# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1099  313 000000000432020  6003     0   4 2022/08/31, 02:19:33,  67.7838  -53.6799  41      0 6,14,14
SBD 11146  1099  340 000000000432020  6004     0   3 2022/08/31, 02:19:42,  67.8354  -54.0425  24      0 13
SBD 11146  1099  268 000000000432020  6005     0   2 2022/08/31, 02:19:52,  67.8158  -53.9656  22      0 2,3,8,15
SBD 11146  1099  335 000000000432020  6006     0   1 2022/08/31, 02:20:04,  67.8231  -53.8330  20      0 7,5,14,0,1,13
SBD 11146  1099  332 000000000432020  6007     0   0 2022/08/31, 02:20:21,  67.8231  -53.8330  18      0 0,4,14,14,9,10,11
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1528           60        1588
# Sensor_ID bytes     % packets Description
          0    48  3.14       2         GPS
          1    69  4.52       1    Pressure
          2    91  5.96       1 Temperature
          3    91  5.96       1    Salinity
          4   138  9.03       1        Fall
          5   103  6.74       1        Rise
          6   213 13.94       1        Pump
          7    32  2.09       1 Drift_times
          8     8  0.52       1     Surface
          9    38  2.49       1  HR/Drift_P
         10    38  2.49       1  HR/Drift_T
         11    38  2.49       1  HR/Drift_S
         13   404 26.44       2  Parameters
         14   151  9.88       5 Engineering
         15    66  4.32       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1099 1 2022/08/30 20:57 +67.80473  -53.86892    0  6  0 30 46  1.0     0
G 1099 2 2022/08/31 02:14 +67.80579  -53.85807   50  6  0 33 46  1.2     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1099   86    1     3         0
#      Pres      Temp    Salinity
      1.040     4.6270    32.9770
      2.000     4.6280    32.9770
      3.000     4.6290    32.9770
      4.080     4.6290    32.9770
      5.040     4.6290    32.9770
      6.000     4.6300    32.9770
      6.960     4.6270    32.9780
      8.040     4.6270    32.9780
      9.080     4.6270    32.9780
     10.040     4.6270    32.9780
     11.040     4.6290    32.9770
     12.040     4.6280    32.9780
     12.960     4.6280    32.9780
     13.960     4.6280    32.9780
     15.040     4.6290    32.9780
     16.040     4.6290    32.9780
     16.960     4.6280    32.9780
     18.000     4.6270    32.9780
     18.960     4.6270    32.9770
     20.000     4.6270    32.9770
     21.080     4.6270    32.9780
     22.000     4.6270    32.9770
     23.040     4.6270    32.9780
     24.080     4.6270    32.9780
     25.000     4.6270    32.9780
     26.040     4.6270    32.9780
     26.960     4.6270    32.9780
     27.960     4.6270    32.9780
     29.080     4.6240    32.9790
     30.040     4.6200    32.9810
     31.000     4.6210    32.9800
     32.080     4.6230    32.9790
     33.080     4.6280    32.9770
     34.080     4.6290    32.9770
     35.040     4.6290    32.9770
     35.960     4.6260    32.9780
     37.000     4.6220    32.9800
     38.080     4.6190    32.9810
     39.040     4.6200    32.9810
     40.040     4.6200    32.9810
     41.040     4.6220    32.9800
     42.000     4.6210    32.9810
     43.000     4.6210    32.9800
     44.040     4.6210    32.9810
     45.040     4.6230    32.9790
     45.960     4.6210    32.9800
     47.000     4.5790    33.0080
     48.040     4.5650    33.0180
     49.000     4.5650    33.0180
     50.000     4.5760    33.0100
     50.960     4.5740    33.0110
     52.040     4.5660    33.0180
     53.040     4.5620    33.0200
     54.040     4.5590    33.0220
     55.040     4.5590    33.0220
     56.080     4.5550    33.0260
     57.040     4.5540    33.0270
     57.960     4.5570    33.0240
     58.960     4.5560    33.0250
     60.000     4.5630    33.0200
     60.920     4.5660    33.0170
     61.960     4.5600    33.0220
     63.000     4.5560    33.0250
     63.960     4.5630    33.0200
     65.000     4.5640    33.0190
     66.000     4.5610    33.0220
     67.000     4.5590    33.0230
     68.000     4.5600    33.0220
     69.000     4.5630    33.0200
     70.000     4.5610    33.0220
     71.040     4.5640    33.0190
     72.040     4.5680    33.0160
     73.040     4.5690    33.0160
     74.040     4.5670    33.0170
     74.960     4.5630    33.0200
     76.040     4.5530    33.0280
     77.000     4.5600    33.0220
     78.000     4.5610    33.0210
     79.000     4.5620    33.0210
     80.000     4.5490    33.0310
     81.000     4.5480    33.0320
     82.000     4.5470    33.0320
     83.040     4.5430    33.0350
     84.080     4.5420    33.0360
     85.160     4.5430    33.0360
     85.680     4.5410    33.0370
# Drift Data
#      C Npts comp n_var
I0  1099   15    1     4
#      Pres       Temp   Salinity     Time
     66.600     4.5430    33.0630        5
     67.320     4.5390    33.0660      911
     68.400     4.5780    33.0340     1816
     69.400     4.5910    33.0200     2722
     69.840     4.5940    33.0160     3627
     70.840     4.6050    32.9960     4532
     71.120     4.5940    33.0080     5438
     71.560     4.4650    33.0960     6343
     71.560     4.4620    33.0980     7249
     72.360     4.4500    33.1070     8154
     74.400     4.4480    33.1090     9059
     77.080     4.4620    33.0970     9965
     83.440     4.4750    33.0880    10870
     85.680     4.5060    33.0650    11776
     85.720     4.5140    33.0580    12681
# Fall Time Series
#  Cycle npts     Start                Date
F00 1099   26 715208275 2022/08/30 20:57:55
# time     Pres phase                Date
     0   -0.040     1 2022/08/30 20:57:55
   188    5.400    10 2022/08/30 21:01:03
   218   10.680    10 2022/08/30 21:01:33
   241   15.520    10 2022/08/30 21:01:56
   263   20.600    10 2022/08/30 21:02:18
   283   25.040    10 2022/08/30 21:02:38
   304   29.440     2 2022/08/30 21:02:59
   309   30.640    10 2022/08/30 21:03:04
   332   35.280    10 2022/08/30 21:03:27
   355   40.080    10 2022/08/30 21:03:50
   381   45.200    10 2022/08/30 21:04:16
   407   50.360    10 2022/08/30 21:04:42
   430   55.080    10 2022/08/30 21:05:05
   456   60.160    10 2022/08/30 21:05:31
   485   65.360    10 2022/08/30 21:06:00
   785   65.880    10 2022/08/30 21:11:00
  1087   65.960    10 2022/08/30 21:16:02
  1388   65.880    10 2022/08/30 21:21:03
  1688   65.840    10 2022/08/30 21:26:03
  1989   65.840    10 2022/08/30 21:31:04
  2289   65.800    10 2022/08/30 21:36:04
  2590   66.000    10 2022/08/30 21:41:05
  2697   66.240     3 2022/08/30 21:42:52
  2890   66.360    10 2022/08/30 21:46:05
  3191   66.560    10 2022/08/30 21:51:06
  3335   66.600    10 2022/08/30 21:53:30
# Rise Time Series
#  Cycle npts     Start                Date
R00 1099   19 715226028 2022/08/31 01:53:48
# time     Pres phase                Date
     0   85.720     7 2022/08/31 01:53:48
   306   84.680    13 2022/08/31 01:58:54
   400   79.200    13 2022/08/31 02:00:28
   463   74.840    13 2022/08/31 02:01:31
   547   69.720    13 2022/08/31 02:02:55
   610   64.800    13 2022/08/31 02:03:58
   663   59.600    13 2022/08/31 02:04:51
   705   54.560    13 2022/08/31 02:05:33
   747   49.240    13 2022/08/31 02:06:15
   820   44.360    13 2022/08/31 02:07:28
   883   38.960    13 2022/08/31 02:08:31
   925   33.560    13 2022/08/31 02:09:13
   967   27.840    13 2022/08/31 02:09:55
   999   22.600    13 2022/08/31 02:10:27
  1030   17.320    13 2022/08/31 02:10:58
  1062   11.880    13 2022/08/31 02:11:30
  1093    5.920    13 2022/08/31 02:12:01
  1146    0.080    13 2022/08/31 02:12:54
  1157    0.120     8 2022/08/31 02:13:05
# Pump Time Series
#   Cycle npts
B02  1099   19
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    -0.040   61   15.05       0    670    -9    14
    29.440  590   15.02       0    633    -9     2
    85.720  127   15.13       0    636    -9     7
    85.760   21   15.13       0    638    -9    13
    85.600   21   15.13       0    640    -9    13
    85.640   21   15.12       0    642    -9    13
    85.280   21   15.12       0    643    -9    13
    84.160   21   15.12       0    645    -9    13
    81.840   21   15.11       0    646    -9    13
    79.200   21   15.12       0    648    -9    13
    76.080   21   15.10       0    650    -9    13
    73.920   21   15.10       0    651    -9    13
    71.440   21   15.10       0    653    -9    13
    68.080   21   15.09       0    654    -9    13
    64.800   21   15.09       0    656    -9    13
    47.400   21   15.08       0    657    -9    13
    45.000   21   15.08       0    658    -9    13
    42.200   21   15.09       0    660    -9    13
    -0.040   60   15.08       0    663    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1099  2.4   26    90    60   600    54    10   240      8     1    4020    5  0.2741
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1099 2022/08/30 20:57:53
# Pcase vacuums
   6.630   6.270
# Oil valve positions
   1.691   0.497   0.663
# Seek attempts and pump time
      1      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1099 2022/08/30 21:53:31 9999/99/99 99:99:99
# Pcase vacuums
   6.270   6.270
# Average 
#   P      T      S
   73.680   4.522  33.061
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1099 2022/08/31 01:53:48 9999/99/99 99:99:99
# Pcase vacuums
   6.270   6.600
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1099 2022/08/31 02:13:05
# number of packets previous packets
      5              45
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.590   6.670
# Air Pump max current time  max pressure
     119       372     0.000
# Air Valve Positions
   1.279   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.519
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1099      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1099  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.120    299   5.870
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1099       197    2022/08/31 02:20:04
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1099	(cycle number) |
     MPS =        0	(phase) |
     MPC =      490	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1096	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       28	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      600	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      587	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        3	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1096	(cycle) |
  PTO[0] =    14400	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   900000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      900	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1096	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       20	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1096	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

