# Message Decoder for Solo2, Alto and Deep Solo Floats; v241223
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11074    54  159 000000000000009   387     0  11 2024/12/20, 05:16:42,  41.5334  -70.6398  22      0 6,14,14
SBD 11074    54  339 000000000000009   388     0  10 2024/12/20, 05:16:52,  41.5603  -70.6398  21      0 13
SBD 11074    54  332 000000000000009   389     0   9 2024/12/20, 05:17:07,  41.5255  -70.5945  19      0 8,15,13
SBD 11074    54  335 000000000000009   390     0   8 2024/12/20, 05:17:27,  41.5413  -70.6851  15      0 8
SBD 11074    54  283 000000000000009   391     0   7 2024/12/20, 05:17:46,  41.5334  -70.6398  13      0 5,14,0,1,2,3
SBD 11074    54  335 000000000000009   392     0   6 2024/12/20, 05:19:04,  41.6349  -68.5555 180      2 6
SBD 11074    54  335 000000000000009   393     0   5 2024/12/20, 05:19:23,  41.5334  -70.6398   6      0 6
SBD 11074    54  324 000000000000009   394     0   4 2024/12/20, 05:19:54,  41.5187  -70.7009   5      0 12,12,14,9,10,11,7
SBD 11074    54  336 000000000000009   395     0   3 2024/12/20, 05:21:15,  41.5413  -70.6851   5      0 12,12,12,12,12,12,12
SBD 11074    54  336 000000000000009   396     0   2 2024/12/20, 05:21:35,  41.5683  -70.6851   6      0 12,12,12,12,12,12,12
SBD 11074    54  336 000000000000009   397     0   1 2024/12/20, 05:21:52,  41.5857  -68.5730 182      2 12,12,12,12,12,12,12
SBD 11074    54  301 000000000000009   398     0   0 2024/12/20, 05:22:03,  41.5857  -68.5730 181      2 0,4,14,12,12,12
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      50         3607          144        3751
# Sensor_ID bytes     % packets Description
          0    48  1.33       2         GPS
          1    29  0.80       1    Pressure
          2    33  0.91       1 Temperature
          3    33  0.91       1    Salinity
          4    83  2.30       1        Fall
          5   133  3.69       1        Rise
          6   705 19.55       3        Pump
          7    46  1.28       1 Drift_times
          8   502 13.92       2     Surface
          9    46  1.28       1  HR/Drift_P
         10    62  1.72       1  HR/Drift_T
         11    54  1.50       1  HR/Drift_S
         12  1214 33.66      26  Aux Sensor
         13   402 11.14       2  Parameters
         14   151  4.19       5 Engineering
         15    66  1.83       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   54 1 2024/12/19 05:20 +41.53433  -70.64687    0  4  0  0  0  2.3     0
G   54 2 2024/12/20 05:05 +41.53431  -70.64683   30  7  0  0  0  1.6     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    54   10    1     3         1
#      Pres      Temp    Salinity
      0.360    17.3630    32.7560
      1.040    17.3290    32.7620
      2.000    17.2180    32.7500
      2.960    17.1700    32.7470
      4.000    17.1280    32.7470
      5.040    17.0880    32.7470
      6.040    17.0550    32.7450
      7.040    17.0320    32.7450
      8.040    17.0000    32.7450
      9.320    16.8890    32.7450
# Drift Data
#      C Npts comp n_var
I0    54   44    1     4
#      Pres       Temp   Salinity     Time
      9.240    16.8850    32.7330        6
      9.240    16.8690    32.7350     1811
      9.240    16.8650    32.7310     3617
      9.240    16.9980    32.7310     5423
      9.240    17.0680    32.7310     7228
      9.280    17.1150    32.7440     9034
      9.280    17.1110    32.7430    10839
      9.280    17.1200    32.7430    12645
      9.240    17.1350    32.7430    14451
      9.280    17.1520    32.7420    16256
      9.280    17.1570    32.7440    18062
      9.280    17.1710    32.7440    19867
      9.280    17.1840    32.7440    21673
      9.280    17.1940    32.7440    23479
      9.280    17.2050    32.7440    25284
      9.320    17.2070    32.7450    27090
      9.320    17.2010    32.7450    28895
      9.320    17.1930    32.7440    30701
      9.320    17.1800    32.7460    32507
      9.320    17.1730    32.7440    34312
      9.320    17.1630    32.7440    36118
      9.320    17.1510    32.7460    37923
      9.280    17.1430    32.7440    39729
      9.320    17.1310    32.7450    41535
      9.320    17.1210    32.7450    43340
      9.320    17.1120    32.7450    45146
      9.320    17.1020    32.7440    46951
      9.320    17.0910    32.7450    48757
      9.360    17.0820    32.7450    50563
      9.360    17.0730    32.7440    52368
      9.360    17.0620    32.7450    54174
      9.360    17.0510    32.7450    55979
      9.360    17.0410    32.7450    57785
      9.360    17.0300    32.7450    59591
      9.360    17.0200    32.7450    61396
      9.360    17.0090    32.7450    63202
      9.360    16.9990    32.7440    65007
      9.360    16.9880    32.7440    66813
      9.360    16.9770    32.7440    68619
      9.360    16.9650    32.7450    70424
      9.360    16.9530    32.7440    72230
      9.360    16.9440    32.7440    74035
      9.360    16.9330    32.7430    75841
      9.360    16.9220    32.7430    77647
# Surface Pressure Time Series
#    C npts n_var               Start
S   54    54     6 2024/12/20 05:06:17
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.100     17.233   -999.000      14 2024/12/20 05:06:17
    13      0.100     17.118   -999.000      14 2024/12/20 05:06:30
    24      0.110     17.071   -999.000      14 2024/12/20 05:06:41
    35      0.100     17.056   -999.000      14 2024/12/20 05:06:52
    46      0.090     17.069   -999.000      14 2024/12/20 05:07:03
    58      0.100     17.026   -999.000      14 2024/12/20 05:07:15
    68      0.110     17.014   -999.000      14 2024/12/20 05:07:25
    79      0.090     17.002   -999.000      14 2024/12/20 05:07:36
    91      0.110     16.965   -999.000      14 2024/12/20 05:07:48
   102      0.110     16.894   -999.000      14 2024/12/20 05:07:59
   113      0.080     16.852   -999.000      14 2024/12/20 05:08:10
   124      0.100     16.728   -999.000      14 2024/12/20 05:08:21
   135      0.080     16.675   -999.000      14 2024/12/20 05:08:32
   146      0.090     16.651   -999.000      14 2024/12/20 05:08:43
   157      0.080     16.601   -999.000      14 2024/12/20 05:08:54
   168      0.070     16.545   -999.000      14 2024/12/20 05:09:05
   179      0.090     16.463   -999.000      14 2024/12/20 05:09:16
   190      0.070     16.400   -999.000      14 2024/12/20 05:09:27
   201      0.080     16.361   -999.000      14 2024/12/20 05:09:38
   213      0.070     16.311   -999.000      14 2024/12/20 05:09:50
   224      0.050     16.246   -999.000      14 2024/12/20 05:10:01
   235      0.070     16.177   -999.000      14 2024/12/20 05:10:12
   246      0.080     16.071   -999.000      14 2024/12/20 05:10:23
   257      0.070     15.986   -999.000      14 2024/12/20 05:10:34
   268      0.060     15.925   -999.000      14 2024/12/20 05:10:45
   279      0.050     15.835   -999.000      14 2024/12/20 05:10:56
   290      0.060     15.751   -999.000      14 2024/12/20 05:11:07
   301      0.040     15.678   -999.000      14 2024/12/20 05:11:18
   312      0.050     15.634   -999.000      14 2024/12/20 05:11:29
   323      0.060     15.581   -999.000      14 2024/12/20 05:11:40
   334      0.040     15.546   -999.000      14 2024/12/20 05:11:51
   346      0.040     15.511   -999.000      14 2024/12/20 05:12:03
   357      0.060     15.500   -999.000      14 2024/12/20 05:12:14
   368      0.030     15.473   -999.000      14 2024/12/20 05:12:25
   379      0.040     15.448   -999.000      14 2024/12/20 05:12:36
     0      0.020     15.412   -999.000      14 2024/12/20 05:12:47
    11      0.040     15.361   -999.000      14 2024/12/20 05:12:58
    22      0.040     15.314   -999.000      14 2024/12/20 05:13:09
    33      0.030     15.292   -999.000      14 2024/12/20 05:13:20
    44      0.020     15.270   -999.000      14 2024/12/20 05:13:31
    55      0.010     15.246   -999.000      14 2024/12/20 05:13:42
    66      0.010     15.201   -999.000      14 2024/12/20 05:13:53
    78      0.030     15.164   -999.000      14 2024/12/20 05:14:05
    89      0.010     15.135   -999.000      14 2024/12/20 05:14:16
   100      0.010     15.095   -999.000      14 2024/12/20 05:14:27
   111      0.010     15.032   -999.000      14 2024/12/20 05:14:38
   122      0.000     14.959   -999.000      14 2024/12/20 05:14:49
   133      0.010     14.917   -999.000      14 2024/12/20 05:15:00
   144      0.000     14.875   -999.000      14 2024/12/20 05:15:11
   155      0.000     14.824   -999.000      14 2024/12/20 05:15:22
   166      0.000     14.797   -999.000      14 2024/12/20 05:15:33
   177     -0.010     14.790   -999.000      14 2024/12/20 05:15:44
   188     -0.010     14.800   -999.000      14 2024/12/20 05:15:55
   199     -0.020     14.836   -999.000      14 2024/12/20 05:16:06
# Fall Time Series
#  Cycle npts     Start                Date
F00   54   15 787900808 2024/12/19 05:20:08
# time     Pres phase                Date
     0   -0.160     1 2024/12/19 05:20:08
   192    1.280    10 2024/12/19 05:23:20
   196    2.040    10 2024/12/19 05:23:24
   204    4.000    10 2024/12/19 05:23:32
   210    5.640     2 2024/12/19 05:23:38
   212    6.360    10 2024/12/19 05:23:40
   217    7.680    10 2024/12/19 05:23:45
   221    9.040    10 2024/12/19 05:23:49
   525    9.240    10 2024/12/19 05:28:53
   826    9.240    10 2024/12/19 05:33:54
  1127    9.240    10 2024/12/19 05:38:55
  1428    9.240    10 2024/12/19 05:43:56
  1730    9.240    10 2024/12/19 05:48:58
  2031    9.240    10 2024/12/19 05:53:59
  2039    9.240    10 2024/12/19 05:54:07
# Rise Time Series
#  Cycle npts     Start                Date
R00   54   25 787982409 2024/12/20 04:00:09
# time     Pres phase                Date
     0    9.400     7 2024/12/20 04:00:09
   316    9.360    13 2024/12/20 04:05:25
   621    9.360    13 2024/12/20 04:10:30
   925    9.320    13 2024/12/20 04:15:34
  1230    9.320    13 2024/12/20 04:20:39
  1534    9.360    13 2024/12/20 04:25:43
  1839    9.360    13 2024/12/20 04:30:48
  2144    9.360    13 2024/12/20 04:35:53
  2448    9.360    13 2024/12/20 04:40:57
  2753    9.320    13 2024/12/20 04:46:02
  3057    9.320    13 2024/12/20 04:51:06
  3362    9.360    13 2024/12/20 04:56:11
  3666    9.080    13 2024/12/20 05:01:15
  3708    9.000    13 2024/12/20 05:01:57
  3719    9.040    13 2024/12/20 05:02:08
  3729    8.960    13 2024/12/20 05:02:18
  3750    9.000    13 2024/12/20 05:02:39
  3761    8.920    13 2024/12/20 05:02:50
  3803    7.680    13 2024/12/20 05:03:32
  3824    6.160    13 2024/12/20 05:03:53
  3834    4.960    13 2024/12/20 05:04:03
  3845    3.680    13 2024/12/20 05:04:14
  3855    2.000    13 2024/12/20 05:04:24
  3876    0.120    13 2024/12/20 05:04:45
  3887    0.120     8 2024/12/20 05:04:56
# Pump Time Series
#   Cycle npts
B02    54   63
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     5.640   22   14.95     161    538    -9     2
     9.400   22   15.06     184    535    -9     7
     9.360   11   15.06     181    536    -9    13
     9.360   11   15.05     191    537    -9    13
     9.320   11   15.06     180    538    -9    13
     9.360   11   15.06     186    539    -9    13
     9.360   11   15.05     185    540    -9    13
     9.360   11   15.05     186    541    -9    13
     9.360   11   15.03     186    543    -9    13
     9.360   11   15.04     178    544    -9    13
     9.360   11   15.03     189    544    -9    13
     9.320   11   15.03     186    545    -9    13
     9.360   11   15.04     175    546    -9    13
     9.360   11   15.03     183    547    -9    13
     9.320   11   15.02     185    548    -9    13
     9.320   11   15.02     177    549    -9    13
     9.360   11   15.02     182    550    -9    13
     9.360   11   15.02     187    551    -9    13
     9.360   11   15.02     178    552    -9    13
     9.360   11   15.02     172    553    -9    13
     9.360   11   15.02     172    554    -9    13
     9.360   11   15.00     188    555    -9    13
     9.360   11   15.00     181    556    -9    13
     9.400   11   15.01     173    556    -9    13
     9.320   11   14.99     183    557    -9    13
     9.360   11   15.01     169    558    -9    13
     9.320   11   15.00     170    559    -9    13
     9.360   11   15.01     162    560    -9    13
     9.360   11   15.00     163    561    -9    13
     9.320   11   15.00     167    562    -9    13
     9.320   10   14.99     166    563    -9    13
     9.360   10   14.99     175    564    -9    13
     9.320   10   14.98     171    564    -9    13
     9.360   10   14.97     170    565    -9    13
     9.360   10   14.99     163    566    -9    13
     9.360   10   14.98     174    567    -9    13
     9.320   10   14.97     172    568    -9    13
     9.320   10   14.97     175    569    -9    13
     9.320   10   14.98     165    570    -9    13
     9.360   10   14.97     169    570    -9    13
     9.320   10   14.97     169    571    -9    13
     9.360   10   14.97     176    572    -9    13
     9.320   10   14.98     162    573    -9    13
     9.360   10   14.98     163    574    -9    13
     9.360   10   14.97     168    575    -9    13
     9.360   10   14.96     170    576    -9    13
     9.360   10   14.97     168    576    -9    13
     9.320   10   14.96     165    578    -9    13
     9.360   10   14.96     175    578    -9    13
     9.360   10   14.96     169    579    -9    13
     9.360   10   14.97     164    580    -9    13
     9.320   10   14.96     169    581    -9    13
     9.320   10   14.96     166    582    -9    13
     9.360   10   14.96     164    583    -9    13
     9.360   10   14.96     174    583    -9    13
     9.280   10   14.95     166    584    -9    13
     9.280   10   14.95     166    585    -9    13
     9.200   10   14.96     163    586    -9    13
     9.120   10   14.96     174    586    -9    13
     9.080   10   14.95     165    587    -9    13
     9.000   10   14.95     168    588    -9    13
     7.680   10   14.94     160    589    -9    13
    -0.160   59   14.93     170    647    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    54  2.4    3    20    10   600    31    30     0      5     2       2   15  0.0678
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   54 2024/12/19 05:20:05
# Pcase vacuums
   8.350   5.370
# Oil valve positions
   2.034   0.586   0.793
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   54 2024/12/19 05:54:08 9999/99/99 99:99:99
# Pcase vacuums
   5.370   5.330
# Average 
#   P      T      S
    9.320  17.074  32.743
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   54 2024/12/20 04:00:09 9999/99/99 99:99:99
# Pcase vacuums
   5.330   5.880
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   54 2024/12/20 05:04:55
# number of packets previous packets
     12              16
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   5.870   8.160
# Air Pump max current time  max pressure
      43       570     7.526
# Air Valve Positions
   1.279   0.000
# Air Valve Calibrations
   1.229   0.154   0.691
# Oil Valve Positions
   2.046   0.607
# Oil Valve Calibrations
   1.965   0.659
# Encountered Error Codes
#    Cycle Errors
EE20    54      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   54  1      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.160    205   5.360
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   54       197    2024/12/20 05:17:07
Parameter Table Param_Table-V18.txt |
     IFV = 10.9.1.34	(IFV) |
     VER =       18	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       54	(cycle number) |
     MPS =        1	(phase) |
     MPC =        5	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        1	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        7	(dBar) |
     CRS =     1000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    21600	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PNB =    43668	(V) |
     PNC =    50958	(inHg) |
     PXC =        0	(inHg) |
     PBL =    25960	(12V cal) |
     PBH =     1790	(14V cal) |
     PP0 =     2089	(oil pump 0 mA cal) |
     PP1 =    61614	(oil pump 1000 mA cal) |
     PVL =    32000	(V12 12V cal) |
     PVH =        0	(V12 14V cal) |
     PV0 =    16600	(vac 0 inHg cal) |
     PV5 =        0	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =     1280	(V12_sig 1000mA cal) |
     ICM =        5	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        1	(ice month mask) |
  ICA[0] =        1	(summer attempts) |
  ICA[1] =       70	(winter attempts) |
  ICB[0] =       70	(summer cycles) |
  ICB[1] =      180	(winter cycles) |
  IMT[0] =      180	(summer message threshold) |
  IMT[1] =       20	(winter message threshold) |
  ILT[0] =       40	(summer time threshold) |
  ILT[1] =    65371	(winter time threshold) |
     ICN =       30	(shallow limit for test dBar) |
     ICX =      514	(deep limit for test dBar) |
     ICT =     2.56	(deg C) |
     ICS =     5120	(sec) |
  ISS[0] =      180	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        0	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     2.56	(PSU) |
     ISO =    25600	(sec) |
     ISP =      100	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =       30	(summer descent depth dBar) |
  ICD[1] =        0	(winter descent depth dBar) |
  ICR[0] =  1966080	(summer minutes) |
  ICR[1] = 16779780	(winter minutes) |
  ICG[0] =      176	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =       30	(attempts) |
     ISD =     1200	(dBar) |
     IAT =       45	(secs) |
     ISV =       45	(secs) |
     ISW =        0	(secs) |
  ICO[0] =        0	(summer secs) |
  ICO[1] =        0	(winter secs) |
     ICF =       51	(summer mask) |
     ICW =  1310723	(winter ice mask) |
     DNS =        3	(cycle number) |
     DRA =        3	(target descent rate dBar/sec) |
  DSD[0] =     1800	(dBar) |
  DSD[1] =     1800	(dBar) |
  DSD[2] =     1800	(dBar) |
  DST[0] =      300	(sec) |
  DST[1] =        0	(sec) |
  DST[2] =        0	(sec) |
     DDF =     2000	(sec) |
     DDC =        1	(sec) |
     DSI = 19660820	(ms) |
     DDD =       20	(dBar) |
     DDT =       20	(sec) |
  DFT[0] =        1	(sec) |
  DFT[1] =     1800	(sec) |
  DFT[2] =     1800	(sec) |
     DCM =        7	(0=dis 1=en) |
     DAL =        8	(0=cal 1=seek) |
  DCT[0] =      512	(sec) |
  DCT[1] =      768	(sec) |
  DCT[2] =        3	(sec) |
     DCA =      252	(max attempts) |
     DRC =        2	(cal rate calculation) |
     DRT =        2	(sec) |
     DCX =      200	(sec) |
     DCP =        0	(sec) |
     DCF =      900	(sec) |
     DCD =       50	(dBar) |
     DCR =       10	(dBar/sec) |
     DSW =       10	(sec) |
     DUC =        5	(under damping factor) |
     DPT =       30	(sec) |
     DUT =        5	(sec) |
     DPD =      900	(sec) |
     DVT =       19	(sec) |
     DVD =       16	(dBar) |
     DVW =        3	(sec) |
     PFR =       60	(cc/min) |
     PFC =       51	(.001 cc/dBar) |
     DDP =        0	(dBar) |
     DDS =    10800	(sec) |
     PNS =        0	(cycle) |
  PTO[0] =        2	(sec) |
  PTO[1] = -1560281088	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =  1048576	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =      256	(24 LSB) |
     PTD = 460800000	(sec) |
     PSA =        0	(0=avg 1=avg+raw) |
     PSI = 16777479	(ms) |
  PFM[0] =        8	(0=dis 1=en) |
  PFM[1] =       42	(0=dis 1=en) |
  PFM[2] =       48	(0=dis 1=en) |
  PFT[0] =      500	(sec) |
  PFT[1] =       20	(sec) |
  PFT[2] =     2000	(sec) |
  PFD[0] =        1	(dBar) |
  PFD[1] =        5	(dBar) |
  PFD[2] =        5	(dBar) |
  PFV[0] =     1800	(sec) |
  PFV[1] =      256	(sec) |
  PFV[2] =      200	(sec) |
     PDT =    10800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =     7682	(dBar) |
     PST =    22535	(sec) |
     PSV =       51	(sec) |
     PSW =        5	(sec) |
     PSS =        9	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =    24576	(cycle number) |
     ARR =     5120	(target rise rate cm/sec) |
     APM =     2560	(sec) |
     APS =    12800	(sec) |
     APN =       39	(sec) |
     APF =     4096	(sec) |
     ASI = 16777473	(ms) |
     ASD =    11268	(dBar) |
     ADD =    19459	(dBar) |
     ADT =    59392	(sec) |
     ARC =      512	(?) |
     AHR =    13056	(cm/sec) |
     AHD =        0	(dBar) |
     SFN =     3840	(cycle number) |
     SDC =    15360	(dBar) |
     SGT =      300	(min) |
     SET =        0	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =       39	(sec) |
     SSA =     4096	(0=dis 1=en) |
     SSI = 167845196	(ms) |
     SDS =     6500	(sec) |
     SAM =        4	(0=dis 1=en) |
     SAX =    14338	(.001 psi) |
     SAN =    22528	(.001 psi) |
     SAO =    15360	(sec) |
     SPT =        0	(sec) |
     SSP =    15360	(sec) |

