# Message Decoder for Solo2, Alto and Deep Solo Floats; v210818
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11070   110  282 000000000000009   970     0   6 2020/02/01, 21:34:03, -67.7956  176.5834   3      0 6,14,14
SBD 11070   110  340 000000000000009   971     0   5 2020/02/01, 21:34:24, -67.8267  176.4829   3      0 13
SBD 11070   110  330 000000000000009   972     0   4 2020/02/01, 21:34:44, -67.8288  176.5777   3      0 12,12,15,13
SBD 11070   110  282 000000000000009   973     0   3 2020/02/01, 21:35:01, -67.8288  176.5777   3      0 3,8
SBD 11070   110  282 000000000000009   974     0   2 2020/02/01, 21:35:18, -67.8267  176.4829   3      0 2
SBD 11070   110  298 000000000000009   975     0   1 2020/02/01, 21:35:30, -67.8267  176.4829   3      0 1,12
SBD 11070   110  304 000000000000009   976     0   0 2020/02/01, 21:36:03, -67.8288  176.5777   4      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         2034           84        2118
# Sensor_ID bytes     % packets Description
          0    48  2.36       2         GPS
          1   199  9.78       1    Pressure
          2   270 13.27       1 Temperature
          3   235 11.55       1    Salinity
          4    68  3.34       1        Fall
          5   113  5.56       1        Rise
          6   158  7.77       1        Pump
          8    35  1.72       1     Surface
         12   253 12.44       3  Aux Sensor
         13   414 20.35       2  Parameters
         14   175  8.60       5 Engineering
         15    66  3.24       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  110 1 2020/02/01 19:55 -67.80916 +176.44306  110  4 25 34 38  2.5     0
G  110 2 2020/02/01 21:22 -67.80355 +176.44258  210  4 29 41 50  2.0     0
# Warning - Unknown C2 Sensor Data received; channel 0x31
# Warning - Unknown C2 Sensor Data received; channel 0x32
# Warning - Unknown C2 Sensor Data received; channel 0x33
# Profile Data
#      C Npts comp n_var
D0   110  283    1     3
#      Pres      Temp    Salinity
      0.320    -0.5870    33.7680
      1.080    -0.5870    33.7700
      2.000    -0.5910    33.7710
      3.080    -0.5910    33.7710
      3.960    -0.5950    33.7740
      4.920    -0.5960    33.7730
      6.040    -0.5950    33.7730
      7.040    -0.5940    33.7720
      7.960    -0.5940    33.7720
      8.960    -0.5960    33.0000
     10.040    -0.5970    33.7730
     11.040    -0.5960    33.7730
     11.960    -0.5960    33.7730
     12.960    -0.5970    33.7730
     13.960    -0.5940    33.7730
     15.040    -0.5940    33.7730
     16.040    -0.5920    33.7730
     17.120    -0.5890    33.7720
     18.040    -0.5900    33.7710
     19.040    -0.5870    33.7710
     20.000    -0.5870    33.7700
     20.960    -0.5880    33.7700
     21.920    -0.5880    33.7710
     22.920    -0.5880    33.7700
     24.040    -0.5880    33.7700
     25.040    -0.5920    33.7700
     25.960    -0.5930    33.7720
     28.080    -0.5930    33.7720
     30.080    -0.5910    33.7720
     31.080    -0.5900    33.7710
     32.000    -0.5900    33.7710
     33.040    -0.5930    33.7710
     34.040    -0.5900    33.7720
     35.000    -0.5910    33.7700
     35.960    -0.5890    33.7710
     37.000    -0.5890    33.7700
     38.080    -0.5860    33.7700
     39.080    -0.5880    33.7680
     40.000    -0.5880    33.7690
     40.960    -0.5930    33.7680
     41.960    -0.5970    33.7710
     42.960    -0.5910    33.7710
     44.040    -0.5920    33.7670
     45.000    -0.5930    33.7690
     45.920    -0.5930    33.7680
     46.960    -0.5960    33.7670
     48.000    -0.5990    33.7680
     48.960    -0.6000    33.7680
     49.880    -0.6010    33.7670
     50.880    -0.5990    33.7670
     51.920    -0.5930    33.7640
     52.920    -0.5970    33.7600
     53.920    -0.5980    33.7600
     54.920    -0.6000    33.7570
     55.960    -0.6120    33.7530
     56.960    -0.6270    33.7550
     57.960    -0.6330    33.7540
     59.080    -0.7210    33.7440
     59.960    -0.9750    33.7770
     60.920    -1.2620    33.8890
     62.000    -1.4330    33.9730
     62.960    -1.6210    34.0370
     63.920    -1.6500    34.1030
     65.040    -1.6520    34.1190
     66.000    -1.6530    34.1200
     66.960    -1.6610    34.1230
     68.000    -1.6680    34.1300
     69.000    -1.6750    34.1360
     70.000    -1.6830    34.1530
     71.040    -1.6880    34.1670
     72.040    -1.6920    34.1750
     73.080    -1.6960    34.1840
     74.000    -1.7030    34.1940
     74.960    -1.7130    34.2070
     76.040    -1.7150    34.2130
     76.960    -1.7170    34.2180
     77.920    -1.7210    34.2240
     78.920    -1.7230    34.2330
     79.960    -1.7220    34.2400
     81.000    -1.7210    34.2430
     82.040    -1.7230    34.2500
     83.000    -1.7240    34.2590
     83.920    -1.7240    34.2660
     85.000    -1.7170    34.2710
     86.080    -1.7030    34.2710
     87.000    -1.7010    34.2770
     87.920    -1.6900    34.2820
     88.960    -1.6760    34.2870
     90.040    -1.6520    34.2920
     91.080    -1.6260    34.2960
     92.080    -1.6060    34.3050
     93.120    -1.5810    34.3110
     94.040    -1.5610    34.3120
     94.920    -1.5580    34.3190
     96.000    -1.5030    34.3230
     97.080    -1.4690    34.3190
     98.000    -1.4120    34.3330
     98.920    -1.3810    34.3460
    100.000    -1.2980    34.3280
    101.040    -1.1750    34.3460
    102.000    -1.1340    34.3620
    102.920    -1.1010    34.3580
    104.000    -1.0250    34.3640
    105.080    -0.9960    34.3710
    106.000    -0.9360    34.3730
    106.960    -0.9130    34.3840
    108.040    -0.8940    34.3850
    109.000    -0.8750    34.3910
    109.960    -0.8600    34.3930
    111.040    -0.8290    34.3950
    111.960    -0.7890    34.4020
    112.920    -0.7080    34.4020
    113.960    -0.6670    34.4150
    115.080    -0.5850    34.4130
    116.000    -0.5050    34.4330
    116.920    -0.4320    34.4320
    118.000    -0.3480    34.4400
    119.080    -0.2780    34.4520
    119.960    -0.2160    34.4650
    120.920    -0.1610    34.4650
    122.000    -0.1210    34.4770
    123.080    -0.0400    34.4790
    124.040     0.0350    34.5020
    125.000     0.1130    34.4930
    125.960     0.1370    34.5060
    126.880     0.1630    34.5050
    127.960     0.1950    34.5130
    129.040     0.2130    34.5160
    129.960     0.2310    34.5100
    130.880     0.2820    34.5230
    131.960     0.3200    34.5250
    133.040     0.3630    34.5360
    134.080     0.3870    34.5340
    135.040     0.4100    34.5310
    135.960     0.4940    34.5470
    137.040     0.5300    34.5510
    138.120     0.5530    34.5560
    139.040     0.5730    34.5610
    140.000     0.5890    34.5600
    142.000     0.6000    34.5660
    142.920     0.6200    34.5670
    144.000     0.6440    34.5710
    145.040     0.6600    34.5750
    146.080     0.6690    34.5740
    147.080     0.6820    34.5800
    148.080     0.7040    34.5830
    149.120     0.7240    34.5820
    150.000     0.7770    34.5920
    150.880     0.8020    34.5890
    151.880     0.8210    34.5960
    152.920     0.8530    34.5990
    153.920     0.8670    34.6020
    154.960     0.8810    34.6040
    156.000     0.8870    34.6040
    157.040     0.9150    34.6090
    158.080     0.9440    34.6140
    159.080     0.9610    34.6090
    160.920     0.9800    34.6230
    161.920     1.0230    34.6230
    162.960     1.0330    34.6270
    164.000     1.0460    34.6270
    165.040     1.0610    34.6310
    166.040     1.0750    34.6320
    167.120     1.0890    34.6330
    168.000     1.0980    34.6340
    168.920     1.1050    34.6350
    169.960     1.1080    34.6360
    171.000     1.1100    34.6360
    172.080     1.1100    34.6350
    173.080     1.1110    34.6360
    174.120     1.1170    34.6400
    175.000     1.1310    34.6400
    175.920     1.1360    34.6410
    176.920     1.1390    34.6410
    177.880     1.1420    34.6430
    179.000     1.1450    34.6440
    180.120     1.1510    34.6440
    181.120     1.1560    34.6460
    182.080     1.1580    34.6460
    183.040     1.1650    34.6470
    184.040     1.1710    34.6480
    185.000     1.1740    34.6490
    185.960     1.1770    34.6500
    186.920     1.1820    34.6510
    188.000     1.1850    34.6520
    189.120     1.1870    34.6520
    190.080     1.1900    34.6550
    191.040     1.1960    34.6560
    192.040     1.2080    34.6580
    193.040     1.2170    34.6580
    194.000     1.2280    34.6610
    195.000     1.2380    34.6630
    195.960     1.2530    34.6630
    196.920     1.2700    34.6640
    198.000     1.2730    34.6650
    199.080     1.2740    34.6650
    200.000     1.2770    34.6670
    200.920     1.2810    34.6680
    202.000     1.2910    34.6690
    203.040     1.3000    34.6700
    204.000     1.3060    34.6720
    205.040     1.3120    34.6740
    206.080     1.3240    34.6740
    206.960     1.3310    34.6750
    208.000     1.3330    34.6760
    209.040     1.3350    34.6760
    210.040     1.3450    34.6790
    211.080     1.3550    34.6780
    213.000     1.3640    34.6780
    214.000     1.3610    34.6800
    214.960     1.3620    34.6800
    215.960     1.3710    34.6830
    216.960     1.3800    34.6820
    218.040     1.3870    34.6820
    219.040     1.3870    34.6840
    220.040     1.3830    34.6850
    221.000     1.3920    34.6850
    222.000     1.4000    34.6870
    223.000     1.4070    34.6870
    223.960     1.4120    34.6880
    225.040     1.4190    34.6890
    226.000     1.4220    34.6890
    226.960     1.4260    34.6900
    228.000     1.4280    34.6890
    229.040     1.4290    34.6910
    230.080     1.4230    34.6910
    231.080     1.4240    34.6920
    232.080     1.4310    34.6920
    233.080     1.4330    34.6930
    234.080     1.4340    34.6930
    235.080     1.4360    34.6940
    236.040     1.4400    34.6940
    237.040     1.4410    34.6940
    238.040     1.4400    34.6940
    239.000     1.4390    34.6950
    240.000     1.4340    34.6970
    240.960     1.4300    34.6980
    242.000     1.4290    34.6980
    243.040     1.4390    34.6980
    244.040     1.4530    34.6980
    245.040     1.4570    34.6980
    246.040     1.4520    34.6990
    247.040     1.4540    34.6990
    248.000     1.4550    34.6990
    249.000     1.4570    34.6990
    250.000     1.4590    34.6990
    251.000     1.4600    34.7000
    251.960     1.4600    34.7000
    253.040     1.4580    34.7000
    254.000     1.4580    34.6990
    254.960     1.4590    34.7000
    255.960     1.4610    34.7000
    256.960     1.4600    34.7000
    258.000     1.4610    34.7000
    259.000     1.4620    34.7000
    260.000     1.4600    34.7000
    260.920     1.4580    34.7010
    261.960     1.4540    34.7010
    262.960     1.4500    34.7020
    264.000     1.4490    34.7030
    265.000     1.4490    34.7030
    265.960     1.4500    34.7040
    266.960     1.4540    34.7040
    268.000     1.4560    34.7050
    269.000     1.4620    34.7050
    269.960     1.4650    34.7050
    270.960     1.4670    34.7060
    272.040     1.4670    34.7060
    273.040     1.4680    34.7070
    274.000     1.4690    34.7080
    275.040     1.4720    34.7080
    276.000     1.4740    34.7080
    277.000     1.4800    34.7080
    278.000     1.4840    34.7090
    279.000     1.4860    34.7080
    280.080     1.4860    34.7090
    281.000     1.4860    34.7100
    282.000     1.4870    34.7100
    283.040     1.4880  -999.0000
    283.760     1.4870  -999.0000
   -999.000     1.4880  -999.0000
   -999.000     1.4880  -999.0000
   -999.000     1.4880  -999.0000
# Surface Pressure Time Series
#    C npts n_var               Start
S  110     3     9 2020/02/01 20:58:03
# time   Pressure       Temp       Psal Mission                Date O2_Molarity O2_Temp O2_Phase
#  sec       dBar       degC        PSU   phase                 UTC      umol/L    degC  degrees
   124      0.090     -0.623   -999.000      14 2020/02/01 21:00:07      -9.000  -9.000   -9.000
  1238     -0.070     -0.656   -999.000      14 2020/02/01 21:18:41      -9.000  -9.000   -9.000
  1673      0.060     -0.645   -999.000      14 2020/02/01 21:25:56      -9.000  -9.000   -9.000
# Fall Time Series
#  Cycle npts     Start                Date
F00  110   12 633902178 2020/02/01 19:56:18
# time     Pres phase                Date
     0    0.000     1 2020/02/01 19:56:18
   611   50.920    10 2020/02/01 20:06:29
   912  116.080    10 2020/02/01 20:11:30
  1216  182.640     2 2020/02/01 20:16:34
  1219  183.400    10 2020/02/01 20:16:37
  1524  231.880    10 2020/02/01 20:21:42
  1665  241.120    10 2020/02/01 20:24:03
  1821  251.000    10 2020/02/01 20:26:39
  1984  261.040    10 2020/02/01 20:29:22
  2150  271.040    10 2020/02/01 20:32:08
  2288 9999.000    10 2020/02/01 20:34:26
  2291  279.280    10 2020/02/01 20:34:29
# Rise Time Series
#  Cycle npts     Start                Date
R00  110   21 633904481 2020/02/01 20:34:41
# time     Pres phase                Date
     0  279.960     7 2020/02/01 20:34:41
   264  269.560    13 2020/02/01 20:39:05
   335  259.120    13 2020/02/01 20:40:16
   405  246.840    13 2020/02/01 20:41:26
   476  233.280    13 2020/02/01 20:42:37
   547  218.280    13 2020/02/01 20:43:48
   617  202.440    13 2020/02/01 20:44:58
   688  185.320    13 2020/02/01 20:46:09
   759  167.480    13 2020/02/01 20:47:20
   829  149.480    13 2020/02/01 20:48:30
   900  130.760    13 2020/02/01 20:49:41
   970  111.960    13 2020/02/01 20:50:51
  1041   93.240    13 2020/02/01 20:52:02
  1076   83.760    13 2020/02/01 20:52:37
  1146   65.440    13 2020/02/01 20:53:47
  1216   47.080    13 2020/02/01 20:54:57
  1252   37.400    13 2020/02/01 20:55:33
  1287   27.120    13 2020/02/01 20:56:08
  1322   17.200    13 2020/02/01 20:56:43
  1357    7.360    13 2020/02/01 20:57:18
  1401    0.000     8 2020/02/01 20:58:02
# Pump Time Series
#   Cycle npts
B02   110   14
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
   182.640  302   14.87     268    612    -9     2
   279.960   17   14.90     321    608    -9     7
   282.400   35   14.88     301    610    -9    13
   283.440   35   14.87     292    613    -9    13
   278.200   36   14.86     297    617    -9    13
   269.560   35   14.86     294    620    -9    13
   259.120   35   14.86     289    623    -9    13
   246.840   36   14.86     282    626    -9    13
   233.280   36   14.84     267    630    -9    13
   218.280   35   14.83     259    633    -9    13
   202.440   35   14.84     261    636    -9    13
   185.320   36   14.83     250    639    -9    13
   149.480   36   14.84     238    643    -9    13
    -0.080  120   14.85     178    657    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   110  2.4    8   500  1500   600   311    60     0     10    12    1245   20  0.4966
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  110 2020/02/01 19:56:14
# Pcase vacuums
   6.920   6.070
# Oil valve positions
   1.688   0.738   0.654
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18  110 2020/02/01 20:34:30 9999/99/99 99:99:99
# Pcase vacuums
   6.070   6.070
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18  110 2020/02/01 20:34:41 9999/99/99 99:99:99
# Pcase vacuums
   6.070   6.470
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20  110 2020/02/01 20:58:03
# number of packets previous packets
      7               8
# CTD Status pres offset
 0x4e410000   0.530
# Pcase vacuums
   6.470   6.930
# Air Pump max current time  max pressure
       0      1107     6.632
# Air Valve Positions
   1.119   0.657
# Air Valve Calibrations
   1.041   0.099   0.570
# Oil Valve Positions
   1.687   0.735
# Oil Valve Calibrations
   1.634   0.786
# Encountered Error Codes
#    Cycle Errors
EE20   110      4
#               Date   Code Description
 2020/02/01 20:34:26 0x0913 DescendMaxDepth
 2020/02/01 21:32:35 0x0116 SurfaceCtd
 2020/02/01 21:32:37 0x0116 SurfaceCtd
 2020/02/01 21:32:37 0x0A16 SurfaceData
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  110  1      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.080   1210   5.880
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  110       198    2020/02/01 21:34:44
Parameter Table Param_Table-V17.txt |
     IFV = 6.8.116.IFV	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =      109	(cycle number) |
     MPS =        0	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =        0	(count) |
  MNM[2] =        0	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        0	(count) |
  MNS[2] =        0	(count) |
  DTD[0] =     1500	(dBar) |
  DTD[1] =      400	(dBar) |
  DTD[2] =      400	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =      300	(dBar) |
  CPB[3] =      100	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        1	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        1	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       50	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =    30000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1800	(V) |
     PNC =       10	(inHg) |
     PXC =       50	(inHg) |
     PBL =       20	(12V cal) |
     PBH =     1200	(14V cal) |
     PP0 =      400	(oil pump 0 mA cal) |
     PP1 =     1500	(oil pump 1000 mA cal) |
     PVL =    43668	(V12 12V cal) |
     PVH =    50958	(V12 14V cal) |
     PV0 =        0	(vac 0 inHg cal) |
     PV5 =    25960	(vac 5 inHg cal) |
     PC0 =     1790	(V12_sig 0mA cal) |
     PC1 =     2089	(V12_sig 1000mA cal) |
     ICM =      240	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =    44669	(ice month mask) |
  ICA[0] =        0	(summer attempts) |
  ICA[1] =       64	(winter attempts) |
  ICB[0] =    55296	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =        5	(summer message threshold) |
  IMT[1] =        5	(winter message threshold) |
  ILT[0] =        1	(summer time threshold) |
  ILT[1] =        1	(winter time threshold) |
     ICN =       70	(shallow limit for test dBar) |
     ICX =       70	(deep limit for test dBar) |
     ICT =     1.80	(deg C) |
     ICS =      180	(sec) |
  ISS[0] =        0	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =       20	(winter 0=disable 1=always 2=urgent) |
     ISM =        0	(bit0=depth bit1=GPS bit2=salinity) |
     IST =   104.95	(PSU) |
     ISO =    23296	(sec) |
     ISP =       30	(0=disable 1=enbable) |
  IUE[0] =        2	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        2	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      256	(summer descent depth dBar) |
  ICD[1] =     5120	(winter descent depth dBar) |
  ICR[0] = -1275068416	(summer minutes) |
  ICR[1] =  6553700	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        0	(attempts) |
     ISD =     7680	(dBar) |
     IAT =        0	(secs) |
     ISV =     7680	(secs) |
     ISW =        1	(secs) |
  ICO[0] =       10	(summer secs) |
  ICO[1] =     1200	(winter secs) |
     ICF =  1967280	(summer mask) |
     ICW =  2949165	(winter ice mask) |
     DNS =        0	(cycle number) |
     DRA =        0	(target descent rate dBar/sec) |
  DSD[0] =        0	(dBar) |
  DSD[1] =        0	(dBar) |
  DSD[2] =      109	(dBar) |
  DST[0] =       20	(sec) |
  DST[1] =     2700	(sec) |
  DST[2] =      400	(sec) |
     DDF =      400	(sec) |
     DDC =     1200	(sec) |
     DSI = 471866400	(ms) |
     DDD =      300	(dBar) |
     DDT =        0	(sec) |
  DFT[0] =        4	(sec) |
  DFT[1] =    37856	(sec) |
  DFT[2] =       10	(sec) |
     DCM =        2	(0=dis 1=en) |
     DAL =       88	(0=cal 1=seek) |
  DCT[0] =       75	(sec) |
  DCT[1] =      130	(sec) |
  DCT[2] =      130	(sec) |
     DCA =        1	(max attempts) |
     DRC =      314	(cal rate calculation) |
     DRT =    38954	(sec) |
     DCX =    12330	(sec) |
     DCP =    12300	(sec) |
     DCF =        3	(sec) |
     DCD =        0	(dBar) |
     DCR =     1020	(dBar/sec) |
     DSW =        5	(sec) |
     DUC =        5	(under damping factor) |
     DPT =      100	(sec) |
     DUT =        0	(sec) |
     DPD =     1200	(sec) |
     DVT =      100	(sec) |
     DVD =        5	(dBar) |
     DVW =        5	(sec) |
     PFR =        5	(cc/min) |
     PFC =       30	(.001 cc/dBar) |
     DDP =       25	(dBar) |
     DDS =      300	(sec) |
     PNS =       19	(cycle) |
  PTO[0] =  1048676	(sec) |
  PTO[1] = 235929709	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        0	(0=avg 1=avg+raw) |
     PSI =        0	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =      256	(sec) |
  PFT[1] =      234	(sec) |
  PFT[2] =    24576	(sec) |
  PFD[0] =        0	(dBar) |
  PFD[1] =     3600	(dBar) |
  PFD[2] =     3600	(dBar) |
  PFV[0] =     3600	(sec) |
  PFV[1] =      500	(sec) |
  PFV[2] =        0	(sec) |
     PDT =        0	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =        0	(dBar) |
     PST =        0	(sec) |
     PSV =        1	(sec) |
     PSW =    11265	(sec) |
     ASN =      100	(cycle number) |
     ARR =     1200	(target rise rate cm/sec) |
     APM =       30	(sec) |
     APS =     1300	(sec) |
     APN =     1792	(sec) |
     APF =    27904	(sec) |
     ASI = 168386560	(ms) |
     ASD =     2560	(dBar) |
     ADD =     7680	(dBar) |
     ADT =     2560	(sec) |
     ARC =      117	(?) |
     AHR =    12288	(cm/sec) |
     AHD =      512	(dBar) |
     SFN =     2562	(cycle number) |
     SDC =    22539	(dBar) |
     SGT =    47107	(min) |
     SET =    59392	(min) |
     SBM =       10	(0=dis 1=en) |
     SBT =      109	(sec) |
     SDT =       15	(interval) |
     SSA =       15	(0=dis 1=en) |
     SSI =    76800	(ms) |
     SDS =        0	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =    10000	(.001 psi) |
     SAN =       60	(.001 psi) |
     SAO =      285	(sec) |
     SAP =    19481	(sec) |
     SPT =    25604	(sec) |
     SSP =    14340	(sec) |

