# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1046  295 000000000432020  5705     0   3 2022/08/18, 20:02:10,  67.7141  -54.2645   5      0 6,14,14
SBD 11146  1046  340 000000000432020  5706     0   2 2022/08/18, 20:02:29,  67.7141  -54.2645   6      0 13
SBD 11146  1046  320 000000000432020  5707     0   1 2022/08/18, 20:02:50,  67.7740  -54.2861  11      0 1,2,3,0,8,15,13
SBD 11146  1046  312 000000000432020  5708     0   0 2022/08/18, 20:03:03,  67.7740  -54.2861  10      0 0,4,14,14,9,10,11,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1219           48        1267
# Sensor_ID bytes     % packets Description
          0    48  3.94       2         GPS
          1    42  3.45       1    Pressure
          2    50  4.10       1 Temperature
          3    42  3.45       1    Salinity
          4    53  4.35       1        Fall
          5    53  4.35       1        Rise
          6   147 12.06       1        Pump
          7    26  2.13       1 Drift_times
          8     8  0.66       1     Surface
          9    27  2.21       1  HR/Drift_P
         10    27  2.21       1  HR/Drift_T
         11    27  2.21       1  HR/Drift_S
         13   404 33.14       2  Parameters
         14   199 16.32       5 Engineering
         15    66  5.41       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1046 1 2022/08/18 14:18 +67.68472  -54.29785    0  8 34 39 42  1.1     0
G 1046 2 2022/08/18 19:56 +67.71639  -54.27038   80  7  0 34 46  1.5     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1046   35    1     3         0
#      Pres      Temp    Salinity
      1.040     4.5680    33.0970
      2.000     4.5710    33.0970
      3.040     4.5660    33.0990
      4.040     4.5630    33.0980
      5.040     4.5640    33.0980
      6.040     4.5640    33.0980
      7.000     4.5620    33.0990
      7.960     4.5620    33.0990
      9.000     4.5630    33.1000
     10.000     4.5640    33.1000
     11.080     4.5630    33.1000
     11.960     4.5620    33.0990
     12.920     4.5610    33.0980
     14.040     4.5610    33.0980
     15.080     4.5610    33.0980
     16.000     4.5620    33.0990
     17.040     4.5630    33.1010
     18.080     4.5640    33.1010
     19.040     4.5630    33.1010
     19.920     4.5620    33.1000
     21.000     4.5630    33.1000
     22.040     4.5640    33.1010
     22.960     4.5670    33.1030
     23.920     4.5700    33.1060
     25.000     4.5700    33.1060
     26.000     4.5700    33.1060
     27.040     4.5710    33.1060
     28.080     4.5710    33.1070
     28.960     4.5710    33.1070
     29.960     4.5700    33.1060
     30.960     4.5690    33.1050
     31.960     4.5690    33.1050
     32.960     4.5710    33.1070
     34.120     4.5700    33.1060
     34.800     4.5720    33.1080
# Drift Data
#      C Npts comp n_var
I0  1046    4    1     4
#      Pres       Temp   Salinity     Time
     24.360     4.7210    33.3090        6
     23.160     4.7180    33.2980     3612
     24.280     4.7000    33.2690     7217
     24.480     4.6450    33.2090    10823
# Fall Time Series
#  Cycle npts     Start                Date
F00 1046    9 714147585 2022/08/18 14:19:45
# time     Pres phase                Date
     0    0.120     1 2022/08/18 14:19:45
   147    5.080    10 2022/08/18 14:22:12
   180   10.400    10 2022/08/18 14:22:45
   209   15.280    10 2022/08/18 14:23:14
   238   20.080    10 2022/08/18 14:23:43
   367   24.040     2 2022/08/18 14:25:52
   538   24.200    10 2022/08/18 14:28:43
   840   24.240    10 2022/08/18 14:33:45
  1036   24.280    10 2022/08/18 14:37:01
# Rise Time Series
#  Cycle npts     Start                Date
R00 1046    9 714166630 2022/08/18 19:37:10
# time     Pres phase                Date
     0   34.600     7 2022/08/18 19:37:10
   306   34.440    13 2022/08/18 19:42:16
   401   29.680    13 2022/08/18 19:43:51
   485   24.280    13 2022/08/18 19:45:15
   569   19.400    13 2022/08/18 19:46:39
   642   14.480    13 2022/08/18 19:47:52
   695    9.880    13 2022/08/18 19:48:45
   747    3.960    13 2022/08/18 19:49:37
  1041    0.160     8 2022/08/18 19:54:31
# Pump Time Series
#   Cycle npts
B02  1046   13
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    24.040  361   15.10       0    617    -9     2
    34.600  128   15.21       0    620    -9     7
    32.720   21   15.21       0    622    -9    13
    33.880   21   15.20       0    624    -9    13
    34.400   21   15.19       0    626    -9    13
    33.200   21   15.18       0    628    -9    13
    28.680   21   15.18       0    629    -9    13
    24.280   21   15.18       0    631    -9    13
    19.400   21   15.18       0    632    -9    13
    15.200   21   15.17       0    634    -9    13
  9999.000   60   15.17       0    637    -9    13
  9999.000   60   15.16       0    642    -9    13
     0.040   61   15.15       0    647    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1046  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1046 2022/08/18 14:19:43
# Pcase vacuums
   6.440   6.140
# Oil valve positions
   1.689   0.501   0.665
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1046 2022/08/18 14:37:01 9999/99/99 99:99:99
# Pcase vacuums
   6.140   6.110
# Average 
#   P      T      S
   24.080   4.696  33.271
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1046 2022/08/18 19:37:10 9999/99/99 99:99:99
# Pcase vacuums
   6.110   6.420
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1046 2022/08/18 19:54:31
# number of packets previous packets
      4               4
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.430   6.490
# Air Pump max current time  max pressure
     119       394     0.000
# Air Valve Positions
   1.279   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.525
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1046      8
#               Date   Code Description
 2022/08/18 19:50:29 0x0114 AscendCtd
 2022/08/18 19:50:59 0x0114 AscendCtd
 2022/08/18 19:51:29 0x0114 AscendCtd
 2022/08/18 19:51:59 0x0114 AscendCtd
 2022/08/18 19:52:29 0x0114 AscendCtd
 2022/08/18 19:52:59 0x0114 AscendCtd
 2022/08/18 19:53:29 0x0114 AscendCtd
 2022/08/18 19:53:59 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1046  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.200    361   5.890
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1046       197    2022/08/18 20:02:50
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1046	(cycle number) |
     MPS =        0	(phase) |
     MPC =      437	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1043	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      360	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1043	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1043	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1043	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

