# Message Decoder for Solo2, Alto and Deep Solo Floats; v220516
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146   610  299 000000000432020  3558     0   4 2022/01/11, 19:56:58,  66.2624  -55.3218  11      0 13,6,14,14
SBD 11146   610  340 000000000432020  3559     0   3 2022/01/11, 19:57:37,  66.2726  -55.5190  16      0 13
SBD 11146   610  312 000000000432020  3560     0   2 2022/01/11, 19:57:57,  66.2726  -55.5190  17      0 2,3,0,8,15
SBD 11146   610  292 000000000432020  3561     0   1 2022/01/11, 19:58:16,  66.2955  -55.3951  20      0 10,11,7,5,14,1
SBD 11146   610  331 000000000432020  3562     0   0 2022/01/11, 19:58:55,  66.2955  -55.3951  16      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1514           60        1574
# Sensor_ID bytes     % packets Description
          0    48  3.17       2         GPS
          1    68  4.49       1    Pressure
          2   118  7.79       1 Temperature
          3    84  5.55       1    Salinity
          4   208 13.74       1        Fall
          5    88  5.81       1        Rise
          6   103  6.80       1        Pump
          7    31  2.05       1 Drift_times
          8     8  0.53       1     Surface
          9    43  2.84       1  HR/Drift_P
         10    37  2.44       1  HR/Drift_T
         11    37  2.44       1  HR/Drift_S
         13   404 26.68       2  Parameters
         14   171 11.29       5 Engineering
         15    66  4.36       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  610 1 2022/01/11 13:49 +66.28402  -55.35559    0  5 35 39 43  1.4     0
G  610 2 2022/01/11 19:55 +66.25717  -55.29124  170  4 25 34 41  2.8     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   610   85    1     3         0
#      Pres      Temp    Salinity
      1.080     0.5700    33.4380
      2.040     0.5700    33.4380
      3.000     0.5690    33.4380
      4.040     0.5670    33.4380
      5.040     0.5680    33.4380
      6.040     0.5720    33.4380
      7.040     0.5750    33.4370
      8.000     0.5730    33.4380
      9.000     0.5740    33.4380
      9.920     0.5740    33.4380
     10.960     0.5750    33.4380
     12.000     0.5760    33.4380
     13.040     0.5770    33.4380
     14.080     0.5770    33.4390
     15.040     0.5830    33.4370
     15.920     0.5810    33.4400
     16.960     0.5760    33.4390
     18.000     0.5780    33.4380
     19.000     0.5930    33.4390
     19.960     0.5960    33.4380
     20.920     0.6000    33.4380
     21.960     0.5970    33.4380
     23.000     0.5830    33.4380
     24.000     0.5790    33.4390
     25.000     0.5860    33.4370
     26.000     0.5920    33.4380
     27.000     0.5850    33.4370
     27.960     0.5830    33.4390
     28.920     0.6000    33.4370
     29.960     0.5950    33.4380
     31.000     0.5920    33.4400
     32.000     0.5880    33.4380
     32.960     0.5820    33.4380
     34.000     0.5850    33.4390
     35.080     0.5970    33.4370
     36.040     0.6080    33.4380
     37.000     0.6070    33.4380
     37.960     0.6080    33.4380
     39.040     0.6080    33.4380
     40.000     0.6140    33.4380
     40.960     0.6110    33.4380
     42.000     0.6070    33.4380
     43.040     0.6050    33.4380
     44.040     0.6040    33.4380
     45.040     0.6030    33.4380
     46.040     0.6030    33.4380
     47.040     0.6020    33.4380
     48.040     0.6020    33.4380
     49.000     0.6030    33.4380
     50.000     0.6080    33.4380
     51.040     0.6080    33.4380
     52.040     0.6090    33.4390
     53.000     0.6070    33.4380
     54.040     0.6140    33.4390
     55.080     0.6200    33.4400
     55.960     0.6180    33.4410
     56.960     0.6230    33.4410
     57.960     0.6390    33.4440
     59.000     0.6520    33.4450
     60.040     0.6730    33.4470
     61.000     0.6970    33.4510
     61.960     0.7040    33.4530
     62.920     0.7250    33.4520
     63.960     0.7570    33.4620
     65.080     0.7830    33.4640
     66.080     0.8250    33.4720
     67.000     0.8580    33.4750
     68.000     0.9310    33.4860
     69.000     0.9690    33.4930
     70.000     0.9820    33.4970
     71.000     1.0180    33.5020
     72.000     1.0630    33.5080
     73.000     1.1160    33.5180
     74.000     1.1580    33.5240
     75.000     1.2270    33.5340
     76.040     1.2920    33.5460
     77.040     1.3330    33.5550
     78.000     1.4260    33.5730
     78.960     1.5130    33.5950
     80.000     1.8380    33.6360
     81.000     2.3160    33.7160
     82.000     2.4600    33.7460
     83.040     2.9520    33.8280
     84.080     3.2770    33.9020
     84.600     3.3070    33.9100
# Drift Data
#      C Npts comp n_var
I0   610   14    1     4
#      Pres       Temp   Salinity     Time
     83.120     3.3160    33.9130        6
     89.120     3.2640    33.9010      611
     87.360     3.2540    33.8970     1217
     84.080     3.3030    33.9060     1822
     84.560     3.3220    33.9120     2428
     88.760     3.3090    33.9120     3033
     90.040     3.2340    33.8940     3639
     90.600     3.2260    33.8920     4244
     84.800     3.2710    33.8990     4849
     82.920     3.3290    33.9090     5455
     80.560     3.3470    33.9200     6060
     83.160     3.3430    33.9200     6666
     85.880     3.2790    33.9070     7271
     87.000     3.2900    33.9080     7877
# Fall Time Series
#  Cycle npts     Start                Date
F00  610   40 695224147 2022/01/11 13:49:07
# time     Pres phase                Date
     0    0.000     1 2022/01/11 13:49:07
   192    5.120    10 2022/01/11 13:52:19
   222   10.000    10 2022/01/11 13:52:49
   251   15.160    10 2022/01/11 13:53:18
   277   20.160    10 2022/01/11 13:53:44
   303   25.280    10 2022/01/11 13:54:10
   329   30.080    10 2022/01/11 13:54:36
   362   35.480    10 2022/01/11 13:55:09
   391   40.440    10 2022/01/11 13:55:38
   417   45.080    10 2022/01/11 13:56:04
   450   50.280    10 2022/01/11 13:56:37
   454   50.920     2 2022/01/11 13:56:41
   483   55.120    10 2022/01/11 13:57:10
   522   60.280    10 2022/01/11 13:57:49
   561   65.160    10 2022/01/11 13:58:28
   603   70.080    10 2022/01/11 13:59:10
   655   75.080    10 2022/01/11 14:00:02
   727   80.200    10 2022/01/11 14:01:14
   975   85.000    10 2022/01/11 14:05:22
  1517   80.960    10 2022/01/11 14:14:24
  2118   78.160    10 2022/01/11 14:24:25
  2578   74.000    10 2022/01/11 14:32:05
  2606   73.440     3 2022/01/11 14:32:33
  3553   81.920    10 2022/01/11 14:48:20
  4053   77.000    10 2022/01/11 14:56:40
  4654   73.000    10 2022/01/11 15:06:41
  5253   78.040    10 2022/01/11 15:16:40
  5354   78.920     3 2022/01/11 15:18:21
  5853   80.080    10 2022/01/11 15:26:40
  6301   76.000    10 2022/01/11 15:34:08
  6902   71.920    10 2022/01/11 15:44:09
  7192   72.320     3 2022/01/11 15:48:59
  8139   79.280    10 2022/01/11 16:04:46
  8740   80.600    10 2022/01/11 16:14:47
  9287   85.000    10 2022/01/11 16:23:54
  9888   89.360    10 2022/01/11 16:33:55
 10150   84.960    10 2022/01/11 16:38:17
 10751   80.480    10 2022/01/11 16:48:18
 11352   82.320    10 2022/01/11 16:58:19
 11741   83.120    10 2022/01/11 17:04:48
# Rise Time Series
#  Cycle npts     Start                Date
R00  610   16 695244898 2022/01/11 19:34:58
# time     Pres phase                Date
     0   84.520     7 2022/01/11 19:34:58
   135   79.680    13 2022/01/11 19:37:13
   217   73.680    13 2022/01/11 19:38:35
   278   68.960    13 2022/01/11 19:39:36
   340   63.800    13 2022/01/11 19:40:38
   401   57.960    13 2022/01/11 19:41:39
   463   51.960    13 2022/01/11 19:42:41
   524   45.960    13 2022/01/11 19:43:42
   586   39.840    13 2022/01/11 19:44:44
   647   33.600    13 2022/01/11 19:45:45
   709   28.320    13 2022/01/11 19:46:47
   770   23.280    13 2022/01/11 19:47:48
   832   16.840    13 2022/01/11 19:48:50
   893   10.520    13 2022/01/11 19:49:51
   955    4.360    13 2022/01/11 19:50:53
  1057    0.240     8 2022/01/11 19:52:35
# Pump Time Series
#   Cycle npts
B02   610    9
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    50.920  349   15.01       0    629    -9     2
    84.520  135   15.22       0    629    -9     7
    72.240   21   15.22       0    635    -9    13
    61.920   21   15.22       0    638    -9    13
    47.880   20   15.22       0    641    -9    13
    33.600   21   15.21       0    643    -9    13
    23.280   21   15.20       0    644    -9    13
  9999.000   32   15.21       0    647    -9    13
     0.000   60   15.20       0    651    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   610  2.4    1    90    95   600    50    10   150      9     3    4020   30  0.2664
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  610 2022/01/11 13:49:05
# Pcase vacuums
   6.570   6.190
# Oil valve positions
   1.678   0.521   0.677
# Seek attempts and pump time
      3     -5
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  610 2022/01/11 17:04:48 9999/99/99 99:99:99
# Pcase vacuums
   6.180   6.180
# Average 
#   P      T      S
   85.840   3.292  33.906
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  610 2022/01/11 19:34:58 9999/99/99 99:99:99
# Pcase vacuums
   6.180   6.470
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  610 2022/01/11 19:52:36
# number of packets previous packets
      5               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.470   6.540
# Air Pump max current time  max pressure
     119       245     0.000
# Air Valve Positions
   1.283   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.554
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20   610      1
#               Date   Code Description
 2022/01/11 19:51:53 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  610  3      0
# MC    Pres      Time  Pcase Vacuum
   1     0.080    449   6.070
   3    72.720      2   6.210
   3    72.200      2   6.200
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  610       197    2022/01/11 19:57:37
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      610	(cycle number) |
     MPS =        0	(phase) |
     MPC =        1	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       95	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      607	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       50	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      600	(sec) |
  DFT[0] =      347	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        5	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      607	(cycle) |
  PTO[0] =     9000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      607	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        2	(dBar) |
     ADD =        5	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      607	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      180	(sec) |
     SSP =       60	(sec) |

