# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1269  254 000000000432020  7174     0   4 2022/10/08, 06:20:25,  68.8034  -52.5946   2      0 6,14,14
SBD 11146  1269  340 000000000432020  7175     0   3 2022/10/08, 06:20:46,  68.8034  -52.5946   2      0 13
SBD 11146  1269  311 000000000432020  7176     0   2 2022/10/08, 06:21:33,  68.8034  -52.5946   2      0 3,8,15,13
SBD 11146  1269  260 000000000432020  7177     0   1 2022/10/08, 06:21:43,  68.8034  -52.5946   2      0 1,2
SBD 11146  1269  289 000000000432020  7178     0   0 2022/10/08, 06:22:00,  68.8034  -52.5946   2      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      17         1394           60        1454
# Sensor_ID bytes     % packets Description
          0    48  3.44       2         GPS
          1    91  6.53       1    Pressure
          2   157 11.26       1 Temperature
          3   149 10.69       1    Salinity
          4   118  8.46       1        Fall
          5    48  3.44       1        Rise
          6   147 10.55       1        Pump
          8     8  0.57       1     Surface
         13   404 28.98       2  Parameters
         14   158 11.33       5 Engineering
         15    66  4.73       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1269 1 2022/10/08 02:06 +68.78545  -52.65280    0  4 33 36 39  2.2     0
G 1269 2 2022/10/08 06:14 +68.78520  -52.65138   40  5  0 30 42 20.5     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1269  134    1     3         0
#      Pres      Temp    Salinity
      1.120     3.9630    32.6010
      2.000     3.9640    32.6020
      3.040     3.9660    32.6020
      4.000     3.9640    32.6020
      5.000     3.9730    32.6030
      6.000     3.9790    32.6030
      7.080     3.9840    32.6030
      8.000     3.9730    32.6040
      9.080     3.9750    32.6030
     10.000     3.9810    32.6040
     11.040     3.9860    32.6040
     11.960     3.9850    32.6040
     13.000     3.9840    32.6050
     14.000     3.9830    32.6040
     15.000     3.9820    32.6050
     16.080     3.9830    32.6050
     17.000     3.9820    32.6060
     18.040     3.9840    32.6070
     19.000     3.9980    32.6100
     20.000     4.0420    32.6140
     21.040     4.0620    32.6240
     22.000     4.0770    32.6270
     23.000     4.0950    32.6330
     23.960     4.1440    32.6440
     25.040     4.2070    32.6640
     25.960     4.2430    32.6790
     27.000     4.2720    32.6910
     28.000     4.2980    32.7060
     29.000     4.3020    32.7110
     29.960     4.3040    32.7140
     31.000     4.2910    32.7170
     32.000     4.2770    32.7140
     33.080     4.2630    32.7150
     34.040     4.2680    32.7190
     35.080     4.2810    32.7320
     35.960     4.2780    32.7400
     36.960     4.2870    32.7520
     37.960     4.3080    32.7700
     39.040     4.3220    32.7770
     40.000     4.3350    32.7800
     41.000     4.3290    32.7840
     42.040     4.3230    32.7890
     43.000     4.3100    32.8010
     44.080     4.3040    32.8010
     44.960     4.3020    32.8000
     45.960     4.3030    32.8020
     46.960     4.3040    32.8030
     48.080     4.3080    32.8070
     49.000     4.3270    32.8170
     50.040     4.3450    32.8340
     51.040     4.3560    32.8520
     51.960     4.3580    32.8700
     52.960     4.3530    32.8800
     54.000     4.3470    32.8880
     55.000     4.3440    32.8900
     56.040     4.3450    32.8900
     57.000     4.3440    32.8890
     57.960     4.3440    32.8880
     59.080     4.3420    32.8890
     60.000     4.3390    32.8920
     60.920     4.3250    32.9030
     61.960     4.3210    32.9110
     63.040     4.3160    32.9270
     64.080     4.3020    32.9500
     65.080     4.3010    32.9530
     66.000     4.3000    32.9530
     67.080     4.2950    32.9670
     67.960     4.2930    32.9790
     68.960     4.2920    32.9840
     70.000     4.2840    32.9920
     71.040     4.2560    33.0140
     71.920     4.2380    33.0530
     73.000     4.2360    33.0840
     74.000     4.2430    33.0950
     74.960     4.2500    33.1110
     76.040     4.2470    33.1280
     77.000     4.2170    33.1670
     78.000     4.1430    33.2040
     79.080     4.1190    33.2200
     80.080     4.0640    33.2300
     81.040     4.0500    33.2440
     82.040     4.0330    33.2450
     83.000     4.0110    33.2410
     83.960     4.0030    33.2540
     85.000     3.9640    33.2680
     86.000     3.9270    33.2710
     87.000     3.9250    33.2700
     88.040     3.9180    33.2700
     88.960     3.8700    33.2860
     89.960     3.8440    33.2960
     91.040     3.8360    33.3020
     92.040     3.8300    33.3040
     93.000     3.8130    33.3100
     94.040     3.7920    33.3150
     95.000     3.7510    33.3280
     95.920     3.6640    33.3610
     96.960     3.6080    33.3780
     97.960     3.5850    33.3840
     98.960     3.5440    33.3910
     99.960     3.4010    33.4310
    100.960     3.2860    33.4580
    102.000     3.1910    33.4800
    103.080     3.1580    33.4870
    104.080     3.1290    33.4930
    105.040     3.1090    33.4980
    106.000     3.0550    33.5130
    106.960     3.0300    33.5180
    107.960     3.0250    33.5180
    108.960     3.0240    33.5180
    110.000     3.0160    33.5180
    111.040     2.9870    33.5270
    112.000     2.9820    33.5280
    112.960     2.9800    33.5280
    114.000     2.9760    33.5270
    115.080     2.9180    33.5420
    116.080     2.8400    33.5610
    117.000     2.7560    33.5820
    118.000     2.6950    33.5980
    119.000     2.6490    33.6090
    119.960     2.6280    33.6140
    121.000     2.5750    33.6270
    122.000     2.5460    33.6350
    123.040     2.5230    33.6400
    124.000     2.5000    33.6460
    124.960     2.5020    33.6450
    125.960     2.5000    33.6460
    127.000     2.4850    33.6490
    128.000     2.4540    33.6570
    129.000     2.4430    33.6600
    130.000     2.4240    33.6650
    131.000     2.4000    33.6710
    132.080     2.3810    33.6750
    133.040     2.3610    33.6810
    133.920     2.4100    33.6720
# Fall Time Series
#  Cycle npts     Start                Date
F00 1269   22 718509996 2022/10/08 02:06:36
# time     Pres phase                Date
     0    0.080     1 2022/10/08 02:06:36
   251   20.040    10 2022/10/08 02:10:47
   321   37.320     2 2022/10/08 02:11:57
   333   40.280    10 2022/10/08 02:12:09
   417   60.720    10 2022/10/08 02:13:33
   505   80.560    10 2022/10/08 02:15:01
   600  100.440    10 2022/10/08 02:16:36
   707  120.080    10 2022/10/08 02:18:23
  1608  134.720    10 2022/10/08 02:33:24
  2508  134.600    10 2022/10/08 02:48:24
  3408  134.440    10 2022/10/08 03:03:24
  4309  134.320    10 2022/10/08 03:18:25
  5209  134.200    10 2022/10/08 03:33:25
  6110  134.120    10 2022/10/08 03:48:26
  7010  134.040    10 2022/10/08 04:03:26
  7910  133.960    10 2022/10/08 04:18:26
  8811  133.920    10 2022/10/08 04:33:27
  9711  133.880    10 2022/10/08 04:48:27
 10110  133.880     3 2022/10/08 04:55:06
 11075  133.840    10 2022/10/08 05:11:11
 11975  133.840    10 2022/10/08 05:26:11
 12876  133.880    10 2022/10/08 05:41:12
# Rise Time Series
#  Cycle npts     Start                Date
R00 1269    8 718522880 2022/10/08 05:41:20
# time     Pres phase                Date
     0  133.880     7 2022/10/08 05:41:20
   730  113.360    13 2022/10/08 05:53:30
   996   92.320    13 2022/10/08 05:57:56
  1222   72.520    13 2022/10/08 06:01:42
  1427   52.960    13 2022/10/08 06:05:07
  1632   32.760    13 2022/10/08 06:08:32
  1837   11.400    13 2022/10/08 06:11:57
  1939    0.800     8 2022/10/08 06:13:39
# Pump Time Series
#   Cycle npts
B02  1269   13
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.320  785   11.83       0    655    -9     2
   133.880   94   12.23       0    655    -9     7
   133.920   20   12.23       0    657    -9    13
   132.520   20   12.22       0    659    -9    13
   128.360   20   12.20       0    660    -9    13
   123.120   20   12.19       0    662    -9    13
   116.160   20   12.18       0    663    -9    13
   107.480   20   12.17       0    664    -9    13
    97.400   20   12.17       0    665    -9    13
    85.280   21   12.15       0    665    -9    13
    74.360   21   12.15       0    665    -9    13
    51.320   21   12.14       0    666    -9    13
    -0.040  120   12.10       0    675    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1269  2.4   41    90   500   600   178    10     0      8     1    4020   15  0.2005
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1269 2022/10/08 02:06:34
# Pcase vacuums
   6.830   6.490
# Oil valve positions
   1.688   0.493   0.669
# Seek attempts and pump time
      1    -18
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1269 2022/10/08 05:41:12 9999/99/99 99:99:99
# Pcase vacuums
   6.490   6.490
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1269 2022/10/08 05:41:20 9999/99/99 99:99:99
# Pcase vacuums
   6.490   6.660
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1269 2022/10/08 06:13:40
# number of packets previous packets
      5               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.660   6.780
# Air Pump max current time  max pressure
     119       377     0.000
# Air Valve Positions
   1.281   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.526
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1269      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1269  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.040    315   5.900
   3   133.880     21   6.490
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1269       197    2022/10/08 06:21:33
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1269	(cycle number) |
     MPS =        0	(phase) |
     MPC =      660	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      500	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1266	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      900	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       20	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =      785	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     9000	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65521	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1266	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1266	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       20	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1266	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =      120	(sec) |

