# Message Decoder for Solo2, Alto and Deep Solo Floats; v220708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11146  1038  214 000000000432020  5673     0   3 2022/08/16, 22:55:23,  67.7258  -54.4733   5      0 6,14,14
SBD 11146  1038  340 000000000432020  5674     0   2 2022/08/16, 22:55:37,  67.7335  -54.3413   5      0 13
SBD 11146  1038  297 000000000432020  5675     0   1 2022/08/16, 22:55:47,  67.7258  -54.4733   5      0 1,2,3,0,8,15,13
SBD 11146  1038  307 000000000432020  5676     0   0 2022/08/16, 22:56:01,  67.7258  -54.4733   5      0 0,4,14,14,9,10,11,7,5,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1110           48        1158
# Sensor_ID bytes     % packets Description
          0    48  4.32       2         GPS
          1    37  3.33       1    Pressure
          2    37  3.33       1 Temperature
          3    37  3.33       1    Salinity
          4    58  5.23       1        Fall
          5    43  3.87       1        Rise
          6   114 10.27       1        Pump
          7    26  2.34       1 Drift_times
          8     8  0.72       1     Surface
          9    27  2.43       1  HR/Drift_P
         10    27  2.43       1  HR/Drift_T
         11    27  2.43       1  HR/Drift_S
         13   404 36.40       2  Parameters
         14   151 13.60       5 Engineering
         15    66  5.95       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G 1038 1 2022/08/16 17:19 +67.74145  -54.44418    0  7 20 23 27  0.9     0
G 1038 2 2022/08/16 22:49 +67.72943  -54.37530   80  7  0 30 43  1.1     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0  1038   25    1     3         0
#      Pres      Temp    Salinity
      0.960     4.5990    33.2080
      1.920     4.5990    33.2080
      3.000     4.5990    33.2080
      4.000     4.5990    33.2080
      5.000     4.5990    33.2080
      6.000     4.5990    33.2070
      6.960     4.5990    33.2070
      8.000     4.6000    33.2070
      9.000     4.6000    33.2070
     10.040     4.6000    33.2070
     11.000     4.6000    33.2070
     12.000     4.5990    33.2070
     13.000     4.6000    33.2070
     14.080     4.6000    33.2070
     15.040     4.6000    33.2070
     16.000     4.5990    33.2070
     17.000     4.5990    33.2070
     18.000     4.5990    33.2070
     19.000     4.5990    33.2070
     20.040     4.5990    33.2070
     21.000     4.5990    33.2070
     22.040     4.5990    33.2070
     22.920     4.5990    33.2070
     24.080     4.5990    33.2060
     25.200     4.5980    33.2060
# Drift Data
#      C Npts comp n_var
I0  1038    4    1     4
#      Pres       Temp   Salinity     Time
     24.480     4.6190    33.2350        5
     26.720     4.5730    33.1990     3611
     24.680     4.5200    33.1570     7216
     24.560     4.5260    33.1540    10822
# Fall Time Series
#  Cycle npts     Start                Date
F00 1038   10 713985633 2022/08/16 17:20:33
# time     Pres phase                Date
     0    0.000     1 2022/08/16 17:20:33
   156    5.240    10 2022/08/16 17:23:09
   182   10.280    10 2022/08/16 17:23:35
   211   15.160    10 2022/08/16 17:24:04
   247   20.120    10 2022/08/16 17:24:40
   293   25.160    10 2022/08/16 17:25:26
   369   25.240     2 2022/08/16 17:26:42
   593   24.680    10 2022/08/16 17:30:26
   895   24.960    10 2022/08/16 17:35:28
  1035   24.560    10 2022/08/16 17:37:48
# Rise Time Series
#  Cycle npts     Start                Date
R00 1038    7 714004678 2022/08/16 22:37:58
# time     Pres phase                Date
     0   25.320     7 2022/08/16 22:37:58
   306   22.520    13 2022/08/16 22:43:04
   390   17.960    13 2022/08/16 22:44:28
   442   12.920    13 2022/08/16 22:45:20
   516    7.560    13 2022/08/16 22:46:34
   579    2.440    13 2022/08/16 22:47:37
   621    0.080     8 2022/08/16 22:48:19
# Pump Time Series
#   Cycle npts
B02  1038   10
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    25.240  362   15.10       0    618    -9     2
    25.320  127   15.20       0    620    -9     7
    24.440   21   15.21       0    623    -9    13
    23.120   21   15.20       0    624    -9    13
    22.640   21   15.19       0    626    -9    13
    20.520   21   15.19       0    628    -9    13
    12.080   21   15.19       0    629    -9    13
     7.560   21   15.18       0    631    -9    13
     2.440   31   15.18       0    633    -9    13
    -0.080   89   15.16       0    640    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2  1038  2.4   21    90    60   600    26    10   300      8     1    4020    5  0.2964
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17 1038 2022/08/16 17:20:30
# Pcase vacuums
   6.440   6.150
# Oil valve positions
   1.689   0.502   0.667
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18 1038 2022/08/16 17:37:49 9999/99/99 99:99:99
# Pcase vacuums
   6.150   6.110
# Average 
#   P      T      S
   25.120   4.559  33.186
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18 1038 2022/08/16 22:37:58 9999/99/99 99:99:99
# Pcase vacuums
   6.110   6.330
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20 1038 2022/08/16 22:48:20
# number of packets previous packets
      4               4
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.330   6.420
# Air Pump max current time  max pressure
     119       394     0.000
# Air Valve Positions
   1.279   0.000
# Air Valve Calibrations
   1.246   0.148   0.697
# Oil Valve Positions
   0.000   0.526
# Oil Valve Calibrations
   1.620   0.570
# Encountered Error Codes
#    Cycle Errors
EE20  1038      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00 1038  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.120    364   5.900
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M 1038       197    2022/08/16 22:55:47
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =     1038	(cycle number) |
     MPS =        0	(phase) |
     MPC =      429	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       60	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        0	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =     1035	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       27	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =      360	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =      360	(sec) |
  DFT[1] =      220	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      300	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        1	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        5	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      300	(sec) |
     DUC =      100	(under damping factor) |
     DPT =        5	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      300	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =     1035	(cycle) |
  PTO[0] =    18000	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  3600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =     1035	(cycle number) |
     ARR =        8	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        5	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        3	(dBar) |
     SFN =     1035	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      300	(sec) |
     SSP =       60	(sec) |

