MOD 3.1 solo/Alto Message_Decoder for SBD messages, format version 0.5-2.1
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11011    45  178 000000000000009  3014     0   0 2019/06/09, 08:02:04,  41.5413  -70.6851   2      0 5,14,14
SBD 11011    45  335 000000000000009  3015     0   4 2019/06/09, 08:02:13,  41.5187  -70.7009   2      0 6
SBD 11011    45  340 000000000000009  3016     0   3 2019/06/09, 08:02:39,  41.5187  -70.7009   2      0 13
SBD 11011    45  266 000000000000009  3017     0   2 2019/06/09, 08:02:51,  41.5334  -70.6398   2      0 11,15,13,14
SBD 11011    45  261 000000000000009  3018     0   1 2019/06/09, 08:03:22,  41.5187  -70.7009   2      0 2,3,9,10
SBD 11011    45  340 000000000000009  3019     0   0 2019/06/09, 08:03:35,  41.5187  -70.7009   2      0 0,4,14,14,9,10,11,7,14,1
# Raw packet information
#  Packets Bytes
P0      23  1648
# Sensor_ID bytes     % packets
          0    24  1.46       1
          1    32  1.94       1
          2    36  2.18       1
          3    32  1.94       1
          4    88  5.34       1
          5   123  7.46       1
          6   323 19.60       1
          7    29  1.76       1
          9   103  6.25       2
         10   136  8.25       2
         11   128  7.77       2
         13   405 24.58       2
         14   123  7.46       6
         15    66  4.00       1
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   45 1 2019/06/08 19:47 +41.53464  -70.64684   10  6 27 34 38  1.3     0
G   45 2 2019/03/27 16:26 +41.53429  -70.64685   30  7 27 35 41  1.0     0
# Profile Data
#     C   Npts comp n_var
D0    45    9    1     3
#      Pres      Temp      Salinity
      1.160    22.0760    30.5930
      2.000    21.0700    30.5830
      3.040    20.4270    30.5480
      4.040    20.0630    30.5230
      5.080    19.7520    30.5070
      5.920    19.5030    30.4890
      6.960    18.9980    30.4790
      8.000    17.8480    30.4710
      9.400    16.8400    30.4410
# High Resolution Data
#     C   Npts comp
H0    45  100    1
#      Pres      Temp      Salinity
      1.320    21.8570    30.5990
      1.400    21.7910    30.6080
      1.480    21.7110    30.6060
      1.480    21.6330    30.6030
      1.520    21.5760    30.5950
      1.560    21.5280    30.5910
      1.600    21.4890    30.5830
      1.640    21.4480    30.5880
      1.680    21.3890    30.5960
      1.720    21.3270    30.5960
      1.760    21.2640    30.5970
      1.840    21.1950    30.5950
      1.920    21.1210    30.6000
      1.960    21.0510    30.5960
      2.080    20.9910    30.5870
      2.120    20.9340    30.5870
      2.160    20.8780    30.5820
      2.240    20.8220    30.5850
      2.240    20.7840    30.5730
      2.280    20.7570    30.5660
      2.360    20.7420    30.5600
      2.400    20.7220    30.5650
      2.440    20.7010    30.5630
      2.480    20.6760    30.5600
      2.560    20.6460    30.5630
      2.640    20.6130    30.5620
      2.720    20.5790    30.5620
      2.800    20.5540    30.5520
      2.840    20.5330    30.5480
      2.920    20.5080    30.5520
      2.920    20.4810    30.5420
      2.960    20.4610    30.5450
      2.960    20.4270    30.5520
      3.120    20.3980    30.5460
      3.080    20.3650    30.5480
      3.160    20.3410    30.5430
      3.240    20.3220    30.5410
      3.280    20.2960    30.5470
      3.360    20.2670    30.5430
      3.440    20.2450    30.5380
      3.480    20.2200    30.5390
      3.560    20.2020    30.5340
      3.560    20.1840    30.5320
      3.600    20.1650    30.5320
      3.600    20.1550    30.5270
      3.680    20.1410    30.5280
      3.720    20.1310    30.5260
      3.720    20.1230    30.5240
      3.800    20.1150    30.5230
      3.840    20.1090    30.5240
      3.920    20.0980    30.5260
      4.000    20.0830    30.5260
      4.040    20.0730    30.5220
      4.080    20.0620    30.5230
      4.160    20.0470    30.5230
      4.200    20.0340    30.5200
      4.240    20.0250    30.5180
      4.280    20.0150    30.5180
      4.280    20.0070    30.5180
      4.320    19.9960    30.5190
      4.400    19.9790    30.5200
      4.360    19.9640    30.5180
      4.400    19.9520    30.5190
      4.440    19.9370    30.5220
      4.480    19.9230    30.5180
      4.520    19.9120    30.5170
      4.600    19.8990    30.5180
      4.640    19.8870    30.5160
      4.680    19.8720    30.5180
      4.720    19.8560    30.5160
      4.760    19.8450    30.5110
      4.800    19.8330    30.5120
      4.840    19.8190    30.5120
      4.880    19.8130    30.5090
      4.880    19.8020    30.5110
      4.880    19.7880    30.5120
      4.920    19.7740    30.5110
      4.920    19.7630    30.5100
      4.960    19.7550    30.5080
      5.000    19.7490    30.5070
      5.040    19.7450    30.5060
      5.040    19.7400    30.5060
      5.080    19.7350    30.5060
      5.120    19.7330    30.5030
      5.160    19.7290    30.5050
      5.200    19.7230    30.5030
      5.240    19.7190    30.5030
      5.280    19.7100    30.5040
      5.320    19.7020    30.5030
      5.320    19.6980    30.5010
      5.360    19.6930    30.5020
      5.360    19.6880    30.5010
      5.360    19.6850    30.5010
      5.400    19.6800    30.5010
      5.360    19.6730    30.5020
      5.360    19.6670    30.5030
      5.360    19.6600    30.5030
      5.400    19.6560    30.5030
      5.440    19.6520    30.5020
      5.440    19.6500    30.5000
# Drift Data
#     C   Npts comp
I0    45   10    1
#      Pres      Temp      Salinity
      9.440    16.8090    30.4420
      9.440    16.8090    30.4420
      9.440    16.8090    30.4420
      9.440    16.8090    30.4420
      9.440    16.8090    30.4420
      9.440    16.8090    30.4420
      9.440    16.8090    30.4420
      9.440    16.8090    30.4420
      9.440    16.8090    30.4420
      9.440    16.8090    30.4420
# Fall Time Series
#  Cycle  npts  Start
#               seconds        Date
F00   45  16    613338475 2019/06/08 19:47:55
# time    Pres   phase Date
     0    0.040    1    2019/06/08 19:47:55
   163    1.400   10    2019/06/08 19:50:38
   171    2.480   10    2019/06/08 19:50:46
   175    3.080   10    2019/06/08 19:50:50
   183    4.320   10    2019/06/08 19:50:58
   192    5.600   10    2019/06/08 19:51:07
   198    6.480    2    2019/06/08 19:51:13
   201    7.000   10    2019/06/08 19:51:16
   209    8.200   10    2019/06/08 19:51:24
   217    9.280   10    2019/06/08 19:51:32
   520    9.440   10    2019/06/08 19:56:35
   822    9.440   10    2019/06/08 20:01:37
  1125    9.440   10    2019/06/08 20:06:40
  1428    9.440   10    2019/06/08 20:11:43
  1731    9.440   10    2019/06/08 20:16:46
  2026    9.440   10    2019/06/08 20:21:41
# Rise Time Series
#  Cycle  npts  Start
#               seconds        Date
R00   45  23    613357218 2019/06/09 01:00:18
# time    Pres   phase Date
     0    9.440    7    2019/06/09 01:00:18
    64    9.360   13    2019/06/09 01:01:22
   130    9.440   13    2019/06/09 01:02:28
   196    9.400   13    2019/06/09 01:03:34
   262    9.480   13    2019/06/09 01:04:40
   328    9.440   13    2019/06/09 01:05:46
   388    9.400   13    2019/06/09 01:06:46
   454    9.440   13    2019/06/09 01:07:52
   520    9.440   13    2019/06/09 01:08:58
   586    9.440   13    2019/06/09 01:10:04
   652    9.440   13    2019/06/09 01:11:10
   679    9.000   13    2019/06/09 01:11:37
   712    7.680   13    2019/06/09 01:12:10
   732    6.720   13    2019/06/09 01:12:30
   758    6.000   13    2019/06/09 01:12:56
   812    4.880   13    2019/06/09 01:13:50
   839    3.720   13    2019/06/09 01:14:17
   859    2.840   13    2019/06/09 01:14:37
   879    1.640   13    2019/06/09 01:14:57
  1196    0.680   13    2019/06/09 01:20:14
  1513    0.680   13    2019/06/09 01:25:31
  1957    0.680   13    2019/06/09 01:32:55
  2822    0.240    8    2019/06/09 01:47:20
# Pump Time Series
#  Cycle  npts
B00   45  13
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    6.480    21   12.23     178  606     -9      2
    9.440    17   12.19     201  619     -9      7
    9.400     7   12.22     200  619     -9     13
    9.360     7   12.23     205  620     -9     13
    9.480     7   12.17     211  623     -9     13
    9.240     6   12.19     190  629     -9     13
    8.640     7   12.15     192  630     -9     13
    7.080     7   12.13     205  629     -9     13
    6.000     7   12.15     204  628     -9     13
    5.560     7   12.17     193  628     -9     13
    4.680     7   12.21     192  627     -9     13
    2.840     7   12.17     191  625     -9     13
    0.680   137   12.28     155  625     -9     13
# Argo Mission Data
#  Cycle Data  Miss Depth       Time                       Ascent Seeks Max Time           Cycle
#        Form  No   Prof  Park  Rise to Park to Prof Drift Speed        Surface  Seek      Time
A2   45  4.2   11    10     5    40    90       1      0    10     3     930      60       0.2001
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   45 2019/06/08 19:47:51
# Pcase vacuums
   7.030   6.150
# Oil valve positions
   1.757   0.848   0.575
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   45  2019/06/08 20:21:42 9999/99/99 99:99:99
# Pcase vacuums
   6.150   6.180
# Average 
#   P      T      S
    9.440  16.809  30.442
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   45  2019/06/09 01:00:19 9999/99/99 99:99:99
# Pcase vacuums
   0.450   6.230
# Alto Engineering: Surface
#       C  Mode  Status
EI17   45  1      4
# urgency attempts SBDs elapsed Time P min T test S test
  0      1     1      20755           0.680  14.166   0.000
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   45  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.240     68   6.020
   1     0.040    191   6.000
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   45       196    2019/06/09 08:02:04
Parameter Table Param_Table-V17.txt |
     IFV = 6.5..0	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor) |
     MSN =       44	(cycle number) |
     MPS =        0	(phase) |
     MPC =       45	(count) |
  MMM[0] =     1000	(count) |
  MMM[1] =        0	(count) |
  MMM[2] =        0	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        0	(count) |
  MNS[2] =        0	(count) |
  DTD[0] =        5	(dBar) |
  DTD[1] =      400	(dBar) |
  DTD[2] =      400	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =      100	(HRD samples) |
     MTD =        1	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        1	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =      255	(ice month mask) |
  ICA[0] =        6	(summer attempts) |
  ICA[1] =        2	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =       30	(winter time threshold) |
     ICN =        4	(shallow limit for test dBar) |
     ICX =        7	(deep limit for test dBar) |
     ICT =    19.00	(deg C) |
     ICS =        5	(sec) |
  ISS[0] =        1	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        0	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.00	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        1	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =        7	(summer descent depth dBar) |
  ICD[1] =        7	(winter descent depth dBar) |
  ICR[0] =       15	(summer minutes) |
  ICR[1] =        5	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       45	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =      400	(dBar) |
  DSD[2] =      400	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     7200	(sec) |
  DST[2] =     7200	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =      130	(sec) |
  DFT[2] =      130	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =    10800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =       60	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        5	(dBar) |
     DDS =     3600	(sec) |
     PNS =       45	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =   532610	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =     1000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     3600	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =       10	(dBar) |
  PFD[1] =        0	(dBar) |
  PFD[2] =        0	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        0	(sec) |
  PFV[2] =        0	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =       45	(cycle number) |
     ARR =       10	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       10	(sec) |
     APN =        5	(sec) |
     APF =       20	(sec) |
     ASI =     1000	(ms) |
     ASD =        2	(dBar) |
     ADD =        1	(dBar) |
     ADT =       60	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       24	(cycle number) |
     SDC =        0	(dBar) |
     SGT =        5	(min) |
     SET =       20	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =     1000	(0=dis 1=en) |
     SSI =   655645	(ms) |
     SDS =    19481	(sec) |
     SAM =      100	(0=dis 1=en) |
     SAX =      420	(.001 psi) |
     SAN =      600	(.001 psi) |
     SAO =      300	(sec) |
     SAP =       30	(sec) |

