# Message Decoder for Solo2, Alto and Deep Solo Floats; v211007
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11145   305  249 000000000337990  1893     0   4 2021/10/29, 22:19:12,  64.0243  -54.2435   6      0 13,6,14,14
SBD 11145   305  340 000000000337990  1894     0   3 2021/10/29, 22:19:22,  64.0243  -54.2435   6      0 13
SBD 11145   305  324 000000000337990  1895     0   2 2021/10/29, 22:19:37,  64.0055  -54.1771   6      0 3,0,8,15
SBD 11145   305  296 000000000337990  1896     0   1 2021/10/29, 22:19:47,  64.0243  -54.2435   6      0 11,7,5,14,1,2
SBD 11145   305  331 000000000337990  1897     0   0 2021/10/29, 22:19:58,  64.0243  -54.2435   6      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1480           60        1540
# Sensor_ID bytes     % packets Description
          0    48  3.24       2         GPS
          1    55  3.72       1    Pressure
          2    69  4.66       1 Temperature
          3    61  4.12       1    Salinity
          4   163 11.01       1        Fall
          5    68  4.59       1        Rise
          6    48  3.24       1        Pump
          7    33  2.23       1 Drift_times
          8   161 10.88       1     Surface
          9    48  3.24       1  HR/Drift_P
         10    40  2.70       1  HR/Drift_T
         11    40  2.70       1  HR/Drift_S
         13   404 27.30       2  Parameters
         14   176 11.89       5 Engineering
         15    66  4.46       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  305 1 2021/10/29 17:34 +64.01279  -54.13771    0  6 32 38 44  1.6     0
G  305 2 2021/10/29 22:15 +64.01926  -54.28357   70  7  0 32 45  1.1     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0   305   62    1     3         0
#      Pres      Temp    Salinity
      1.240     2.6750    33.1450
      2.000     2.6750    33.1450
      2.960     2.6750    33.1450
      3.960     2.6750    33.1450
      5.000     2.6740    33.1450
      6.000     2.6730    33.1460
      7.040     2.6750    33.1450
      8.040     2.6760    33.1450
      9.000     2.6760    33.1450
     10.000     2.6760    33.1460
     11.000     2.6770    33.1460
     12.000     2.6780    33.1460
     13.000     2.6800    33.1460
     14.000     2.6820    33.1470
     15.040     2.6850    33.1480
     15.960     2.6880    33.1490
     17.000     2.6910    33.1510
     18.000     2.6970    33.1530
     19.000     2.7100    33.1590
     20.040     2.7160    33.1630
     20.960     2.7170    33.1640
     22.000     2.7180    33.1650
     23.000     2.7240    33.1690
     23.960     2.7280    33.1720
     25.000     2.7340    33.1760
     26.040     2.7410    33.1800
     27.000     2.7450    33.1830
     28.040     2.7500    33.1860
     29.040     2.7540    33.1900
     30.000     2.7570    33.1920
     30.960     2.7630    33.1960
     31.960     2.7660    33.1990
     33.000     2.7700    33.2020
     34.000     2.7760    33.2070
     34.960     2.7760    33.2060
     36.000     2.7760    33.2060
     37.040     2.7770    33.2070
     38.000     2.7780    33.2080
     39.000     2.7940    33.2140
     40.040     2.8140    33.2240
     41.000     2.8450    33.2400
     41.960     2.8610    33.2500
     43.000     2.8920    33.2670
     44.040     2.9350    33.2910
     45.040     2.9610    33.3090
     46.040     2.9930    33.3250
     47.040     3.0120    33.3380
     48.040     3.0390    33.3540
     49.040     3.0680    33.3690
     50.000     3.0980    33.3890
     51.040     3.1180    33.3980
     51.960     3.1490    33.4170
     52.960     3.1700    33.4260
     54.040     3.1880    33.4370
     55.040     3.2170    33.4480
     56.040     3.2370    33.4610
     57.000     3.2510    33.4680
     58.000     3.2740    33.4780
     59.000     3.3140    33.4980
     60.040     3.3540    33.5180
     61.040     3.3810    33.5350
     61.840     3.3840    33.5380
# Drift Data
#      C Npts comp n_var
I0   305   17    1     4
#      Pres       Temp   Salinity     Time
     55.720     3.3760    33.5330        6
     51.600     3.4130    33.5550      611
     47.160     3.4220    33.5590     1216
     52.760     3.4130    33.5580     1822
     51.360     3.4050    33.5500     2427
     51.680     3.4030    33.5510     3033
     51.960     3.4060    33.5500     3638
     55.280     3.4050    33.5500     4243
     51.960     3.4100    33.5540     4849
     50.560     3.4160    33.5560     5454
     50.400     3.4250    33.5590     6060
     51.960     3.4210    33.5590     6665
     56.160     3.4180    33.5580     7271
     53.400     3.4110    33.5530     7876
     54.480     3.4220    33.5550     8481
     64.840     3.3770    33.5360     9087
     56.120     3.3960    33.5380     9692
# Surface Pressure Time Series
#    C npts n_var               Start
S  305    17     6 2021/10/29 22:14:29
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    78      0.090      2.653   -999.000      14 2021/10/29 22:15:47
    91      0.010      2.652   -999.000      14 2021/10/29 22:16:00
   102      0.060      2.664   -999.000      14 2021/10/29 22:16:11
   113      0.060      2.612   -999.000      14 2021/10/29 22:16:22
   124      0.010      2.656   -999.000      14 2021/10/29 22:16:33
   135      0.070      2.660   -999.000      14 2021/10/29 22:16:44
   147      0.020      2.658   -999.000      14 2021/10/29 22:16:56
   158      0.100      2.647   -999.000      14 2021/10/29 22:17:07
   169      0.080      2.636   -999.000      14 2021/10/29 22:17:18
   180      0.020      2.654   -999.000      14 2021/10/29 22:17:29
   191      0.010      2.652   -999.000      14 2021/10/29 22:17:40
   202      0.030      2.647   -999.000      14 2021/10/29 22:17:51
   213     -0.020      2.643   -999.000      14 2021/10/29 22:18:02
   224      0.000      2.636   -999.000      14 2021/10/29 22:18:13
   235      0.030      2.635   -999.000      14 2021/10/29 22:18:24
   246      0.060      2.537   -999.000      14 2021/10/29 22:18:35
   257      0.000      2.493   -999.000      14 2021/10/29 22:18:46
# Fall Time Series
#  Cycle npts     Start                Date
F00  305   31 688844128 2021/10/29 17:35:28
# time     Pres phase                Date
     0    0.080     1 2021/10/29 17:35:28
   285    5.320    10 2021/10/29 17:40:13
   321   10.440    10 2021/10/29 17:40:49
   353   15.240    10 2021/10/29 17:41:21
   386   20.120    10 2021/10/29 17:41:54
   418   25.040    10 2021/10/29 17:42:26
   454   30.400    10 2021/10/29 17:43:02
   490   35.200    10 2021/10/29 17:43:38
   526   40.080    10 2021/10/29 17:44:14
   562   45.160    10 2021/10/29 17:44:50
   598   50.160    10 2021/10/29 17:45:26
   602   50.800     2 2021/10/29 17:45:30
   634   55.160    10 2021/10/29 17:46:02
   670   60.120    10 2021/10/29 17:46:38
   709   65.040    10 2021/10/29 17:47:17
   748   70.000    10 2021/10/29 17:47:56
   790   75.160    10 2021/10/29 17:48:38
   839   80.240    10 2021/10/29 17:49:27
   901   85.160    10 2021/10/29 17:50:29
  1306   80.960    10 2021/10/29 17:57:14
  1473   75.960    10 2021/10/29 18:00:01
  1586   70.920    10 2021/10/29 18:01:54
  1683   65.920    10 2021/10/29 18:03:31
  1775   60.960    10 2021/10/29 18:05:03
  1893   55.960    10 2021/10/29 18:07:01
  2494   55.960    10 2021/10/29 18:17:02
  2727   56.200     3 2021/10/29 18:20:55
  3677   57.200    10 2021/10/29 18:36:45
  4278   53.800    10 2021/10/29 18:46:46
  4756   58.000    10 2021/10/29 18:54:44
  5258   55.880    10 2021/10/29 19:03:06
# Rise Time Series
#  Cycle npts     Start                Date
R00  305   12 688860219 2021/10/29 22:03:39
# time     Pres phase                Date
     0   62.160     7 2021/10/29 22:03:39
   114   56.520    13 2021/10/29 22:05:33
   176   50.280    13 2021/10/29 22:06:35
   237   43.960    13 2021/10/29 22:07:36
   299   37.960    13 2021/10/29 22:08:38
   360   31.600    13 2021/10/29 22:09:39
   422   25.200    13 2021/10/29 22:10:41
   463   20.760    13 2021/10/29 22:11:22
   524   14.160    13 2021/10/29 22:12:23
   565    9.840    13 2021/10/29 22:13:04
   627    3.360    13 2021/10/29 22:14:06
   648    1.240     8 2021/10/29 22:14:27
# Pump Time Series
#   Cycle npts
B02   305    4
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    50.800  323   15.21     196    619    -9     2
    62.160  135   15.29     213    626    -9     7
    42.040   21   15.29     210    629    -9    13
     0.040   76   15.26     195    699    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   305  2.4   15    90   100   600    50    10   180      9     3    4020   30  0.2873
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  305 2021/10/29 17:35:25
# Pcase vacuums
   7.750   6.150
# Oil valve positions
   1.679   0.577   0.547
# Seek attempts and pump time
      1     -4
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  305 2021/10/29 19:03:07 9999/99/99 99:99:99
# Pcase vacuums
   6.150   6.150
# Average 
#   P      T      S
   53.360   3.408  33.551
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  305 2021/10/29 22:03:39 9999/99/99 99:99:99
# Pcase vacuums
   6.150   6.300
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  305 2021/10/29 22:14:28
# number of packets previous packets
      5               6
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.300   7.640
# Air Pump max current time  max pressure
      47       261     7.336
# Air Valve Positions
   1.213   0.000
# Air Valve Calibrations
   1.165   0.132   0.649
# Oil Valve Positions
   1.689   0.586
# Oil Valve Calibrations
   1.598   0.657
# Encountered Error Codes
#    Cycle Errors
EE20   305      3
#               Date   Code Description
 2021/10/29 19:03:04 0x0113 DescendCtd
 2021/10/29 19:03:04 0x0913 DescendMaxDepth
 2021/10/29 19:03:04 0x0013 Descend
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  305  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.080    597   5.930
   3    56.040      5   6.160
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  305       197    2021/10/29 22:19:22
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      305	(cycle number) |
     MPS =        0	(phase) |
     MPC =      305	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      304	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       50	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      600	(sec) |
  DFT[0] =      320	(sec) |
  DFT[1] =      320	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65525	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =      304	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =      304	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =    20000	(ms) |
     ASD =        2	(dBar) |
     ADD =        5	(dBar) |
     ADT =     1800	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      304	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      180	(sec) |
     SSP =       60	(sec) |

