# Message Decoder for Solo2, Alto and Deep Solo Floats; v210824
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11145    80   88 000000000337990   529     0   6 2021/09/03, 05:50:30,  59.7780  -45.6551   2      0 14
SBD 11145    80  340 000000000337990   530     0   5 2021/09/03, 05:50:38,  59.7558  -45.7048   2      0 13
SBD 11145    80  320 000000000337990   531     0   4 2021/09/03, 05:50:51,  59.7341  -45.6539   2      0 15,13,6,14
SBD 11145    80  309 000000000337990   532     0   3 2021/09/03, 05:51:02,  59.7558  -45.7048   7      0 3,8
SBD 11145    80  296 000000000337990   533     0   2 2021/09/03, 05:51:24,  59.7558  -45.7048   2      0 1,2
SBD 11145    80  338 000000000337990   534     0   1 2021/09/03, 05:51:37,  59.7341  -45.6539   2      0 9,10,11,7,5,14
SBD 11145    80  332 000000000337990   535     0   0 2021/09/03, 05:51:59,  59.7558  -45.7048   3      0 0,4,14,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1939           84        2023
# Sensor_ID bytes     % packets Description
          0    48  2.48       2         GPS
          1    88  4.54       1    Pressure
          2   196 10.11       1 Temperature
          3   136  7.01       1    Salinity
          4   228 11.76       1        Fall
          5   138  7.12       1        Rise
          6   147  7.58       1        Pump
          7    33  1.70       1 Drift_times
          8   161  8.30       1     Surface
          9    48  2.48       1  HR/Drift_P
         10    48  2.48       1  HR/Drift_T
         11    40  2.06       1  HR/Drift_S
         13   404 20.84       2  Parameters
         14   158  8.15       5 Engineering
         15    66  3.40       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   80 1 2021/09/02 23:21 +59.74261  -45.61483    0  6 31 38 47  1.3     0
G   80 2 2021/09/03 05:46 +59.76087  -45.68182   50  6  0 27 36  1.5     0
# Profile Data
#      C Npts comp n_var
D0    80  121    1     3
#      Pres      Temp    Salinity
      2.080     3.8350    31.1580
      3.000     3.7350    31.2210
      4.000     3.6200    31.2590
      5.000     3.6070    31.2690
      6.040     3.5980    31.2750
      7.040     3.5280    31.2950
      8.080     3.4290    31.3170
      9.040     3.2940    31.3630
     10.000     3.1370    31.4240
     11.000     2.9260    31.5380
     12.040     2.7220    31.6410
     13.000     2.7200    31.6470
     13.960     2.6730    31.6730
     15.000     2.5250    31.7740
     15.960     2.5820    31.8320
     17.000     2.6280    31.9870
     18.000     2.5900    32.0290
     19.040     2.5420    32.0650
     20.040     2.4660    32.1100
     21.040     2.3550    32.1750
     22.040     2.3550    32.1780
     23.000     2.3480    32.1810
     23.960     2.3430    32.1860
     25.000     2.3370    32.2000
     26.000     2.3410    32.2180
     27.000     2.3580    32.2370
     28.000     2.3320    32.2520
     29.040     2.0590    32.3030
     30.040     1.9330    32.3950
     31.040     1.8910    32.4220
     32.000     1.8340    32.4300
     33.000     1.7400    32.4430
     34.000     1.7340    32.4450
     35.000     1.7390    32.4650
     36.000     1.9230    32.5160
     37.040     2.0030    32.5820
     38.000     2.0230    32.6330
     38.960     2.0970    32.6730
     40.000     2.1410    32.6890
     41.040     2.1880    32.7110
     42.040     2.2280    32.7330
     43.040     2.2830    32.7430
     44.040     2.2890    32.7760
     45.040     2.3970    32.8130
     46.000     2.3470    32.8190
     47.000     2.2960    32.8210
     48.040     2.2450    32.8170
     49.000     2.2120    32.8130
     50.000     2.1490    32.8170
     51.040     2.1250    32.8170
     52.040     2.1260    32.8160
     53.040     2.1300    32.8170
     54.040     2.1200    32.8200
     55.040     2.0870    32.8220
     56.000     2.0810    32.8210
     57.000     2.0800    32.8230
     58.000     2.0780    32.8240
     59.000     2.0760    32.8250
     60.000     2.0760    32.8270
     61.000     2.0850    32.8290
     62.000     2.0840    32.8370
     62.960     2.0450    32.8410
     64.000     2.0380    32.8410
     65.000     2.0380    32.8430
     66.000     2.0460    32.8520
     67.000     2.0430    32.8600
     68.000     2.1510    32.8810
     69.000     2.2160    32.9170
     70.000     2.2640    32.9230
     71.000     2.2990    32.9400
     72.000     2.3850    32.9510
     73.000     2.4260    32.9670
     74.040     2.4310    32.9710
     75.000     2.4570    32.9740
     76.000     2.5020    32.9970
     77.000     2.5480    33.0140
     77.960     2.5700    33.0320
     78.960     2.6020    33.0440
     80.000     2.7020    33.0760
     81.000     2.8820    33.1310
     82.000     3.3240    33.2250
     83.000     3.5450    33.2840
     83.960     3.6870    33.3300
     85.000     3.7580    33.3480
     86.040     3.8200    33.3700
     87.040     3.8340    33.3810
     88.040     3.9200    33.4040
     89.000     4.0450    33.4420
     90.000     4.1930    33.5370
     91.040     4.4150    33.6000
     92.040     4.6050    33.6620
     93.040     4.5810    33.7250
     94.040     4.5460    33.7520
     95.040     4.5480    33.7720
     96.000     4.5440    33.7790
     97.000     4.5520    33.7890
     97.960     4.5630    33.8030
     98.960     4.5720    33.8170
    100.000     4.5820    33.8340
    101.000     4.5960    33.8490
    102.040     4.6270    33.8720
    103.000     4.7320    33.9320
    104.000     4.7680    33.9550
    105.040     4.8280    33.9720
    106.000     4.9580    34.0240
    107.000     4.9200    34.0810
    108.000     4.6320    34.0600
    109.000     4.5790    34.0620
    110.040     4.5520    34.0670
    111.040     4.4340    34.0670
    112.040     3.8840    34.0660
    113.040     3.4590    34.0220
    114.040     3.2770    34.0070
    115.040     3.1060    33.9900
    116.040     2.8840    33.9840
    117.040     2.7710    33.9710
    118.000     2.7980    33.9770
    119.080     2.8320    33.9830
    120.080     2.8660    33.9910
    121.160     2.8890    33.9970
    121.520     2.8920    33.9970
# Drift Data
#      C Npts comp n_var
I0    80   17    1     4
#      Pres       Temp   Salinity     Time
    124.960     2.7380    34.0000        5
    126.280     2.7570    33.9920      611
    125.160     2.8240    33.9850     1216
    125.920     2.8510    33.9940     1822
    124.720     2.8690    34.0000     2427
    123.680     2.8510    33.9910     3032
    124.560     2.8130    33.9950     3638
    125.920     2.8130    33.9970     4243
    124.240     2.8510    33.9980     4849
    125.960     2.8900    33.9980     5454
    121.680     2.8830    34.0000     6059
    126.240     2.8860    33.9970     6665
    122.040     2.8650    33.9890     7270
    126.320     2.8860    33.9970     7876
    126.640     2.8080    33.9710     8481
    124.080     2.8620    33.9880     9087
    121.520     2.8880    33.9950     9692
# Surface Pressure Time Series
#    C npts n_var               Start
S   80    17     6 2021/09/03 05:45:36
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    79      0.150      4.282   -999.000      14 2021/09/03 05:46:55
    92      0.170      4.119   -999.000      14 2021/09/03 05:47:08
   103      0.100      4.229   -999.000      14 2021/09/03 05:47:19
   114      0.110      4.316   -999.000      14 2021/09/03 05:47:30
   125      0.160      4.322   -999.000      14 2021/09/03 05:47:41
   136      0.180      4.327   -999.000      14 2021/09/03 05:47:52
   147      0.150      4.317   -999.000      14 2021/09/03 05:48:03
   158      0.150      4.243   -999.000      14 2021/09/03 05:48:14
   169      0.140      4.309   -999.000      14 2021/09/03 05:48:25
   180      0.150      4.307   -999.000      14 2021/09/03 05:48:36
   191      0.130      4.318   -999.000      14 2021/09/03 05:48:47
   202      0.090      4.285   -999.000      14 2021/09/03 05:48:58
   213      0.100      4.317   -999.000      14 2021/09/03 05:49:09
   225      0.120      4.255   -999.000      14 2021/09/03 05:49:21
   236      0.120      4.300   -999.000      14 2021/09/03 05:49:32
   247      0.120      4.304   -999.000      14 2021/09/03 05:49:43
   258      0.090      4.333   -999.000      14 2021/09/03 05:49:54
# Fall Time Series
#  Cycle npts     Start                Date
F00   80   44 683940147 2021/09/02 23:22:27
# time     Pres phase                Date
     0    0.080     1 2021/09/02 23:22:27
   217    5.160    10 2021/09/02 23:26:04
   253   10.440    10 2021/09/02 23:26:40
   286   15.240    10 2021/09/02 23:27:13
   318   20.080    10 2021/09/02 23:27:45
   351   25.320    10 2021/09/02 23:28:18
   383   30.360    10 2021/09/02 23:28:50
   416   35.440    10 2021/09/02 23:29:23
   448   40.320    10 2021/09/02 23:29:55
   481   45.200    10 2021/09/02 23:30:28
   513   50.360    10 2021/09/02 23:31:00
   518   51.040     2 2021/09/02 23:31:05
   546   55.280    10 2021/09/02 23:31:33
   579   60.040    10 2021/09/02 23:32:06
   615   65.000    10 2021/09/02 23:32:42
   654   70.040    10 2021/09/02 23:33:21
   699   75.040    10 2021/09/02 23:34:06
   752   80.000    10 2021/09/02 23:34:59
  1096   75.960    10 2021/09/02 23:40:43
  1600   80.040    10 2021/09/02 23:49:07
  1925   85.040    10 2021/09/02 23:54:32
  2526   86.720    10 2021/09/03 00:04:33
  2665   84.800     3 2021/09/03 00:06:52
  2929   81.960    10 2021/09/03 00:11:16
  3529   83.600    10 2021/09/03 00:21:16
  4130   81.440    10 2021/09/03 00:31:17
  4375   76.960    10 2021/09/03 00:35:22
  4502   76.120     3 2021/09/03 00:37:29
  5450   96.800    10 2021/09/03 00:53:17
  5648  101.080    10 2021/09/03 00:56:35
  5757  106.080    10 2021/09/03 00:58:24
  5879  111.000    10 2021/09/03 01:00:26
  6308  116.000    10 2021/09/03 01:07:35
  6438  121.040    10 2021/09/03 01:09:45
  6692  126.000    10 2021/09/03 01:13:59
  7250  127.880     3 2021/09/03 01:23:17
  7292  127.680    10 2021/09/03 01:23:59
  7544  122.960    10 2021/09/03 01:28:11
  8145  123.240    10 2021/09/03 01:38:12
  8746  123.080    10 2021/09/03 01:48:13
  9347  122.920    10 2021/09/03 01:58:14
  9947  121.840    10 2021/09/03 02:08:14
 10548  124.480    10 2021/09/03 02:18:15
 10892  124.960    10 2021/09/03 02:23:59
# Rise Time Series
#  Cycle npts     Start                Date
R00   80   26 683961862 2021/09/03 05:24:22
# time     Pres phase                Date
     0  121.560     7 2021/09/03 05:24:22
   104  116.920    13 2021/09/03 05:26:06
   156  111.880    13 2021/09/03 05:26:58
   212  106.920    13 2021/09/03 05:27:54
   270  102.000    13 2021/09/03 05:28:52
   332   96.960    13 2021/09/03 05:29:54
   389   91.920    13 2021/09/03 05:30:51
   447   86.920    13 2021/09/03 05:31:49
   498   81.920    13 2021/09/03 05:32:40
   546   77.000    13 2021/09/03 05:33:28
   594   72.000    13 2021/09/03 05:34:16
   638   66.880    13 2021/09/03 05:35:00
   681   61.960    13 2021/09/03 05:35:43
   725   56.760    13 2021/09/03 05:36:27
   767   51.840    13 2021/09/03 05:37:09
   810   46.920    13 2021/09/03 05:37:52
   857   41.840    13 2021/09/03 05:38:39
   902   36.960    13 2021/09/03 05:39:24
   957   31.960    13 2021/09/03 05:40:19
  1013   26.920    13 2021/09/03 05:41:15
  1070   21.960    13 2021/09/03 05:42:12
  1121   16.800    13 2021/09/03 05:43:03
  1173   11.920    13 2021/09/03 05:43:55
  1223    7.000    13 2021/09/03 05:44:45
  1272    1.920    13 2021/09/03 05:45:34
  1273    1.800     8 2021/09/03 05:45:35
# Pump Time Series
#   Cycle npts
B02    80   13
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
    51.040  345   15.26     209    627    -9     2
   127.720    2   15.32     299    616    -9     3
   121.560  121   15.38     248    625    -9     7
   106.200   21   15.37     249    624    -9    13
    96.120   20   15.37     247    624    -9    13
    85.600   20   15.37     233    626    -9    13
    73.240   20   15.36     224    631    -9    13
    59.480   20   15.36     225    635    -9    13
    45.600   20   15.36     216    637    -9    13
    33.280   20   15.36     212    639    -9    13
    22.640   20   15.35     204    640    -9    13
    11.080   20   15.35     196    641    -9    13
     0.080   59   15.33     194    693    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    80  2.4   13    90   100   600    50    10   180      7     3    4020   30  0.2906
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   80 2021/09/02 23:22:25
# Pcase vacuums
   7.870   6.160
# Oil valve positions
   1.660   0.595   0.548
# Seek attempts and pump time
      3     -1
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   80 2021/09/03 02:24:00 9999/99/99 99:99:99
# Pcase vacuums
   6.160   6.160
# Average 
#   P      T      S
  124.720   2.843  33.993
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   80 2021/09/03 05:24:22 9999/99/99 99:99:99
# Pcase vacuums
   6.160   6.390
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   80 2021/09/03 05:45:35
# number of packets previous packets
      6               7
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.390   7.730
# Air Pump max current time  max pressure
      45       261     7.404
# Air Valve Positions
   1.212   0.000
# Air Valve Calibrations
   1.165   0.132   0.649
# Oil Valve Positions
   1.667   0.602
# Oil Valve Calibrations
   1.598   0.657
# Encountered Error Codes
#    Cycle Errors
EE20    80      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   80  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.040    513   5.970
   3    75.880      3   6.180
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   80       197    2021/09/03 05:50:51
Parameter Table Param_Table-V17.txt |
     IFV =  9.0.0.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       80	(cycle number) |
     MPS =        0	(phase) |
     MPC =       80	(count) |
  MNM[0] =      999	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =      100	(dBar) |
  DTD[2] =     1000	(dBar) |
     MDS =     2100	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        0	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  2592000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       79	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       50	(dBar) |
  DSD[1] =       90	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      420	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        5	(dBar) |
     DDT =      600	(sec) |
  DFT[0] =      344	(sec) |
  DFT[1] =      320	(sec) |
  DFT[2] =      180	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    21600	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        7	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =        6	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =      900	(sec) |
     PNS =       79	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   777600	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =      600	(sec) |
  PFT[1] =      600	(sec) |
  PFT[2] =    21600	(sec) |
  PFD[0] =       90	(dBar) |
  PFD[1] =      300	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =       10	(sec) |
  PFV[1] =       10	(sec) |
  PFV[2] =      420	(sec) |
     PDT =      600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       79	(cycle number) |
     ARR =        7	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =      120	(sec) |
     APN =       20	(sec) |
     APF =      100	(sec) |
     ASI =     1000	(ms) |
     ASD =        2	(dBar) |
     ADD =        5	(dBar) |
     ADT =     1800	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       79	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      180	(sec) |
     SSP =       60	(sec) |

