# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1228  148 000000000432020  6969     0   5 2022/09/30, 13:41:02,  68.7971  -52.7336   4      0 14
SBD 11146  1228  233 000000000432020  6970     0   4 2022/09/30, 13:41:14,  68.7971  -52.7336   3      0 6,14
SBD 11146  1228  340 000000000432020  6971     0   3 2022/09/30, 13:41:27,  68.7768  -52.6548   4      0 13
SBD 11146  1228  334 000000000432020  6972     0   2 2022/09/30, 13:41:38,  68.7971  -52.7336   4      0 3,8,15,13
SBD 11146  1228  285 000000000432020  6973     0   1 2022/09/30, 13:42:17,  68.7768  -52.6548   6      0 1,2
SBD 11146  1228  299 000000000432020  6974     0   0 2022/09/30, 13:42:32,  68.7971  -52.7336   4      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      17         1567           72        1639
# Sensor_ID bytes     % packets Description
          0    48  3.06       2         GPS
          1    99  6.32       1    Pressure
          2   174 11.10       1 Temperature
          3   172 10.98       1    Salinity
          4   123  7.85       1        Fall
          5    53  3.38       1        Rise
          6   202 12.89       1        Pump
          8     8  0.51       1     Surface
         13   404 25.78       2  Parameters
         14   218 13.91       5 Engineering
         15    66  4.21       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1228 1 2022/09/30 09:01 +68.78193  -52.73469    0  8 23 35 43  0.9     0
G 1228 2 2022/09/30 13:34 +68.78362  -52.72546   60  9  0 37 45  1.1     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1228  149    1     3         0
#      Pres      Temp    Salinity
      1.040     4.2430    32.4480
      2.000     4.2430    32.4470
      3.000     4.2430    32.4470
      4.000     4.2480    32.4500
      5.000     4.2540    32.4520
      6.040     4.2740    32.4560
      7.000     4.2930    32.4630
      8.040     4.3060    32.4700
      9.000     4.3070    32.4750
     10.000     4.3070    32.4790
     11.000     4.3040    32.4830
     12.000     4.3000    32.4860
     13.040     4.2930    32.4880
     14.000     4.2900    32.4920
     14.960     4.2970    32.4970
     16.040     4.3170    32.5230
     17.000     4.3350    32.5440
     17.960     4.3520    32.5840
     19.040     4.3640    32.6160
     20.080     4.3690    32.6200
     20.960     4.3700    32.6220
     21.960     4.3690    32.6210
     23.000     4.3710    32.6220
     24.000     4.3700    32.6210
     25.040     4.3710    32.6230
     26.000     4.3720    32.6240
     27.040     4.3730    32.6240
     28.040     4.3760    32.6260
     29.040     4.3830    32.6300
     30.000     4.3910    32.6370
     31.000     4.3950    32.6410
     32.000     4.4040    32.6500
     33.040     4.4170    32.6650
     33.960     4.4200    32.6690
     35.040     4.4230    32.6740
     35.960     4.4220    32.6870
     36.960     4.4210    32.6760
     37.960     4.4210    32.6710
     39.040     4.4210    32.7040
     40.000     4.4220    32.7090
     41.040     4.4220    32.7120
     42.040     4.4210    32.7160
     43.000     4.4200    32.7210
     44.040     4.4170    32.7290
     45.040     4.4130    32.7330
     46.000     4.4080    32.7360
     46.920     4.4060    32.7380
     48.000     4.4040    32.7400
     49.000     4.4030    32.7430
     50.000     4.4010    32.7540
     50.960     4.4020    32.7570
     52.080     4.4020    32.7590
     53.000     4.4050    32.7620
     54.000     4.4070    32.7670
     55.080     4.4080    32.7710
     56.000     4.4080    32.7770
     56.960     4.4050    32.7900
     58.000     4.4010    32.8060
     59.040     4.3990    32.8130
     59.960     4.3970    32.8200
     61.040     4.3970    32.8220
     62.040     4.3980    32.8260
     63.000     4.3980    32.8260
     64.040     4.3980    32.8260
     65.040     4.3980    32.8260
     65.960     4.3980    32.8260
     67.040     4.3970    32.8270
     68.040     4.3950    32.8320
     68.960     4.3960    32.8300
     70.040     4.3920    32.8370
     71.040     4.3880    32.8450
     72.000     4.3840    32.8480
     73.040     4.3760    32.8510
     74.080     4.3700    32.8530
     74.960     4.3630    32.8550
     76.000     4.3610    32.8570
     77.040     4.3480    32.8650
     78.040     4.3410    32.8680
     79.040     4.3380    32.8760
     80.000     4.3570    32.8900
     80.960     4.3710    32.8990
     82.000     4.3740    32.9070
     82.960     4.3810    32.9240
     83.920     4.3870    32.9350
     84.920     4.3810    32.9510
     85.960     4.4280    32.9640
     87.000     4.4830    32.9840
     88.000     4.5080    33.0290
     89.040     4.4120    33.0470
     90.040     4.2840    33.1050
     91.040     4.2120    33.1680
     92.080     4.1860    33.1870
     93.040     4.1660    33.1990
     94.040     4.1450    33.2100
     95.040     4.1280    33.2240
     96.040     4.0660    33.2610
     97.040     3.9920    33.2940
     98.040     3.9000    33.3270
     99.040     3.8610    33.3310
    100.040     3.8510    33.3310
    101.040     3.8390    33.3320
    102.000     3.8330    33.3330
    102.960     3.8280    33.3340
    103.960     3.8200    33.3350
    105.000     3.8180    33.3340
    105.960     3.8170    33.3340
    106.960     3.7910    33.3460
    108.040     3.7850    33.3490
    109.040     3.7790    33.3510
    110.040     3.7780    33.3510
    111.080     3.7890    33.3540
    112.080     3.7910    33.3550
    113.000     3.7870    33.3590
    113.960     3.7750    33.3620
    115.000     3.7690    33.3640
    116.040     3.7720    33.3630
    117.000     3.7500    33.3690
    117.960     3.7300    33.3690
    119.000     3.7200    33.3700
    120.000     3.7120    33.3730
    121.040     3.6690    33.3880
    122.040     3.6220    33.3950
    123.040     3.6010    33.4020
    124.040     3.5420    33.4130
    125.040     3.5310    33.4150
    126.000     3.5260    33.4220
    127.040     3.5180    33.4290
    128.000     3.5150    33.4290
    129.000     3.5040    33.4330
    130.000     3.3010    33.5090
    130.960     3.2350    33.5130
    132.000     3.1380    33.5360
    133.040     3.0770    33.5410
    134.040     3.0230    33.5570
    134.960     2.9970    33.5590
    135.960     2.9950    33.5590
    136.960     2.9880    33.5610
    137.960     2.9850    33.5610
    139.000     2.9830    33.5620
    140.000     2.9850    33.5620
    141.040     2.9830    33.5630
    142.000     2.9790    33.5630
    143.000     2.9800    33.5620
    144.000     2.9730    33.5630
    145.000     2.9620    33.5630
    146.000     2.9310    33.5740
    147.000     2.6790    33.6510
    147.920     2.6250    33.6650
    149.000     2.6060    33.6870
# Fall Time Series
#  Cycle npts     Start                Date
F00 1228   23 717843721 2022/09/30 09:02:01
# time     Pres phase                Date
     0   -0.080     1 2022/09/30 09:02:01
   264   20.600    10 2022/09/30 09:06:25
   334   37.720     2 2022/09/30 09:07:35
   346   40.720    10 2022/09/30 09:07:47
   427   60.560    10 2022/09/30 09:09:08
   515   80.400    10 2022/09/30 09:10:36
   613  100.000    10 2022/09/30 09:12:14
   720  120.000    10 2022/09/30 09:14:01
   847  140.240    10 2022/09/30 09:16:08
  1748  143.240    10 2022/09/30 09:31:09
  2648  145.840    10 2022/09/30 09:46:09
  3549  147.640    10 2022/09/30 10:01:10
  4449  147.760    10 2022/09/30 10:16:10
  5350  147.880    10 2022/09/30 10:31:11
  6250  147.960    10 2022/09/30 10:46:11
  7150  148.120    10 2022/09/30 11:01:11
  8051  148.240    10 2022/09/30 11:16:12
  8951  148.360    10 2022/09/30 11:31:12
  9851  148.520    10 2022/09/30 11:46:12
 10122  148.520     3 2022/09/30 11:50:43
 11085  148.640    10 2022/09/30 12:06:46
 11986  148.800    10 2022/09/30 12:21:47
 12886  148.880    10 2022/09/30 12:36:47
# Rise Time Series
#  Cycle npts     Start                Date
R00 1228    9 717856616 2022/09/30 12:36:56
# time     Pres phase                Date
     0  148.880     7 2022/09/30 12:36:56
  1139  128.040    13 2022/09/30 12:55:55
  1344  106.880    13 2022/09/30 12:59:20
  1549   86.480    13 2022/09/30 13:02:45
  1754   66.240    13 2022/09/30 13:06:10
  1980   45.360    13 2022/09/30 13:09:56
  2185   24.760    13 2022/09/30 13:13:21
  2410    3.320    13 2022/09/30 13:17:06
  3395   -0.080     8 2022/09/30 13:33:31
# Pump Time Series
#   Cycle npts
B02  1228   18
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    37.720  785   13.28       0    655    -9     2
   148.880   94   13.47       0    655    -9     7
   149.040   21   13.47       0    658    -9    13
   148.960   21   13.45       0    659    -9    13
   149.040   21   13.46       0    661    -9    13
   149.040   21   13.44       0    662    -9    13
   149.080   21   13.44       0    664    -9    13
   149.040   21   13.43       0    665    -9    13
   148.440   21   13.43       0    667    -9    13
    53.480   20   13.42       0    662    -9    13
    13.240   20   13.41       0    664    -9    13
  9999.000   40   13.41       0    668    -9    13
  9999.000   40   13.41       0    671    -9    13
  9999.000   41   13.40       0    674    -9    13
  9999.000   40   13.40       0    677    -9    13
  9999.000   40   13.40       0    680    -9    13
  9999.000   41   13.39       0    683    -9    13
    -0.040  120   13.37       0    692    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1228  2.4   41    90   500   600   178    10     0      8     1    4020   15  0.2005
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1228 2022/09/30 09:01:59
# Pcase vacuums
   6.860   6.490
# Oil valve positions
   1.690   0.493   0.669
# Seek attempts and pump time
      1    -17
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1228 2022/09/30 12:36:48 9999/99/99 99:99:99
# Pcase vacuums
   6.490   6.490
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1228 2022/09/30 12:36:56 9999/99/99 99:99:99
# Pcase vacuums
   6.490   6.830
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1228 2022/09/30 13:33:31
# number of packets previous packets
      5               5
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.840   6.950
# Air Pump max current time  max pressure
     119       409     0.000
# Air Valve Positions
   1.281   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.528
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1228     10
#               Date   Code Description
 2022/09/30 13:18:07 0x0114 AscendCtd
 2022/09/30 13:18:47 0x0114 AscendCtd
 2022/09/30 13:19:27 0x0114 AscendCtd
 2022/09/30 13:20:07 0x0114 AscendCtd
 2022/09/30 13:20:47 0x0114 AscendCtd
 2022/09/30 13:21:27 0x0114 AscendCtd
 2022/09/30 13:22:07 0x0114 AscendCtd
 2022/09/30 13:22:47 0x0114 AscendCtd
 2022/09/30 13:23:27 0x0114 AscendCtd
 2022/09/30 13:24:08 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1228  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.000    328   5.890
   3   148.520     18   6.490
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1228       197    2022/09/30 13:41:38
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1228	(cycle number) |
     MPS =        0	(phase) |
     MPC =      619	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      500	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1225	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       36	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      900	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       20	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =      785	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     9000	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65519	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1225	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1225	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       80	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        1	(dBar) |
     ADD =       20	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1225	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =      120	(sec) |

