# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1242  148 000000000432020  7040     0   4 2022/10/03, 04:18:23,  68.7971  -52.7336   4      0 14
SBD 11146  1242  331 000000000432020  7041     0   3 2022/10/03, 04:18:51,  68.7971  -52.7336   3      0 13,6,14
SBD 11146  1242  340 000000000432020  7042     0   2 2022/10/03, 04:19:02,  68.7971  -52.7336   3      0 13
SBD 11146  1242  280 000000000432020  7043     0   1 2022/10/03, 04:19:16,  68.7768  -52.6548   3      0 2,3,8,15
SBD 11146  1242  322 000000000432020  7044     0   0 2022/10/03, 04:19:26,  68.7704  -52.7935   4      0 0,4,14,14,5,14,0,1
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      17         1361           60        1421
# Sensor_ID bytes     % packets Description
          0    48  3.53       2         GPS
          1    63  4.63       1    Pressure
          2   101  7.42       1 Temperature
          3    93  6.83       1    Salinity
          4   103  7.57       1        Fall
          5    33  2.42       1        Rise
          6   224 16.46       1        Pump
          8     8  0.59       1     Surface
         13   404 29.68       2  Parameters
         14   218 16.02       5 Engineering
         15    66  4.85       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1242 1 2022/10/02 23:50 +68.78057  -52.70111    0  9 30 40 44  0.8     0
G 1242 2 2022/10/03 04:12 +68.78153  -52.70607   50  6  0 31 44  1.4     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1242   78    1     3         0
#      Pres      Temp    Salinity
      1.040     4.2070    32.5810
      1.920     4.2230    32.5790
      3.000     4.2360    32.5820
      4.040     4.2460    32.5850
      5.000     4.2530    32.5860
      6.000     4.2860    32.5940
      7.000     4.3010    32.6010
      7.960     4.3010    32.6010
      9.000     4.2990    32.6010
      9.960     4.3000    32.6010
     11.000     4.2990    32.6000
     11.960     4.2990    32.6000
     13.000     4.3000    32.6010
     14.000     4.3000    32.6010
     15.080     4.3000    32.6020
     16.000     4.3000    32.6020
     17.040     4.3010    32.6020
     17.960     4.3030    32.6030
     18.960     4.3030    32.6040
     20.000     4.3030    32.6050
     20.960     4.3050    32.6080
     22.080     4.3070    32.6120
     23.000     4.3080    32.6130
     23.960     4.3190    32.6300
     25.080     4.3230    32.6380
     26.000     4.3260    32.6400
     27.000     4.3310    32.6460
     28.040     4.3320    32.6480
     29.000     4.3420    32.6530
     30.000     4.3440    32.6540
     31.080     4.3460    32.6550
     32.000     4.3520    32.6630
     33.000     4.3650    32.6870
     34.000     4.3730    32.7130
     35.000     4.3730    32.7440
     36.040     4.3730    32.7410
     37.040     4.3720    32.7420
     37.960     4.3720    32.7410
     39.000     4.3720    32.7610
     40.000     4.3690    32.8110
     41.000     4.3670    32.8390
     42.040     4.3670    32.8300
     43.040     4.3670    32.8380
     43.960     4.3670    32.8480
     44.960     4.3660    32.8560
     46.040     4.3660    32.8550
     47.000     4.3660    32.8620
     48.040     4.3660    32.8720
     48.960     4.3660    32.8730
     50.000     4.3610    32.8860
     50.920     4.3620    32.8830
     52.000     4.3600    32.8890
     53.080     4.3540    32.9080
     54.040     4.3530    32.9120
     54.960     4.3530    32.9110
     56.040     4.3520    32.9150
     57.040     4.3460    32.9290
     58.040     4.3510    32.9180
     59.040     4.3500    32.9190
     60.040     4.3490    32.9210
     61.040     4.3470    32.9270
     62.000     4.3490    32.9230
     63.000     4.3460    32.9280
     64.000     4.3420    32.9360
     64.960     4.3400    32.9400
     66.000     4.3350    32.9480
     67.000     4.3220    32.9710
     68.040     4.3170    32.9810
     69.000     4.3140    32.9850
     70.000     4.3110    32.9900
     71.000     4.3000    33.0050
     72.000     4.2840    33.0270
     73.000     4.2580    33.0670
     73.960     4.2510    33.0790
     74.960     4.2480    33.0840
     76.000     4.2420    33.0940
     77.040     4.2320    33.1120
     77.880     4.2200    33.1330
# Fall Time Series
#  Cycle npts     Start                Date
F00 1242   19 718069817 2022/10/02 23:50:17
# time     Pres phase                Date
     0    0.040     1 2022/10/02 23:50:17
   270   20.200    10 2022/10/02 23:54:47
   343   37.440     2 2022/10/02 23:56:00
   355   40.320    10 2022/10/02 23:56:12
   440   60.320    10 2022/10/02 23:57:37
  1340   77.520    10 2022/10/03 00:12:37
  2241   77.560    10 2022/10/03 00:27:38
  3141   77.560    10 2022/10/03 00:42:38
  4042   77.600    10 2022/10/03 00:57:39
  4942   77.600    10 2022/10/03 01:12:39
  5842   77.640    10 2022/10/03 01:27:39
  6743   77.680    10 2022/10/03 01:42:40
  7643   77.680    10 2022/10/03 01:57:40
  8543   77.720    10 2022/10/03 02:12:40
  9444   77.760    10 2022/10/03 02:27:41
 10130   77.800     3 2022/10/03 02:39:07
 11097   77.800    10 2022/10/03 02:55:14
 11998   77.800    10 2022/10/03 03:10:15
 12898   77.840    10 2022/10/03 03:25:15
# Rise Time Series
#  Cycle npts     Start                Date
R00 1242    5 718082724 2022/10/03 03:25:24
# time     Pres phase                Date
     0   77.840     7 2022/10/03 03:25:24
   913   56.480    13 2022/10/03 03:40:37
  1180   36.280    13 2022/10/03 03:45:04
  1405   15.680    13 2022/10/03 03:48:49
  2793   -0.040     8 2022/10/03 04:11:57
# Pump Time Series
#   Cycle npts
B02  1242   20
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.440  785   12.16       0    650    -9     2
    77.840   93   12.48       0    649    -9     7
    77.840   21   12.47       0    652    -9    13
    77.880   21   12.46       0    653    -9    13
    76.840   21   12.45       0    655    -9    13
    73.440   21   12.44       0    656    -9    13
    67.240   21   12.43       0    658    -9    13
    61.960   21   12.43       0    659    -9    13
    54.920   21   12.42       0    660    -9    13
    45.640   21   12.41       0    662    -9    13
    36.280   21   12.41       0    663    -9    13
  9999.000   40   12.40       0    667    -9    13
  9999.000   41   12.40       0    670    -9    13
  9999.000   40   12.39       0    673    -9    13
  9999.000   40   12.38       0    676    -9    13
  9999.000   41   12.38       0    679    -9    13
  9999.000   40   12.38       0    682    -9    13
  9999.000   40   12.37       0    684    -9    13
  9999.000   41   12.37       0    687    -9    13
    -0.040  120   12.34       0    697    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1242  2.4   41    90   500   600   178    10     0      8     1    4020   15  0.2005
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1242 2022/10/02 23:50:15
# Pcase vacuums
   6.810   6.430
# Oil valve positions
   1.690   0.494   0.663
# Seek attempts and pump time
      1    -21
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1242 2022/10/03 03:25:16 9999/99/99 99:99:99
# Pcase vacuums
   6.430   6.430
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1242 2022/10/03 03:25:24 9999/99/99 99:99:99
# Pcase vacuums
   6.430   6.880
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1242 2022/10/03 04:11:58
# number of packets previous packets
      4               4
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.870   6.980
# Air Pump max current time  max pressure
     119       371     0.000
# Air Valve Positions
   1.280   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.529
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1242     10
#               Date   Code Description
 2022/10/03 03:51:53 0x0114 AscendCtd
 2022/10/03 03:52:33 0x0114 AscendCtd
 2022/10/03 03:53:13 0x0114 AscendCtd
 2022/10/03 03:53:53 0x0114 AscendCtd
 2022/10/03 03:54:33 0x0114 AscendCtd
 2022/10/03 03:55:14 0x0114 AscendCtd
 2022/10/03 03:55:54 0x0114 AscendCtd
 2022/10/03 03:56:34 0x0114 AscendCtd
 2022/10/03 03:57:14 0x0114 AscendCtd
 2022/10/03 03:57:54 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1242  2      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.120    338   5.880
   3    77.800     23   6.430
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1242       197    2022/10/03 04:19:02
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1242	(cycle number) |
     MPS =        0	(phase) |
     MPC =      633	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      500	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1239	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      900	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       20	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =      785	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     9000	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65514	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1239	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1239	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       20	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1239	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =      120	(sec) |

