# Message Decoder for Solo2, Alto and Deep Solo Floats; v241112
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11076    -4   49 000000000000009    11     0   0 2024/11/15, 20:50:06,  41.5334  -70.6398  25      0 0,15
SBD 11076    -4  276 000000000000009    22     0   3 2024/11/15, 20:53:00,  41.5187  -70.7009   5      0 14
SBD 11076    -4  339 000000000000009    23     0   2 2024/11/15, 20:53:12,  41.5187  -70.7009   5      0 13
SBD 11076    -4  111 000000000000009    24     0   1 2024/11/15, 20:53:20,  41.5187  -70.7009   4      0 0,13
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       6          727           48         775
# Sensor_ID bytes     % packets Description
          0    48  6.60       2         GPS
         13   402 55.30       2  Parameters
         14   264 36.31       1 Engineering
         15    13  1.79       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   -4 5 2024/11/15 20:52 +41.52460  -70.67164   10  6 23 36 43  1.4     0
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -4  2.0    1   150   150   180   360     0     0     11     1    2160    0  0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS           25      10      1000       5      1000       1 
# Engineering Data
#       Alto BIT Test
#    cycle software   BIT         Total
#          version    result      time
E011   -4  0x0a090122 0x00000483  0
 # Skyview           BIT_Start             BIT_End
       255 1970/01/01 00:00:03 2024/11/15 20:30:14
# BIT error 0x01: Air pump failure
# BIT error 0x02: Oil pump failure
# BIT Error 0x80: Case pressure failure
# Air pump current high
# Oil pump current low
# Case pressure low
# CTD Status temp    salinity
 0x3bfc5048  -0.020  14.798
# Pressure offsets 
# before   now     after [dBar]
 867523856591954255020032.000   -0.220   -0.050
# Pcase vacuum [inHg]
 809500672.000   0.000
# Battery Voltages [V]
   0.000  14.680
# Air Pump test
# Air Valve Positions [V]
  14.620   1.249   0.164
# Air Valve Calibrations [V]
   1.309   0.104   0.651
# duration count max P max currents      max voltage
   16180   219 164041929828597424821499544469504.00    7.52    0.05    0.07
# Oil Pump test
# Oil Valve Positions [V]
  14.680   1.876
# Oil Valve Calibrations [V]
   1.939   0.698
# duration max currents [mAmp]      max voltage [V]
   16194   33500638317744756883456.00    0.14         0.09
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   -4         0    2024/11/15 20:53:20
