# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1058  236 000000000432020  5756     0   4 2022/08/21, 15:51:41,  67.6347  -54.1666   3      0 6,14,14
SBD 11146  1058  340 000000000432020  5757     0   3 2022/08/21, 15:51:51,  67.6616  -54.1114   3      0 13
SBD 11146  1058  329 000000000432020  5758     0   2 2022/08/21, 15:52:03,  67.6153  -54.0903   3      0 8,15
SBD 11146  1058  206 000000000432020  5759     0   1 2022/08/21, 15:52:14,  67.6347  -54.1666   3      0 2,3,0,13
SBD 11146  1058  340 000000000432020  5760     0   0 2022/08/21, 15:52:25,  67.6616  -54.1114   3      0 0,4,14,14,9,10,11,7,5,14,1
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1391           60        1451
# Sensor_ID bytes     % packets Description
          0    48  3.45       2         GPS
          1    39  2.80       1    Pressure
          2    47  3.38       1 Temperature
          3    47  3.38       1    Salinity
          4    53  3.81       1        Fall
          5    43  3.09       1        Rise
          6   136  9.78       1        Pump
          7    26  1.87       1 Drift_times
          8   251 18.04       1     Surface
          9    26  1.87       1  HR/Drift_P
         10    27  1.94       1  HR/Drift_T
         11    27  1.94       1  HR/Drift_S
         13   404 29.04       2  Parameters
         14   151 10.86       5 Engineering
         15    66  4.74       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1058 1 2022/08/21 10:14 +67.64412  -54.08664    0  4 30 36 44  2.7     0
G 1058 2 2022/08/21 15:46 +67.64429  -54.08537  100  5  0 30 47  3.2     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1058   24    1     3         0
#      Pres      Temp    Salinity
      1.120     4.8750    33.2880
      2.040     4.8800    33.2880
      3.000     4.8690    33.2900
      4.000     4.8610    33.2900
      4.960     4.8510    33.2950
      6.040     4.8430    33.3000
      7.040     4.8350    33.3080
      8.000     4.8170    33.3420
      8.960     4.8090    33.3570
     10.040     4.8020    33.3700
     11.040     4.7960    33.3800
     12.000     4.7920    33.3860
     12.960     4.7890    33.3900
     14.000     4.7850    33.3980
     15.000     4.7840    33.4050
     16.040     4.7790    33.4070
     17.000     4.7680    33.4080
     18.000     4.7680    33.4080
     19.040     4.7680    33.4080
     20.000     4.7650    33.4070
     21.000     4.7480    33.4110
     22.040     4.6910    33.4230
     23.080     4.6900    33.4220
     23.640     4.6900    33.4230
# Drift Data
#      C Npts comp n_var
I0  1058    4    1     4
#      Pres       Temp   Salinity     Time
     23.560     4.7870    33.4150        5
     23.320     4.7670    33.4100     3611
     23.160     4.7620    33.4130     7217
     23.280     4.7370    33.4120    10822
# Surface Pressure Time Series
#    C npts n_var               Start
S 1058    27     6 2022/08/21 15:44:39
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    91      0.190      4.890   -999.000      14 2022/08/21 15:46:10
   104      0.120      4.891   -999.000      14 2022/08/21 15:46:23
   113      0.230      4.890   -999.000      14 2022/08/21 15:46:32
   124      0.200      4.897   -999.000      14 2022/08/21 15:46:43
   135      0.100      4.885   -999.000      14 2022/08/21 15:46:54
   146      0.110      4.883   -999.000      14 2022/08/21 15:47:05
   157      0.190      4.896   -999.000      14 2022/08/21 15:47:16
   168      0.130      4.905   -999.000      14 2022/08/21 15:47:27
   179      0.150      4.891   -999.000      14 2022/08/21 15:47:38
   190      0.030      4.890   -999.000      14 2022/08/21 15:47:49
   202      0.150      4.886   -999.000      14 2022/08/21 15:48:01
   213      0.160      4.892   -999.000      14 2022/08/21 15:48:12
   224      0.170      4.895   -999.000      14 2022/08/21 15:48:23
   235      0.160      4.892   -999.000      14 2022/08/21 15:48:34
   246      0.130      4.883   -999.000      14 2022/08/21 15:48:45
   257      0.110      4.883   -999.000      14 2022/08/21 15:48:56
   268      0.160      4.893   -999.000      14 2022/08/21 15:49:07
   279      0.170      4.891   -999.000      14 2022/08/21 15:49:18
   290      0.160      4.893   -999.000      14 2022/08/21 15:49:29
   301      0.150      4.900   -999.000      14 2022/08/21 15:49:40
   312      0.140      4.895   -999.000      14 2022/08/21 15:49:51
   323      0.160      4.896   -999.000      14 2022/08/21 15:50:02
   335      0.160      4.901   -999.000      14 2022/08/21 15:50:14
   346      0.090      4.905   -999.000      14 2022/08/21 15:50:25
   357      0.180      4.900   -999.000      14 2022/08/21 15:50:36
   368      0.150      4.898   -999.000      14 2022/08/21 15:50:47
   379      0.150      4.896   -999.000      14 2022/08/21 15:50:58
# Fall Time Series
#  Cycle npts     Start                Date
F00 1058    9 714392127 2022/08/21 10:15:27
# time     Pres phase                Date
     0    0.120     1 2022/08/21 10:15:27
   137    5.240    10 2022/08/21 10:17:44
   169   10.480    10 2022/08/21 10:18:16
   198   15.520    10 2022/08/21 10:18:45
   228   20.520    10 2022/08/21 10:19:15
   369   23.520     2 2022/08/21 10:21:36
   528   23.520    10 2022/08/21 10:24:15
   830   23.440    10 2022/08/21 10:29:17
  1035   23.520    10 2022/08/21 10:32:42
# Rise Time Series
#  Cycle npts     Start                Date
R00 1058    7 714411172 2022/08/21 15:32:52
# time     Pres phase                Date
     0   23.640     7 2022/08/21 15:32:52
   306   23.600    13 2022/08/21 15:37:58
   463   18.480    13 2022/08/21 15:40:35
   537   13.640    13 2022/08/21 15:41:49
   600    8.920    13 2022/08/21 15:42:52
   673    3.280    13 2022/08/21 15:44:05
   705    0.720     8 2022/08/21 15:44:37
# Pump Time Series
#   Cycle npts
B02  1058   12
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    23.520  362   15.08       0    617    -9     2
    23.640  127   15.19       0    620    -9     7
    23.720   21   15.20       0    622    -9    13
    23.680   21   15.19       0    624    -9    13
    23.600   21   15.19       0    626    -9    13
    23.040   21   15.19       0    627    -9    13
    20.760   21   15.17       0    629    -9    13
    17.280   21   15.18       0    630    -9    13
    12.840   21   15.17       0    632    -9    13
     7.280   21   15.17       0    633    -9    13
     2.480   21   15.16       0    635    -9    13
     0.080   89   15.16       0    640    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1058  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1058 2022/08/21 10:15:24
# Pcase vacuums
   6.440   6.140
# Oil valve positions
   1.689   0.500   0.664
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1058 2022/08/21 10:32:43 9999/99/99 99:99:99
# Pcase vacuums
   6.140   6.110
# Average 
#   P      T      S
   23.320   4.763  33.412
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1058 2022/08/21 15:32:52 9999/99/99 99:99:99
# Pcase vacuums
   6.100   6.350
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1058 2022/08/21 15:44:36
# number of packets previous packets
      5               5
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.350   6.430
# Air Pump max current time  max pressure
     119       390     0.000
# Air Valve Positions
   1.279   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.523
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1058      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1058  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.080    364   5.890
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1058       197    2022/08/21 15:52:14
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1058	(cycle number) |
     MPS =        0	(phase) |
     MPC =      449	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1055	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      360	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1055	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1055	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1055	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

