# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1052  214 000000000432020  5730     0   3 2022/08/20, 05:56:04,  67.6752  -54.1114   4      0 6,14,14
SBD 11146  1052  340 000000000432020  5731     0   2 2022/08/20, 05:56:14,  67.6752  -54.1114   4      0 13
SBD 11146  1052  312 000000000432020  5732     0   1 2022/08/20, 05:56:25,  67.6752  -54.1114   4      0 1,2,3,0,8,15,13
SBD 11146  1052  306 000000000432020  5733     0   0 2022/08/20, 05:56:34,  67.6752  -54.1114   4      0 0,4,14,14,9,10,11,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1124           48        1172
# Sensor_ID bytes     % packets Description
          0    48  4.27       2         GPS
          1    42  3.74       1    Pressure
          2    38  3.38       1 Temperature
          3    46  4.09       1    Salinity
          4    53  4.72       1        Fall
          5    48  4.27       1        Rise
          6   114 10.14       1        Pump
          7    26  2.31       1 Drift_times
          8     8  0.71       1     Surface
          9    27  2.40       1  HR/Drift_P
         10    26  2.31       1  HR/Drift_T
         11    27  2.40       1  HR/Drift_S
         13   404 35.94       2  Parameters
         14   151 13.43       5 Engineering
         15    66  5.87       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1052 1 2022/08/20 00:18 +67.68339  -54.13081    0  5 35 38 42  5.1     0
G 1052 2 2022/08/20 05:50 +67.65904  -54.11852   80  5 34 40 43  2.7     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1052   27    1     3         0
#      Pres      Temp    Salinity
      1.120     4.8280    33.3190
      2.040     4.8270    33.3190
      3.000     4.8270    33.3190
      4.000     4.8280    33.3110
      4.920     4.8290    33.3190
      6.040     4.8300    33.3200
      7.000     4.8300    33.3200
      8.000     4.8300    33.3200
      9.000     4.8310    33.3190
     10.040     4.8310    33.3190
     10.920     4.8310    33.3190
     12.000     4.8310    33.3200
     13.000     4.8310    33.3200
     14.000     4.8310    33.3200
     15.000     4.8310    33.3210
     16.000     4.8310    33.3210
     16.960     4.8310    33.3210
     18.040     4.8310    33.3210
     18.960     4.8310    33.3200
     19.920     4.8310    33.3210
     21.000     4.8310    33.3230
     21.960     4.8310    33.3230
     23.040     4.8310    33.3230
     24.040     4.8310    33.3220
     24.960     4.8310    33.3220
     26.360     4.8310    33.3200
     26.560     4.8310    33.3200
# Drift Data
#      C Npts comp n_var
I0  1052    4    1     4
#      Pres       Temp   Salinity     Time
     24.480     4.7510    33.1800        6
     27.240     4.7750    33.2110     3611
     26.360     4.7990    33.2430     7217
     25.400     4.8240    33.2880    10822
# Fall Time Series
#  Cycle npts     Start                Date
F00 1052    9 714269989 2022/08/20 00:19:49
# time     Pres phase                Date
     0    0.080     1 2022/08/20 00:19:49
   137    5.040    10 2022/08/20 00:22:06
   166   10.120    10 2022/08/20 00:22:35
   196   15.360    10 2022/08/20 00:23:05
   232   20.480    10 2022/08/20 00:23:41
   366   23.640     2 2022/08/20 00:25:55
   532   23.760    10 2022/08/20 00:28:41
   834   24.480    10 2022/08/20 00:33:43
  1032   24.480    10 2022/08/20 00:37:01
# Rise Time Series
#  Cycle npts     Start                Date
R00 1052    8 714289031 2022/08/20 05:37:11
# time     Pres phase                Date
     0   26.520     7 2022/08/20 05:37:11
   306   25.720    13 2022/08/20 05:42:17
   442   20.320    13 2022/08/20 05:44:33
   505   15.960    13 2022/08/20 05:45:36
   568   10.240    13 2022/08/20 05:46:39
   621    5.280    13 2022/08/20 05:47:32
   695    0.080    13 2022/08/20 05:48:46
   705    0.120     8 2022/08/20 05:48:56
# Pump Time Series
#   Cycle npts
B02  1052   10
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    23.640  362   15.10       0    617    -9     2
    26.520  128   15.20       0    619    -9     7
    26.560   22   15.20       0    622    -9    13
    26.360   22   15.19       0    623    -9    13
    25.920   22   15.19       0    625    -9    13
    24.280   22   15.18       0    627    -9    13
    21.520   22   15.19       0    628    -9    13
    17.560   22   15.19       0    630    -9    13
     2.560   22   15.18       0    631    -9    13
     0.200   60   15.17       0    635    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1052  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1052 2022/08/20 00:19:47
# Pcase vacuums
   6.420   6.140
# Oil valve positions
   1.688   0.500   0.665
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1052 2022/08/20 00:37:02 9999/99/99 99:99:99
# Pcase vacuums
   6.140   6.100
# Average 
#   P      T      S
   25.880   4.787  33.230
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1052 2022/08/20 05:37:11 9999/99/99 99:99:99
# Pcase vacuums
   6.100   6.310
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1052 2022/08/20 05:48:57
# number of packets previous packets
      4               5
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.320   6.380
# Air Pump max current time  max pressure
     119       395     0.000
# Air Valve Positions
   1.278   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.523
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1052      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1052  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.080    361   5.890
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1052       197    2022/08/20 05:56:25
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1052	(cycle number) |
     MPS =        0	(phase) |
     MPC =      443	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1049	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      360	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1049	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1049	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1049	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

