# Message Decoder for Solo2, Alto and Deep Solo Floats; v230727
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11042     6  340 000000000000009    73     0   4 2023/04/08, 13:20:02,  41.5683  -70.6851   4      0 13
SBD 11042     6  302 000000000000009    74     0   3 2023/04/08, 13:20:19,  41.5413  -70.6851   4      0 15,13,6,14,14
SBD 11042     6  335 000000000000009    75     0   2 2023/04/08, 13:20:29,  41.5413  -70.6851   4      0 8
SBD 11042     6  327 000000000000009    76     0   1 2023/04/08, 13:21:44,  41.5683  -70.6851   4      0 5,14,0,1,2,3,8
SBD 11042     6  328 000000000000009    77     0   0 2023/04/08, 13:21:57,  41.5683  -70.6851   4      0 0,4,14,14,9,10,11,7
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      22         1572           60        1632
# Sensor_ID bytes     % packets Description
          0    48  3.05       2         GPS
          1    25  1.59       1    Pressure
          2    25  1.59       1 Temperature
          3    25  1.59       1    Salinity
          4   138  8.78       1        Fall
          5    18  1.15       1        Rise
          6    37  2.35       1        Pump
          7    27  1.72       1 Drift_times
          8   502 31.93       2     Surface
          9    27  1.72       1  HR/Drift_P
         10    29  1.84       1  HR/Drift_T
         11    27  1.72       1  HR/Drift_S
         13   421 26.78       2  Parameters
         14   157  9.99       5 Engineering
         15    66  4.20       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G    6 1 2023/04/08 08:21 +41.53429  -70.64761    0  6 27 39 51  2.4     0
G    6 2 2023/04/08 13:09 +41.53440  -70.64754   20  6 33 41 49  1.5     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0     6    1    1     3         0
#      Pres      Temp    Salinity
      6.960    21.5490     0.3030
# Drift Data
#      C Npts comp n_var
I0     6    6    1     4
#      Pres       Temp   Salinity     Time
     19.800     0.0500     0.0100        6
     19.760     0.0500     0.0100     1812
     19.760     0.0500     0.0100     3617
     19.760     0.0700     0.0100     5423
     19.840     0.0600     0.0100     7229
     19.880     0.0800     0.0100     9034
# Surface Pressure Time Series
#    C npts n_var               Start
S    6    54     6 2023/04/08 13:16:16
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
     0      0.070     19.954   -999.000      14 2023/04/08 13:09:41
    13      0.080     19.953   -999.000      14 2023/04/08 13:09:54
    24      0.070     19.953   -999.000      14 2023/04/08 13:10:05
    35      0.070     19.954   -999.000      14 2023/04/08 13:10:16
    47      0.080     19.954   -999.000      14 2023/04/08 13:10:28
    58      0.070     19.954   -999.000      14 2023/04/08 13:10:39
    69      0.070     19.954   -999.000      14 2023/04/08 13:10:50
    80      0.080     19.954   -999.000      14 2023/04/08 13:11:01
    91      0.080     19.953   -999.000      14 2023/04/08 13:11:12
   102      0.080     19.955   -999.000      14 2023/04/08 13:11:23
   114      0.070     19.954   -999.000      14 2023/04/08 13:11:35
   125      0.090     19.955   -999.000      14 2023/04/08 13:11:46
   137      0.070     19.954   -999.000      14 2023/04/08 13:11:58
   149      0.060     19.955   -999.000      14 2023/04/08 13:12:10
   160      0.060     19.954   -999.000      14 2023/04/08 13:12:21
   172      0.080     19.955   -999.000      14 2023/04/08 13:12:33
   183      0.070     19.954   -999.000      14 2023/04/08 13:12:44
   194      0.070     19.955   -999.000      14 2023/04/08 13:12:55
   205      0.070     19.954   -999.000      14 2023/04/08 13:13:06
   217      0.070     19.955   -999.000      14 2023/04/08 13:13:18
   228      0.080     19.954   -999.000      14 2023/04/08 13:13:29
   239      0.070     19.954   -999.000      14 2023/04/08 13:13:40
   250      0.070     19.955   -999.000      14 2023/04/08 13:13:51
   262      0.070     19.955   -999.000      14 2023/04/08 13:14:03
   273      0.070     19.954   -999.000      14 2023/04/08 13:14:14
   284      0.070     19.955   -999.000      14 2023/04/08 13:14:25
   295      0.080     19.954   -999.000      14 2023/04/08 13:14:36
   307      0.070     19.954   -999.000      14 2023/04/08 13:14:48
   318      0.060     19.954   -999.000      14 2023/04/08 13:14:59
   329      0.070     19.954   -999.000      14 2023/04/08 13:15:10
   340      0.060     19.953   -999.000      14 2023/04/08 13:15:21
   351      0.080     19.954   -999.000      14 2023/04/08 13:15:32
   362      0.070     19.954   -999.000      14 2023/04/08 13:15:43
   373      0.080     19.955   -999.000      14 2023/04/08 13:15:54
   384      0.070     19.955   -999.000      14 2023/04/08 13:16:05
     0      0.070     19.954   -999.000      14 2023/04/08 13:16:16
    12      0.070     19.955   -999.000      14 2023/04/08 13:16:28
    23      0.080     19.954   -999.000      14 2023/04/08 13:16:39
    34      0.070     19.955   -999.000      14 2023/04/08 13:16:50
    45      0.080     19.955   -999.000      14 2023/04/08 13:17:01
    56      0.070     19.955   -999.000      14 2023/04/08 13:17:12
    67      0.060     19.956   -999.000      14 2023/04/08 13:17:23
    78      0.060     19.956   -999.000      14 2023/04/08 13:17:34
    89      0.070     19.956   -999.000      14 2023/04/08 13:17:45
   100      0.070     19.957   -999.000      14 2023/04/08 13:17:56
   111      0.070     19.957   -999.000      14 2023/04/08 13:18:07
   122      0.080     19.957   -999.000      14 2023/04/08 13:18:18
   133      0.060     19.957   -999.000      14 2023/04/08 13:18:29
   145      0.070     19.958   -999.000      14 2023/04/08 13:18:41
   156      0.070     19.958   -999.000      14 2023/04/08 13:18:52
   167      0.070     19.959   -999.000      14 2023/04/08 13:19:03
   178      0.080     19.958   -999.000      14 2023/04/08 13:19:14
   189      0.070     19.959   -999.000      14 2023/04/08 13:19:25
   200      0.070     19.960   -999.000      14 2023/04/08 13:19:36
# Fall Time Series
#  Cycle npts     Start                Date
F00    6   26 734257294 2023/04/08 08:21:34
# time     Pres phase                Date
     0    0.000     1 2023/04/08 08:21:34
   303    0.040    10 2023/04/08 08:26:37
   604    0.040    10 2023/04/08 08:31:38
   905    0.080    10 2023/04/08 08:36:39
  1206    0.040    10 2023/04/08 08:41:40
  1508    0.040    10 2023/04/08 08:46:42
  1806    0.080     2 2023/04/08 08:51:40
  1808    0.040    10 2023/04/08 08:51:42
  2108    0.040    10 2023/04/08 08:56:42
  2409    0.080    10 2023/04/08 09:01:43
  2709    0.040    10 2023/04/08 09:06:43
  3009    0.040    10 2023/04/08 09:11:43
  3309    0.040    10 2023/04/08 09:16:43
  3611    0.040    10 2023/04/08 09:21:45
  3630    0.080     3 2023/04/08 09:22:04
  3911    0.080    10 2023/04/08 09:26:45
  4211    0.040    10 2023/04/08 09:31:45
  4511    0.040    10 2023/04/08 09:36:45
  4567    0.080     3 2023/04/08 09:37:41
  4813    0.080    10 2023/04/08 09:41:47
  5113    0.080    10 2023/04/08 09:46:47
  5413    0.040    10 2023/04/08 09:51:47
  5713    0.080    10 2023/04/08 09:56:47
  6013    0.080    10 2023/04/08 10:01:47
  6313    0.080    10 2023/04/08 10:06:47
  6404    0.080    10 2023/04/08 10:08:18
# Rise Time Series
#  Cycle npts     Start                Date
R00    6    2 734274507 2023/04/08 13:08:27
# time     Pres phase                Date
     0    0.080     7 2023/04/08 13:08:27
     5    0.080     8 2023/04/08 13:08:32
# Pump Time Series
#   Cycle npts
B02     6    3
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     0.080   21   15.18     148    525    -9     2
     0.080    3   15.25     175    525    -9     7
     0.080   60   15.23     151    573    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2     6  2.4    2   500    10   600    31     0   180      5     2       2   15  0.2831
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17    6 2023/04/08 08:21:32
# Pcase vacuums
   7.150   5.240
# Oil valve positions
   1.806   0.715   0.635
# Seek attempts and pump time
      2      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18    6 2023/04/08 10:08:18 9999/99/99 99:99:99
# Pcase vacuums
   5.240   5.240
# Average 
#   P      T      S
   19.800   0.060   0.010
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18    6 2023/04/08 13:08:27 9999/99/99 99:99:99
# Pcase vacuums
   5.240   5.250
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20    6 2023/04/08 13:08:33
# number of packets previous packets
      8              53
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   5.250   8.180
# Air Pump max current time  max pressure
      41       670     6.926
# Air Valve Positions
   1.180   0.000
# Air Valve Calibrations
   1.122   0.088   0.605
# Oil Valve Positions
   1.810   0.711
# Oil Valve Calibrations
   1.746   0.773
# Encountered Error Codes
#    Cycle Errors
EE20     6      1
#               Date   Code Description
 2023/04/08 13:08:25 0x0615 ParkDepthMin
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00    6  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.040   1802   5.240
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M    6       201    2023/04/08 13:20:19
Parameter Table Param_Table-V17.txt |
     IFV = 10.5.0.4	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =        6	(cycle number) |
     MPS =        0	(phase) |
     MPC =        3	(count) |
  MNM[0] =        6	(count) |
  MNM[1] =        6	(count) |
  MNM[2] =      999	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        1	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        7	(dBar) |
     CRS =     1000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS =    21600	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =      800	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =    -1.65	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =        4	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        3	(dBar) |
  DSD[1] =        3	(dBar) |
  DSD[2] =        3	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =       20	(sec) |
  DFT[2] =       20	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        2	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =      900	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =       60	(sec) |
     PNS =        4	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =    65552	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     1800	(sec) |
  PFT[2] =    10800	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =       20	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =        5	(sec) |
     PDT =     1800	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =        4	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       20	(sec) |
     APN =       10	(sec) |
     APF =       50	(sec) |
     ASI =    10000	(ms) |
     ASD =        1	(dBar) |
     ADD =        1	(dBar) |
     ADT =      300	(sec) |
     ARC =     1100	(?) |
     AHR =     1000	(cm/sec) |
     AHD =        2	(dBar) |
     SFN =        4	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SSA =        0	(0=dis 1=en) |
     SSI =    10000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |
     PPT =   0.1300	(1) |
     PPA =   5.9000	(1) |
     PPD =   0.3500	(sec) |
     PPC =   0.0102	(1) |

