# Message Decoder for Solo2, Alto and Deep Solo Floats; v220516
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   617  257 000000000432020  3595     0   4 2022/01/13, 12:05:16,  66.3282  -54.8983   4      0 13,6,14,14
SBD 11146   617  340 000000000432020  3596     0   3 2022/01/13, 12:05:29,  66.3884  -54.9200   9      0 13
SBD 11146   617  338 000000000432020  3597     0   2 2022/01/13, 12:06:27,  66.3362  -54.7738   5      0 2,3,0,8,15
SBD 11146   617  314 000000000432020  3598     0   1 2022/01/13, 12:06:44,  66.3282  -54.8983   5      0 10,11,7,5,14,1
SBD 11146   617  336 000000000432020  3599     0   0 2022/01/13, 12:06:55,  66.3551  -54.8469   4      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1525           60        1585
# Sensor_ID bytes     % packets Description
          0    48  3.15       2         GPS
          1    74  4.85       1    Pressure
          2   130  8.52       1 Temperature
          3    98  6.43       1    Salinity
          4   213 13.97       1        Fall
          5    98  6.43       1        Rise
          6    81  5.31       1        Pump
          7    31  2.03       1 Drift_times
          8     8  0.52       1     Surface
          9    43  2.82       1  HR/Drift_P
         10    43  2.82       1  HR/Drift_T
         11    37  2.43       1  HR/Drift_S
         13   404 26.49       2  Parameters
         14   151  9.90       5 Engineering
         15    66  4.33       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  617 1 2022/01/13 06:28 +66.29060  -54.92491   10  5  0 32 42 11.9     0
G  617 2 2022/01/13 12:02 +66.32151  -54.86648   80  6  0 32 44  1.3     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   617   99    1     3         0
#      Pres      Temp    Salinity
      1.080     0.5870    33.4550
      1.960     0.5930    33.4540
      3.000     0.6030    33.4550
      4.000     0.5930    33.4560
      5.080     0.6060    33.4550
      6.000     0.6090    33.4550
      7.040     0.6140    33.4550
      8.000     0.6140    33.4570
      8.920     0.6170    33.4550
     10.000     0.6120    33.4550
     10.960     0.6130    33.4540
     12.000     0.6040    33.4570
     13.080     0.6050    33.4560
     14.040     0.6160    33.4550
     15.000     0.6230    33.4560
     16.040     0.6200    33.4570
     17.000     0.6190    33.4570
     18.000     0.6220    33.4560
     19.080     0.6190    33.4560
     20.040     0.6170    33.4560
     20.920     0.6180    33.4570
     21.920     0.6180    33.4570
     22.920     0.6240    33.4570
     23.960     0.6240    33.4590
     24.920     0.6290    33.4580
     26.040     0.6310    33.4590
     27.080     0.6320    33.4600
     28.080     0.6420    33.4590
     29.000     0.6440    33.4600
     30.000     0.6430    33.4600
     30.960     0.6430    33.4600
     32.000     0.6420    33.4600
     33.040     0.6420    33.4600
     34.040     0.6420    33.4600
     35.000     0.6430    33.4600
     36.040     0.6430    33.4600
     36.960     0.6420    33.4600
     38.040     0.6410    33.4610
     39.080     0.6400    33.4610
     39.960     0.6410    33.4600
     41.040     0.6420    33.4610
     42.040     0.6420    33.4610
     43.000     0.6400    33.4610
     43.960     0.6430    33.4610
     45.040     0.6430    33.4620
     46.000     0.6430    33.4610
     46.960     0.6430    33.4610
     48.000     0.6420    33.4620
     49.080     0.6450    33.4620
     50.080     0.6480    33.4620
     51.000     0.6530    33.4630
     52.000     0.6670    33.4640
     53.000     0.6680    33.4650
     53.960     0.6670    33.4650
     54.920     0.6670    33.4650
     56.000     0.6620    33.4660
     57.080     0.6600    33.4670
     58.080     0.6630    33.4690
     59.000     0.6690    33.4710
     60.000     0.7300    33.4730
     61.080     0.7960    33.4860
     62.040     0.8220    33.4880
     62.920     0.8500    33.4950
     63.960     0.8720    33.5050
     65.000     1.1160    33.5090
     66.040     1.3520    33.5550
     67.000     1.4050    33.5680
     67.960     1.4360    33.5790
     68.960     1.5320    33.5820
     69.960     1.5980    33.6000
     71.000     1.7590    33.6070
     72.000     1.9460    33.6530
     73.040     2.0080    33.6610
     74.080     2.1700    33.6840
     75.080     2.2660    33.7110
     75.960     2.2670    33.7100
     76.960     2.2900    33.7200
     78.040     2.3270    33.7250
     79.000     2.4730    33.7420
     79.960     2.5700    33.7700
     81.000     2.5700    33.7720
     82.040     2.5910    33.7730
     83.080     2.5990    33.7780
     84.080     2.6250    33.7790
     84.960     2.6660    33.7890
     85.960     2.6690    33.7910
     87.000     2.6640    33.7930
     88.000     2.6760    33.7940
     89.000     2.6950    33.7990
     90.000     2.7220    33.8020
     91.000     2.7770    33.8120
     91.960     2.8240    33.8210
     92.960     2.8710    33.8330
     93.960     2.9200    33.8420
     95.000     2.9440    33.8490
     96.000     3.0050    33.8610
     97.040     3.0280    33.8670
     98.160     3.0410    33.8710
     98.520     3.0450    33.8720
# Drift Data
#      C Npts comp n_var
I0   617   14    1     4
#      Pres       Temp   Salinity     Time
    115.320     2.9600    33.8520        6
    118.720     2.8490    33.8310      611
    109.840     2.8310    33.8190     1217
     99.640     2.9460    33.8490     1822
    101.520     3.0400    33.8780     2428
    102.240     2.9820    33.8570     3033
     93.680     3.0710    33.8790     3638
     97.080     3.0580    33.8740     4244
     97.560     3.0600    33.8770     4849
     99.920     3.0340    33.8720     5455
     97.400     3.0510    33.8700     6060
     95.560     3.0720    33.8770     6666
     90.720     3.0970    33.8920     7271
     92.440     3.1110    33.8920     7876
# Fall Time Series
#  Cycle npts     Start                Date
F00  617   41 695370541 2022/01/13 06:29:01
# time     Pres phase                Date
     0    0.520     1 2022/01/13 06:29:01
   150    5.000    10 2022/01/13 06:31:31
   186   10.280    10 2022/01/13 06:32:07
   218   15.080    10 2022/01/13 06:32:39
   251   20.120    10 2022/01/13 06:33:12
   284   25.040    10 2022/01/13 06:33:45
   316   30.280    10 2022/01/13 06:34:17
   345   35.040    10 2022/01/13 06:34:46
   378   40.520    10 2022/01/13 06:35:19
   407   45.200    10 2022/01/13 06:35:48
   437   50.080    10 2022/01/13 06:36:18
   441   50.800     2 2022/01/13 06:36:22
   466   55.080    10 2022/01/13 06:36:47
   499   60.320    10 2022/01/13 06:37:20
   532   65.440    10 2022/01/13 06:37:53
   564   70.320    10 2022/01/13 06:38:25
   597   75.000    10 2022/01/13 06:38:58
   639   80.320    10 2022/01/13 06:39:40
   681   85.000    10 2022/01/13 06:40:22
   737   90.160    10 2022/01/13 06:41:18
   842   95.040    10 2022/01/13 06:43:03
  1163  100.000    10 2022/01/13 06:48:24
  1764   98.640    10 2022/01/13 06:58:25
  2365  100.920    10 2022/01/13 07:08:26
  2586  101.440     3 2022/01/13 07:12:07
  2966  104.200    10 2022/01/13 07:18:27
  3345  109.000    10 2022/01/13 07:24:46
  3946  106.200    10 2022/01/13 07:34:47
  4425  105.040     3 2022/01/13 07:42:46
  4547  104.240    10 2022/01/13 07:44:48
  5148  106.800    10 2022/01/13 07:54:49
  5747  111.040    10 2022/01/13 08:04:48
  6263  107.200     3 2022/01/13 08:13:24
  6348  107.080    10 2022/01/13 08:14:49
  6949  104.840    10 2022/01/13 08:24:50
  7550  103.880    10 2022/01/13 08:34:51
  8151  105.800    10 2022/01/13 08:44:52
  8627  110.000    10 2022/01/13 08:52:48
  9228  111.040    10 2022/01/13 09:02:49
  9829  114.120    10 2022/01/13 09:12:50
  9905  115.240    10 2022/01/13 09:14:06
# Rise Time Series
#  Cycle npts     Start                Date
R00  617   18 695389457 2022/01/13 11:44:17
# time     Pres phase                Date
     0   98.520     7 2022/01/13 11:44:17
   114   93.240    13 2022/01/13 11:46:11
   176   87.280    13 2022/01/13 11:47:13
   237   80.960    13 2022/01/13 11:48:14
   299   74.720    13 2022/01/13 11:49:16
   360   68.560    13 2022/01/13 11:50:17
   422   62.120    13 2022/01/13 11:51:19
   463   57.760    13 2022/01/13 11:52:00
   524   51.360    13 2022/01/13 11:53:01
   565   46.960    13 2022/01/13 11:53:42
   627   40.600    13 2022/01/13 11:54:44
   668   36.000    13 2022/01/13 11:55:25
   729   28.880    13 2022/01/13 11:56:26
   791   22.600    13 2022/01/13 11:57:28
   852   15.680    13 2022/01/13 11:58:29
   914    9.560    13 2022/01/13 11:59:31
   975    2.800    13 2022/01/13 12:00:32
   996    0.680     8 2022/01/13 12:00:53
# Pump Time Series
#   Cycle npts
B02   617    7
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    50.800  343   15.12       0    627    -9     2
   107.080    1   15.19       0    619    -9     3
    98.520  135   15.22       0    629    -9     7
    70.680   20   15.21       0    634    -9    13
    45.000   20   15.21       0    639    -9    13
     9.560   20   15.20       0    641    -9    13
     0.000   60   15.20       0    646    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   617  2.4    1    90    95   600    50    10   150      9     3    4020   30  0.2664
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  617 2022/01/13 06:28:59
# Pcase vacuums
   6.500   6.190
# Oil valve positions
   1.680   0.520   0.679
# Seek attempts and pump time
      3      1
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  617 2022/01/13 09:14:06 9999/99/99 99:99:99
# Pcase vacuums
   6.190   6.190
# Average 
#   P      T      S
  100.840   3.012  33.866
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  617 2022/01/13 11:44:17 9999/99/99 99:99:99
# Pcase vacuums
   6.190   6.420
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  617 2022/01/13 12:00:54
# number of packets previous packets
      5              11
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.420   6.490
# Air Pump max current time  max pressure
     119       245     0.000
# Air Valve Positions
   1.282   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.542
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   617      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  617  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.480    436   6.070
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  617       197    2022/01/13 12:05:29
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      617	(cycle number) |
     MPS =        0	(phase) |
     MPC =        8	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       95	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      614	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       50	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      600	(sec) |
  DFT[0] =      342	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      614	(cycle) |
  PTO[0] =     9000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      614	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        2	(dBar) |
     ADD =        5	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      614	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      180	(sec) |
     SSP =       60	(sec) |

