# Message Decoder for Solo2, Alto and Deep Solo Floats; v211115
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11031    -1  178 000000000750080     4     0   0 2020/02/28, 16:12:32,  41.9643  -88.0683   4      0 0,15,14
SBD 11031    -1  178 000000000750080     7     0   0 2021/11/15, 18:01:54,  41.5334  -70.6398   6      0 0,15,14
SBD 11031    -1  178 000000000750080    10     0   0 2021/11/19, 15:32:57,  41.5187  -70.7009   3      0 0,15,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0       9          498           36         534
# Sensor_ID bytes     % packets Description
          0    72 14.46       3         GPS
         14   228 45.78       3 Engineering
         15   198 39.76       3     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   -1 5 2021/11/19 15:31 +41.53397  -70.64686   30  6 37 40 46  1.6     0
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    -1  2.4    1  1000  1000   600   543     0    60     10     3    1545  180  0.9274
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
#       Alto BIT Test
#    cycle software   BIT         Total
#          version    result      time
E008   -1  0x06000100 0x00000000  1260
# Successful Bit Test
# CTD Status temp    salinity
 0x4e410000  12.098   0.006
# Pressure offsets 
# before   now     after [dBar]
    0.000    0.000    0.000
# Pcase vacuum [inHg]
   5.240   5.790
# Battery Voltages [V]
  13.390  13.330
# Air Pump test
# Air Valve Positions [V]
   1.137   0.045   0.614
# Air Valve Calibrations [V]
   1.092   0.097   0.594
# duration count max P max currents      max voltage
       0     0    0.00    0.00    0.00    0.00
# Oil Pump test
# Oil Valve Positions [V]
   1.832   0.854
# Oil Valve Calibrations [V]
   1.786   0.883
# duration max currents [mAmp]      max voltage [V]
     600    226.00   99.00         1.23
