# Message Decoder for Solo2, Alto and Deep Solo Floats; v220516
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   834  340 000000000432020  4743     0   3 2022/07/01, 08:20:11,  67.2215  -54.2040  17      0 13
SBD 11146   834  308 000000000432020  4744     0   2 2022/07/01, 08:20:22,  67.2676  -54.2250  14      0 15,13,6,14,14
SBD 11146   834  309 000000000432020  4745     0   1 2022/07/01, 08:20:32,  67.2408  -54.2792  13      0 7,5,14,1,2,3,0,8
SBD 11146   834  337 000000000432020  4746     0   0 2022/07/01, 08:20:46,  67.2408  -54.2792  12      0 0,4,14,14,9,10,11
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1246           48        1294
# Sensor_ID bytes     % packets Description
          0    48  3.85       2         GPS
          1    56  4.49       1    Pressure
          2    48  3.85       1 Temperature
          3    48  3.85       1    Salinity
          4    98  7.87       1        Fall
          5    48  3.85       1        Rise
          6    59  4.74       1        Pump
          7    37  2.97       1 Drift_times
          8    17  1.36       1     Surface
          9    61  4.90       1  HR/Drift_P
         10    49  3.93       1  HR/Drift_T
         11    49  3.93       1  HR/Drift_S
         13   404 32.42       2  Parameters
         14   158 12.68       5 Engineering
         15    66  5.30       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  834 1 2022/07/01 02:39 +67.27564  -54.14637    0  4 37 39 42  1.6     0
G  834 2 2022/07/01 08:14 +67.21785  -54.12822   60  5 35 40 45  3.3     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   834   33    1     3         0
#      Pres      Temp    Salinity
      1.440     1.6740    33.4000
      2.040     1.6740    33.4000
      3.000     1.6750    33.4000
      4.000     1.6730    33.4000
      5.080     1.6760    33.3990
      6.040     1.6760    33.4000
      7.000     1.6760    33.4000
      8.000     1.6750    33.4000
      9.000     1.6740    33.4000
     10.000     1.6730    33.4000
     11.080     1.6710    33.4000
     12.040     1.6700    33.4010
     13.000     1.6710    33.4010
     14.000     1.6710    33.4010
     15.000     1.6710    33.4010
     16.000     1.6700    33.4010
     17.080     1.6710    33.4010
     18.040     1.6720    33.4010
     19.040     1.6720    33.4010
     20.000     1.6730    33.4010
     21.120     1.6750    33.4030
     21.920     1.6820    33.4110
     23.000     1.6810    33.4100
     23.960     1.6810    33.4100
     24.960     1.6790    33.4090
     26.040     1.6790    33.4080
     27.040     1.6790    33.4080
     28.040     1.6800    33.4080
     28.960     1.6860    33.4130
     29.960     1.6870    33.4140
     31.000     1.6920    33.4170
     32.040     1.6920    33.4180
     32.920     1.6950    33.4200
# Drift Data
#      C Npts comp n_var
I0   834   26    1     4
#      Pres       Temp   Salinity     Time
     47.640     1.7410    33.4360        6
     43.920     1.7560    33.4500      611
     44.160     1.7610    33.4580     1217
     44.200     1.7570    33.4540     1822
     42.120     1.7550    33.4530     2428
     43.120     1.7530    33.4490     3033
     43.520     1.7290    33.4360     3638
     40.840     1.7130    33.4320     4244
     44.000     1.7550    33.4500     4849
     43.320     1.7100    33.4340     5455
     37.200     1.6940    33.4260     6060
     31.440     1.6410    33.4180     6665
     34.560     1.6520    33.4190     7271
     35.840     1.6980    33.4350     7876
     36.520     1.6840    33.4270     8482
     36.760     1.6690    33.4310     9087
     40.240     1.7080    33.4420     9693
     32.440     1.6620    33.4250    10298
     31.480     1.6830    33.4290    10903
     31.360     1.6600    33.4200    11509
     40.200     1.6960    33.4370    12114
     39.760     1.7050    33.4400    12720
     42.320     1.6930    33.4310    13325
     35.000     1.6790    33.4190    13931
     27.160     1.6750    33.4080    14536
     30.400     1.6860    33.4160    15141
# Surface Pressure Time Series
#    C npts n_var               Start
S  834     1     6 2022/07/01 08:13:56
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    62      0.430      1.693   -999.000      14 2022/07/01 08:14:58
# Fall Time Series
#  Cycle npts     Start                Date
F00  834   18 709958374 2022/07/01 02:39:34
# time     Pres phase                Date
     0    0.440     1 2022/07/01 02:39:34
   141    5.000    10 2022/07/01 02:41:55
   176   10.080    10 2022/07/01 02:42:30
   212   15.320    10 2022/07/01 02:43:06
   245   20.240    10 2022/07/01 02:43:39
   269   23.920     2 2022/07/01 02:44:03
   278   25.320    10 2022/07/01 02:44:12
   310   30.120    10 2022/07/01 02:44:44
   346   35.040    10 2022/07/01 02:45:20
   392   40.320    10 2022/07/01 02:46:06
   434   45.040    10 2022/07/01 02:46:48
   490   50.160    10 2022/07/01 02:47:44
  1090   47.160    10 2022/07/01 02:57:44
  1691   47.480    10 2022/07/01 03:07:45
  2292   49.400    10 2022/07/01 03:17:46
  2434   48.920     3 2022/07/01 03:20:08
  2893   48.480    10 2022/07/01 03:27:47
  3386   47.720    10 2022/07/01 03:36:00
# Rise Time Series
#  Cycle npts     Start                Date
R00  834    8 709977970 2022/07/01 08:06:10
# time     Pres phase                Date
     0   33.560     7 2022/07/01 08:06:10
    85   28.120    13 2022/07/01 08:07:35
   148   23.320    13 2022/07/01 08:08:38
   264   18.680    13 2022/07/01 08:10:34
   337   13.160    13 2022/07/01 08:11:47
   390    8.400    13 2022/07/01 08:12:40
   442    3.600    13 2022/07/01 08:13:32
   464    1.720     8 2022/07/01 08:13:54
# Pump Time Series
#   Cycle npts
B02   834    5
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    23.920  362   15.15       0    633    -9     2
    33.560  107   15.23       0    634    -9     7
    21.560   21   15.23       0    637    -9    13
    18.040   22   15.22       0    638    -9    13
     0.400   20   15.23       0    640    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   834  2.4   14    90   120   600    51    10   270      8     1    4020    5  0.2929
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  834 2022/07/01 02:39:32
# Pcase vacuums
   6.420   6.270
# Oil valve positions
   1.684   0.514   0.676
# Seek attempts and pump time
      1     -5
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  834 2022/07/01 03:36:00 9999/99/99 99:99:99
# Pcase vacuums
   6.270   6.280
# Average 
#   P      T      S
   38.440   1.704  33.434
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  834 2022/07/01 08:06:10 9999/99/99 99:99:99
# Pcase vacuums
   6.270   6.380
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  834 2022/07/01 08:13:55
# number of packets previous packets
      4               4
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.380   6.430
# Air Pump max current time  max pressure
     119       360     0.000
# Air Valve Positions
   1.284   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.549
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   834      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  834  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.440    263   6.050
   3    48.640      3   6.280
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  834       197    2022/07/01 08:20:22
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      834	(cycle number) |
     MPS =        0	(phase) |
     MPC =      225	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      831	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       23	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      600	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65531	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =       80	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      831	(cycle) |
  PTO[0] =    16200	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      831	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      100	(sec) |
     APN =       20	(sec) |
     APF =       70	(sec) |
     ASI =    10000	(ms) |
     ASD =        2	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      831	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    20000	(ms) |
     SDS =       20	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       20	(sec) |

