# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1257  335 000000000432020  7115     0   4 2022/10/06, 00:17:40,  68.7768  -52.6548   2      0 6
SBD 11146  1257  280 000000000432020  7116     0   3 2022/10/06, 00:17:52,  68.7768  -52.6548   2      0 13,6,14,14
SBD 11146  1257  340 000000000432020  7117     0   2 2022/10/06, 00:18:03,  68.7971  -52.7336   2      0 13
SBD 11146  1257  270 000000000432020  7118     0   1 2022/10/06, 00:18:16,  68.7768  -52.6548   2      0 2,3,8,15
SBD 11146  1257  332 000000000432020  7119     0   0 2022/10/06, 00:18:30,  68.7768  -52.6548   2      0 0,4,14,14,5,14,0,1
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      18         1497           60        1557
# Sensor_ID bytes     % packets Description
          0    48  3.21       2         GPS
          1    68  4.54       1    Pressure
          2    92  6.15       1 Temperature
          3    92  6.15       1    Salinity
          4   103  6.88       1        Fall
          5    38  2.54       1        Rise
          6   360 24.05       2        Pump
          8     8  0.53       1     Surface
         13   404 26.99       2  Parameters
         14   218 14.56       5 Engineering
         15    66  4.41       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1257 1 2022/10/05 19:16 +68.77835  -52.68018    0  4 28 33 38  1.6     0
G 1257 2 2022/10/06 00:11 +68.77823  -52.67465   30  7  0 36 45  1.9     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1257   77    1     3         0
#      Pres      Temp    Salinity
      1.000     4.0760    32.5460
      1.960     4.0820    32.5460
      3.000     4.0860    32.5460
      4.000     4.0980    32.5470
      5.000     4.1170    32.5550
      6.040     4.1210    32.5570
      6.960     4.1200    32.5590
      8.000     4.1210    32.5610
      8.960     4.1310    32.5640
      9.960     4.1250    32.5670
     11.000     4.1200    32.5680
     11.960     4.1200    32.5730
     12.960     4.1270    32.5800
     14.000     4.1360    32.5830
     14.960     4.1470    32.5880
     16.000     4.1490    32.5900
     17.000     4.1530    32.5940
     18.080     4.1570    32.5950
     19.000     4.1810    32.6010
     20.040     4.1980    32.6080
     21.000     4.1980    32.6090
     22.040     4.1870    32.6090
     22.960     4.1800    32.6090
     24.040     4.1820    32.6110
     25.040     4.1760    32.6150
     26.080     4.1900    32.6250
     26.960     4.2210    32.6470
     27.960     4.2420    32.6600
     28.960     4.2480    32.6640
     29.960     4.2520    32.6670
     31.040     4.2570    32.6690
     32.000     4.2590    32.6700
     33.040     4.2630    32.6700
     33.960     4.2750    32.6790
     35.000     4.2780    32.6870
     35.960     4.3000    32.7060
     37.080     4.3080    32.7260
     38.000     4.3080    32.7300
     39.000     4.3080    32.7300
     40.040     4.3080    32.7310
     41.040     4.3100    32.7320
     42.040     4.3100    32.7340
     43.000     4.3120    32.7370
     44.000     4.3120    32.7380
     44.960     4.3130    32.7390
     45.920     4.3130    32.7380
     46.960     4.3130    32.7390
     47.960     4.3150    32.7420
     49.000     4.3180    32.7560
     50.040     4.3180    32.7660
     51.040     4.3160    32.7680
     52.000     4.3150    32.7690
     53.120     4.3150    32.7760
     54.000     4.3170    32.7810
     55.040     4.3230    32.7860
     55.960     4.3300    32.8040
     57.000     4.3310    32.8060
     58.040     4.3300    32.8100
     58.960     4.3290    32.8100
     59.960     4.3260    32.8090
     61.040     4.3250    32.8120
     62.000     4.3260    32.8180
     63.040     4.3270    32.8270
     64.000     4.3270    32.8300
     65.080     4.3290    32.8390
     66.000     4.3290    32.8440
     67.040     4.3300    32.8500
     68.040     4.3320    32.8580
     69.000     4.3350    32.8710
     70.000     4.3360    32.8800
     70.960     4.3360    32.8890
     71.960     4.3340    32.9120
     73.000     4.3280    32.9370
     73.960     4.3170    32.9570
     74.960     4.3210    32.9510
     76.040     4.3200    32.9530
     76.760     4.3000    32.9690
# Fall Time Series
#  Cycle npts     Start                Date
F00 1257   19 718312582 2022/10/05 19:16:22
# time     Pres phase                Date
     0   -0.120     1 2022/10/05 19:16:22
   303   20.000    10 2022/10/05 19:21:25
   373   37.680     2 2022/10/05 19:22:35
   385   40.720    10 2022/10/05 19:22:47
   463   60.120    10 2022/10/05 19:24:05
  1363   76.920    10 2022/10/05 19:39:05
  2263   77.000    10 2022/10/05 19:54:05
  3164   77.040    10 2022/10/05 20:09:06
  4064   77.040    10 2022/10/05 20:24:06
  4964   77.080    10 2022/10/05 20:39:06
  5865   77.080    10 2022/10/05 20:54:07
  6765   77.120    10 2022/10/05 21:09:07
  7666   77.080    10 2022/10/05 21:24:08
  8566   77.080    10 2022/10/05 21:39:08
  9466   77.040    10 2022/10/05 21:54:08
 10162   77.040     3 2022/10/05 22:05:44
 11129   76.960    10 2022/10/05 22:21:51
 12030   76.920    10 2022/10/05 22:36:52
 12930   76.840    10 2022/10/05 22:51:52
# Rise Time Series
#  Cycle npts     Start                Date
R00 1257    6 718325521 2022/10/05 22:52:01
# time     Pres phase                Date
     0   76.840     7 2022/10/05 22:52:01
  1795   56.680    13 2022/10/05 23:21:56
  1959   34.240    13 2022/10/05 23:24:40
  2123   12.840    13 2022/10/05 23:27:24
  4721   -0.120    13 2022/10/06 00:10:42
  4732   -0.160     8 2022/10/06 00:10:53
# Pump Time Series
#   Cycle npts
B02  1257   32
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.680  785   12.01       0    652    -9     2
    76.840   94   12.38       0    649    -9     7
    76.920   21   12.37       0    651    -9    13
    76.840   21   12.36       0    653    -9    13
    76.800   21   12.35       0    654    -9    13
    76.800   21   12.35       0    656    -9    13
    76.760   21   12.33       0    657    -9    13
    76.800   21   12.32       0    659    -9    13
    76.800   21   12.32       0    660    -9    13
    76.680   21   12.31       0    662    -9    13
    76.720   21   12.29       0    663    -9    13
    76.560   21   12.29       0    665    -9    13
    76.600   21   12.28       0    666    -9    13
    73.640   21   12.27       0    667    -9    13
    68.160   21   12.27       0    669    -9    13
  9999.000   40   12.27       0    673    -9    13
  9999.000   40   12.27       0    676    -9    13
  9999.000   41   12.26       0    679    -9    13
  9999.000   40   12.26       0    682    -9    13
  9999.000   40   12.26       0    685    -9    13
  9999.000   41   12.26       0    688    -9    13
  9999.000   40   12.24       0    691    -9    13
  9999.000   40   12.24       0    694    -9    13
  9999.000   41   12.24       0    696    -9    13
  9999.000   40   12.24       0    699    -9    13
  9999.000   40   12.24       0    702    -9    13
  9999.000   40   12.23       0    705    -9    13
  9999.000   40   12.22       0    708    -9    13
  9999.000   40   12.22       0    710    -9    13
  9999.000   40   12.21       0    713    -9    13
  9999.000   40   12.21       0    716    -9    13
    -0.040  120   12.19       0    725    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1257  2.4   41    90   500   600   178    10     0      8     1    4020   15  0.2005
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1257 2022/10/05 19:16:20
# Pcase vacuums
   6.930   6.430
# Oil valve positions
   1.689   0.493   0.667
# Seek attempts and pump time
      1    -21
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1257 2022/10/05 22:51:52 9999/99/99 99:99:99
# Pcase vacuums
   6.430   6.430
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1257 2022/10/05 22:52:01 9999/99/99 99:99:99
# Pcase vacuums
   6.420   7.160
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1257 2022/10/06 00:10:53
# number of packets previous packets
      5               5
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   7.160   7.270
# Air Pump max current time  max pressure
     119       375     0.000
# Air Valve Positions
   1.281   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.530
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1257     10
#               Date   Code Description
 2022/10/05 23:29:26 0x0114 AscendCtd
 2022/10/05 23:30:06 0x0114 AscendCtd
 2022/10/05 23:30:46 0x0114 AscendCtd
 2022/10/05 23:31:26 0x0114 AscendCtd
 2022/10/05 23:32:06 0x0114 AscendCtd
 2022/10/05 23:32:46 0x0114 AscendCtd
 2022/10/05 23:33:26 0x0114 AscendCtd
 2022/10/05 23:34:06 0x0114 AscendCtd
 2022/10/05 23:34:46 0x0114 AscendCtd
 2022/10/05 23:35:27 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1257  2      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.080    368   5.890
   3    77.040     23   6.430
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1257       197    2022/10/06 00:18:03
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1257	(cycle number) |
     MPS =        0	(phase) |
     MPC =      648	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      500	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1254	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      900	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       20	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =      785	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     9000	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65517	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1254	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1254	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       20	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1254	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =      120	(sec) |

