# Message Decoder for Solo2, Alto and Deep Solo Floats; v210818
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11070    97  304 000000000000009   876     0   6 2020/01/31, 23:19:59, -67.9573  176.6560   6      0 6,14,14
SBD 11070    97  340 000000000000009   877     0   5 2020/01/31, 23:20:11, -67.9573  176.6560   6      0 13
SBD 11070    97  331 000000000000009   878     0   4 2020/01/31, 23:20:24, -67.9905  176.6504   7      0 12,12,15,13
SBD 11070    97  276 000000000000009   879     0   3 2020/01/31, 23:20:38, -67.9086  176.5664   7      0 3,8
SBD 11070    97  297 000000000000009   880     0   2 2020/01/31, 23:21:05, -67.9086  176.5664   7      0 2
SBD 11070    97  289 000000000000009   881     0   1 2020/01/31, 23:21:20, -67.9106  176.6615   7      0 1,12
SBD 11070    97  299 000000000000009   882     0   0 2020/01/31, 23:21:32, -67.9086  176.5664   7      0 0,4,14,14,5,14,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         2052           84        2136
# Sensor_ID bytes     % packets Description
          0    48  2.34       2         GPS
          1   198  9.65       1    Pressure
          2   285 13.89       1 Temperature
          3   229 11.16       1    Salinity
          4    68  3.31       1        Fall
          5   108  5.26       1        Rise
          6   180  8.77       1        Pump
          8    35  1.71       1     Surface
         12   246 11.99       3  Aux Sensor
         13   414 20.18       2  Parameters
         14   175  8.53       5 Engineering
         15    66  3.22       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   97 1 2020/01/31 21:45 -67.91044 +176.62379   10  4 36 37 40  0.8     0
G   97 2 2020/01/31 23:08 -67.90463 +176.61334  150  4 32 39 50  1.7     0
# Warning - Unknown C2 Sensor Data received; channel 0x31
# Warning - Unknown C2 Sensor Data received; channel 0x32
# Warning - Unknown C2 Sensor Data received; channel 0x33
# Profile Data
#      C Npts comp n_var
D0    97  266    1     3
#      Pres      Temp    Salinity
      0.360    -0.4170    33.6610
      1.000    -0.4170    33.6610
      1.920    -0.4170    33.6610
      3.000    -0.4180    33.6620
      4.040    -0.4220    33.6640
      5.040    -0.4220    33.6640
      6.000    -0.4220    33.6640
      7.000    -0.4200    33.6630
      8.000    -0.4180    33.6630
      9.040    -0.4170    33.6610
     10.000    -0.4180    33.6620
     11.040    -0.4240    33.6650
     12.000    -0.4230    33.6640
     13.040    -0.4170    33.6590
     14.000    -0.4150    33.6580
     14.920    -0.4150    33.6580
     15.960    -0.4150    33.6580
     17.000    -0.4140    33.6570
     17.960    -0.4140    33.6570
     19.000    -0.4140    33.6570
     20.040    -0.4140    33.6560
     21.000    -0.4150    33.6560
     22.000    -0.4180    33.6560
     23.960    -0.4230    33.6590
     25.040    -0.4180    33.6600
     26.040    -0.4180    33.6570
     27.080    -0.4260    33.6570
     28.880    -0.4290    33.6620
     29.960    -0.4300    33.6620
     30.960    -0.4290    33.6620
     32.000    -0.4290    33.6590
     33.080    -0.4300    33.6590
     34.000    -0.4330    33.6580
     35.080    -0.4370    33.6570
     36.120    -0.4370    33.6560
     37.000    -0.4330    33.6530
     37.960    -0.4490    33.6480
     39.000    -0.4960    33.6490
     39.960    -0.5790    33.6710
     41.040    -0.6220    33.7080
     42.120    -0.6620    33.7020
     43.040    -0.9520    33.7160
     43.960    -1.4580    33.8800
     45.000    -1.5930    34.1210
     46.000    -1.6150    34.1570
     46.960    -1.6360    34.1660
     47.960    -1.6570    34.1830
     48.960    -1.6650    34.2100
     50.000    -1.6800    34.2260
     50.960    -1.6900    34.2350
     51.880    -1.6990    34.2430
     52.920    -1.7040    34.2470
     53.920     1.7090    34.2500
     54.960    -1.7120    34.2530
     56.000    -1.7180    34.2590
     57.000    -1.7310    34.2620
     58.040    -1.7360    34.2670
     59.080    -1.7400    34.2650
     60.080    -1.7460    34.2670
     61.080    -1.7480    34.2670
     62.120    -1.7510    34.2680
     63.000    -1.7550    34.2680
     63.920    -1.7600    34.2710
     64.960    -1.7680    34.2710
     66.000    -1.7740    34.2730
     67.000    -1.7730    34.2750
     68.040    -1.7720    34.2760
     69.080    -1.7700    34.2760
     70.000    -1.7580    34.2770
     70.880    -1.7520    34.2750
     71.920    -1.7700    34.2800
     72.960    -1.7770    34.2850
     74.040    -1.7790    34.2830
     75.080    -1.7670    34.2840
     76.000    -1.7590    34.2820
     76.920    -1.7580    34.2850
     77.960    -1.7520    34.2870
     78.920    -1.7510    34.2900
     79.960    -1.7470    34.2910
     81.000    -1.7420    34.2930
     82.000    -1.7390    34.2950
     83.040    -1.7330    34.2950
     84.080    -1.7260    34.3010
     85.000    -1.7270    34.2970
     85.920    -1.7180    34.3030
     86.960    -1.7110    34.3070
     89.080    -1.7160    34.3070
     91.000    -1.7150    34.3090
     91.880    -1.7080    34.3120
     92.960    -1.7080    34.3140
     94.000    -1.7090    34.3160
     95.040    -1.7090    34.3160
     96.080    -1.7030    34.3180
     97.120    -1.6620    34.3180
     98.040    -1.6210    34.3200
     98.920    -1.5590    34.3300
     99.960    -1.5420    34.3340
    101.040    -1.5280    34.3310
    103.120    -1.4840    34.3410
    104.040    -1.4500    34.3300
    104.920    -1.3630    34.3510
    105.960    -1.2440    34.3600
    107.040    -1.2340    34.3580
    107.960    -1.2040    34.3620
    108.880    -1.1620    34.3550
    109.960    -1.0670    34.3770
    111.040    -1.0120    34.3820
    111.960    -1.0070    34.3830
    112.920    -1.0030    34.3800
    113.960    -0.9780    34.3890
    115.040    -0.9280    34.3910
    116.120    -0.9090    34.3980
    117.040    -0.8870    34.3970
    117.960    -0.8460    34.3940
    119.000    -0.7410    34.4110
    120.080    -0.6680    34.4150
    120.960    -0.6270    34.4240
    121.880    -0.5910    34.4190
    122.960    -0.5310    34.4240
    124.000    -0.4270    34.4260
    125.040    -0.3140    34.4520
    126.000    -0.2900    34.4530
    126.920    -0.2900    34.4570
    128.000    -0.2760    34.4580
    129.040    -0.2570    34.4270
    130.080    -0.1810    34.4690
    131.000    -0.0810    34.4770
    131.920    -0.1020    34.4750
    132.920    -0.1130    34.4710
    133.960    -0.0850    34.4810
    134.960    -0.0380    34.4800
    136.040     0.0020    34.4780
    137.040     0.0570    34.4940
    138.120     0.1060    34.5000
    139.000     0.1070    34.4940
    139.920     0.1340    34.5000
    140.960     0.1920    34.5020
    141.960     0.2950    34.5190
    142.960     0.3170    34.5120
    145.040     0.3590    34.5240
    146.000     0.3900    34.5230
    147.040     0.4000    34.5270
    148.000     0.4300    34.5300
    149.080     0.4410    34.5280
    150.080     0.4760    34.5370
    150.960     0.5030    34.5310
    151.880     0.5450    34.5460
    152.920     0.6020    34.5490
    153.960     0.6170    34.5470
    155.000     0.6370    34.5510
    156.040     0.6660    34.5570
    157.040     0.6840    34.5600
    158.080     0.6970    34.5620
    159.120     0.7000    34.5600
    160.000     0.7130    34.5610
    160.960     0.7530    34.5710
    161.960     0.7770    34.5700
    162.960     0.7830    34.5730
    164.000     0.7960    34.5750
    165.000     0.7970    34.5760
    165.960     0.7960    34.5750
    166.960     0.8010    34.5770
    167.960     0.8170    34.5800
    168.920     0.8250    34.5820
    169.920     0.8320    34.5800
    171.000     0.8460    34.5860
    172.120     0.8580    34.5840
    173.080     0.8750    34.5880
    174.080     0.8920    34.5920
    175.040     0.9050    34.5910
    176.040     0.9130    34.5950
    177.000     0.9390    34.5950
    177.960     0.9500    34.6010
    178.960     0.9650    34.6020
    180.040     0.9680    34.6030
    181.120     0.9670    34.6050
    182.080     0.9600    34.6070
    183.000     0.9460    34.6100
    183.960     0.9320    34.6060
    185.040     0.9120    34.6130
    186.080     0.9250    34.6130
    187.000     0.9380    34.6160
    187.920     0.9470    34.6160
    188.960     0.9600    34.6250
    190.040     0.9930    34.6230
    190.960     1.0360    34.6260
    192.000     1.0680    34.6340
    193.040     1.0960    34.6360
    194.000     1.1020    34.6370
    195.040     1.1060    34.6360
    196.080     1.1230    34.6400
    196.960     1.1380    34.6430
    198.000     1.1470    34.6420
    199.000     1.1540    34.6440
    200.000     1.1650    34.6460
    201.080     1.1670    34.6460
    202.040     1.1730    34.6460
    203.040     1.1850    34.6480
    204.000     1.1930    34.6490
    204.960     1.2000    34.6520
    206.040     1.2120    34.6510
    207.000     1.2190    34.6550
    207.960     1.2200    34.6550
    209.040     1.2230    34.6560
    210.040     1.2420    34.6560
    210.960     1.2610    34.6590
    212.040     1.2770    34.6630
    213.080     1.2830    34.6640
    214.000     1.2980    34.6640
    214.920     1.3110    34.6660
    215.960     1.3170    34.6680
    216.960     1.3250    34.6680
    217.960     1.3320    34.6690
    218.960     1.3380    34.6690
    219.920     1.3410    34.6710
    221.000     1.3380    34.6720
    222.000     1.3380    34.6720
    222.920     1.3350    34.6720
    224.000     1.3310    34.6710
    225.120     1.3190    34.6740
    226.040     1.3120    34.6720
    227.040     1.3080    34.6720
    228.040     1.3060    34.6720
    229.040     1.3110    34.6720
    230.000     1.3160    34.6730
    231.000     1.3210    34.6740
    232.000     1.3220    34.6750
    233.960     1.3260    34.6750
    234.920     1.3270    34.6760
    236.040     1.3310    34.6760
    237.080     1.3330    34.6770
    238.040     1.3340    34.6770
    239.000     1.3350    34.6780
    240.000     1.3360    34.6780
    241.040     1.3410    34.6790
    241.960     1.3410    34.6790
    243.000     1.3410    34.6790
    244.040     1.3430    34.6800
    244.960     1.3470    34.6810
    245.960     1.3480    34.6820
    247.000     1.3490    34.6820
    248.040     1.3490    34.6820
    249.040     1.3490    34.6820
    250.040     1.3490    34.6820
    251.040     1.3490    34.6830
    252.040     1.3480    34.6830
    253.000     1.3470    34.6840
    254.000     1.3470    34.6840
    255.000     1.3490    34.6840
    255.960     1.3530    34.6840
    257.040     1.3590    34.6850
    257.960     1.3630    34.6860
    259.000     1.3660    34.6870
    260.000     1.3690    34.6870
    261.000     1.3700    34.6880
    262.040     1.3700    34.6880
    263.000     1.3710    34.6880
    264.000     1.3730    34.6880
    265.040     1.3730    34.6880
    266.000     1.3730    34.6890
    267.000     1.3740    34.6890
    268.120     1.3750    34.6890
    268.520     1.3760    34.6890
   -999.000     1.3780    34.6900
   -999.000     1.3820    34.6910
   -999.000     1.3830  -999.0000
# Surface Pressure Time Series
#    C npts n_var               Start
S   97     3     9 2020/01/31 22:45:18
# time   Pressure       Temp       Psal Mission                Date O2_Molarity O2_Temp O2_Phase
#  sec       dBar       degC        PSU   phase                 UTC      umol/L    degC  degrees
   125     -0.020     -0.449   -999.000      14 2020/01/31 22:47:23      -9.000  -9.000   -9.000
  1226      0.000     -0.453   -999.000      14 2020/01/31 23:05:44      -9.000  -9.000   -9.000
  1606     -0.070     -0.429   -999.000      14 2020/01/31 23:12:04      -9.000  -9.000   -9.000
# Fall Time Series
#  Cycle npts     Start                Date
F00   97   12 633822382 2020/01/31 21:46:22
# time     Pres phase                Date
     0   -0.080     1 2020/01/31 21:46:22
   611   42.880    10 2020/01/31 21:56:33
   912  106.680    10 2020/01/31 22:01:34
  1216  171.960     2 2020/01/31 22:06:38
  1219  172.680    10 2020/01/31 22:06:41
  1524  221.160    10 2020/01/31 22:11:46
  1668  231.200    10 2020/01/31 22:14:10
  1815  241.040    10 2020/01/31 22:16:37
  1965  251.080    10 2020/01/31 22:19:07
  2122  261.200    10 2020/01/31 22:21:44
  2154 9999.000    10 2020/01/31 22:22:16
  2157  263.400    10 2020/01/31 22:22:19
# Rise Time Series
#  Cycle npts     Start                Date
R00   97   20 633824551 2020/01/31 22:22:31
# time     Pres phase                Date
     0  264.120     7 2020/01/31 22:22:31
   264  254.840    13 2020/01/31 22:26:55
   334  244.440    13 2020/01/31 22:28:05
   405  232.520    13 2020/01/31 22:29:16
   476  218.960    13 2020/01/31 22:30:27
   547  204.120    13 2020/01/31 22:31:38
   617  188.040    13 2020/01/31 22:32:48
   688  171.280    13 2020/01/31 22:33:59
   759  153.320    13 2020/01/31 22:35:10
   829  135.080    13 2020/01/31 22:36:20
   900  116.760    13 2020/01/31 22:37:31
   970   98.280    13 2020/01/31 22:38:41
  1005   88.920    13 2020/01/31 22:39:16
  1076   70.760    13 2020/01/31 22:40:27
  1146   52.760    13 2020/01/31 22:41:37
  1217   35.480    13 2020/01/31 22:42:48
  1252   25.480    13 2020/01/31 22:43:23
  1288   14.880    13 2020/01/31 22:43:59
  1323    4.440    13 2020/01/31 22:44:34
  1366   -0.080     8 2020/01/31 22:45:17
# Pump Time Series
#   Cycle npts
B02    97   16
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
   171.960  302   14.82     261    613    -9     2
   264.120   17   14.86     329    608    -9     7
   266.800   35   14.84     295    611    -9    13
   268.160   35   14.84     297    614    -9    13
   263.240   36   14.84     282    617    -9    13
   254.840   35   14.82     281    621    -9    13
   244.440   35   14.83     267    624    -9    13
   232.520   36   14.81     268    627    -9    13
   218.960   36   14.81     274    630    -9    13
   204.120   35   14.81     265    633    -9    13
   188.040   35   14.80     246    637    -9    13
   171.280   36   14.80     236    640    -9    13
   144.400   36   14.80     225    644    -9    13
    52.760   36   14.81     207    653    -9    13
    35.480   35   14.81     199    653    -9    13
     0.000  120   14.80     184    663    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    97  2.4    8   500  1500   600   311    60     0     10    12    1245   20  0.4966
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   97 2020/01/31 21:46:18
# Pcase vacuums
   6.910   6.080
# Oil valve positions
   1.687   0.739   0.657
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   97 2020/01/31 22:22:20 9999/99/99 99:99:99
# Pcase vacuums
   6.080   6.080
# Average 
#   P      T      S
    0.000   0.000   0.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   97 2020/01/31 22:22:31 9999/99/99 99:99:99
# Pcase vacuums
   6.080   6.530
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   97 2020/01/31 22:45:18
# number of packets previous packets
      7              10
# CTD Status pres offset
 0x4e410000   0.510
# Pcase vacuums
   6.540   6.940
# Air Pump max current time  max pressure
       0      1094     6.673
# Air Valve Positions
   1.118   0.658
# Air Valve Calibrations
   1.041   0.099   0.570
# Oil Valve Positions
   1.687   0.736
# Oil Valve Calibrations
   1.634   0.786
# Encountered Error Codes
#    Cycle Errors
EE20    97      4
#               Date   Code Description
 2020/01/31 22:22:16 0x0913 DescendMaxDepth
 2020/01/31 23:18:43 0x0116 SurfaceCtd
 2020/01/31 23:18:45 0x0116 SurfaceCtd
 2020/01/31 23:18:46 0x0A16 SurfaceData
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   97  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.080   1210   5.890
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   97       198    2020/01/31 23:20:24
Parameter Table Param_Table-V17.txt |
     IFV = 6.8.116.IFV	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       96	(cycle number) |
     MPS =        0	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =        0	(count) |
  MNM[2] =        0	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        0	(count) |
  MNS[2] =        0	(count) |
  DTD[0] =     1500	(dBar) |
  DTD[1] =      400	(dBar) |
  DTD[2] =      400	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        1	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =      300	(dBar) |
  CPB[3] =      100	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        1	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        1	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       50	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =    30000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1800	(V) |
     PNC =       10	(inHg) |
     PXC =       50	(inHg) |
     PBL =       20	(12V cal) |
     PBH =     1200	(14V cal) |
     PP0 =      400	(oil pump 0 mA cal) |
     PP1 =     1500	(oil pump 1000 mA cal) |
     PVL =    43668	(V12 12V cal) |
     PVH =    50958	(V12 14V cal) |
     PV0 =        0	(vac 0 inHg cal) |
     PV5 =    25960	(vac 5 inHg cal) |
     PC0 =     1790	(V12_sig 0mA cal) |
     PC1 =     2089	(V12_sig 1000mA cal) |
     ICM =      240	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =    44669	(ice month mask) |
  ICA[0] =        0	(summer attempts) |
  ICA[1] =       64	(winter attempts) |
  ICB[0] =    55296	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =        5	(summer message threshold) |
  IMT[1] =        5	(winter message threshold) |
  ILT[0] =        1	(summer time threshold) |
  ILT[1] =        1	(winter time threshold) |
     ICN =       70	(shallow limit for test dBar) |
     ICX =       70	(deep limit for test dBar) |
     ICT =     1.80	(deg C) |
     ICS =      180	(sec) |
  ISS[0] =        0	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =       20	(winter 0=disable 1=always 2=urgent) |
     ISM =        0	(bit0=depth bit1=GPS bit2=salinity) |
     IST =   104.95	(PSU) |
     ISO =    23296	(sec) |
     ISP =       30	(0=disable 1=enbable) |
  IUE[0] =        2	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        2	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      256	(summer descent depth dBar) |
  ICD[1] =     5120	(winter descent depth dBar) |
  ICR[0] = -1275068416	(summer minutes) |
  ICR[1] =  6553700	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        0	(attempts) |
     ISD =     7680	(dBar) |
     IAT =        0	(secs) |
     ISV =     7680	(secs) |
     ISW =        1	(secs) |
  ICO[0] =       10	(summer secs) |
  ICO[1] =     1200	(winter secs) |
     ICF =  1967280	(summer mask) |
     ICW =  2949165	(winter ice mask) |
     DNS =        0	(cycle number) |
     DRA =        0	(target descent rate dBar/sec) |
  DSD[0] =        0	(dBar) |
  DSD[1] =        0	(dBar) |
  DSD[2] =       96	(dBar) |
  DST[0] =       20	(sec) |
  DST[1] =     2700	(sec) |
  DST[2] =      400	(sec) |
     DDF =      400	(sec) |
     DDC =     1200	(sec) |
     DSI = 471866400	(ms) |
     DDD =      300	(dBar) |
     DDT =        0	(sec) |
  DFT[0] =        4	(sec) |
  DFT[1] =    37856	(sec) |
  DFT[2] =       10	(sec) |
     DCM =        2	(0=dis 1=en) |
     DAL =       88	(0=cal 1=seek) |
  DCT[0] =       75	(sec) |
  DCT[1] =      130	(sec) |
  DCT[2] =      130	(sec) |
     DCA =        1	(max attempts) |
     DRC =      314	(cal rate calculation) |
     DRT =    38954	(sec) |
     DCX =    12330	(sec) |
     DCP =    12300	(sec) |
     DCF =        3	(sec) |
     DCD =        0	(dBar) |
     DCR =     1020	(dBar/sec) |
     DSW =        5	(sec) |
     DUC =        5	(under damping factor) |
     DPT =      100	(sec) |
     DUT =        0	(sec) |
     DPD =     1200	(sec) |
     DVT =      100	(sec) |
     DVD =        5	(dBar) |
     DVW =        5	(sec) |
     PFR =        5	(cc/min) |
     PFC =       30	(.001 cc/dBar) |
     DDP =       25	(dBar) |
     DDS =      300	(sec) |
     PNS =       19	(cycle) |
  PTO[0] =  1048676	(sec) |
  PTO[1] = 235929696	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        0	(0=avg 1=avg+raw) |
     PSI =        0	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =      256	(sec) |
  PFT[1] =      234	(sec) |
  PFT[2] =    24576	(sec) |
  PFD[0] =        0	(dBar) |
  PFD[1] =     3600	(dBar) |
  PFD[2] =     3600	(dBar) |
  PFV[0] =     3600	(sec) |
  PFV[1] =      500	(sec) |
  PFV[2] =        0	(sec) |
     PDT =        0	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =        0	(dBar) |
     PST =        0	(sec) |
     PSV =        1	(sec) |
     PSW =    11265	(sec) |
     ASN =      100	(cycle number) |
     ARR =     1200	(target rise rate cm/sec) |
     APM =       30	(sec) |
     APS =     1300	(sec) |
     APN =     1792	(sec) |
     APF =    24576	(sec) |
     ASI = 168386560	(ms) |
     ASD =     2560	(dBar) |
     ADD =     7680	(dBar) |
     ADT =     2560	(sec) |
     ARC =      117	(?) |
     AHR =    12288	(cm/sec) |
     AHD =      512	(dBar) |
     SFN =     2562	(cycle number) |
     SDC =    22539	(dBar) |
     SGT =    47107	(min) |
     SET =    59392	(min) |
     SBM =       10	(0=dis 1=en) |
     SBT =       96	(sec) |
     SDT =       15	(interval) |
     SSA =       15	(0=dis 1=en) |
     SSI =    76800	(ms) |
     SDS =        0	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =    10000	(.001 psi) |
     SAN =       60	(.001 psi) |
     SAO =      285	(sec) |
     SAP =    19481	(sec) |
     SPT =    25604	(sec) |
     SSP =    14340	(sec) |

