# Message Decoder for Solo2, Alto and Deep Solo Floats; v241112
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11076    -1  276 000000000000009     4     0   2 2024/11/15, 20:22:29,  41.5187  -70.7009   3      0 14
SBD 11076    -1  339 000000000000009     5     0   1 2024/11/15, 20:22:39,  41.5187  -70.7009   3      0 13
SBD 11076    -1  148 000000000000009     6     0   0 2024/11/15, 20:22:58,  41.5413  -70.6851   8      0 0,15,0,13
# Delayed Transmission - (3/3) No surface time available, transmission was most likely delayed
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       6          727           36         763
# Sensor_ID bytes     % packets Description
          0    48  6.60       2         GPS
         13   402 55.30       2  Parameters
         14   264 36.31       1 Engineering
         15    13  1.79       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   -1 5 9999/99/99 99:99 +99.99000 +999.99000  900  0  0  0  0  0.0     0
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -1  2.0    1   150   150   180   360     0     0     11     1    2160    0  0.2908
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS           25      10      1000       5      1000       1 
# Engineering Data
#       Alto BIT Test
#    cycle software   BIT         Total
#          version    result      time
E011   -1  0x0a090122 0x000004b3  0
 # Skyview           BIT_Start             BIT_End
       255 1970/01/01 00:00:03 1970/01/01 00:03:15
# BIT error 0x01: Air pump failure
# BIT error 0x02: Oil pump failure
# BIT error 0x10: GPS failure
# BIT error 0x20: CTD Sensor failure
# BIT Error 0x80: Case pressure failure
# Air pump current high
# Oil pump current low
# GPS no fix
# GPS failure
# CTD unable to communicate
# Satellite first transmit failure
# Case pressure low
# CTD Status temp    salinity
 0x4f800000 4294967296.000 4294967296.000
# Pressure offsets 
# before   now     after [dBar]
    0.000 4294967296.000    0.000
# Pcase vacuum [inHg]
 809500672.000   0.000
# Battery Voltages [V]
   0.000  15.310
# Air Pump test
# Air Valve Positions [V]
  15.110   1.248   0.162
# Air Valve Calibrations [V]
   1.309   0.102   0.642
# duration count max P max currents      max voltage
   16180   128    0.00    7.58    0.06    0.07
# Oil Pump test
# Oil Valve Positions [V]
  15.280   1.879
# Oil Valve Calibrations [V]
   1.940   0.699
# duration max currents [mAmp]      max voltage [V]
   16195      0.00    0.16         0.10
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   -1         0    2024/11/15 20:22:58
