MOD 3.1 solo/Alto Message_Decoder for SBD messages, format version 0.5-2.1
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11011    59  238 000000000000009  3103     0   0 2019/06/12, 20:04:33,  41.5334  -70.6398   5      0 5,14,14
SBD 11011    59  237 000000000000009  3104     0   5 2019/06/12, 20:04:42,  41.5334  -70.6398   5      0 6,14
SBD 11011    59  335 000000000000009  3105     0   4 2019/06/12, 20:04:51,  41.5413  -70.6851   5      0 6
SBD 11011    59  340 000000000000009  3106     0   3 2019/06/12, 20:05:14,  41.5683  -70.6851   5      0 13
SBD 11011    59  229 000000000000009  3107     0   2 2019/06/12, 20:05:33,  41.5762  -70.7303   6      0 11,15,13
SBD 11011    59  269 000000000000009  3108     0   1 2019/06/12, 20:05:49,  41.5683  -70.6851   6      0 1,2,3,9,10
SBD 11011    59  313 000000000000009  3109     0   0 2019/06/12, 20:06:11,  41.5379  -72.8091 182      0 0,4,14,14,9,10,11,7,14
# Raw packet information
#  Packets Bytes
P0      24  1877
# Sensor_ID bytes     % packets
          0    24  1.28       1
          1    32  1.70       1
          2    36  1.92       1
          3    32  1.70       1
          4    93  4.95       1
          5   183  9.75       1
          6   536 28.56       2
          7    29  1.55       1
          9   103  5.49       2
         10   112  5.97       2
         11   103  5.49       2
         13   405 21.58       2
         14   123  6.55       6
         15    66  3.52       1
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   59 1 9999/99/99 99:99 +100.00000 +200.00000   50  0 255  0  0  0.0     0
G   59 2 2019/04/18 13:27 +41.53423  -70.64695   10  9 26 33 43  0.9     0
# Profile Data
#     C   Npts comp n_var
D0    59    9    1     3
#      Pres      Temp      Salinity
      1.280    21.8370    30.6350
      2.040    21.2240    30.6180
      3.040    20.6450    30.5670
      4.040    20.3000    30.5440
      5.320    19.9460    30.5100
      5.960    19.7410    30.5040
      6.960    19.2880    30.4970
      8.000    18.2750    30.4920
      9.440    17.3070    30.4560
# High Resolution Data
#     C   Npts comp
H0    59  100    1
#      Pres      Temp      Salinity
      5.320    19.9590    30.5210
      5.320    19.9520    30.5210
      5.280    19.9540    30.5190
      5.320    19.9520    30.5200
      5.320    19.9500    30.5190
      5.320    19.9490    30.5170
      5.320    19.9480    30.5160
      5.320    19.9480    30.5150
      5.360    19.9470    30.5140
      5.320    19.9470    30.5140
      5.400    19.9480    30.5130
      5.400    19.9480    30.5130
      5.400    19.9480    30.5120
      5.400    19.9480    30.5120
      5.400    19.9480    30.5130
      5.400    19.9480    30.5130
      5.400    19.9470    30.5130
      5.400    19.9470    30.5130
      5.360    19.9460    30.5140
      5.360    19.9450    30.5150
      5.320    19.9450    30.5150
      5.320    19.9440    30.5160
      5.320    19.9430    30.5160
      5.320    19.9440    30.5160
      5.320    19.9440    30.5150
      5.360    19.9450    30.5150
      5.360    19.9450    30.5140
      5.400    19.9460    30.5130
      5.400    19.9470    30.5130
      5.400    19.9480    30.5120
      5.400    19.9480    30.5120
      5.360    19.9480    30.5110
      5.360    19.9490    30.5110
      5.320    19.9480    30.5110
      5.320    19.9480    30.5110
      5.320    19.9480    30.5120
      5.320    19.9480    30.5110
      5.360    19.9480    30.5110
      5.360    19.9490    30.5110
      5.400    19.9490    30.5110
      5.400    19.9490    30.5110
      5.400    19.9490    30.5100
      5.400    19.9490    30.5110
      5.400    19.9480    30.5110
      5.400    19.9470    30.5110
      5.400    19.9460    30.5110
      5.360    19.9450    30.5110
      5.320    19.9450    30.5110
      5.320    19.9440    30.5110
      5.320    19.9430    30.5110
      5.360    19.9440    30.5100
      5.360    19.9440    30.5100
      5.360    19.9440    30.5100
      5.400    19.9450    30.5100
      5.400    19.9450    30.5110
      5.440    19.9450    30.5120
      5.400    19.9450    30.5120
      5.360    19.9440    30.5120
      5.360    19.9430    30.5130
      5.320    19.9420    30.5130
      5.320    19.9420    30.5130
      5.360    19.9410    30.5130
      5.360    19.9400    30.5130
      5.400    19.9400    30.5120
      5.400    19.9400    30.5110
      5.400    19.9400    30.5100
      5.400    19.9410    30.5100
      5.400    19.9400    30.5100
      5.400    19.9400    30.5090
      5.360    19.9400    30.5090
      5.360    19.9390    30.5090
      5.360    19.9400    30.5080
      5.320    19.9390    30.5080
      5.320    19.9390    30.5070
      5.320    19.9390    30.5060
      5.320    19.9390    30.5060
      5.360    19.9380    30.5060
      5.360    19.9390    30.5050
      5.360    19.9390    30.5030
      5.400    19.9390    30.5030
      5.400    19.9400    30.5010
      5.400    19.9400    30.5000
      5.400    19.9400    30.4980
      5.400    19.9400    30.4970
      5.400    19.9400    30.4960
      5.400    19.9390    30.4950
      5.360    19.9390    30.4930
      5.360    19.9380    30.4920
      5.400    19.9380    30.4900
      5.320    19.9370    30.4880
      5.320    19.9370    30.4850
      5.320    19.9370    30.4820
      5.360    19.9360    30.4790
      5.360    19.9350    30.4760
      5.400    19.9330    30.4730
      5.400    19.9290    30.4720
      5.400    19.9200    30.4730
      5.440    19.9090    30.4780
      5.400    19.8970    30.4840
      5.400    19.8930    30.4860
# Drift Data
#     C   Npts comp
I0    59    9    1
#      Pres      Temp      Salinity
      9.440    17.2990    30.4540
      9.440    17.2990    30.4540
      9.440    17.2990    30.4540
      9.440    17.2990    30.4540
      9.440    17.2990    30.4540
      9.440    17.2990    30.4540
      9.440    17.2990    30.4540
      9.440    17.2990    30.4540
      9.440    17.2990    30.4540
# Fall Time Series
#  Cycle  npts  Start
#               seconds        Date
F00   59  17    613641831 2019/06/12 08:03:51
# time    Pres   phase Date
     0    0.040    1    2019/06/12 08:03:51
   171    1.200   10    2019/06/12 08:06:42
   180    2.280   10    2019/06/12 08:06:51
   188    3.480   10    2019/06/12 08:06:59
   192    4.120   10    2019/06/12 08:07:03
   200    5.400   10    2019/06/12 08:07:11
   206    6.320    2    2019/06/12 08:07:17
   209    6.840   10    2019/06/12 08:07:20
   214    7.480   10    2019/06/12 08:07:25
   218    8.080   10    2019/06/12 08:07:29
   226    9.200   10    2019/06/12 08:07:37
   528    9.440   10    2019/06/12 08:12:39
   831    9.440   10    2019/06/12 08:17:42
  1134    9.400   10    2019/06/12 08:22:45
  1437    9.440   10    2019/06/12 08:27:48
  1739    9.440   10    2019/06/12 08:32:50
  2034    9.440   10    2019/06/12 08:37:45
# Rise Time Series
#  Cycle  npts  Start
#               seconds        Date
R00   59  35    613659619 2019/06/12 13:00:19
# time    Pres   phase Date
     0    9.440    7    2019/06/12 13:00:19
    63    9.440   13    2019/06/12 13:01:22
   129    9.440   13    2019/06/12 13:02:28
   195    9.400   13    2019/06/12 13:03:34
   260    9.440   13    2019/06/12 13:04:39
   320    9.440   13    2019/06/12 13:05:39
   386    9.400   13    2019/06/12 13:06:45
   451    9.400   13    2019/06/12 13:07:50
   511    9.440   13    2019/06/12 13:08:50
   577    9.440   13    2019/06/12 13:09:56
   642    9.440   13    2019/06/12 13:11:01
   702    9.480   13    2019/06/12 13:12:01
   767    9.400   13    2019/06/12 13:13:06
   833    9.480   13    2019/06/12 13:14:12
   893    9.400   13    2019/06/12 13:15:12
   958    9.440   13    2019/06/12 13:16:17
  1018    9.400   13    2019/06/12 13:17:17
  1084    9.440   13    2019/06/12 13:18:23
  1149    9.360   13    2019/06/12 13:19:28
  1209    9.160   13    2019/06/12 13:20:28
  1215    8.840   13    2019/06/12 13:20:34
  1235    7.800   13    2019/06/12 13:20:54
  1255    6.680   13    2019/06/12 13:21:14
  1275    5.880   13    2019/06/12 13:21:34
  1335    5.360   13    2019/06/12 13:22:34
  1400    5.400   13    2019/06/12 13:23:39
  1420    4.960   13    2019/06/12 13:23:59
  1440    3.600   13    2019/06/12 13:24:19
  1453    2.480   13    2019/06/12 13:24:32
  1460    1.880   13    2019/06/12 13:24:39
  1775    1.280   13    2019/06/12 13:29:54
  2089    1.240   13    2019/06/12 13:35:08
  2404    1.240   13    2019/06/12 13:40:23
  2847    1.240   13    2019/06/12 13:47:46
  3736    1.320    8    2019/06/12 14:02:35
# Pump Time Series
#  Cycle  npts
B01   59  15
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    9.400     6   12.19     198  630     -9     13
    9.400     7   12.18     196  631     -9     13
    9.440     7   12.18     202  631     -9     13
    9.440     7   12.18     203  631     -9     13
    9.360     7   12.20     184  634     -9     13
    8.840     7   12.18     186  635     -9     13
    7.120     7   12.18     190  634     -9     13
    5.720     7   12.17     200  633     -9     13
    5.400     7   12.17     185  633     -9     13
    5.400     7   12.16     203  633     -9     13
    5.280     7   12.16     197  632     -9     13
    3.160     7   12.14     194  631     -9     13
    1.240   443   12.22     141  657     -9     13
    6.320    21   12.22     176  603     -9      2
    9.480     7   12.21     213  619     -9     13
# Argo Mission Data
#  Cycle Data  Miss Depth       Time                       Ascent Seeks Max Time           Cycle
#        Form  No   Prof  Park  Rise to Park to Prof Drift Speed        Surface  Seek      Time
A2   59  4.2   11    10     5    40    90       1      0    10     3     930      60       0.2001
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   59 2019/06/12 08:03:48
# Pcase vacuums
   7.110   6.120
# Oil valve positions
   1.756   0.849   0.570
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   59  2019/06/12 08:37:46 9999/99/99 99:99:99
# Pcase vacuums
   6.120   6.160
# Average 
#   P      T      S
    9.440  17.299  30.454
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   59  2019/06/12 13:00:19 9999/99/99 99:99:99
# Pcase vacuums
   0.590   6.550
# Alto Engineering: Surface
#       C  Mode  Status
EI17   59  0      4
# urgency attempts SBDs elapsed Time P min T test S test
  0      1     1      20989           1.240  -5.000   0.000
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   59  2      0
# MC    Pres      Time  Pcase Vacuum
   1     1.320    109   6.000
   1     0.040    200   5.990
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   59       196    2019/06/12 20:04:33
Parameter Table Param_Table-V17.txt |
     IFV = 6.5..0	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor) |
     MSN =       58	(cycle number) |
     MPS =        0	(phase) |
     MPC =       59	(count) |
  MMM[0] =     1000	(count) |
  MMM[1] =        0	(count) |
  MMM[2] =        0	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        0	(count) |
  MNS[2] =        0	(count) |
  DTD[0] =        5	(dBar) |
  DTD[1] =      400	(dBar) |
  DTD[2] =      400	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =      100	(HRD samples) |
     MTD =        1	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        1	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        0	(summer attempts) |
  ICA[1] =        0	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =       30	(winter time threshold) |
     ICN =        4	(shallow limit for test dBar) |
     ICX =        7	(deep limit for test dBar) |
     ICT =    19.00	(deg C) |
     ICS =        5	(sec) |
  ISS[0] =        0	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        0	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.00	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        1	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =        7	(summer descent depth dBar) |
  ICD[1] =        7	(winter descent depth dBar) |
  ICR[0] =       15	(summer minutes) |
  ICR[1] =        5	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       59	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =      400	(dBar) |
  DSD[2] =      400	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     7200	(sec) |
  DST[2] =     7200	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =      130	(sec) |
  DFT[2] =      130	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =    10800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =       60	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        5	(dBar) |
     DDS =     3600	(sec) |
     PNS =       59	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =   532610	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =     1000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     3600	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =       10	(dBar) |
  PFD[1] =        0	(dBar) |
  PFD[2] =        0	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        0	(sec) |
  PFV[2] =        0	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =       59	(cycle number) |
     ARR =       10	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       10	(sec) |
     APN =        5	(sec) |
     APF =       20	(sec) |
     ASI =     1000	(ms) |
     ASD =        2	(dBar) |
     ADD =        1	(dBar) |
     ADT =       60	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       33	(cycle number) |
     SDC =        0	(dBar) |
     SGT =        5	(min) |
     SET =       20	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =     1000	(0=dis 1=en) |
     SSI =   655645	(ms) |
     SDS =    19481	(sec) |
     SAM =      100	(0=dis 1=en) |
     SAX =      420	(.001 psi) |
     SAN =      600	(.001 psi) |
     SAO =      300	(sec) |
     SAP =       30	(sec) |

