# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1040  244 000000000432020  5681     0   3 2022/08/17, 10:11:31,  67.7335  -54.3413   4      0 6,14,14
SBD 11146  1040  340 000000000432020  5682     0   2 2022/08/17, 10:11:43,  67.7335  -54.3413   3      0 13
SBD 11146  1040  309 000000000432020  5683     0   1 2022/08/17, 10:12:04,  67.7065  -54.3964   4      0 1,2,3,0,8,15,13
SBD 11146  1040  302 000000000432020  5684     0   0 2022/08/17, 10:12:16,  67.7065  -54.3964  17      0 0,4,14,14,9,10,11,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1147           48        1195
# Sensor_ID bytes     % packets Description
          0    48  4.18       2         GPS
          1    40  3.49       1    Pressure
          2    37  3.23       1 Temperature
          3    37  3.23       1    Salinity
          4    53  4.62       1        Fall
          5    43  3.75       1        Rise
          6   114  9.94       1        Pump
          7    26  2.27       1 Drift_times
          8    17  1.48       1     Surface
          9    27  2.35       1  HR/Drift_P
         10    27  2.35       1  HR/Drift_T
         11    27  2.35       1  HR/Drift_S
         13   404 35.22       2  Parameters
         14   181 15.78       5 Engineering
         15    66  5.75       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1040 1 2022/08/17 04:34 +67.72813  -54.35455    0  7 27 35 44  1.0     0
G 1040 2 2022/08/17 10:07 +67.71392  -54.34604  170  7  0 32 43  0.9     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1040   25    1     3         0
#      Pres      Temp    Salinity
      1.160     4.5210    33.1760
      1.960     4.5210    33.1750
      3.000     4.5210    33.1750
      4.040     4.5210    33.1750
      5.080     4.5210    33.1760
      6.000     4.5210    33.1760
      6.960     4.5220    33.1760
      8.040     4.5220    33.1760
      9.000     4.5220    33.1760
     10.080     4.5220    33.1760
     11.040     4.5220    33.1770
     12.000     4.5230    33.1770
     13.000     4.5230    33.1770
     14.000     4.5230    33.1770
     14.920     4.5230    33.1770
     15.960     4.5230    33.1770
     17.040     4.5220    33.1760
     18.040     4.5190    33.1730
     18.960     4.5190    33.1730
     19.960     4.5180    33.1730
     21.080     4.5180    33.1730
     22.160     4.5180    33.1730
     22.880     4.5180    33.1720
     24.120     4.5170    33.1720
     24.640     4.5150    33.1700
# Drift Data
#      C Npts comp n_var
I0  1040    4    1     4
#      Pres       Temp   Salinity     Time
     25.120     4.5690    33.1920        5
     28.760     4.5370    33.1670     3611
     27.920     4.5030    33.1440     7217
     27.240     4.4920    33.1420    10822
# Surface Pressure Time Series
#    C npts n_var               Start
S 1040     1     6 2022/08/17 10:04:31
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    94      0.050      4.470   -999.000      14 2022/08/17 10:06:05
# Fall Time Series
#  Cycle npts     Start                Date
F00 1040    9 714026107 2022/08/17 04:35:07
# time     Pres phase                Date
     0    0.080     1 2022/08/17 04:35:07
   147    5.080    10 2022/08/17 04:37:34
   179   10.440    10 2022/08/17 04:38:06
   209   15.320    10 2022/08/17 04:38:36
   238   20.400    10 2022/08/17 04:39:05
   366   24.680     2 2022/08/17 04:41:13
   538   24.880    10 2022/08/17 04:44:05
   840   25.080    10 2022/08/17 04:49:07
  1032   25.080    10 2022/08/17 04:52:19
# Rise Time Series
#  Cycle npts     Start                Date
R00 1040    7 714045149 2022/08/17 09:52:29
# time     Pres phase                Date
     0   24.720     7 2022/08/17 09:52:29
   306   20.720    13 2022/08/17 09:57:35
   369   15.040    13 2022/08/17 09:58:38
   421   10.600    13 2022/08/17 09:59:30
   474    5.840    13 2022/08/17 10:00:23
   709    0.080    13 2022/08/17 10:04:18
   719    0.200     8 2022/08/17 10:04:28
# Pump Time Series
#   Cycle npts
B02  1040   10
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    24.680  362   15.10       0    618    -9     2
    24.720  127   15.21       0    620    -9     7
    22.760   21   15.20       0    622    -9    13
    22.200   21   15.20       0    624    -9    13
    21.400   21   15.20       0    626    -9    13
    14.280   21   15.19       0    628    -9    13
     5.840   21   15.19       0    629    -9    13
     1.200   60   15.18       0    633    -9    13
  9999.000   52   15.17       0    638    -9    13
     0.040   88   15.16       0    644    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1040  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1040 2022/08/17 04:35:05
# Pcase vacuums
   6.420   6.150
# Oil valve positions
   1.688   0.501   0.667
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1040 2022/08/17 04:52:20 9999/99/99 99:99:99
# Pcase vacuums
   6.150   6.110
# Average 
#   P      T      S
   27.280   4.525  33.161
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1040 2022/08/17 09:52:29 9999/99/99 99:99:99
# Pcase vacuums
   6.110   6.370
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1040 2022/08/17 10:04:29
# number of packets previous packets
      4               9
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.370   6.460
# Air Pump max current time  max pressure
     119       392     0.000
# Air Valve Positions
   1.279   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.526
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1040      5
#               Date   Code Description
 2022/08/17 10:01:56 0x0114 AscendCtd
 2022/08/17 10:02:26 0x0114 AscendCtd
 2022/08/17 10:02:56 0x0114 AscendCtd
 2022/08/17 10:03:26 0x0114 AscendCtd
 2022/08/17 10:03:56 0x0114 AscendCtd
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1040  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.080    361   5.900
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1040       197    2022/08/17 10:12:04
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1040	(cycle number) |
     MPS =        0	(phase) |
     MPC =      431	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1037	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      360	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1037	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1037	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1037	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

