:21
   284   25.360    10 2022/07/03 17:45:30
   314   30.080    10 2022/07/03 17:46:00
   346   35.120    10 2022/07/03 17:46:32
   379   40.080    10 2022/07/03 17:47:05
   979   43.480    10 2022/07/03 17:57:05
  1580   42.880    10 2022/07/03 18:07:06
  2181   42.200    10 2022/07/03 18:17:07
  2440   42.000     3 2022/07/03 18:21:26
  2788   43.240    10 2022/07/03 18:27:14
  3389   43.280    10 2022/07/03 18:37:15
# Rise Time Series
#  Cycle npts     Start                Date
R00  845   10 710204845 2022/07/03 23:07:25
# time     Pres phase                Date
     0   45.200     7 2022/07/03 23:07:25
   127   40.360    13 2022/07/03 23:09:32
   191   35.120    13 2022/07/03 23:10:36
   243   30.280    13 2022/07/03 23:11:28
   295   25.600    13 2022/07/03 23:12:20
   348   20.600    13 2022/07/03 23:13:13
   400   15.360    13 2022/07/03 23:14:05
   453   10.440    13 2022/07/03 23:14:58
   505    5.480    13 2022/07/03 23:15:50
   737    1.920     8 2022/07/03 23:19:42
# Pump Time Series
#   Cycle npts
B02   845    8
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    23.960  362   15.14       0    633    -9     2
    45.200  107   15.23       0    633    -9     7
    36.080   21   15.23       0    636    -9    13
    24.760   21   15.23       0    638    -9    13
     5.480   22   15.22       0    638    -9    13
     2.560   21   15.21       0    634    -9    13
     2.320   22   15.21       0    633    -9    13
     0.440   20   15.21       0    634    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   845  2.4   15    90   100   600    51    10   270      8     1    4020    5  0.2929
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  845 2022/07/03 17:40:44
# Pcase vacuums
   6.420   6.270
# Oil valve positions
   1.684   0.514   0.677
# Seek attempts and pump time
      1     -4
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  845 2022/07/03 18:37:16 9999/99/99 99:99:99
# Pcase vacuums
   6.270   6.260
# Average 
#   P      T      S
   43.400   1.824  33.437
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  845 2022/07/03 23:07:25 9999/99/99 99:99:99
# Pcase vacuums
   6.260   6.330
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  845 2022/07/03 23:19:44
# number of packets previous packets
      4               5
# CTD Status pres offset
 0x4e410000   0.230
# Pcase vacuums
   6.330   6.350
# Air Pump max current time  max pressure
     119       605     0.000
# Air Valve Positions
   1.284   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.548
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   845      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  845  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.480    270   6.050
   3    42.040      3   6.270
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  845       197    2022/07/03 23:31:33
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      845	(cycle number) |
     MPS =        0	(phase) |
     MPC =      236	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      842	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       23	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      600	(sec) |
  DFT[0] =      362	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65531	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =       80	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      842	(cycle) |
  PTO[0] =    16200	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      842	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      100	(sec) |
     APN =       20	(sec) |
     APF =       70	(sec) |
     ASI =    10000	(ms) |
     ASD =        2	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      842	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    20000	(ms) |
     SDS =       20	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       20	(sec) |

