# Message Decoder for Solo2, Alto and Deep Solo Floats; v211007
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11145   332  340 000000000337990  2054     0   4 2021/11/05, 16:28:20,  65.4731  -54.9443  13      0 13
SBD 11145   332  330 000000000337990  2055     0   3 2021/11/05, 16:28:49,  65.4543  -54.8736  16      0 15,13,6,14,14
SBD 11145   332  283 000000000337990  2056     0   2 2021/11/05, 16:29:39,  65.4196  -55.0433  15      0 3,0,8
SBD 11145   332  316 000000000337990  2057     0   1 2021/11/05, 16:30:22,  65.4196  -55.0433  11      0 7,5,14,1,2
SBD 11145   332  321 000000000337990  2058     0   0 2021/11/05, 16:30:33,  65.4196  -55.0433  10      0 0,4,14,14,9,10,11
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1530           60        1590
# Sensor_ID bytes     % packets Description
          0    48  3.14       2         GPS
          1    66  4.31       1    Pressure
          2    98  6.41       1 Temperature
          3    86  5.62       1    Salinity
          4   113  7.39       1        Fall
          5    88  5.75       1        Rise
          6    70  4.58       1        Pump
          7    33  2.16       1 Drift_times
          8   161 10.52       1     Surface
          9    48  3.14       1  HR/Drift_P
         10    40  2.61       1  HR/Drift_T
         11    40  2.61       1  HR/Drift_S
         13   404 26.41       2  Parameters
         14   169 11.05       5 Engineering
         15    66  4.31       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  332 1 2021/11/05 12:34 +65.40233  -54.96907    0  6 27 35 44  2.1     0
G  332 2 2021/11/05 16:26 +65.40128  -55.01724  170  5 31 37 47  1.3     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   332   75    1     3         0
#      Pres      Temp    Salinity
      1.040     2.3340    32.9460
      2.000     2.3310    32.9460
      3.000     2.3300    32.9460
      4.000     2.3260    32.9450
      5.000     2.3300    32.9460
      6.040     2.3290    32.9460
      7.040     2.3290    32.9460
      7.960     2.3320    32.9460
      9.000     2.3370    32.9470
     10.040     2.3440    32.9490
     11.000     2.3460    32.9490
     11.960     2.3460    32.9500
     13.000     2.3470    32.9500
     14.000     2.3540    32.9520
     15.040     2.3690    32.9570
     16.040     2.3770    32.9590
     17.000     2.3780    32.9610
     18.000     2.3810    32.9640
     19.000     2.3860    32.9660
     19.960     2.3930    32.9710
     21.000     2.4080    32.9800
     22.000     2.4340    32.9920
     22.960     2.5180    33.0200
     24.000     2.6810    33.1010
     25.000     2.7420    33.1450
     26.040     2.8350    33.1910
     27.040     2.8720    33.2140
     28.000     2.8970    33.2320
     29.040     2.9260    33.2540
     30.000     2.9730    33.2880
     31.040     3.1170    33.3770
     32.040     3.2320    33.4720
     33.000     3.2470    33.4850
     34.000     3.3330    33.5280
     35.000     3.4740    33.6150
     36.040     3.5430    33.6490
     37.000     3.6240    33.7000
     38.040     3.6310    33.7050
     39.080     3.6330    33.7060
     40.080     3.6370    33.7090
     41.040     3.6620    33.7200
     42.040     3.6760    33.7310
     43.040     3.6940    33.7400
     44.000     3.7150    33.7500
     45.000     3.7400    33.7620
     46.000     3.7610    33.7750
     47.040     3.7660    33.7790
     48.000     3.7790    33.7880
     49.000     3.7900    33.7930
     50.000     3.8020    33.8000
     51.040     3.8100    33.8060
     52.040     3.8130    33.8080
     52.960     3.8150    33.8080
     54.000     3.8350    33.8230
     55.040     3.8440    33.8280
     56.000     3.8550    33.8340
     56.960     3.8620    33.8390
     57.960     3.8720    33.8450
     59.000     3.8780    33.8490
     59.960     3.8860    33.8540
     61.000     3.9030    33.8620
     62.040     3.9080    33.8650
     63.000     3.9070    33.8650
     63.960     3.9150    33.8680
     64.960     3.9230    33.8740
     65.960     3.9290    33.8780
     67.000     3.9360    33.8840
     68.000     3.9500    33.8930
     69.000     3.9660    33.9010
     70.000     3.9740    33.9060
     71.000     3.9890    33.9130
     72.000     3.9890    33.9140
     73.000     3.9930    33.9150
     74.120     4.0050    33.9210
     74.600     4.0070    33.9220
# Drift Data
#      C Npts comp n_var
I0   332   17    1     4
#      Pres       Temp   Salinity     Time
     91.040     3.9750    33.9030        6
     93.440     3.9690    33.9030      611
     82.200     3.9610    33.8990     1216
     78.880     3.9920    33.9160     1822
     83.280     4.0010    33.9160     2427
     87.200     3.9760    33.9060     3033
     90.920     3.9730    33.9050     3638
     85.640     3.9600    33.8980     4244
     77.760     3.9860    33.9110     4849
     76.280     4.0040    33.9200     5454
     68.680     4.0440    33.9470     6060
     80.440     4.0610    33.9460     6665
     71.880     4.0040    33.9210     7271
     74.880     4.0570    33.9450     7876
     85.720     3.9960    33.9140     8481
     80.920     3.9880    33.9120     9087
     76.840     3.9920    33.9150     9692
# Surface Pressure Time Series
#    C npts n_var               Start
S  332    17     6 2021/11/05 16:23:27
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    77      0.140      2.286   -999.000      14 2021/11/05 16:24:44
    90      0.240      2.293   -999.000      14 2021/11/05 16:24:57
   101      0.120      2.275   -999.000      14 2021/11/05 16:25:08
   112      0.090      2.281   -999.000      14 2021/11/05 16:25:19
   123      0.110      2.292   -999.000      14 2021/11/05 16:25:30
   135      0.100      2.272   -999.000      14 2021/11/05 16:25:42
   146      0.140      2.269   -999.000      14 2021/11/05 16:25:53
   157      0.120      2.283   -999.000      14 2021/11/05 16:26:04
   168      0.120      2.274   -999.000      14 2021/11/05 16:26:15
   179      0.120      2.298   -999.000      14 2021/11/05 16:26:26
   190      0.120      2.270   -999.000      14 2021/11/05 16:26:37
   201      0.110      2.297   -999.000      14 2021/11/05 16:26:48
   212      0.110      2.299   -999.000      14 2021/11/05 16:26:59
   223      0.210      2.292   -999.000      14 2021/11/05 16:27:10
   234      0.200      2.250   -999.000      14 2021/11/05 16:27:21
   245      0.130      2.268   -999.000      14 2021/11/05 16:27:32
   256      0.080      2.259   -999.000      14 2021/11/05 16:27:43
# Fall Time Series
#  Cycle npts     Start                Date
F00  332   21 689430916 2021/11/05 12:35:16
# time     Pres phase                Date
     0    0.080     1 2021/11/05 12:35:16
   269    5.160    10 2021/11/05 12:39:45
   305   10.360    10 2021/11/05 12:40:21
   338   15.440    10 2021/11/05 12:40:54
   367   20.360    10 2021/11/05 12:41:23
   396   25.560    10 2021/11/05 12:41:52
   422   30.160    10 2021/11/05 12:42:18
   452   35.280    10 2021/11/05 12:42:48
   481   40.560    10 2021/11/05 12:43:17
   507   45.360    10 2021/11/05 12:43:43
   533   50.200    10 2021/11/05 12:44:09
   537   51.040     2 2021/11/05 12:44:13
   563   55.520    10 2021/11/05 12:44:39
   589   60.000    10 2021/11/05 12:45:05
   621   65.000    10 2021/11/05 12:45:37
   660   70.080    10 2021/11/05 12:46:16
   706   75.120    10 2021/11/05 12:47:02
   765   80.040    10 2021/11/05 12:48:01
  1167   85.040    10 2021/11/05 12:54:43
  1495   90.040    10 2021/11/05 13:00:11
  1973   91.080    10 2021/11/05 13:08:09
# Rise Time Series
#  Cycle npts     Start                Date
R00  332   16 689443731 2021/11/05 16:08:51
# time     Pres phase                Date
     0   74.240     7 2021/11/05 16:08:51
   135   69.920    13 2021/11/05 16:11:06
   217   63.520    13 2021/11/05 16:12:28
   278   58.880    13 2021/11/05 16:13:29
   340   53.840    13 2021/11/05 16:14:31
   401   48.440    13 2021/11/05 16:15:32
   463   42.760    13 2021/11/05 16:16:34
   524   37.080    13 2021/11/05 16:17:35
   586   31.120    13 2021/11/05 16:18:37
   627   26.360    13 2021/11/05 16:19:18
   668   21.160    13 2021/11/05 16:19:59
   709   16.360    13 2021/11/05 16:20:40
   750   11.400    13 2021/11/05 16:21:21
   791    6.920    13 2021/11/05 16:22:02
   863    0.200    13 2021/11/05 16:23:14
   873    0.000     8 2021/11/05 16:23:24
# Pump Time Series
#   Cycle npts
B02   332    6
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    51.040  294   15.20     199    616    -9     2
    74.240  135   15.27     211    621    -9     7
    61.960   21   15.28     206    624    -9    13
    52.120   21   15.28     215    626    -9    13
    40.880   21   15.27     208    628    -9    13
     0.160   60   15.24     185    686    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   332  2.4   17    90   100   600    49    10   180      9     3    4020   30  0.2866
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  332 2021/11/05 12:35:14
# Pcase vacuums
   7.720   6.110
# Oil valve positions
   1.678   0.574   0.544
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  332 2021/11/05 13:08:10 9999/99/99 99:99:99
# Pcase vacuums
   6.110   6.100
# Average 
#   P      T      S
   81.520   3.996  33.916
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  332 2021/11/05 16:08:51 9999/99/99 99:99:99
# Pcase vacuums
   6.100   6.320
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  332 2021/11/05 16:23:25
# number of packets previous packets
      5               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.320   7.650
# Air Pump max current time  max pressure
      47       260     7.346
# Air Valve Positions
   1.214   0.000
# Air Valve Calibrations
   1.165   0.132   0.649
# Oil Valve Positions
   1.694   0.572
# Oil Valve Calibrations
   1.598   0.657
# Encountered Error Codes
#    Cycle Errors
EE20   332      3
#               Date   Code Description
 2021/11/05 13:08:07 0x0113 DescendCtd
 2021/11/05 13:08:07 0x0913 DescendMaxDepth
 2021/11/05 13:08:07 0x0013 Descend
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  332  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.120    532   5.920
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  332       197    2021/11/05 16:28:49
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      332	(cycle number) |
     MPS =        0	(phase) |
     MPC =      332	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      331	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       50	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      600	(sec) |
  DFT[0] =      293	(sec) |
  DFT[1] =      320	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      331	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      331	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        2	(dBar) |
     ADD =        5	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      331	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      180	(sec) |
     SSP =       60	(sec) |

