# Message Decoder for Solo2, Alto and Deep Solo Floats; v220516
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   818   88 000000000432020  4667    30   4 2022/06/27, 15:11:57,  67.1614  -54.1832   4      0 14
SBD 11146   818   61 000000000432020  4668     0   5 2022/06/27, 15:13:09,  67.1614  -54.1832   6      0 13
SBD 11146   818  340 000000000432020  4669     0   3 2022/06/27, 15:13:31,  67.1420  -54.1083   5      0 13
SBD 11146   818  339 000000000432020  4670     0   2 2022/06/27, 15:13:44,  67.1614  -54.1832   4      0 3,0,8,15,13,6,14
SBD 11146   818  299 000000000432020  4671     0   1 2022/06/27, 15:14:05,  67.1882  -54.1290   4      0 7,5,14,1,2
SBD 11146   818  329 000000000432020  4672     0   0 2022/06/27, 15:14:17,  67.1614  -54.1832   3      0 0,4,14,14,9,10,11
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      22         1384           72        1456
# Sensor_ID bytes     % packets Description
          0    48  3.47       2         GPS
          1    58  4.19       1    Pressure
          2    91  6.58       1 Temperature
          3    75  5.42       1    Salinity
          4   123  8.89       1        Fall
          5    83  6.00       1        Rise
          6    59  4.26       1        Pump
          7    36  2.60       1 Drift_times
          8     8  0.58       1     Surface
          9    36  2.60       1  HR/Drift_P
         10    46  3.32       1  HR/Drift_T
         11    44  3.18       1  HR/Drift_S
         13   453 32.73       3  Parameters
         14   158 11.42       5 Engineering
         15    66  4.77       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  818 1 2022/06/27 09:24 +67.15787  -54.15561    0  6 29 36 44  2.3     0
G  818 2 2022/06/27 15:06 +67.15893  -54.15590   90  6  0 30 45  1.2     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   818   68    1     3         0
#      Pres      Temp    Salinity
      1.160     2.5560    33.3850
      2.040     2.5550    33.3840
      2.960     2.5590    33.3790
      4.000     2.5140    33.3810
      5.040     2.3210    33.3880
      6.000     2.1040    33.3910
      7.000     1.7970    33.4180
      8.040     1.6170    33.4150
      8.960     1.5970    33.4130
      9.920     1.5920    33.4130
     11.000     1.5910    33.4130
     12.040     1.5900    33.4140
     13.000     1.5790    33.4150
     14.000     1.5720    33.4140
     15.000     1.5700    33.4140
     16.040     1.5670    33.4150
     17.080     1.5650    33.4150
     18.040     1.5640    33.4150
     19.000     1.5620    33.4150
     19.960     1.5560    33.4160
     21.000     1.5540    33.4160
     22.040     1.5540    33.4160
     23.040     1.5540    33.4160
     24.040     1.5530    33.4160
     24.960     1.5530    33.4160
     25.920     1.5530    33.4160
     26.960     1.5440    33.4170
     28.040     1.5420    33.4160
     29.000     1.5440    33.4180
     29.960     1.5500    33.4180
     31.000     1.5670    33.4180
     31.960     1.5870    33.4220
     32.960     1.5890    33.4220
     34.000     1.5860    33.4220
     34.960     1.5860    33.4220
     36.000     1.5860    33.4220
     37.000     1.5840    33.4220
     37.960     1.5820    33.4220
     38.960     1.5820    33.4220
     39.960     1.5820    33.4210
     41.000     1.5830    33.4220
     42.000     1.5810    33.4220
     42.960     1.5800    33.4220
     44.040     1.5750    33.4210
     45.000     1.5720    33.4210
     46.000     1.5650    33.4220
     46.960     1.5590    33.4220
     47.960     1.5590    33.4200
     49.000     1.5620    33.4220
     50.040     1.5680    33.4210
     51.000     1.5690    33.4210
     51.960     1.5710    33.4220
     52.960     1.5750    33.4230
     53.960     1.5780    33.4230
     55.000     1.5880    33.4230
     56.000     1.5980    33.4250
     57.000     1.5960    33.4230
     57.960     1.5980    33.4240
     59.000     1.5910    33.4240
     60.000     1.5810    33.4220
     61.040     1.5610    33.4210
     62.040     1.5410    33.4210
     63.040     1.5450    33.4180
     64.000     1.5280    33.4220
     64.960     1.4950    33.4160
     66.000     1.4960    33.3720
     67.040     1.4840    33.4170
     68.200     1.4810    33.4160
# Drift Data
#      C Npts comp n_var
I0   818   23    1     4
#      Pres       Temp   Salinity     Time
     71.160     1.5890    33.4220        6
     71.120     1.5680    33.4220      611
     70.880     1.5650    33.4190     1217
     70.560     1.5590    33.4210     1822
     70.160     1.5110    33.4240     2428
     69.800     1.5450    33.4170     3033
     69.440     1.4810    33.4160     3639
     69.000     1.4170    33.4100     4244
     68.560     1.4110    33.4110     4849
     68.240     1.4070    33.4080     5455
     68.000     1.4100    33.4110     6060
     67.760     1.4180    33.4090     6666
     67.560     1.4220    33.4120     7271
     67.480     1.4240    33.4130     7877
     67.360     1.4230    33.4120     8482
     67.400     1.4370    33.4060     9087
     67.400     1.4320    33.4100     9693
     67.480     1.4270    33.4120    10298
     67.680     1.4320    33.4110    10904
     67.800     1.4440    33.4130    11509
     67.840     1.4640    33.4120    12114
     68.000     1.4620    33.4140    12720
     68.200     1.4700    33.4140    13325
# Fall Time Series
#  Cycle npts     Start                Date
F00  818   23 709637103 2022/06/27 09:25:03
# time     Pres phase                Date
     0    0.400     1 2022/06/27 09:25:03
   140    5.080    10 2022/06/27 09:27:23
   176   10.280    10 2022/06/27 09:27:59
   208   15.120    10 2022/06/27 09:28:31
   241   20.080    10 2022/06/27 09:29:04
   265   23.800     2 2022/06/27 09:29:28
   274   25.160    10 2022/06/27 09:29:37
   310   30.280    10 2022/06/27 09:30:13
   346   35.280    10 2022/06/27 09:30:49
   385   40.160    10 2022/06/27 09:31:28
   424   45.280    10 2022/06/27 09:32:07
   463   50.120    10 2022/06/27 09:32:46
   512   55.080    10 2022/06/27 09:33:35
   567   60.040    10 2022/06/27 09:34:30
   702   65.040    10 2022/06/27 09:36:45
   933   70.000    10 2022/06/27 09:40:36
  1534   71.760    10 2022/06/27 09:50:37
  2135   71.760    10 2022/06/27 10:00:38
  2420   71.840     3 2022/06/27 10:05:23
  3367   71.560    10 2022/06/27 10:21:10
  3968   71.440    10 2022/06/27 10:31:11
  4569   71.200    10 2022/06/27 10:41:12
  5168   71.160    10 2022/06/27 10:51:11
# Rise Time Series
#  Cycle npts     Start                Date
R00  818   15 709656723 2022/06/27 14:52:03
# time     Pres phase                Date
     0   68.400     7 2022/06/27 14:52:03
   127   63.440    13 2022/06/27 14:54:10
   180   58.760    13 2022/06/27 14:55:03
   232   53.840    13 2022/06/27 14:55:55
   295   48.120    13 2022/06/27 14:56:58
   348   43.200    13 2022/06/27 14:57:51
   401   38.120    13 2022/06/27 14:58:44
   442   33.960    13 2022/06/27 14:59:25
   495   28.840    13 2022/06/27 15:00:18
   547   23.520    13 2022/06/27 15:01:10
   600   18.240    13 2022/06/27 15:02:03
   652   13.040    13 2022/06/27 15:02:55
   694    8.840    13 2022/06/27 15:03:37
   747    3.800    13 2022/06/27 15:04:30
   779    1.040     8 2022/06/27 15:05:02
# Pump Time Series
#   Cycle npts
B02   818    5
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    23.800  352   15.13       0    634    -9     2
    68.400  128   15.23       0    636    -9     7
    41.280   21   15.23       0    638    -9    13
     3.800   21   15.22       0    638    -9    13
     0.400   20   15.22       0    639    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   818  2.4   11    90   120   600    50    10   240      8     1    4020   15  0.2783
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  818 2022/06/27 09:25:01
# Pcase vacuums
   6.430   6.270
# Oil valve positions
   1.683   0.514   0.680
# Seek attempts and pump time
      1     -2
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  818 2022/06/27 10:51:11 9999/99/99 99:99:99
# Pcase vacuums
   6.270   6.280
# Average 
#   P      T      S
   68.640   1.466  33.414
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  818 2022/06/27 14:52:03 9999/99/99 99:99:99
# Pcase vacuums
   6.270   6.380
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  818 2022/06/27 15:05:01
# number of packets previous packets
      4               5
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.380   6.420
# Air Pump max current time  max pressure
     119       384     0.000
# Air Valve Positions
   1.285   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.547
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   818      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  818  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.480    260   6.060
   3    71.880      3   6.280
# Upload command:
C 1dsd 0 21;dsw 300;dvw 300;duc 75;MIC 1
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  818       197    2022/06/27 15:13:44
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      818	(cycle number) |
     MPS =        0	(phase) |
     MPC =      209	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      815	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       23	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      600	(sec) |
  DFT[0] =      352	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65534	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      815	(cycle) |
  PTO[0] =    14400	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      815	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    10000	(ms) |
     ASD =        2	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      815	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    20000	(ms) |
     SDS =       20	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       20	(sec) |

