# Message Decoder for Solo2, Alto and Deep Solo Floats; v210818
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11070    99  326 000000000000009   890     0   6 2020/02/01, 02:36:37, -67.9086  176.5664   3      0 6,14,14
SBD 11070    99  340 000000000000009   891     0   5 2020/02/01, 02:36:58, -67.8755  176.5721   2      0 13
SBD 11070    99  331 000000000000009   892     0   4 2020/02/01, 02:37:12, -67.9106  176.6615   2      0 12,12,15,13
SBD 11070    99  270 000000000000009   893     0   3 2020/02/01, 02:37:20, -67.9086  176.5664   2      0 3,8
SBD 11070    99  291 000000000000009   894     0   2 2020/02/01, 02:37:31, -67.9106  176.6615   2      0 2
SBD 11070    99  316 000000000000009   895     0   1 2020/02/01, 02:37:46, -67.8755  176.5721   2      0 1,12
SBD 11070    99  284 000000000000009   896     0   0 2020/02/01, 02:37:58, -67.9086  176.5664   2      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         2074           84        2158
# Sensor_ID bytes     % packets Description
          0    48  2.31       2         GPS
          1   217 10.46       1    Pressure
          2   279 13.45       1 Temperature
          3   223 10.75       1    Salinity
          4    58  2.80       1        Fall
          5   103  4.97       1        Rise
          6   202  9.74       1        Pump
          8    35  1.69       1     Surface
         12   254 12.25       3  Aux Sensor
         13   414 19.96       2  Parameters
         14   175  8.44       5 Engineering
         15    66  3.18       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   99 1 2020/02/01 01:03 -67.89342 +176.58602   10  4 36 40 44  0.9     0
G   99 2 2020/02/01 02:25 -67.88879 +176.57046   50  4 32 39 47  0.3     0
# Warning - Unknown C2 Sensor Data received; channel 0x31
# Warning - Unknown C2 Sensor Data received; channel 0x32
# Warning - Unknown C2 Sensor Data received; channel 0x33
# Profile Data
#      C Npts comp n_var
D0    99  269    1     3
#      Pres      Temp    Salinity
      0.240    -0.4120    33.6630
      1.080    -0.4120    33.6630
      2.040    -0.4120    33.6630
      3.080    -0.4120    33.6630
      3.920    -0.4120    33.6630
      4.920    -0.4120    33.6630
      6.040     0.4120    33.6620
      7.000    -0.4120    33.6630
      8.040    -0.4120    33.6620
      9.040    -0.4120    33.6620
      9.880    -0.4120    33.6620
     11.000    -0.4120    33.6620
     11.960    -0.4120    33.6620
     12.920    -0.4120    33.6620
     14.000    -0.4120    33.6620
     14.920    -0.4120    33.6620
     16.000    -0.4120    33.6620
     16.960    -0.4140    33.6620
     17.920    -0.4180    33.6640
     18.920    -0.4190    33.6630
     20.040    -0.4190    33.6630
     20.960    -0.4190    33.6630
     22.000    -0.4180    33.6620
     23.040    -0.4190    33.6630
     23.960    -0.4180    33.6620
     24.960    -0.4200    33.6620
     26.040    -0.4180    33.6620
     26.920    -0.4180    33.6610
     27.920    -0.4200    33.6610
     29.000    -0.4210    33.6620
     30.000    -0.4200    33.6600
     31.040    -0.4210    33.6600
     32.000    -0.4250    33.6620
     33.000    -0.4250    33.6600
     34.040    -0.4240    33.6590
     34.960    -0.4250    33.6590
     35.880    -0.4240    33.6570
     37.000    -0.4240    33.6550
     38.120    -0.4250    33.6530
     39.080    -0.4250    33.6510
     39.960    -0.4220    33.6460
     40.960    -0.4210    33.6400
     41.960    -0.4430    33.6440
     42.920    -0.4940    33.6640
     44.040    -0.8090    33.8040
     45.120    -0.9590    33.8530
     45.960    -1.2730    34.0260
     46.960    -1.5220    34.1640
     47.960    -1.5820    34.2200
     49.000    -1.6780    34.2220
     50.080    -1.6820    34.2310
     51.080    -1.6910    34.2400
     51.960    -1.7010    34.2510
     52.960    -1.7150    34.2610
     53.880    -1.7360    34.2620
     55.000    -1.7420    34.2640
     56.040    -1.7450    34.2660
     57.040    -1.7530    34.2660
     58.120    -1.7540    34.2660
     59.000    -1.7530    34.2670
     60.000    -1.7540    34.2670
     61.040    -1.7560    34.2670
     61.960    -1.7560    34.2680
     63.000    -1.7570    34.2680
     64.040    -1.7580    34.2680
     65.040    -1.7590    34.2680
     65.960    -1.7590    34.2680
     67.000    -1.7600    34.2700
     67.960    -1.7640    34.2700
     68.920    -1.7660    34.2710
     69.960    -1.7670    34.2710
     71.000    -1.7680    34.2720
     72.080    -1.7700    34.2720
     73.000    -1.7720    34.2730
     73.960    -1.7740    34.2730
     75.040    -1.7760    34.2730
     76.120    -1.7780    34.2740
     77.000    -1.7800    34.2740
     77.920    -1.7810    34.2750
     78.960    -1.7850    34.2760
     80.000    -1.7880    34.2780
     80.960    -1.7930    34.2780
     82.000    -1.7930    34.2780
     83.000    -1.7940    34.2790
     83.960    -1.7940    34.2790
     85.000    -1.7940    34.2800
     86.000    -1.7940    34.2810
     87.040    -1.7940    34.2810
     88.000    -1.7930    34.2830
     89.000    -1.7910    34.2840
     90.040    -1.7910    34.2850
     91.080    -1.7900    34.2860
     92.080    -1.7870    34.2880
     93.120    -1.7840    34.2880
     94.000    -1.7780    34.2920
     94.920    -1.7650    34.2940
     95.960    -1.7530    34.2970
     97.000    -1.7450    34.2960
     98.080    -1.7270    34.3050
     99.920    -1.7120    34.2990
    101.000    -1.6800    34.3040
    102.040    -1.6100    34.3120
    103.000    -1.5940    34.3100
    103.880    -1.5610    34.3170
    104.960    -1.5350    34.3210
    105.960    -1.5120    34.3180
    107.000    -1.4690    34.3280
    108.040    -1.4220    34.3310
    109.080    -1.3740    34.3330
    109.960    -1.3420    34.3360
    110.960    -1.3160    34.3440
    112.000    -1.2960    34.3470
    113.080    -1.2590    34.3470
    114.000    -1.2030    34.3480
    114.920    -1.1240    34.3550
    115.960    -1.0630    34.3680
    117.080    -0.9740    34.3590
    118.000    -0.9360    34.3800
    119.000    -0.9090    34.3800
    121.000    -0.8440    34.3760
    121.920    -0.8230    34.3960
    122.960    -0.8140    34.3980
    124.040    -0.7940    34.3980
    125.000    -0.7250    34.3930
    125.920    -0.7080    34.4170
    127.000    -0.6180    34.3870
    128.080    -0.4500    34.4230
    129.000    -0.3800    34.4240
    129.920    -0.3400    34.4360
    130.960    -0.2890    34.4390
    132.040    -0.2590    34.4530
    133.080    -0.2440    34.4490
    134.000    -0.2020    34.4620
    134.920    -0.1900    34.4690
    135.960    -0.1200    34.4550
    137.000    -0.0370    34.4810
    138.000     0.0490    34.4750
    139.040     0.0870    34.4890
    140.040     0.1340    34.4940
    141.080     0.1880    34.4910
    142.080     0.2560    34.5020
    143.120     0.3190    34.5130
    144.120     0.3520    34.5210
    145.120     0.3670    34.5260
    147.040     0.4180    34.5230
    147.960     0.4860    34.5210
    150.040     0.5500    34.5340
    151.080     0.5830    34.5450
    152.080     0.5890    34.5470
    153.000     0.5960    34.5510
    153.920     0.5910    34.5530
    154.960     0.5900    34.5530
    156.000     0.5970    34.5530
    157.040     0.6030    34.5560
    158.080     0.6080    34.5550
    159.040     0.6250    34.5550
    159.920     0.6820    34.5520
    160.960     0.7020    34.5640
    162.000     0.6980    34.5640
    163.040     0.6890    34.5670
    164.080     0.6800    34.5670
    165.080     0.6750    34.5670
    166.120     0.6880    34.5670
    167.120     0.6970    34.5690
    168.120     0.7210    34.5710
    169.120     0.7360    34.5750
    170.120     0.7550    34.5780
    171.000     0.7680    34.5810
    171.880     0.7740    34.5830
    173.000     0.7860    34.5810
    174.080     0.8060    34.5840
    175.080     0.8140    34.5890
    177.080     0.8200    34.5890
    178.080     0.8250    34.5910
    179.080     0.8320    34.5920
    180.040     0.8420    34.5960
    181.000     0.9140    34.5920
    182.000     0.9290    34.6000
    182.960     0.9300    34.6050
    183.960     0.9300    34.6070
    184.920     0.9330    34.6070
    186.000     0.9420    34.6070
    187.080     0.9520    34.6070
    188.920     0.9630    34.6100
    190.000     0.9700    34.6120
    191.040     0.9770    34.6140
    192.040     1.0060    34.6150
    193.080     1.0220    34.6130
    194.000     1.0250    34.6210
    194.920     1.0410    34.6220
    195.960     1.0550    34.6220
    197.000     1.0710    34.6240
    198.040     1.0790    34.6250
    199.080     1.0820    34.6290
    199.960     1.0810    34.6310
    201.000     1.0820    34.6320
    202.080     1.0890    34.6320
    202.960     1.0940    34.6320
    204.000     1.1120    34.6340
    205.000     1.1270    34.6330
    206.000     1.1320    34.6380
    207.000     1.1430    34.6390
    207.960     1.1520    34.6400
    209.040     1.1640    34.6410
    210.000     1.1740    34.6430
    210.960     1.1800    34.6430
    211.960     1.1800    34.6470
    213.040     1.1800    34.6480
    214.080     1.1900    34.6480
    215.040     1.1990    34.6470
    217.040     1.2060    34.6510
    218.080     1.2170    34.6520
    219.000     1.2310    34.6550
    219.920     1.2400    34.6560
    221.000     1.2480    34.6580
    222.000     1.2590    34.6580
    222.960     1.2660    34.6600
    224.040     1.2680    34.6610
    225.000     1.2700    34.6610
    226.000     1.2740    34.6620
    227.000     1.2790    34.6620
    227.960     1.2730    34.6640
    229.040     1.2740    34.6640
    230.080     1.2750    34.6640
    231.040     1.2790    34.6640
    232.000     1.2820    34.6650
    233.000     1.2870    34.6660
    234.040     1.2910    34.6660
    235.000     1.3000    34.6670
    236.000     1.3010    34.6690
    237.000     1.3020    34.6690
    237.960     1.3090    34.6690
    238.960     1.3170    34.6710
    240.000     1.3240    34.6720
    241.040     1.3280    34.6730
    242.040     1.3300    34.6740
    243.000     1.3310    34.6740
    243.920     1.3330    34.6750
    245.000     1.3390    34.6760
    246.040     1.3430    34.6770
    247.000     1.3490    34.6770
    248.040     1.3560    34.6790
    249.000     1.3600    34.6800
    250.040     1.3590    34.6810
    251.040     1.3660    34.6800
    252.040     1.3710    34.6820
    253.080     1.3740    34.6820
    254.040     1.3750    34.6830
    254.960     1.3750    34.6830
    255.960     1.3750    34.6830
    257.000     1.3750    34.6840
    257.960     1.3780    34.6840
    258.960     1.3800    34.6850
    260.000     1.3800    34.6850
    261.000     1.3910    34.6860
    262.000     1.4010    34.6880
    262.960     1.4040    34.6890
    264.040     1.4080    34.6890
    265.080     1.4130    34.6900
    266.000     1.4170    34.6910
    267.120     1.4220    34.6920
    268.000     1.4210    34.6930
    269.000     1.4230    34.6930
    270.000     1.4250    34.6930
    271.080     1.4230    34.6930
    271.600     1.4220    34.6940
   -999.000     1.4210    34.6940
   -999.000     1.4220    34.6940
   -999.000     1.4230    34.6950
# Surface Pressure Time Series
#    C npts n_var               Start
S   99     3     9 2020/02/01 01:59:13
# time   Pressure       Temp       Psal Mission                Date O2_Molarity O2_Temp O2_Phase
#  sec       dBar       degC        PSU   phase                 UTC      umol/L    degC  degrees
   125     -0.100     -0.436   -999.000      14 2020/02/01 02:01:18      -9.000  -9.000   -9.000
  1238     -0.140     -0.439   -999.000      14 2020/02/01 02:19:51      -9.000  -9.000   -9.000
  1774     -0.120     -0.468   -999.000      14 2020/02/01 02:28:47      -9.000  -9.000   -9.000
# Fall Time Series
#  Cycle npts     Start                Date
F00   99   10 633834233 2020/02/01 01:03:53
# time     Pres phase                Date
     0   -0.080     1 2020/02/01 01:03:53
   611   50.880    10 2020/02/01 01:14:04
   912 9999.000    10 2020/02/01 01:19:05
  1216  186.920     2 2020/02/01 01:24:09
  1219  187.640    10 2020/02/01 01:24:12
  1524  238.240    10 2020/02/01 01:29:17
  1655  248.120    10 2020/02/01 01:31:28
  1793  258.040    10 2020/02/01 01:33:46
  1901 9999.000    10 2020/02/01 01:35:34
  1905  265.760    10 2020/02/01 01:35:38
# Rise Time Series
#  Cycle npts     Start                Date
R00   99   19 633836149 2020/02/01 01:35:49
# time     Pres phase                Date
     0  266.560     7 2020/02/01 01:35:49
   300  254.520    13 2020/02/01 01:40:49
   370  243.520    13 2020/02/01 01:41:59
   441  230.840    13 2020/02/01 01:43:10
   512  216.840    13 2020/02/01 01:44:21
   583  201.720    13 2020/02/01 01:45:32
   653  185.280    13 2020/02/01 01:46:42
   724  167.760    13 2020/02/01 01:47:53
   795  149.640    13 2020/02/01 01:49:04
   865  131.360    13 2020/02/01 01:50:14
   936  112.640    13 2020/02/01 01:51:25
  1006   94.000    13 2020/02/01 01:52:35
  1076   76.240    13 2020/02/01 01:53:45
  1112   66.880    13 2020/02/01 01:54:21
  1182   49.000    13 2020/02/01 01:55:31
  1253   33.160    13 2020/02/01 01:56:42
  1324   14.360    13 2020/02/01 01:57:53
  1396    0.120    13 2020/02/01 01:59:05
  1403    0.120     8 2020/02/01 01:59:12
# Pump Time Series
#   Cycle npts
B02    99   18
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
   186.920  302   14.81     289    613    -9     2
   266.560   17   14.84     331    609    -9     7
   269.520   35   14.81     310    612    -9    13
   271.480   36   14.81     311    615    -9    13
   267.240   35   14.81     302    618    -9    13
   259.120   36   14.79     308    621    -9    13
   249.120   36   14.79     289    624    -9    13
   237.440   35   14.79     284    627    -9    13
   224.080   35   14.79     283    630    -9    13
   209.440   36   14.77     269    634    -9    13
   193.400   36   14.78     264    637    -9    13
   176.720   35   14.77     256    641    -9    13
   140.720   35   14.78     237    645    -9    13
    85.120   36   14.78     220    653    -9    13
    49.000   36   14.78     217    655    -9    13
    33.160   35   14.78     207    656    -9    13
     7.160   38   14.78     199    659    -9    13
    -0.040  120   14.76     194    669    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    99  2.4    8   500  1500   600   311    60     0     10    12    1245   20  0.4966
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   99 2020/02/01 01:03:49
# Pcase vacuums
   6.940   6.090
# Oil valve positions
   1.685   0.738   0.655
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   99 2020/02/01 01:35:39 9999/99/99 99:99:99
# Pcase vacuums
   6.090   6.090
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   99 2020/02/01 01:35:49 9999/99/99 99:99:99
# Pcase vacuums
   6.090   6.590
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   99 2020/02/01 01:59:13
# number of packets previous packets
      7              11
# CTD Status pres offset
 0x4e410000   0.480
# Pcase vacuums
   6.590   6.960
# Air Pump max current time  max pressure
       0      1107     6.635
# Air Valve Positions
   1.118   0.654
# Air Valve Calibrations
   1.041   0.099   0.570
# Oil Valve Positions
   1.687   0.736
# Oil Valve Calibrations
   1.634   0.786
# Encountered Error Codes
#    Cycle Errors
EE20    99      4
#               Date   Code Description
 2020/02/01 01:35:35 0x0913 DescendMaxDepth
 2020/02/01 02:35:21 0x0116 SurfaceCtd
 2020/02/01 02:35:24 0x0116 SurfaceCtd
 2020/02/01 02:35:24 0x0A16 SurfaceData
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   99  1      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.120   1210   5.890
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   99       198    2020/02/01 02:37:12
Parameter Table Param_Table-V17.txt |
     IFV = 6.8.116.IFV	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       98	(cycle number) |
     MPS =        0	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =        0	(count) |
  MNM[2] =        0	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        0	(count) |
  MNS[2] =        0	(count) |
  DTD[0] =     1500	(dBar) |
  DTD[1] =      400	(dBar) |
  DTD[2] =      400	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =      300	(dBar) |
  CPB[3] =      100	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        1	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        1	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       50	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =    30000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1800	(V) |
     PNC =       10	(inHg) |
     PXC =       50	(inHg) |
     PBL =       20	(12V cal) |
     PBH =     1200	(14V cal) |
     PP0 =      400	(oil pump 0 mA cal) |
     PP1 =     1500	(oil pump 1000 mA cal) |
     PVL =    43668	(V12 12V cal) |
     PVH =    50958	(V12 14V cal) |
     PV0 =        0	(vac 0 inHg cal) |
     PV5 =    25960	(vac 5 inHg cal) |
     PC0 =     1790	(V12_sig 0mA cal) |
     PC1 =     2089	(V12_sig 1000mA cal) |
     ICM =      240	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =    44669	(ice month mask) |
  ICA[0] =        0	(summer attempts) |
  ICA[1] =       64	(winter attempts) |
  ICB[0] =    55296	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =        5	(summer message threshold) |
  IMT[1] =        5	(winter message threshold) |
  ILT[0] =        1	(summer time threshold) |
  ILT[1] =        1	(winter time threshold) |
     ICN =       70	(shallow limit for test dBar) |
     ICX =       70	(deep limit for test dBar) |
     ICT =     1.80	(deg C) |
     ICS =      180	(sec) |
  ISS[0] =        0	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =       20	(winter 0=disable 1=always 2=urgent) |
     ISM =        0	(bit0=depth bit1=GPS bit2=salinity) |
     IST =   104.95	(PSU) |
     ISO =    23296	(sec) |
     ISP =       30	(0=disable 1=enbable) |
  IUE[0] =        2	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        2	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      256	(summer descent depth dBar) |
  ICD[1] =     5120	(winter descent depth dBar) |
  ICR[0] = -1275068416	(summer minutes) |
  ICR[1] =  6553700	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        0	(attempts) |
     ISD =     7680	(dBar) |
     IAT =        0	(secs) |
     ISV =     7680	(secs) |
     ISW =        1	(secs) |
  ICO[0] =       10	(summer secs) |
  ICO[1] =     1200	(winter secs) |
     ICF =  1967280	(summer mask) |
     ICW =  2949165	(winter ice mask) |
     DNS =        0	(cycle number) |
     DRA =        0	(target descent rate dBar/sec) |
  DSD[0] =        0	(dBar) |
  DSD[1] =        0	(dBar) |
  DSD[2] =       98	(dBar) |
  DST[0] =       20	(sec) |
  DST[1] =     2700	(sec) |
  DST[2] =      400	(sec) |
     DDF =      400	(sec) |
     DDC =     1200	(sec) |
     DSI = 471866400	(ms) |
     DDD =      300	(dBar) |
     DDT =        0	(sec) |
  DFT[0] =        4	(sec) |
  DFT[1] =    37856	(sec) |
  DFT[2] =       10	(sec) |
     DCM =        2	(0=dis 1=en) |
     DAL =       88	(0=cal 1=seek) |
  DCT[0] =       75	(sec) |
  DCT[1] =      130	(sec) |
  DCT[2] =      130	(sec) |
     DCA =        1	(max attempts) |
     DRC =      314	(cal rate calculation) |
     DRT =    38954	(sec) |
     DCX =    12330	(sec) |
     DCP =    12300	(sec) |
     DCF =        3	(sec) |
     DCD =        0	(dBar) |
     DCR =     1020	(dBar/sec) |
     DSW =        5	(sec) |
     DUC =        5	(under damping factor) |
     DPT =      100	(sec) |
     DUT =        0	(sec) |
     DPD =     1200	(sec) |
     DVT =      100	(sec) |
     DVD =        5	(dBar) |
     DVW =        5	(sec) |
     PFR =        5	(cc/min) |
     PFC =       30	(.001 cc/dBar) |
     DDP =       25	(dBar) |
     DDS =      300	(sec) |
     PNS =       19	(cycle) |
  PTO[0] =  1048676	(sec) |
  PTO[1] = 235929698	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        0	(0=avg 1=avg+raw) |
     PSI =        0	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =      256	(sec) |
  PFT[1] =      234	(sec) |
  PFT[2] =    24576	(sec) |
  PFD[0] =        0	(dBar) |
  PFD[1] =     3600	(dBar) |
  PFD[2] =     3600	(dBar) |
  PFV[0] =     3600	(sec) |
  PFV[1] =      500	(sec) |
  PFV[2] =        0	(sec) |
     PDT =        0	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =        0	(dBar) |
     PST =        0	(sec) |
     PSV =        1	(sec) |
     PSW =    11265	(sec) |
     ASN =      100	(cycle number) |
     ARR =     1200	(target rise rate cm/sec) |
     APM =       30	(sec) |
     APS =     1300	(sec) |
     APN =     1792	(sec) |
     APF =    25088	(sec) |
     ASI = 168386560	(ms) |
     ASD =     2560	(dBar) |
     ADD =     7680	(dBar) |
     ADT =     2560	(sec) |
     ARC =      117	(?) |
     AHR =    12288	(cm/sec) |
     AHD =      512	(dBar) |
     SFN =     2562	(cycle number) |
     SDC =    22539	(dBar) |
     SGT =    47107	(min) |
     SET =    59392	(min) |
     SBM =       10	(0=dis 1=en) |
     SBT =       98	(sec) |
     SDT =       15	(interval) |
     SSA =       15	(0=dis 1=en) |
     SSI =    76800	(ms) |
     SDS =        0	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =    10000	(.001 psi) |
     SAN =       60	(.001 psi) |
     SAO =      285	(sec) |
     SAP =    19481	(sec) |
     SPT =    25604	(sec) |
     SSP =    14340	(sec) |

