# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1240  255 000000000432020  7032     0   3 2022/10/02, 19:41:29,  68.7768  -52.6548   3      0 6,14,14
SBD 11146  1240  340 000000000432020  7033     0   2 2022/10/02, 19:41:39,  68.7768  -52.6548   2      0 13
SBD 11146  1240  316 000000000432020  7034     0   1 2022/10/02, 19:41:49,  68.7971  -52.7336   2      0 2,3,8,15,13
SBD 11146  1240  324 000000000432020  7035     0   0 2022/10/02, 19:41:59,  68.7971  -52.7336   2      0 0,4,14,14,5,14,0,1
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      17         1187           48        1235
# Sensor_ID bytes     % packets Description
          0    48  4.04       2         GPS
          1    65  5.48       1    Pressure
          2    65  5.48       1 Temperature
          3    89  7.50       1    Salinity
          4   103  8.68       1        Fall
          5    33  2.78       1        Rise
          6   136 11.46       1        Pump
          8     8  0.67       1     Surface
         13   404 34.04       2  Parameters
         14   170 14.32       5 Engineering
         15    66  5.56       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1240 1 2022/10/02 15:34 +68.77914  -52.69855   50  6  0 31 43  5.2     0
G 1240 2 2022/10/02 19:35 +68.77944  -52.69740   20  4  0 29 46  1.9     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1240   66    1     3         0
#      Pres      Temp    Salinity
      1.080     4.3130    32.5760
      2.000     4.3130    32.5770
      2.960     4.3100    32.5760
      4.000     4.3170    32.5790
      5.000     4.3240    32.5850
      6.040     4.3240    32.5880
      7.000     4.3230    32.5980
      8.000     4.3230    32.5960
      9.040     4.3230    32.5950
     10.040     4.3220    32.5990
     11.000     4.3220    32.6000
     12.040     4.3210    32.6010
     12.960     4.3200    32.6020
     14.000     4.3200    32.6020
     15.000     4.3180    32.6040
     15.960     4.3130    32.6040
     16.960     4.3120    32.6050
     17.960     4.3120    32.6050
     19.000     4.3120    32.6050
     20.000     4.3120    32.6050
     21.000     4.3130    32.6040
     21.960     4.3130    32.6040
     23.040     4.3120    32.6050
     24.040     4.3120    32.6050
     25.000     4.3100    32.6080
     26.000     4.3100    32.6100
     27.040     4.3090    32.6120
     28.000     4.3070    32.6160
     29.000     4.3040    32.6190
     29.960     4.3060    32.6220
     31.000     4.3090    32.6360
     31.960     4.3120    32.6410
     33.000     4.3140    32.6500
     34.040     4.3220    32.6820
     35.000     4.3240    32.6900
     36.080     4.3260    32.6960
     37.040     4.3270    32.7020
     38.000     4.3330    32.7120
     38.960     4.3360    32.7170
     40.080     4.3370    32.7190
     41.000     4.3380    32.7190
     42.000     4.3410    32.7220
     43.040     4.3460    32.7250
     43.960     4.3500    32.7280
     45.000     4.3500    32.7290
     46.040     4.3500    32.7290
     47.000     4.3510    32.7340
     48.040     4.3510    32.7350
     49.080     4.3520    32.7370
     49.960     4.3530    32.7410
     50.920     4.3550    32.7470
     51.960     4.3560    32.7480
     52.960     4.3580    32.7510
     54.000     4.3600    32.7560
     55.000     4.3620    32.7580
     56.000     4.3640    32.7620
     57.040     4.3690    32.7740
     57.960     4.3720    32.7840
     59.000     4.3710    32.7820
     60.040     4.3760    32.8030
     61.000     4.3790    32.8190
     62.040     4.3790    32.8190
     63.040     4.3790    32.8250
     64.040     4.3790    32.8310
     65.320     4.3770    32.8440
     65.520     4.3770    32.8450
# Fall Time Series
#  Cycle npts     Start                Date
F00 1240   19 718040077 2022/10/02 15:34:37
# time     Pres phase                Date
     0   -0.080     1 2022/10/02 15:34:37
   309   20.480    10 2022/10/02 15:39:46
   376   37.240     2 2022/10/02 15:40:53
   388   40.240    10 2022/10/02 15:41:05
   472   60.520    10 2022/10/02 15:42:29
  1373   65.680    10 2022/10/02 15:57:30
  2273   65.760    10 2022/10/02 16:12:30
  3173   65.800    10 2022/10/02 16:27:30
  4074   65.800    10 2022/10/02 16:42:31
  4974   65.800    10 2022/10/02 16:57:31
  5875   65.760    10 2022/10/02 17:12:32
  6775   65.760    10 2022/10/02 17:27:32
  7675   65.720    10 2022/10/02 17:42:32
  8576   65.680    10 2022/10/02 17:57:33
  9476   65.680    10 2022/10/02 18:12:33
 10164   65.600     3 2022/10/02 18:24:01
 11132   65.560    10 2022/10/02 18:40:09
 12032   65.520    10 2022/10/02 18:55:09
 12933   65.440    10 2022/10/02 19:10:10
# Rise Time Series
#  Cycle npts     Start                Date
R00 1240    5 718053018 2022/10/02 19:10:18
# time     Pres phase                Date
     0   65.480     7 2022/10/02 19:10:18
   874   45.800    13 2022/10/02 19:24:52
  1120   24.680    13 2022/10/02 19:28:58
  1324    4.720    13 2022/10/02 19:32:22
  1488   -0.040     8 2022/10/02 19:35:06
# Pump Time Series
#   Cycle npts
B02  1240   12
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.240  785   12.30       0    651    -9     2
    65.480   95   12.56       0    649    -9     7
    65.480   21   12.55       0    651    -9    13
    65.480   21   12.54       0    653    -9    13
    64.960   21   12.54       0    654    -9    13
    62.240   21   12.52       0    656    -9    13
    57.520   21   12.52       0    657    -9    13
    50.440   21   12.51       0    659    -9    13
    40.920   21   12.51       0    661    -9    13
    28.440   20   12.50       0    662    -9    13
     1.280   40   12.48       0    665    -9    13
    -0.040  120   12.46       0    675    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1240  2.4   41    90   500   600   178    10     0      8     1    4020   15  0.2005
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1240 2022/10/02 15:34:34
# Pcase vacuums
   6.890   6.430
# Oil valve positions
   1.688   0.493   0.664
# Seek attempts and pump time
      1    -21
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1240 2022/10/02 19:10:10 9999/99/99 99:99:99
# Pcase vacuums
   6.430   6.430
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1240 2022/10/02 19:10:18 9999/99/99 99:99:99
# Pcase vacuums
   6.430   6.660
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1240 2022/10/02 19:35:06
# number of packets previous packets
      4               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.660   6.770
# Air Pump max current time  max pressure
     119       370     0.000
# Air Valve Positions
   1.280   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.529
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1240      2
#               Date   Code Description
 2022/10/02 19:33:44 0x0114 AscendCtd
 2022/10/02 19:34:24 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1240  2      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.040    370   5.880
   3    65.640     23   6.430
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1240       197    2022/10/02 19:41:49
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1240	(cycle number) |
     MPS =        0	(phase) |
     MPC =      631	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      500	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1237	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      900	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       20	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =      785	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     9000	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65515	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1237	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1237	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       20	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1237	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =      120	(sec) |

