# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1104  269 000000000432020  6030     0   4 2022/09/01, 05:17:30,  67.7963  -53.8889   5      0 6,14,14
SBD 11146  1104  340 000000000432020  6031     0   3 2022/09/01, 05:17:40,  67.8758  -53.9866  13      0 13
SBD 11146  1104  310 000000000432020  6032     0   2 2022/09/01, 05:17:50,  67.8758  -53.9866  14      0 2,3,8,15,13
SBD 11146  1104  289 000000000432020  6033     0   1 2022/09/01, 05:18:03,  67.8758  -53.9866  15      0 11,7,5,14,0,1
SBD 11146  1104  309 000000000432020  6034     0   0 2022/09/01, 05:18:15,  67.8231  -53.8330   9      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1457           60        1517
# Sensor_ID bytes     % packets Description
          0    48  3.29       2         GPS
          1    66  4.53       1    Pressure
          2    82  5.63       1 Temperature
          3    66  4.53       1    Salinity
          4   153 10.50       1        Fall
          5    98  6.73       1        Rise
          6   169 11.60       1        Pump
          7    32  2.20       1 Drift_times
          8     8  0.55       1     Surface
          9    38  2.61       1  HR/Drift_P
         10    38  2.61       1  HR/Drift_T
         11    38  2.61       1  HR/Drift_S
         13   404 27.73       2  Parameters
         14   151 10.36       5 Engineering
         15    66  4.53       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1104 1 2022/08/31 23:56 +67.80414  -53.85511    0  6 29 35 45  2.3     0
G 1104 2 2022/09/01 05:11 +67.80689  -53.85837   30  5  0 29 42  1.3     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1104   84    1     3         0
#      Pres      Temp    Salinity
      1.040     4.5960    32.9580
      1.960     4.6100    32.9650
      3.000     4.6090    32.9660
      4.000     4.6080    32.9660
      5.040     4.6140    32.9660
      6.000     4.6180    32.9700
      7.000     4.6190    32.9710
      8.080     4.6200    32.9720
      9.000     4.6210    32.9720
     10.040     4.6210    32.9730
     10.960     4.6210    32.9720
     12.000     4.6220    32.9730
     12.960     4.6230    32.9730
     14.040     4.6240    32.9740
     15.000     4.6240    32.9740
     16.000     4.6240    32.9730
     17.080     4.6230    32.9740
     18.080     4.6250    32.9750
     18.960     4.6260    32.9750
     19.920     4.6230    32.9740
     21.040     4.6180    32.9690
     22.000     4.6190    32.9710
     23.000     4.6170    32.9690
     24.040     4.6210    32.9710
     25.040     4.6210    32.9720
     25.960     4.6220    32.9710
     27.000     4.6240    32.9730
     28.040     4.6250    32.9740
     29.040     4.6260    32.9740
     30.040     4.6260    32.9750
     31.000     4.6270    32.9750
     32.000     4.6260    32.9750
     33.000     4.6260    32.9750
     33.960     4.6250    32.9750
     34.960     4.6250    32.9740
     35.960     4.6270    32.9750
     36.960     4.6270    32.9750
     37.960     4.6270    32.9750
     39.000     4.6270    32.9750
     40.080     4.6270    32.9750
     40.960     4.6270    32.9750
     41.960     4.6270    32.9740
     43.000     4.6270    32.9750
     44.040     4.6270    32.9750
     44.960     4.6280    32.9750
     46.040     4.6280    32.9760
     46.960     4.6280    32.9750
     48.040     4.6290    32.9760
     49.080     4.6290    32.9760
     50.040     4.6290    32.9760
     51.040     4.6290    32.9760
     51.960     4.6280    32.9760
     52.960     4.6240    32.9720
     54.000     4.6280    32.9750
     55.000     4.6280    32.9760
     56.080     4.6290    32.9760
     57.080     4.6280    32.9750
     58.000     4.6290    32.9760
     58.960     4.6290    32.9760
     59.960     4.6290    32.9760
     60.960     4.6280    32.9750
     61.960     4.6280    32.9750
     63.000     4.6280    32.9750
     63.960     4.6290    32.9750
     65.000     4.6290    32.9750
     66.000     4.6290    32.9750
     67.040     4.6290    32.9750
     68.000     4.6280    32.9740
     69.000     4.6280    32.9740
     70.000     4.6270    32.9730
     71.000     4.6280    32.9740
     71.960     4.6280    32.9740
     72.960     4.6290    32.9740
     73.960     4.6290    32.9750
     75.000     4.6290    32.9750
     76.040     4.6290    32.9750
     77.040     4.6280    32.9740
     78.040     4.6280    32.9750
     78.960     4.6300    32.9760
     79.880     4.6320    32.9770
     81.040     4.6310    32.9770
     82.080     4.6310    32.9770
     82.840     4.6300    32.9760
     83.680     4.6280    32.9740
# Drift Data
#      C Npts comp n_var
I0  1104   15    1     4
#      Pres       Temp   Salinity     Time
     84.520     4.5990    33.0190        6
     84.840     4.5980    33.0190      911
     85.360     4.6010    33.0180     1817
     85.440     4.6030    33.0180     2722
     85.320     4.6240    33.0080     3628
     85.360     4.6260    33.0080     4533
     85.360     4.5920    33.0290     5438
     85.320     4.5650    33.0440     6344
     85.240     4.6060    33.0190     7249
     85.240     4.6190    33.0120     8155
     85.160     4.6250    33.0070     9060
     85.040     4.6250    33.0070     9965
     84.960     4.6410    32.9930    10871
     84.840     4.6420    32.9840    11776
     84.680     4.6350    32.9820    12682
# Fall Time Series
#  Cycle npts     Start                Date
F00 1104   29 715305412 2022/08/31 23:56:52
# time     Pres phase                Date
     0    0.280     1 2022/08/31 23:56:52
   143    5.480    10 2022/08/31 23:59:15
   169   10.120    10 2022/08/31 23:59:41
   195   15.080    10 2022/09/01 00:00:07
   221   20.640    10 2022/09/01 00:00:33
   244   25.480    10 2022/09/01 00:00:56
   262   29.040     2 2022/09/01 00:01:14
   267   30.200    10 2022/09/01 00:01:19
   293   35.400    10 2022/09/01 00:01:45
   319   40.640    10 2022/09/01 00:02:11
   342   45.400    10 2022/09/01 00:02:34
   368   50.360    10 2022/09/01 00:03:00
   394   55.320    10 2022/09/01 00:03:26
   427   60.400    10 2022/09/01 00:03:59
   459   65.440    10 2022/09/01 00:04:31
   492   70.400    10 2022/09/01 00:05:04
   521   75.040    10 2022/09/01 00:05:33
   554   80.440    10 2022/09/01 00:06:06
   854   83.840    10 2022/09/01 00:11:06
  1154   83.880    10 2022/09/01 00:16:06
  1456   83.920    10 2022/09/01 00:21:08
  1757   84.000    10 2022/09/01 00:26:09
  2057   84.080    10 2022/09/01 00:31:09
  2358   84.160    10 2022/09/01 00:36:10
  2658   84.320     3 2022/09/01 00:41:10
  2660   84.320    10 2022/09/01 00:41:12
  2961   84.520    10 2022/09/01 00:46:13
  3263   84.520    10 2022/09/01 00:51:15
  3303   84.520    10 2022/09/01 00:51:55
# Rise Time Series
#  Cycle npts     Start                Date
R00 1104   18 715323133 2022/09/01 04:52:13
# time     Pres phase                Date
     0   83.680     7 2022/09/01 04:52:13
   306   82.520    13 2022/09/01 04:57:19
   411   77.720    13 2022/09/01 04:59:04
   485   72.880    13 2022/09/01 05:00:18
   548   67.280    13 2022/09/01 05:01:21
   611   62.480    13 2022/09/01 05:02:24
   663   57.320    13 2022/09/01 05:03:16
   705   52.600    13 2022/09/01 05:03:58
   747   47.920    13 2022/09/01 05:04:40
   789   41.880    13 2022/09/01 05:05:22
   821   36.880    13 2022/09/01 05:05:54
   863   31.200    13 2022/09/01 05:06:36
   905   25.520    13 2022/09/01 05:07:18
   947   20.160    13 2022/09/01 05:08:00
   989   15.800    13 2022/09/01 05:08:42
  1041   10.040    13 2022/09/01 05:09:34
  1083    4.960    13 2022/09/01 05:10:16
  1126    0.640     8 2022/09/01 05:10:59
# Pump Time Series
#   Cycle npts
B02  1104   15
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    29.040  593   14.97       0    635    -9     2
    84.360    2   15.02       0    627    -9     3
    83.680  128   15.12       0    636    -9     7
    83.040   21   15.12       0    638    -9    13
    82.720   21   15.11       0    640    -9    13
    82.560   21   15.11       0    641    -9    13
    82.840   21   15.10       0    643    -9    13
    82.120   21   15.11       0    645    -9    13
    79.720   21   15.10       0    647    -9    13
    78.120   21   15.10       0    648    -9    13
    76.040   21   15.09       0    650    -9    13
    70.240   22   15.08       0    652    -9    13
    65.520   21   15.08       0    653    -9    13
    62.480   21   15.08       0    654    -9    13
    -0.040   59   15.07       0    657    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1104  2.4   26    90    60   600    54    10   240      8     1    4020    5  0.2741
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1104 2022/08/31 23:56:49
# Pcase vacuums
   6.620   6.270
# Oil valve positions
   1.690   0.497   0.664
# Seek attempts and pump time
      1      2
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1104 2022/09/01 00:51:55 9999/99/99 99:99:99
# Pcase vacuums
   6.270   6.270
# Average 
#   P      T      S
   85.120   4.613  33.011
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1104 2022/09/01 04:52:13 9999/99/99 99:99:99
# Pcase vacuums
   6.270   6.530
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1104 2022/09/01 05:10:59
# number of packets previous packets
      5               8
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.530   6.610
# Air Pump max current time  max pressure
     119       372     0.000
# Air Valve Positions
   1.279   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.518
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1104      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1104  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.280    256   5.900
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1104       197    2022/09/01 05:17:50
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1104	(cycle number) |
     MPS =        0	(phase) |
     MPC =      495	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1101	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       28	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      590	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        2	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        3	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1101	(cycle) |
  PTO[0] =    14400	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   900000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      900	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1101	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       20	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1101	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

