# Message Decoder for Solo2, Alto and Deep Solo Floats; v220516
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   769  340 000000000432020  4419    23   3 2022/06/15, 20:18:31,  66.9883  -57.3530 137      2 13
SBD 11146   769  137 000000000432020  4420     0   4 2022/06/15, 20:18:50,  67.0161  -57.3034 138      2 13
SBD 11146   769  300 000000000432020  4421     0   2 2022/06/15, 20:19:00,  67.0161  -57.3034 143      2 15,13,6,14,14
SBD 11146   769  301 000000000432020  4422     0   1 2022/06/15, 20:19:13,  67.0161  -57.3034 137      2 5,14,1,2,3,0,8
SBD 11146   769  308 000000000432020  4423     0   0 2022/06/15, 20:19:44,  67.1307  -54.6982  20      0 0,4,14,14,9,10,11,7
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      22         1326           60        1386
# Sensor_ID bytes     % packets Description
          0    48  3.62       2         GPS
          1    60  4.52       1    Pressure
          2    76  5.73       1 Temperature
          3    60  4.52       1    Salinity
          4   128  9.65       1        Fall
          5    33  2.49       1        Rise
          6    37  2.79       1        Pump
          7    25  1.89       1 Drift_times
          8    17  1.28       1     Surface
          9    25  1.89       1  HR/Drift_P
         10    25  1.89       1  HR/Drift_T
         11    25  1.89       1  HR/Drift_S
         13   529 39.89       3  Parameters
         14   172 12.97       5 Engineering
         15    66  4.98       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  769 1 2022/06/15 11:25 +67.09063  -54.29872    0  4 28 37 44  1.4     0
G  769 2 2022/06/15 20:15 +67.09769  -54.29812   60  6  0 30 43  1.8     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   769   71    1     3         0
#      Pres      Temp    Salinity
      1.120     1.4780    33.3440
      2.000     1.4770    33.3450
      2.960     1.4760    33.3450
      3.920     1.4760    33.3450
      5.000     1.4720    33.3450
      6.040     1.4640    33.3450
      7.040     1.4620    33.3430
      8.040     1.4620    33.3460
      9.040     1.4340    33.3440
     10.040     1.4310    33.3430
     11.040     1.4380    33.3420
     12.000     1.4160    33.3440
     13.000     1.4000    33.3430
     14.000     1.4000    33.3420
     15.000     1.3970    33.3420
     16.000     1.3700    33.3440
     16.960     1.3550    33.3410
     17.920     1.3410    33.3420
     19.000     1.3250    33.3400
     20.040     1.2970    33.3390
     21.000     1.2740    33.3410
     22.000     1.2650    33.3400
     23.000     1.2580    33.3400
     24.040     1.2170    33.3440
     25.080     1.2080    33.3450
     26.040     1.2040    33.3460
     27.040     1.1940    33.3490
     28.040     1.1890    33.3500
     29.040     1.1860    33.3500
     30.000     1.1850    33.3510
     31.000     1.1830    33.3510
     32.000     1.1830    33.3510
     33.000     1.1820    33.3520
     34.000     1.1810    33.3530
     35.000     1.1800    33.3530
     36.000     1.1790    33.3540
     37.000     1.1790    33.3550
     38.040     1.1770    33.3550
     39.000     1.1780    33.3560
     39.960     1.1790    33.3560
     41.000     1.1800    33.3560
     41.960     1.1800    33.3570
     43.000     1.1790    33.3570
     43.960     1.1780    33.3570
     44.920     1.1780    33.3570
     46.000     1.1790    33.3580
     47.040     1.1780    33.3590
     48.040     1.1780    33.3590
     49.040     1.1770    33.3600
     50.000     1.1760    33.3600
     51.040     1.1750    33.3600
     52.080     1.1750    33.3600
     53.000     1.1750    33.3600
     53.920     1.1740    33.3610
     55.000     1.1720    33.3610
     56.040     1.1710    33.3610
     56.960     1.1690    33.3610
     57.920     1.1670    33.3610
     58.960     1.1650    33.3610
     60.040     1.1630    33.3610
     61.080     1.1630    33.3610
     62.000     1.1640    33.3610
     62.960     1.1640    33.3610
     64.000     1.1610    33.3610
     65.120     1.1610    33.3610
     66.080     1.1610    33.3610
     67.000     1.1610    33.3610
     68.000     1.1600    33.3610
     69.040     1.1600    33.3610
     70.000     1.1600    33.3610
     70.840     1.1600    33.3620
# Drift Data
#      C Npts comp n_var
I0   769    1    1     4
#      Pres       Temp   Salinity     Time
     70.200     1.1600    33.3630     3611
# Surface Pressure Time Series
#    C npts n_var               Start
S  769     1     6 2022/06/15 20:13:58
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    65      0.370      1.508   -999.000      14 2022/06/15 20:15:03
# Fall Time Series
#  Cycle npts     Start                Date
F00  769   24 708607592 2022/06/15 11:26:32
# time     Pres phase                Date
     0    0.280     1 2022/06/15 11:26:32
   235   10.440    10 2022/06/15 11:30:27
   297   20.080    10 2022/06/15 11:31:29
   362   30.080    10 2022/06/15 11:32:34
   402   36.040     2 2022/06/15 11:33:14
   431   40.400    10 2022/06/15 11:33:43
   503   50.320    10 2022/06/15 11:34:55
   577   60.040    10 2022/06/15 11:36:09
   685   70.200    10 2022/06/15 11:37:57
  2486   75.520    10 2022/06/15 12:07:58
  4287   74.920    10 2022/06/15 12:37:59
  4374   75.320     3 2022/06/15 12:39:26
  6088   74.960    10 2022/06/15 13:08:00
  7888   71.560    10 2022/06/15 13:38:00
  8923   72.320     3 2022/06/15 13:55:15
  9870   72.160    10 2022/06/15 14:11:02
 11671   71.440    10 2022/06/15 14:41:03
 13472   70.680     3 2022/06/15 15:11:04
 13474   70.680    10 2022/06/15 15:11:06
 15274   70.400    10 2022/06/15 15:41:06
 17075   70.120    10 2022/06/15 16:11:07
 18876   70.080    10 2022/06/15 16:41:08
 20676   70.000    10 2022/06/15 17:11:08
 21622   69.880    10 2022/06/15 17:26:54
# Rise Time Series
#  Cycle npts     Start                Date
R00  769    5 708638223 2022/06/15 19:57:03
# time     Pres phase                Date
     0   70.760     7 2022/06/15 19:57:03
   422   49.400    13 2022/06/15 20:04:05
   648   28.800    13 2022/06/15 20:07:51
   894    7.840    13 2022/06/15 20:11:57
   976    1.160     8 2022/06/15 20:13:19
# Pump Time Series
#   Cycle npts
B02   769    3
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    36.040  369   15.13       0    635    -9     2
    70.760  136   15.23       0    638    -9     7
     0.360   60   15.22       0    642    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   769  2.4    5    90   120   600    81    10   150      7     3    4020   60  0.3538
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  769 2022/06/15 11:26:30
# Pcase vacuums
   6.570   6.290
# Oil valve positions
   1.683   0.518   0.685
# Seek attempts and pump time
      3     -6
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  769 2022/06/15 17:26:54 9999/99/99 99:99:99
# Pcase vacuums
   6.290   6.290
# Average 
#   P      T      S
   70.200   1.160  33.363
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  769 2022/06/15 19:57:03 9999/99/99 99:99:99
# Pcase vacuums
   6.290   6.370
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  769 2022/06/15 20:13:57
# number of packets previous packets
      4               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.370   6.440
# Air Pump max current time  max pressure
     119       245     0.000
# Air Valve Positions
   1.284   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.553
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   769      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  769  4      0
# MC    Pres      Time  Pcase Vacuum
   1     0.240    396   6.070
   3    75.480      2   6.290
   3    72.320      3   6.290
   3    70.760      2   6.290
# Upload command:
C }dsd 0 25;dct 0 1800;dsw 1800;psi 600000;pdt 600;add 5;asi 10000;ddt 600;adt 300;ddd 5;spt 300;set 10;sgt 10;MIC 1
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  769       197    2022/06/15 20:19:00
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      769	(cycle number) |
     MPS =        0	(phase) |
     MPC =      160	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      120	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      766	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       35	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     1800	(sec) |
  DFT[0] =      367	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     3600	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65532	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     3600	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      766	(cycle) |
  PTO[0] =     9000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     7200	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      766	(cycle number) |
     ARR =        7	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        2	(dBar) |
     ADD =       20	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      766	(cycle number) |
     SDC =        0	(dBar) |
     SGT =        5	(min) |
     SET =        5	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =     5000	(ms) |
     SDS =        5	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      180	(sec) |
     SSP =       60	(sec) |

