# Message Decoder for Solo2, Alto and Deep Solo Floats; v210818
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11070   133  112 000000000000009  1138     0   7 2020/02/03, 14:26:50, -67.6806  176.5061  13      0 14
SBD 11070   133  279 000000000000009  1139     0   6 2020/02/03, 14:27:00, -67.6785  176.4119  12      0 13,6,14
SBD 11070   133  340 000000000000009  1140     0   5 2020/02/03, 14:27:16, -67.6806  176.5061  10      0 13
SBD 11070   133  323 000000000000009  1141     0   4 2020/02/03, 14:27:24, -67.6806  176.5061  10      0 12,12,12,15
SBD 11070   133  324 000000000000009  1142     0   3 2020/02/03, 14:27:32, -67.6806  176.5061   9      0 3,3
SBD 11070   133  316 000000000000009  1143     0   2 2020/02/03, 14:27:45, -67.6785  176.4119   8      0 2,2
SBD 11070   133  263 000000000000009  1144     0   1 2020/02/03, 14:27:59, -67.6785  176.4119   7      0 1,1,8
SBD 11070   133  319 000000000000009  1145     0   0 2020/02/03, 14:28:13, -68.1840 -177.3931 274      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      23         2180           96        2276
# Sensor_ID bytes     % packets Description
          0    48  2.20       2         GPS
          1   216  9.91       2    Pressure
          2   304 13.94       2 Temperature
          3   312 14.31       2    Salinity
          4    78  3.58       1        Fall
          5   118  5.41       1        Rise
          6   169  7.75       1        Pump
          8    35  1.61       1     Surface
         12   245 11.24       3  Aux Sensor
         13   414 18.99       2  Parameters
         14   175  8.03       5 Engineering
         15    66  3.03       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  133 1 2020/02/03 12:41 -67.66948 +176.41025   10  4 31 36 41  0.0     0
G  133 2 2020/02/03 14:15 -67.66587 +176.40510  150  4 39 43 47  0.1     0
# Warning - Unknown C2 Sensor Data received; channel 0x31
# Warning - Unknown C2 Sensor Data received; channel 0x32
# Warning - Unknown C2 Sensor Data received; channel 0x33
# Profile Data
#      C Npts comp n_var
D0   133  310    1     3
#      Pres      Temp    Salinity
      0.200    -0.5540    26.2120
      1.040    -0.5530    33.8010
      2.000    -0.5530    33.8010
      3.000    -0.5540    33.8020
      4.040    -0.5550    33.8030
      5.040    -0.5540    33.8020
      6.080    -0.5530    33.8010
      7.000    -0.5520    33.8010
      8.000    -0.5520    33.8010
      9.080    -0.5510    33.8000
     10.080    -0.5500    33.8000
     11.000    -0.5500    33.8000
     11.920    -0.5510    33.8000
     13.000    -0.5510    33.8000
     14.080    -0.5510    33.8000
     14.960    -0.5510    33.8000
     15.920    -0.5510    33.7990
     16.920    -0.5500    33.7990
     17.920    -0.5510    33.7990
     18.960    -0.5550    33.8020
     19.960    -0.5540    33.7990
     20.960    -0.5650    33.8060
     22.040    -0.5690    33.8060
     23.080    -0.5750    33.8090
     24.000    -0.5780    33.8060
     24.960    -0.5760    33.8060
     26.040    -0.5880    33.8130
     27.000    -0.6050    33.8180
     27.920    -0.6030    33.8160
     29.000    -0.6100    33.8180
     30.040    -0.6160    33.8200
     31.000    -0.6180    33.8200
     31.920    -0.6220    33.8210
     32.960    -0.6270    33.8210
     34.040    -0.6310    33.8220
     35.080    -0.6330    33.8220
     36.000    -0.6340    33.8210
     36.920    -0.6350    33.8210
     37.960    -0.6390    33.8210
     38.960    -0.6430    33.8210
     40.000    -0.6500    33.8220
     41.080    -0.6610    33.8220
     42.080    -0.6670    33.8210
     43.000    -0.6740    33.8200
     43.920    -0.6880    33.8230
     44.920    -0.7320    33.8350
     45.920    -0.7660    33.8410
     46.960    -0.8250    33.8560
     47.920    -0.8670    33.8620
     48.920    -0.9460    33.8990
     49.960    -1.0410    33.9290
     51.000    -1.1260    33.9550
     52.040    -1.2670    34.0210
     52.960    -1.4100    34.0890
     53.920    -1.5420    34.1540
     55.000    -1.6170    34.1780
     56.120    -1.6720    34.2080
     57.040    -1.7220    34.2300
     58.040    -1.7260    34.2360
     59.040    -1.7260    34.2400
     61.000    -1.7280    34.2480
     62.040    -1.7360    34.2570
     63.040    -1.7400    34.2610
     64.040    -1.7440    34.2650
     65.040    -1.7450    34.2660
     66.040    -1.7460    34.2670
     67.040    -1.7470    34.2670
     68.040    -1.7490    34.2690
     69.040    -1.7530    34.2720
     70.040    -1.7590    34.2740
     71.040    -1.7650    34.2740
     72.040    -1.7650    34.2760
     73.040    -1.7700    34.2770
     74.040    -1.7690    34.2770
     75.040    -1.7790    34.2780
     76.040    -1.7800    34.2790
     77.040    -1.7770    34.2790
     78.040    -1.7750    34.2800
     79.040    -1.7740    34.2810
     80.000    -1.7730    34.2810
     81.000    -1.7720    34.2820
     81.960    -1.7710    34.2830
     82.960    -1.7690    34.2840
     83.920    -1.7690    34.2850
     85.040    -1.7690    34.2850
     86.000    -1.7680    34.2860
     87.000    -1.7680    34.2870
     87.960    -1.7680    34.2880
     88.960    -1.7680    34.2880
     90.040    -1.7680    34.2890
     91.040    -1.7690    34.2890
     92.000    -1.7680    34.2900
     93.080    -1.7670    34.2900
     94.040    -1.7650    34.2920
     94.960    -1.7640    34.2930
     96.080    -1.7620    34.2940
     97.000    -1.7610    34.2950
     97.920    -1.7610    34.2950
     99.000    -1.7600    34.2950
    100.080    -1.7570    34.2960
    101.000    -1.7550    34.2970
    101.960    -1.7540    34.2980
    103.080    -1.7540    34.2980
    104.000    -1.7520    34.2990
    104.920    -1.7500    34.3000
    106.000    -1.7480    34.3000
    107.040    -1.7460    34.3010
    108.040    -1.7440    34.3020
    108.920    -1.7410    34.3030
    110.000    -1.7400    34.3050
    111.080    -1.7380    34.3060
    112.000    -1.7380    34.3070
    112.920    -1.7330    34.3080
    114.000    -1.7230    34.3090
    115.040    -1.7220    34.3100
    116.080    -1.7150    34.3140
    117.000    -1.7030    34.3130
    117.920    -1.6930    34.3170
    118.960    -1.6740    34.3180
    120.000    -1.6530    34.3210
    121.080    -1.6260    34.3250
    122.000    -1.6100    34.3240
    122.920    -1.5640    34.3330
    123.960    -1.4970    34.3280
    125.040    -1.4550    34.3360
    126.000    -1.3140    34.3430
    126.880    -1.2150    34.3430
    127.960    -1.1550    34.3430
    129.000    -1.1360    34.3490
    130.080    -1.0740    34.3680
    131.000    -1.0140    34.3700
    131.880    -0.9640    34.3690
    132.960    -0.9510    34.3770
    134.000    -0.9280    34.3790
    135.040    -0.8070    34.3950
    136.080    -0.7510    34.3890
    136.960    -0.6750    34.3840
    137.880    -0.5770    34.4050
    138.920    -0.5320    34.4040
    139.920    -0.4770    34.4120
    140.960    -0.4460    34.4260
    141.960    -0.4100    34.4280
    143.000    -0.3440    34.4360
    144.000    -0.2960    34.4370
    145.000    -0.2340    34.4400
    146.000    -0.1940    34.4490
    147.000    -0.1600    34.4480
    148.000    -0.1410    34.4600
    149.040    -0.0740    34.4630
    150.040    -0.0420    34.4660
    151.040    -0.0280    34.4640
    152.080    -0.0180    34.0000
    153.080     0.0000    34.4810
    154.080     0.0300    34.4810
    155.120     0.0900    34.4840
    156.000     0.1240    34.4810
    156.880     0.1340    34.4870
    157.920     0.1380    34.4930
    158.920     0.1370    34.4990
    159.960     0.1500    34.5020
    160.960     0.1950    34.5050
    162.000     0.2570    34.5020
    163.040     0.3080    34.5050
    164.080     0.3420    34.5060
    165.000     0.3880    34.5180
    165.920     0.4220    34.5210
    166.960     0.4380    34.5260
    168.000     0.4930    34.5310
    169.000     0.5280    34.5380
    170.040     0.5540    34.5380
    171.040     0.5820    34.5370
    172.040     0.5990    34.5480
    173.080     0.6100    34.5450
    174.120     0.6160    34.5530
    175.120     0.6240    34.5530
    176.040     0.6350    34.5580
    176.960     0.6430    34.5580
    178.000     0.6510    34.5590
    179.040     0.6720    34.5590
    180.080     0.6990    34.5640
    181.120     0.7130    34.5640
    182.000     0.7190    34.5620
    182.920     0.7230    34.5670
    184.000     0.7270    34.5700
    185.040     0.7260    34.5730
    186.080     0.7270    34.5730
    187.120     0.7290    34.5750
    188.040     0.7380    34.5750
    188.920     0.7520    34.5760
    189.960     0.7650    34.5770
    191.000     0.7850    34.5770
    192.040     0.8010    34.5780
    193.080     0.8430    34.5820
    194.000     0.8530    34.5840
    194.880     0.8540    34.5880
    195.880     0.8690    34.5850
    196.920     0.8870    34.5970
    197.880     0.8970    34.5970
    198.920     0.9060    34.6030
    199.920     0.9330    34.6040
    200.920     0.9520    34.6060
    201.920     0.9750    34.6050
    202.960     0.9900    34.6110
    203.960     1.0140    34.6120
    204.960     1.0350    34.6130
    205.920     1.0470    34.6210
    206.920     1.0620    34.6230
    208.040     1.0770    34.6250
    209.000     1.0830    34.6270
    209.960     1.0910    34.6290
    211.080     1.1020    34.6290
    212.040     1.1030    34.6310
    213.000     1.1030    34.6340
    213.920     1.1080    34.6340
    215.000     1.1170    34.6320
    216.040     1.1210    34.6360
    216.960     1.1280    34.6370
    217.920     1.1350    34.6350
    218.960     1.1410    34.6390
    220.040     1.1480    34.6400
    220.960     1.1580    34.6400
    221.920     1.1610    34.6430
    222.960     1.1670    34.6420
    224.040     1.1670    34.6450
    224.960     1.1740    34.6460
    226.040     1.1900    34.6450
    227.080     1.2100    34.6470
    227.960     1.2170    34.6500
    228.960     1.2230    34.6530
    229.960     1.2280    34.6540
    230.960     1.2310    34.6540
    231.960     1.2330    34.6560
    232.960     1.2440    34.6560
    233.920     1.2490    34.6560
    234.920     1.2490    34.6590
    236.040     1.2490    34.6600
    237.000     1.2500    34.6600
    238.000     1.2520    34.6600
    239.080     1.2560    34.6610
    240.040     1.2580    34.6620
    241.000     1.2640    34.6620
    242.000     1.2760    34.6630
    242.920     1.2850    34.6640
    243.960     1.2880    34.6670
    245.000     1.2880    34.6690
    246.000     1.2910    34.6690
    247.040     1.2950    34.6690
    248.040     1.3030    34.6700
    249.040     1.3080    34.6700
    250.040     1.3140    34.6720
    251.000     1.3180    34.6720
    252.000     1.3190    34.6730
    253.040     1.3190    34.6740
    254.040     1.3210    34.6740
    255.040     1.3270    34.6740
    256.040     1.3290    34.6760
    257.040     1.3290    34.6770
    258.000     1.3290    34.6770
    259.000     1.3390    34.6780
    260.040     1.3470    34.6780
    260.960     1.3490    34.6800
    261.960     1.3540    34.6810
    263.000     1.3600    34.6810
    264.000     1.3600    34.6820
    265.000     1.3650    34.6830
    266.000     1.3650    34.6830
    267.000     1.3650    34.6840
    267.960     1.3650    34.6840
    269.080     1.3650    34.6840
    270.000     1.3660    34.6840
    271.960     1.3690    34.6850
    272.960     1.3720    34.6850
    274.000     1.3730    34.6860
    275.080     1.3730    34.6860
    276.040     1.3770    34.6870
    277.040     1.3840    34.6870
    278.000     1.3860    34.6880
    279.000     1.3880    34.6890
    280.040     1.3890    34.6900
    281.040     1.3920    34.6900
    282.040     1.3950    34.6910
    283.040     1.3980    34.6910
    284.040     1.3990    34.0000
    285.000     1.4010    34.6920
    286.000     1.4020    34.6930
    286.960     1.4040    34.6930
    287.960     1.4040    34.6940
    289.000     1.4040    34.6940
    290.000     1.4050    34.6940
    291.040     1.4060    34.6940
    292.000     1.4060    34.6940
    293.000     1.4060    34.6950
    294.000     1.4070    34.6950
    295.000     1.4080    34.6950
    296.000     1.4100    34.6950
    297.000     1.4110    34.6960
    298.040     1.4130    34.6960
    299.000     1.4160    34.6970
    300.000  -999.0000    34.6970
    301.280  -999.0000  -999.0000
   -999.000  -999.0000  -999.0000
   -999.000  -999.0000  -999.0000
   -999.000  -999.0000  -999.0000
   -999.000  -999.0000  -999.0000
   -999.000  -999.0000  -999.0000
    301.280     1.4160    34.6970
    300.000     1.4100    34.6960
    301.280     1.4110    34.6970
   -999.000     1.4130    34.6970
   -999.000     1.4160  -999.0000
# Surface Pressure Time Series
#    C npts n_var               Start
S  133     3     9 2020/02/03 13:52:02
# time   Pressure       Temp       Psal Mission                Date O2_Molarity O2_Temp O2_Phase
#  sec       dBar       degC        PSU   phase                 UTC      umol/L    degC  degrees
   125     -0.010     -0.660   -999.000      14 2020/02/03 13:54:07      -9.000  -9.000   -9.000
  1225     -0.040     -0.607   -999.000      14 2020/02/03 14:12:27      -9.000  -9.000   -9.000
  1606     -0.090     -0.621   -999.000      14 2020/02/03 14:18:48      -9.000  -9.000   -9.000
# Fall Time Series
#  Cycle npts     Start                Date
F00  133   14 634048928 2020/02/03 12:42:08
# time     Pres phase                Date
     0   -0.080     1 2020/02/03 12:42:08
   611   54.240    10 2020/02/03 12:52:19
   912  120.760    10 2020/02/03 12:57:20
  1216  184.400     2 2020/02/03 13:02:24
  1219  185.080    10 2020/02/03 13:02:27
  1524  231.200    10 2020/02/03 13:07:32
  1690  241.000    10 2020/02/03 13:10:18
  1866  251.160    10 2020/02/03 13:13:14
  2042  261.080    10 2020/02/03 13:16:10
  2221  271.120    10 2020/02/03 13:19:09
  2403  281.120    10 2020/02/03 13:22:11
  2588  291.040    10 2020/02/03 13:25:16
  2706 9999.000    10 2020/02/03 13:27:14
  2710  297.440    10 2020/02/03 13:27:18
# Rise Time Series
#  Cycle npts     Start                Date
R00  133   22 634051649 2020/02/03 13:27:29
# time     Pres phase                Date
     0  298.040     7 2020/02/03 13:27:29
   264  286.000    13 2020/02/03 13:31:53
   335  275.160    13 2020/02/03 13:33:04
   406  262.800    13 2020/02/03 13:34:15
   477  249.200    13 2020/02/03 13:35:26
   547  234.280    13 2020/02/03 13:36:36
   618  218.040    13 2020/02/03 13:37:47
   689  200.800    13 2020/02/03 13:38:58
   759  182.800    13 2020/02/03 13:40:08
   830  164.480    13 2020/02/03 13:41:19
   900  146.600    13 2020/02/03 13:42:29
   971  128.360    13 2020/02/03 13:43:40
  1041  109.880    13 2020/02/03 13:44:50
  1111   91.040    13 2020/02/03 13:46:00
  1147   81.280    13 2020/02/03 13:46:36
  1182   71.040    13 2020/02/03 13:47:11
  1217   61.040    13 2020/02/03 13:47:46
  1252   51.360    13 2020/02/03 13:48:21
  1323   33.360    13 2020/02/03 13:49:32
  1358   23.760    13 2020/02/03 13:50:07
  1428    5.640    13 2020/02/03 13:51:17
  1472    0.040     8 2020/02/03 13:52:01
# Pump Time Series
#   Cycle npts
B02   133   15
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
   184.400  302   14.82     267    613    -9     2
   298.040   17   14.87     330    607    -9     7
   300.240   36   14.86     316    610    -9    13
   300.560   36   14.85     308    613    -9    13
   294.840   35   14.84     296    617    -9    13
   286.000   35   14.85     311    620    -9    13
   275.160   36   14.83     290    623    -9    13
   262.800   35   14.83     287    627    -9    13
   249.200   35   14.82     269    630    -9    13
   234.280   36   14.82     271    633    -9    13
   218.040   36   14.81     251    637    -9    13
   200.800   35   14.82     259    640    -9    13
   146.600   36   14.82     233    646    -9    13
    42.480   36   14.84     200    651    -9    13
     0.160  121   14.83     180    660    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   133  2.4    8   500  1500   600   311    60     0     10    12    1245   20  0.4966
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  133 2020/02/03 12:42:04
# Pcase vacuums
   6.880   6.070
# Oil valve positions
   1.691   0.737   0.656
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  133 2020/02/03 13:27:19 9999/99/99 99:99:99
# Pcase vacuums
   6.070   6.070
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  133 2020/02/03 13:27:29 9999/99/99 99:99:99
# Pcase vacuums
   6.070   6.510
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  133 2020/02/03 13:52:02
# number of packets previous packets
      7               8
# CTD Status pres offset
 0x4e410000   0.370
# Pcase vacuums
   6.510   6.880
# Air Pump max current time  max pressure
       0      1094     6.643
# Air Valve Positions
   1.120   0.656
# Air Valve Calibrations
   1.041   0.099   0.570
# Oil Valve Positions
   1.688   0.734
# Oil Valve Calibrations
   1.634   0.786
# Encountered Error Codes
#    Cycle Errors
EE20   133      4
#               Date   Code Description
 2020/02/03 13:27:15 0x0913 DescendMaxDepth
 2020/02/03 14:25:27 0x0116 SurfaceCtd
 2020/02/03 14:25:29 0x0116 SurfaceCtd
 2020/02/03 14:25:30 0x0A16 SurfaceData
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  133  1      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.120   1210   5.900
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  133       198    2020/02/03 14:27:16
Parameter Table Param_Table-V17.txt |
     IFV = 6.8.116.IFV	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      132	(cycle number) |
     MPS =        0	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =        0	(count) |
  MNM[2] =        0	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        0	(count) |
  MNS[2] =        0	(count) |
  DTD[0] =     1500	(dBar) |
  DTD[1] =      400	(dBar) |
  DTD[2] =      400	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =      300	(dBar) |
  CPB[3] =      100	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        1	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        1	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       50	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =    30000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1800	(V) |
     PNC =       10	(inHg) |
     PXC =       50	(inHg) |
     PBL =       20	(12V cal) |
     PBH =     1200	(14V cal) |
     PP0 =      400	(oil pump 0 mA cal) |
     PP1 =     1500	(oil pump 1000 mA cal) |
     PVL =    43668	(V12 12V cal) |
     PVH =    50958	(V12 14V cal) |
     PV0 =        0	(vac 0 inHg cal) |
     PV5 =    25960	(vac 5 inHg cal) |
     PC0 =     1790	(V12_sig 0mA cal) |
     PC1 =     2089	(V12_sig 1000mA cal) |
     ICM =      240	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =    44669	(ice month mask) |
  ICA[0] =        0	(summer attempts) |
  ICA[1] =       64	(winter attempts) |
  ICB[0] =    55296	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =        5	(summer message threshold) |
  IMT[1] =        5	(winter message threshold) |
  ILT[0] =        1	(summer time threshold) |
  ILT[1] =        1	(winter time threshold) |
     ICN =       70	(shallow limit for test dBar) |
     ICX =       70	(deep limit for test dBar) |
     ICT =     1.80	(deg C) |
     ICS =      180	(sec) |
  ISS[0] =        0	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =       20	(winter 0=disable 1=always 2=urgent) |
     ISM =        0	(bit0=depth bit1=GPS bit2=salinity) |
     IST =   104.95	(PSU) |
     ISO =    23296	(sec) |
     ISP =       30	(0=disable 1=enbable) |
  IUE[0] =        2	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        2	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      256	(summer descent depth dBar) |
  ICD[1] =     5120	(winter descent depth dBar) |
  ICR[0] = -1275068416	(summer minutes) |
  ICR[1] =  6553700	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        0	(attempts) |
     ISD =     7680	(dBar) |
     IAT =        0	(secs) |
     ISV =     7680	(secs) |
     ISW =        1	(secs) |
  ICO[0] =       10	(summer secs) |
  ICO[1] =     1200	(winter secs) |
     ICF =  1967280	(summer mask) |
     ICW =  2949165	(winter ice mask) |
     DNS =        0	(cycle number) |
     DRA =        0	(target descent rate dBar/sec) |
  DSD[0] =        0	(dBar) |
  DSD[1] =        0	(dBar) |
  DSD[2] =      132	(dBar) |
  DST[0] =       20	(sec) |
  DST[1] =     2700	(sec) |
  DST[2] =      400	(sec) |
     DDF =      400	(sec) |
     DDC =     1200	(sec) |
     DSI = 471866400	(ms) |
     DDD =      300	(dBar) |
     DDT =        0	(sec) |
  DFT[0] =        4	(sec) |
  DFT[1] =    37856	(sec) |
  DFT[2] =       10	(sec) |
     DCM =        2	(0=dis 1=en) |
     DAL =       88	(0=cal 1=seek) |
  DCT[0] =       75	(sec) |
  DCT[1] =      130	(sec) |
  DCT[2] =      130	(sec) |
     DCA =        1	(max attempts) |
     DRC =      314	(cal rate calculation) |
     DRT =    38954	(sec) |
     DCX =    12330	(sec) |
     DCP =    12300	(sec) |
     DCF =        3	(sec) |
     DCD =        0	(dBar) |
     DCR =     1020	(dBar/sec) |
     DSW =        5	(sec) |
     DUC =        5	(under damping factor) |
     DPT =      100	(sec) |
     DUT =        0	(sec) |
     DPD =     1200	(sec) |
     DVT =      100	(sec) |
     DVD =        5	(dBar) |
     DVW =        5	(sec) |
     PFR =        5	(cc/min) |
     PFC =       30	(.001 cc/dBar) |
     DDP =       25	(dBar) |
     DDS =      300	(sec) |
     PNS =       19	(cycle) |
  PTO[0] =  1048676	(sec) |
  PTO[1] = 235929732	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        0	(0=avg 1=avg+raw) |
     PSI =        0	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =      256	(sec) |
  PFT[1] =      234	(sec) |
  PFT[2] =    24576	(sec) |
  PFD[0] =        0	(dBar) |
  PFD[1] =     3600	(dBar) |
  PFD[2] =     3600	(dBar) |
  PFV[0] =     3600	(sec) |
  PFV[1] =      500	(sec) |
  PFV[2] =        0	(sec) |
     PDT =        0	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =        0	(dBar) |
     PST =        0	(sec) |
     PSV =        1	(sec) |
     PSW =    11265	(sec) |
     ASN =      100	(cycle number) |
     ARR =     1200	(target rise rate cm/sec) |
     APM =       30	(sec) |
     APS =     1300	(sec) |
     APN =     1792	(sec) |
     APF =    33792	(sec) |
     ASI = 168386560	(ms) |
     ASD =     2560	(dBar) |
     ADD =     7680	(dBar) |
     ADT =     2560	(sec) |
     ARC =      117	(?) |
     AHR =    12288	(cm/sec) |
     AHD =      512	(dBar) |
     SFN =     2562	(cycle number) |
     SDC =    22539	(dBar) |
     SGT =    47107	(min) |
     SET =    59392	(min) |
     SBM =       10	(0=dis 1=en) |
     SBT =      132	(sec) |
     SDT =       15	(interval) |
     SSA =       15	(0=dis 1=en) |
     SSI =    76800	(ms) |
     SDS =        0	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =    10000	(.001 psi) |
     SAN =       60	(.001 psi) |
     SAO =      285	(sec) |
     SAP =    19481	(sec) |
     SPT =    25604	(sec) |
     SSP =    14340	(sec) |

