# Message Decoder for Solo2, Alto and Deep Solo Floats; v241223
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11074    53   88 000000000000009   375     0  11 2024/12/19, 05:15:24,  41.5334  -70.6398   5      0 14
SBD 11074    53  339 000000000000009   376     0  10 2024/12/19, 05:15:47,  41.5334  -70.6398   4      0 13
SBD 11074    53  332 000000000000009   377     0   9 2024/12/19, 05:16:04,  41.5413  -70.6851   3      0 8,15,13
SBD 11074    53  335 000000000000009   378     0   8 2024/12/19, 05:16:26,  41.5413  -70.6851   3      0 8
SBD 11074    53  326 000000000000009   379     0   7 2024/12/19, 05:16:39,  41.5334  -70.6398   3      0 5,14,0,1,2,3,6,14
SBD 11074    53  335 000000000000009   380     0   6 2024/12/19, 05:17:55,  41.5413  -70.6851   3      0 6
SBD 11074    53  335 000000000000009   381     0   5 2024/12/19, 05:18:24,  41.5683  -70.6851   3      0 6
SBD 11074    53  340 000000000000009   382     0   4 2024/12/19, 05:18:43,  41.5603  -70.6398   4      0 12,12,12,14,9,10,7
SBD 11074    53  295 000000000000009   383     0   3 2024/12/19, 05:18:55,  41.5334  -70.6398   4      0 12,12,12,12,12,12
SBD 11074    53  295 000000000000009   384     0   2 2024/12/19, 05:19:13,  41.5334  -70.6398   5      0 12,12,12,12,12,12
SBD 11074    53  295 000000000000009   385     0   1 2024/12/19, 05:19:30,  41.5334  -70.6398   5      0 12,12,12,12,12,12
SBD 11074    53  323 000000000000009   386     0   0 2024/12/19, 05:19:49,  41.5683  -70.6851   7      0 0,4,14,12,12,12,11
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      48         3494          144        3638
# Sensor_ID bytes     % packets Description
          0    48  1.37       2         GPS
          1    29  0.83       1    Pressure
          2    32  0.92       1 Temperature
          3    32  0.92       1    Salinity
          4    88  2.52       1        Fall
          5   118  3.38       1        Rise
          6   694 19.86       3        Pump
          7    46  1.32       1 Drift_times
          8   502 14.37       2     Surface
          9    46  1.32       1  HR/Drift_P
         10    54  1.55       1  HR/Drift_T
         11    54  1.55       1  HR/Drift_S
         12  1132 32.40      24  Aux Sensor
         13   402 11.51       2  Parameters
         14   151  4.32       5 Engineering
         15    66  1.89       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   53 1 2024/12/18 05:18 +41.53430  -70.64655    0  7  0  0  0  1.2     0
G   53 2 2024/12/19 05:04 +41.53427  -70.64679   30  4  0  0  0  2.1     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    53    9    1     3         1
#      Pres      Temp    Salinity
      1.000    17.3900    32.7590
      2.080    17.2410    32.7360
      3.040    17.1990    32.7340
      4.000    17.1600    32.7340
      5.000    17.1150    32.7350
      6.040    17.0730    32.7340
      7.040    17.0430    32.7330
      8.080    17.0150    32.7330
      9.200    16.9130    32.7310
# Drift Data
#      C Npts comp n_var
I0    53   44    1     4
#      Pres       Temp   Salinity     Time
      9.360    16.9690    32.7230        6
      9.320    16.9610    32.7230     1811
      9.360    16.9510    32.7230     3617
      9.360    17.0630    32.7230     5423
      9.320    17.1040    32.7250     7228
      9.360    17.1320    32.7320     9034
      9.360    17.1320    32.7300    10839
      9.360    17.1420    32.7300    12645
      9.360    17.1580    32.7310    14451
      9.360    17.1710    32.7310    16256
      9.360    17.1780    32.7320    18062
      9.360    17.1910    32.7320    19867
      9.360    17.2050    32.7310    21673
      9.360    17.2140    32.7320    23479
      9.360    17.2280    32.7330    25284
      9.360    17.2340    32.7320    27090
      9.360    17.2230    32.7340    28895
      9.360    17.2140    32.7330    30701
      9.360    17.2040    32.7320    32507
      9.360    17.1930    32.7320    34312
      9.360    17.1800    32.7340    36118
      9.360    17.1710    32.7330    37923
      9.360    17.1610    32.7320    39729
      9.320    17.1510    32.7320    41535
      9.360    17.1390    32.7330    43340
      9.320    17.1280    32.7330    45146
      9.320    17.1180    32.7320    46951
      9.320    17.1070    32.7320    48757
      9.320    17.0970    32.7330    50563
      9.320    17.0870    32.7320    52368
      9.320    17.0760    32.7330    54174
      9.280    17.0670    32.7320    55979
      9.320    17.0560    32.7330    57785
      9.280    17.0470    32.7320    59591
      9.280    17.0350    32.7330    61396
      9.280    17.0260    32.7310    63202
      9.280    17.0140    32.7320    65007
      9.280    17.0040    32.7320    66813
      9.240    16.9930    32.7320    68619
      9.280    16.9820    32.7320    70424
      9.240    16.9720    32.7310    72230
      9.240    16.9610    32.7310    74035
      9.240    16.9520    32.7310    75841
      9.240    16.9420    32.7300    77647
# Surface Pressure Time Series
#    C npts n_var               Start
S   53    54     6 2024/12/19 05:04:42
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0     -0.020     17.451   -999.000      14 2024/12/19 05:04:42
    13     -0.020     17.364   -999.000      14 2024/12/19 05:04:55
    24     -0.040     17.329   -999.000      14 2024/12/19 05:05:06
    35     -0.030     17.302   -999.000      14 2024/12/19 05:05:17
    46     -0.030     17.286   -999.000      14 2024/12/19 05:05:28
    58     -0.010     17.194   -999.000      14 2024/12/19 05:05:40
    68     -0.020     17.151   -999.000      14 2024/12/19 05:05:50
    80     -0.030     17.121   -999.000      14 2024/12/19 05:06:02
    91     -0.040     17.112   -999.000      14 2024/12/19 05:06:13
   102     -0.040     17.108   -999.000      14 2024/12/19 05:06:24
   113     -0.040     16.984   -999.000      14 2024/12/19 05:06:35
   124     -0.040     16.887   -999.000      14 2024/12/19 05:06:46
   135     -0.040     16.809   -999.000      14 2024/12/19 05:06:57
   146     -0.060     16.757   -999.000      14 2024/12/19 05:07:08
   157     -0.070     16.712   -999.000      14 2024/12/19 05:07:19
   168     -0.050     16.665   -999.000      14 2024/12/19 05:07:30
   179     -0.050     16.618   -999.000      14 2024/12/19 05:07:41
   190     -0.060     16.575   -999.000      14 2024/12/19 05:07:52
   201     -0.050     16.515   -999.000      14 2024/12/19 05:08:03
   213     -0.050     16.423   -999.000      14 2024/12/19 05:08:15
   224     -0.070     16.346   -999.000      14 2024/12/19 05:08:26
   235     -0.080     16.286   -999.000      14 2024/12/19 05:08:37
   246     -0.060     16.235   -999.000      14 2024/12/19 05:08:48
   257     -0.070     16.186   -999.000      14 2024/12/19 05:08:59
   268     -0.080     16.159   -999.000      14 2024/12/19 05:09:10
   279     -0.080     16.138   -999.000      14 2024/12/19 05:09:21
   290     -0.070     16.062   -999.000      14 2024/12/19 05:09:32
   301     -0.090     15.989   -999.000      14 2024/12/19 05:09:43
   312     -0.090     15.944   -999.000      14 2024/12/19 05:09:54
   323     -0.090     15.935   -999.000      14 2024/12/19 05:10:05
   335     -0.090     15.915   -999.000      14 2024/12/19 05:10:17
   346     -0.100     15.903   -999.000      14 2024/12/19 05:10:28
   357     -0.090     15.872   -999.000      14 2024/12/19 05:10:39
   368     -0.110     15.846   -999.000      14 2024/12/19 05:10:50
   379     -0.110     15.790   -999.000      14 2024/12/19 05:11:01
     0     -0.110     15.765   -999.000      14 2024/12/19 05:11:12
    11     -0.120     15.747   -999.000      14 2024/12/19 05:11:23
    22     -0.120     15.744   -999.000      14 2024/12/19 05:11:34
    33     -0.130     15.711   -999.000      14 2024/12/19 05:11:45
    44     -0.110     15.700   -999.000      14 2024/12/19 05:11:56
    55     -0.140     15.674   -999.000      14 2024/12/19 05:12:07
    67     -0.150     15.635   -999.000      14 2024/12/19 05:12:19
    78     -0.130     15.608   -999.000      14 2024/12/19 05:12:30
    89     -0.160     15.578   -999.000      14 2024/12/19 05:12:41
   100     -0.150     15.516   -999.000      14 2024/12/19 05:12:52
   111     -0.140     15.439   -999.000      14 2024/12/19 05:13:03
   122     -0.140     15.407   -999.000      14 2024/12/19 05:13:14
   133     -0.150     15.396   -999.000      14 2024/12/19 05:13:25
   144     -0.160     15.342   -999.000      14 2024/12/19 05:13:36
   155     -0.160     15.298   -999.000      14 2024/12/19 05:13:47
   166     -0.150     15.267   -999.000      14 2024/12/19 05:13:58
   177     -0.150     15.236   -999.000      14 2024/12/19 05:14:09
   189     -0.150     15.217   -999.000      14 2024/12/19 05:14:21
   200     -0.160     15.225   -999.000      14 2024/12/19 05:14:32
# Fall Time Series
#  Cycle npts     Start                Date
F00   53   16 787814316 2024/12/18 05:18:36
# time     Pres phase                Date
     0   -0.040     1 2024/12/18 05:18:36
   175    1.040    10 2024/12/18 05:21:31
   183    2.480    10 2024/12/18 05:21:39
   187    3.440    10 2024/12/18 05:21:43
   193    5.000     2 2024/12/18 05:21:49
   195    5.680    10 2024/12/18 05:21:51
   200    6.960    10 2024/12/18 05:21:56
   204    8.280    10 2024/12/18 05:22:00
   208    9.320    10 2024/12/18 05:22:04
   508    9.360    10 2024/12/18 05:27:04
   809    9.360    10 2024/12/18 05:32:05
  1110    9.360    10 2024/12/18 05:37:06
  1411    9.360    10 2024/12/18 05:42:07
  1713    9.320    10 2024/12/18 05:47:09
  2014    9.360    10 2024/12/18 05:52:10
  2022    9.320    10 2024/12/18 05:52:18
# Rise Time Series
#  Cycle npts     Start                Date
R00   53   22 787896009 2024/12/19 04:00:09
# time     Pres phase                Date
     0    9.200     7 2024/12/19 04:00:09
   316    9.240    13 2024/12/19 04:05:25
   621    9.200    13 2024/12/19 04:10:30
   925    9.200    13 2024/12/19 04:15:34
  1230    9.200    13 2024/12/19 04:20:39
  1534    9.240    13 2024/12/19 04:25:43
  1839    9.240    13 2024/12/19 04:30:48
  2144    9.240    13 2024/12/19 04:35:53
  2448    9.240    13 2024/12/19 04:40:57
  2753    9.240    13 2024/12/19 04:46:02
  3057    9.240    13 2024/12/19 04:51:06
  3362    9.120    13 2024/12/19 04:56:11
  3540    9.000    13 2024/12/19 04:59:09
  3561    9.000    13 2024/12/19 04:59:30
  3572    8.960    13 2024/12/19 04:59:41
  3729    7.840    13 2024/12/19 05:02:18
  3750    6.560    13 2024/12/19 05:02:39
  3761    5.600    13 2024/12/19 05:02:50
  3771    4.240    13 2024/12/19 05:03:00
  3782    2.760    13 2024/12/19 05:03:11
  3792    0.920    13 2024/12/19 05:03:21
  3793    0.920     8 2024/12/19 05:03:22
# Pump Time Series
#   Cycle npts
B02    53   62
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     5.000   22   14.93     173    537    -9     2
     9.200   22   15.09     193    535    -9     7
     9.200   11   15.09     190    536    -9    13
     9.200   11   15.09     181    537    -9    13
     9.240   11   15.07     192    538    -9    13
     9.240   11   15.07     180    539    -9    13
     9.200   11   15.09     170    540    -9    13
     9.240   11   15.09     169    541    -9    13
     9.240   11   15.08     171    542    -9    13
     9.200   11   15.07     171    543    -9    13
     9.200   11   15.06     182    544    -9    13
     9.240   11   15.08     168    545    -9    13
     9.240   11   15.06     173    546    -9    13
     9.240   11   15.07     171    547    -9    13
     9.240   11   15.06     165    548    -9    13
     9.240   11   15.05     170    549    -9    13
     9.240   11   15.05     174    550    -9    13
     9.200   11   15.06     166    551    -9    13
     9.240   11   15.04     172    551    -9    13
     9.200   11   15.05     168    552    -9    13
     9.240   11   15.04     170    553    -9    13
     9.200   11   15.03     171    554    -9    13
     9.200   11   15.03     171    555    -9    13
     9.240   11   15.04     165    556    -9    13
     9.200   11   15.04     164    556    -9    13
     9.240   11   15.03     170    557    -9    13
     9.240   11   15.02     177    559    -9    13
     9.240   11   15.02     180    560    -9    13
     9.200   11   15.02     177    560    -9    13
     9.200   11   15.00     161    561    -9    13
     9.240   10   15.00     167    562    -9    13
     9.240   10   15.02     168    563    -9    13
     9.200   10   15.01     165    564    -9    13
     9.200   10   15.02     179    564    -9    13
     9.200   10   15.01     162    566    -9    13
     9.240   10   15.00     162    566    -9    13
     9.200   10   15.02     167    567    -9    13
     9.240   10   15.00     172    568    -9    13
     9.200   10   15.01     161    569    -9    13
     9.240   10   14.99     174    570    -9    13
     9.200   10   15.01     161    571    -9    13
     9.200   10   14.99     170    571    -9    13
     9.240   10   15.00     159    572    -9    13
     9.200   10   14.98     164    573    -9    13
     9.200   10   15.00     166    574    -9    13
     9.200   10   14.99     160    575    -9    13
     9.240   10   14.98     172    576    -9    13
     9.200   10   14.98     173    577    -9    13
     9.200   10   14.99     161    578    -9    13
     9.240   10   14.99     161    578    -9    13
     9.200   10   14.98     169    579    -9    13
     9.200   10   14.99     158    580    -9    13
     9.200   10   14.98     159    581    -9    13
     9.200   10   14.96     170    582    -9    13
     9.120   10   14.98     165    583    -9    13
     9.080   10   14.97     166    583    -9    13
     9.040   10   14.96     170    584    -9    13
     9.000   10   14.96     171    585    -9    13
     8.960   10   14.96     168    586    -9    13
     8.880   10   14.98     158    586    -9    13
     7.320   10   14.97     167    587    -9    13
    -0.040   60   14.93     166    645    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    53  2.4    3    20    10   600    31    30     0      5     2       2   15  0.0678
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   53 2024/12/18 05:18:33
# Pcase vacuums
   8.200   5.370
# Oil valve positions
   2.035   0.587   0.791
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   53 2024/12/18 05:52:19 9999/99/99 99:99:99
# Pcase vacuums
   5.370   5.330
# Average 
#   P      T      S
    9.320  17.098  32.731
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   53 2024/12/19 04:00:09 9999/99/99 99:99:99
# Pcase vacuums
   5.330   5.860
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   53 2024/12/19 05:03:21
# number of packets previous packets
     11              15
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   5.860   8.360
# Air Pump max current time  max pressure
      43       655     7.506
# Air Valve Positions
   1.278   0.000
# Air Valve Calibrations
   1.229   0.154   0.691
# Oil Valve Positions
   2.046   0.608
# Oil Valve Calibrations
   1.965   0.659
# Encountered Error Codes
#    Cycle Errors
EE20    53      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   53  1      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.040    188   5.360
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   53       197    2024/12/19 05:16:04
Parameter Table Param_Table-V18.txt |
     IFV = 10.9.1.34	(IFV) |
     VER =       18	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       53	(cycle number) |
     MPS =        1	(phase) |
     MPC =        4	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =      999	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        1	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        7	(dBar) |
     CRS =     1000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    21600	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        0	(sec) |
    MAT2 =        0	(sec) |
     MAP =        0	(sec) |
     ACA =        0	(gain) |
     ACG =        0	(scale) |
     ACR =        0	(sampling rate) |
     PNB =    43668	(V) |
     PNC =    50958	(inHg) |
     PXC =        0	(inHg) |
     PBL =    25960	(12V cal) |
     PBH =     1790	(14V cal) |
     PP0 =     2089	(oil pump 0 mA cal) |
     PP1 =    61614	(oil pump 1000 mA cal) |
     PVL =    32000	(V12 12V cal) |
     PVH =        0	(V12 14V cal) |
     PV0 =    16600	(vac 0 inHg cal) |
     PV5 =        0	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =     1280	(V12_sig 1000mA cal) |
     ICM =        5	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        1	(ice month mask) |
  ICA[0] =        1	(summer attempts) |
  ICA[1] =       70	(winter attempts) |
  ICB[0] =       70	(summer cycles) |
  ICB[1] =      180	(winter cycles) |
  IMT[0] =      180	(summer message threshold) |
  IMT[1] =       20	(winter message threshold) |
  ILT[0] =       40	(summer time threshold) |
  ILT[1] =    65371	(winter time threshold) |
     ICN =       30	(shallow limit for test dBar) |
     ICX =      514	(deep limit for test dBar) |
     ICT =     2.56	(deg C) |
     ICS =     5120	(sec) |
  ISS[0] =      180	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        0	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     2.56	(PSU) |
     ISO =    25600	(sec) |
     ISP =      100	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =       30	(summer descent depth dBar) |
  ICD[1] =        0	(winter descent depth dBar) |
  ICR[0] =  1966080	(summer minutes) |
  ICR[1] = 16779780	(winter minutes) |
  ICG[0] =      176	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =       30	(attempts) |
     ISD =     1200	(dBar) |
     IAT =       45	(secs) |
     ISV =       45	(secs) |
     ISW =        0	(secs) |
  ICO[0] =        0	(summer secs) |
  ICO[1] =        0	(winter secs) |
     ICF =       50	(summer mask) |
     ICW =  1310723	(winter ice mask) |
     DNS =        3	(cycle number) |
     DRA =        3	(target descent rate dBar/sec) |
  DSD[0] =     1800	(dBar) |
  DSD[1] =     1800	(dBar) |
  DSD[2] =     1800	(dBar) |
  DST[0] =      300	(sec) |
  DST[1] =        0	(sec) |
  DST[2] =        0	(sec) |
     DDF =     2000	(sec) |
     DDC =        1	(sec) |
     DSI = 19660820	(ms) |
     DDD =       20	(dBar) |
     DDT =       20	(sec) |
  DFT[0] =        1	(sec) |
  DFT[1] =     1800	(sec) |
  DFT[2] =     1800	(sec) |
     DCM =        7	(0=dis 1=en) |
     DAL =        8	(0=cal 1=seek) |
  DCT[0] =      512	(sec) |
  DCT[1] =      768	(sec) |
  DCT[2] =        3	(sec) |
     DCA =      252	(max attempts) |
     DRC =        2	(cal rate calculation) |
     DRT =        2	(sec) |
     DCX =      200	(sec) |
     DCP =        0	(sec) |
     DCF =      900	(sec) |
     DCD =       50	(dBar) |
     DCR =       10	(dBar/sec) |
     DSW =       10	(sec) |
     DUC =        5	(under damping factor) |
     DPT =       30	(sec) |
     DUT =        5	(sec) |
     DPD =      900	(sec) |
     DVT =       19	(sec) |
     DVD =       16	(dBar) |
     DVW =        3	(sec) |
     PFR =       60	(cc/min) |
     PFC =       50	(.001 cc/dBar) |
     DDP =        0	(dBar) |
     DDS =    10800	(sec) |
     PNS =        0	(cycle) |
  PTO[0] =        2	(sec) |
  PTO[1] = -1560281088	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =  1048576	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =      256	(24 LSB) |
     PTD = 460800000	(sec) |
     PSA =        0	(0=avg 1=avg+raw) |
     PSI = 16777479	(ms) |
  PFM[0] =        8	(0=dis 1=en) |
  PFM[1] =       42	(0=dis 1=en) |
  PFM[2] =       48	(0=dis 1=en) |
  PFT[0] =      500	(sec) |
  PFT[1] =       20	(sec) |
  PFT[2] =     2000	(sec) |
  PFD[0] =        1	(dBar) |
  PFD[1] =        5	(dBar) |
  PFD[2] =        5	(dBar) |
  PFV[0] =     1800	(sec) |
  PFV[1] =      256	(sec) |
  PFV[2] =      200	(sec) |
     PDT =    10800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =     7682	(dBar) |
     PST =    22535	(sec) |
     PSV =       50	(sec) |
     PSW =        5	(sec) |
     PSS =        9	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =    24576	(cycle number) |
     ARR =     5120	(target rise rate cm/sec) |
     APM =     2560	(sec) |
     APS =    12800	(sec) |
     APN =       39	(sec) |
     APF =     4096	(sec) |
     ASI = 16777473	(ms) |
     ASD =    11268	(dBar) |
     ADD =    19459	(dBar) |
     ADT =    59392	(sec) |
     ARC =      512	(?) |
     AHR =    12800	(cm/sec) |
     AHD =        0	(dBar) |
     SFN =     3840	(cycle number) |
     SDC =    15360	(dBar) |
     SGT =      300	(min) |
     SET =        0	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =       39	(sec) |
     SSA =     4096	(0=dis 1=en) |
     SSI = 167845196	(ms) |
     SDS =     6500	(sec) |
     SAM =        4	(0=dis 1=en) |
     SAX =    14338	(.001 psi) |
     SAN =    22528	(.001 psi) |
     SAO =    15360	(sec) |
     SPT =        0	(sec) |
     SSP =    15360	(sec) |

