MOD 3.1 solo/Alto Message_Decoder for SBD messages, format version 0.5-2.1
#     SN Cycle Size   IMEI         MOMSN MTMSN PID   Date     Time      Latitude Longitude Err Status
SBD 11011   20  329 000000000000009  2842     0  1 2019/06/03, 20:13:25,  41.5334  -70.6398   2   0
SBD 11011   20  340 000000000000009  2843     0  0 2019/06/03, 20:14:03,  41.5334  -70.6398   2   0
SBD 11011   20  335 000000000000009  2844     0  8 2019/06/03, 20:14:14,  41.5683  -70.6851   2   0
SBD 11011   20  335 000000000000009  2845     0  7 2019/06/03, 20:14:58,  41.5683  -70.6851   3   0
SBD 11011   20  339 000000000000009  2846     0  6 2019/06/03, 20:15:11,  41.5683  -70.6851   3   0
SBD 11011   20  340 000000000000009  2847     0  5 2019/06/03, 20:15:23,  41.5334  -70.6398   3   0
SBD 11011   20  305 000000000000009  2848     0  4 2019/06/03, 20:16:24,  41.5334  -70.6398   4   0
SBD 11011   20  277 000000000000009  2849     0  3 2019/06/03, 20:16:44,  41.5334  -70.6398   4   0
SBD 11011   20  308 000000000000009  2850     0  2 2019/06/03, 20:16:58,  41.5334  -70.6398   4   0
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   20 1 2019/04/08 15:09 +41.53421  -70.64682   10  9 25 34 41  0.8     0
G   20 2 2019/04/08 19:24 +41.53425  -70.64683   10 10 31 35 41  0.8     0
# Profile Data
#     C   Npts comp n_var
D0    20    9    1     3
#      Pres      Temp      Salinity
      1.240    20.7760    30.5510
      2.000    19.7370    30.5380
      3.000    19.1040    30.4840
      4.040    18.7900    30.4660
      5.000    18.5140    30.4550
      6.000    18.1850    30.4460
      7.000    17.5970    30.4510
      8.000    16.6640    30.4520
      9.400    15.8700    30.4270
# High Resolution Data
#     C   Npts comp
H0    20  100    1
#      Pres      Temp      Salinity
      1.240    20.7580    30.4940
      1.240    20.7180    30.5110
      1.240    20.6830    30.5220
      1.240    20.6800    30.5240
      1.240    20.6260    30.5600
      1.240    20.5950    30.5700
      1.240    20.5970    30.5620
      1.280    20.5940    30.5640
      1.240    20.5650    30.5800
      1.280    20.5800    30.5650
      1.280    20.6200    30.5380
      1.280    20.5990    30.5520
      1.280    20.5840    30.5550
      1.280    20.7040    30.4600
      1.320    20.7040    30.4630
      1.320    20.6720    30.4750
      1.320    20.6610    30.4730
      1.320    20.7120    30.4300
      1.320    20.6590    30.4700
      1.280    20.5580    30.5170
      1.280    20.5650    30.4990
      1.280    20.6870    30.4220
      1.240    20.6690    30.4490
      1.240    20.5430    30.5420
      1.240    20.5740    30.4990
      1.200    20.7210    30.3980
      1.200    20.7240    30.5260
      1.240    20.6730    30.5590
      1.280    20.6210    30.5470
      1.360    20.5820    30.5500
      1.440    20.5300    30.5520
      1.520    20.4630    30.5580
      1.560    20.3640    30.5720
      1.640    20.2290    30.5850
      1.720    20.0640    30.5900
      1.760    19.9260    30.5830
      1.840    19.8220    30.5560
      1.920    19.7430    30.5420
      1.960    19.6830    30.5300
      2.040    19.6430    30.5160
      2.080    19.6050    30.5180
      2.160    19.5630    30.5190
      2.200    19.5170    30.5170
      2.280    19.4760    30.5140
      2.280    19.4240    30.5170
      2.360    19.3700    30.5110
      2.400    19.3390    30.5030
      2.440    19.2960    30.5070
      2.520    19.2690    30.4990
      2.560    19.2450    30.4930
      2.600    19.2220    30.4950
      2.680    19.1910    30.4930
      2.720    19.1680    30.4910
      2.800    19.1500    30.4870
      2.840    19.1300    30.4860
      2.920    19.1110    30.4850
      2.960    19.0930    30.4850
      3.040    19.0760    30.4810
      3.080    19.0600    30.4790
      3.160    19.0450    30.4800
      3.240    19.0330    30.4760
      3.320    19.0180    30.4770
      3.360    19.0020    30.4750
      3.440    18.9860    30.4740
      3.480    18.9690    30.4750
      3.520    18.9520    30.4730
      3.600    18.9360    30.4730
      3.680    18.9200    30.4690
      3.720    18.9060    30.4710
      3.760    18.8900    30.4700
      3.800    18.8700    30.4710
      3.880    18.8500    30.4700
      3.920    18.8280    30.4710
      3.960    18.8060    30.4680
      4.040    18.7880    30.4660
      4.080    18.7690    30.4660
      4.160    18.7530    30.4630
      4.160    18.7360    30.4660
      4.200    18.7180    30.4640
      4.240    18.7060    30.4590
      4.280    18.6940    30.4600
      4.360    18.6840    30.4580
      4.400    18.6760    30.4570
      4.440    18.6620    30.4610
      4.480    18.6500    30.4580
      4.520    18.6400    30.4570
      4.600    18.6260    30.4570
      4.640    18.6150    30.4560
      4.680    18.6020    30.4570
      4.720    18.5860    30.4580
      4.800    18.5690    30.4580
      4.840    18.5530    30.4590
      4.880    18.5370    30.4570
      4.960    18.5230    30.4550
      5.000    18.5110    30.4540
      5.080    18.4980    30.4530
      5.120    18.4840    30.4540
      5.160    18.4670    30.4550
      5.240    18.4470    30.4550
      5.320    18.4280    30.4540
# Drift Data
#     C   Npts comp
I0    20    5    1
#      Pres      Temp      Salinity
      9.440    15.8440    30.4320
      9.440    15.8440    30.4320
      9.440    15.8440    30.4320
      9.440    15.8440    30.4320
      9.440    15.8440    30.4320
# Fall Time Series
#  Cycle  npts  Start
#               seconds        Date
F00   20  16    612806777 2019/06/02 16:06:17
# time    Pres   phase Date
     0    1.280    1    2019/06/02 16:06:17
    40    2.160   10    2019/06/02 16:06:57
    49    3.200   10    2019/06/02 16:07:06
    57    4.320   10    2019/06/02 16:07:14
    65    5.480   10    2019/06/02 16:07:22
    71    6.320    2    2019/06/02 16:07:28
    74    6.800   10    2019/06/02 16:07:31
    79    7.360   10    2019/06/02 16:07:36
    87    8.440   10    2019/06/02 16:07:44
    98    9.360   10    2019/06/02 16:07:55
   401    9.440   10    2019/06/02 16:12:58
   704    9.440   10    2019/06/02 16:18:01
  1007    9.440   10    2019/06/02 16:23:04
  1310    9.440   10    2019/06/02 16:28:07
  1613    9.440   10    2019/06/02 16:33:10
  1899    9.440   10    2019/06/02 16:37:56
# Rise Time Series
#  Cycle  npts  Start
#               seconds        Date
R00   20 122    612817219 2019/06/02 19:00:19
# time    Pres   phase Date
     0    9.400    7    2019/06/02 19:00:19
    63    9.400   13    2019/06/02 19:01:22
   123    9.360   13    2019/06/02 19:02:22
   189    9.440   13    2019/06/02 19:03:28
   255    9.400   13    2019/06/02 19:04:34
   321    9.400   13    2019/06/02 19:05:40
   387    9.360   13    2019/06/02 19:06:46
   447    9.440   13    2019/06/02 19:07:46
   513    9.360   13    2019/06/02 19:08:52
   579    9.360   13    2019/06/02 19:09:58
   625    8.880   13    2019/06/02 19:10:44
   645    7.560   13    2019/06/02 19:11:04
   658    6.680   13    2019/06/02 19:11:17
   672    5.880   13    2019/06/02 19:11:31
   692    4.680   13    2019/06/02 19:11:51
   712    3.760   13    2019/06/02 19:12:11
   725    2.920   13    2019/06/02 19:12:24
   745    1.760   13    2019/06/02 19:12:44
  1060    1.280   13    2019/06/02 19:17:59
  1375    1.200   13    2019/06/02 19:23:14
  1689    1.280   13    2019/06/02 19:28:28
  2132    1.240   13    2019/06/02 19:35:51
  2446    1.240   13    2019/06/02 19:41:05
  2594    1.240    9    2019/06/02 19:43:33
  3327    1.280   13    2019/06/02 19:55:46
  3388    1.280   13    2019/06/02 19:56:47
  3450    1.280   13    2019/06/02 19:57:49
  3511    1.280   13    2019/06/02 19:58:50
  3572    1.280   13    2019/06/02 19:59:51
  3634    1.280   13    2019/06/02 20:00:53
  3695    1.280   13    2019/06/02 20:01:54
  3757    1.280   13    2019/06/02 20:02:56
  3818    1.280   13    2019/06/02 20:03:57
  3880    1.280   13    2019/06/02 20:04:59
  3941    1.280   13    2019/06/02 20:06:00
  4003    1.280   13    2019/06/02 20:07:02
  4064    1.280   13    2019/06/02 20:08:03
  4125    1.280   13    2019/06/02 20:09:04
  4187    1.280   13    2019/06/02 20:10:06
  4538    1.280   13    2019/06/02 20:15:57
  4979    1.280   13    2019/06/02 20:23:18
  5301    9.440    9    2019/06/02 20:28:40
  5304    9.400   13    2019/06/02 20:28:43
  5365    9.440   13    2019/06/02 20:29:44
  5427    9.440   13    2019/06/02 20:30:46
  5488    9.440   13    2019/06/02 20:31:47
  5550    9.440   13    2019/06/02 20:32:49
  5611    9.440   13    2019/06/02 20:33:50
  5672    9.400   13    2019/06/02 20:34:51
  5734    9.440   13    2019/06/02 20:35:53
  5795    9.440   13    2019/06/02 20:36:54
  5857    9.440   13    2019/06/02 20:37:56
  5918    9.440   13    2019/06/02 20:38:57
  5980    9.440   13    2019/06/02 20:39:59
  6041    9.440   13    2019/06/02 20:41:00
  6103    9.400   13    2019/06/02 20:42:02
  6164    9.440   13    2019/06/02 20:43:03
  6224    9.440   13    2019/06/02 20:44:03
  6286    9.400   13    2019/06/02 20:45:05
  6346    9.440   13    2019/06/02 20:46:05
  6409    9.440   13    2019/06/02 20:47:08
  6474    9.440   13    2019/06/02 20:48:13
  6536    9.440   13    2019/06/02 20:49:15
  6598    9.440   13    2019/06/02 20:50:17
  6658    9.440   13    2019/06/02 20:51:17
  6718    9.440   13    2019/06/02 20:52:17
  6789    9.440   13    2019/06/02 20:53:28
  6852    9.440   13    2019/06/02 20:54:31
  6914    9.400   13    2019/06/02 20:55:33
  6931    8.840   13    2019/06/02 20:55:50
  6951    7.520   13    2019/06/02 20:56:10
  6963    6.720   13    2019/06/02 20:56:22
  6974    5.960   13    2019/06/02 20:56:33
  6994    4.840   13    2019/06/02 20:56:53
  7011    4.000   13    2019/06/02 20:57:10
  7034    2.800   13    2019/06/02 20:57:33
  7054    1.720   13    2019/06/02 20:57:53
  7367    1.280   13    2019/06/02 21:03:06
  7681    1.280   13    2019/06/02 21:08:20
  8122    1.280   13    2019/06/02 21:15:41
  8436    1.280   13    2019/06/02 21:20:55
  8757    9.400    9    2019/06/02 21:26:16
  8760    9.440   13    2019/06/02 21:26:19
  8822    9.400   13    2019/06/02 21:27:21
  8883    9.400   13    2019/06/02 21:28:22
  8945    9.440   13    2019/06/02 21:29:24
  9006    9.440   13    2019/06/02 21:30:25
  9068    9.400   13    2019/06/02 21:31:27
  9129    9.400   13    2019/06/02 21:32:28
  9190    9.440   13    2019/06/02 21:33:29
  9252    9.400   13    2019/06/02 21:34:31
  9313    9.400   13    2019/06/02 21:35:32
  9375    9.440   13    2019/06/02 21:36:34
  9436    9.440   13    2019/06/02 21:37:35
  9498    9.400   13    2019/06/02 21:38:37
  9559    9.400   13    2019/06/02 21:39:38
  9621    9.440   13    2019/06/02 21:40:40
  9681    9.400   13    2019/06/02 21:41:40
  9746    9.440   13    2019/06/02 21:42:45
  9808    9.400   13    2019/06/02 21:43:47
  9871    9.400   13    2019/06/02 21:44:50
  9933    9.440   13    2019/06/02 21:45:52
  9993    9.440   13    2019/06/02 21:46:52
 10055    9.400   13    2019/06/02 21:47:54
 10114    9.400   13    2019/06/02 21:48:53
 10180    9.440   13    2019/06/02 21:49:59
 10242    9.440   13    2019/06/02 21:51:01
 10305    9.440   13    2019/06/02 21:52:04
 10367    9.400   13    2019/06/02 21:53:06
 10429    9.400   13    2019/06/02 21:54:08
 10469    8.760   13    2019/06/02 21:54:48
 10486    7.800   13    2019/06/02 21:55:05
 10506    6.520   13    2019/06/02 21:55:25
 10517    5.800   13    2019/06/02 21:55:36
 10534    4.840   13    2019/06/02 21:55:53
 10551    3.880   13    2019/06/02 21:56:10
 10568    2.880   13    2019/06/02 21:56:27
 10585    1.800   13    2019/06/02 21:56:44
 10899    1.280   13    2019/06/02 22:01:58
 11213    1.280   13    2019/06/02 22:07:12
 11654    1.240   13    2019/06/02 22:14:33
 11803    1.280    8    2019/06/02 22:17:02
# Pump Time Series
#  Cycle  npts
B01   20  15
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    9.440     5   12.12     202  619     -9     13
    9.400     6   12.12     200  619     -9     13
    8.840     6   12.13     196  630     -9     13
    7.120     6   12.12     196  630     -9     13
    4.840     6   12.16     191  628     -9     13
    3.400     6   12.14     186  627     -9     13
    1.280   442   12.24     154  650     -9     13
    6.320    21   12.19     181  612     -9      2
    9.360     7   12.18     200  626     -9     13
    9.360     7   12.21     197  626     -9     13
    8.880     7   12.17     207  633     -9     13
    7.160     7   12.19     202  632     -9     13
    5.080     7   12.21     187  631     -9     13
    3.760     7   12.18     183  630     -9     13
    1.760   314   12.18     168  644     -9     13
# Argo Mission Data
#  Cycle Data  Miss Depth       Time                       Ascent Seeks Max Time           Cycle
#        Form  No   Prof  Park  Rise to Park to Prof Drift Speed        Surface  Seek      Time
A2   20  4.2   11    10     5    40    90       1      0    10     3     930      60       0.2001
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   20 2019/06/02 16:06:14
# Pcase vacuums
   6.280   6.200
# Oil valve positions
   1.759   0.850   0.582
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   20  2019/06/02 16:37:59 9999/99/99 99:99:99
# Pcase vacuums
   6.200   6.230
# Average 
#   P      T      S
    9.440  15.844  30.432
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   20  2019/06/02 19:00:19 9999/99/99 99:99:99
# Pcase vacuums
   0.200   6.400
# Alto Engineering: Surface
#       C  Mode  Status
EI17   20  0      4
# urgency attempts SBDs elapsed Time P min T test S test
  1      1    37     107852           1.200  -5.000   0.000
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   20  4      0
# MC    Pres      Time  Pcase Vacuum
   1     1.280     65   6.070
   9     1.320    183   6.130
   9     1.320    183   6.140
   1     1.240     77   6.050
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   20       196    2019/06/03 20:13:25
Parameter Table Param_Table-V17.txt |
     IFV = 6.5..0	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor) |
     MSN =       19	(cycle number) |
     MPS =        0	(phase) |
     MPC =       20	(count) |
  MMM[0] =     1000	(count) |
  MMM[1] =        0	(count) |
  MMM[2] =        0	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        0	(count) |
  MNS[2] =        0	(count) |
  DTD[0] =        5	(dBar) |
  DTD[1] =      400	(dBar) |
  DTD[2] =      400	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =      100	(HRD samples) |
     MTD =        1	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        1	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =       31	(ice month mask) |
  ICA[0] =        4	(summer attempts) |
  ICA[1] =        0	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =        1	(summer time threshold) |
  ILT[1] =       30	(winter time threshold) |
     ICN =        4	(shallow limit for test dBar) |
     ICX =        7	(deep limit for test dBar) |
     ICT =    19.00	(deg C) |
     ICS =        5	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        0	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.00	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        1	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =        7	(summer descent depth dBar) |
  ICD[1] =        7	(winter descent depth dBar) |
  ICR[0] =       15	(summer minutes) |
  ICR[1] =        5	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       20	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =      400	(dBar) |
  DSD[2] =      400	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     7200	(sec) |
  DST[2] =     7200	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =      130	(sec) |
  DFT[2] =      130	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =    10800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =       60	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        5	(dBar) |
     DDS =     3600	(sec) |
     PNS =       20	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =   532610	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =     1000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     3600	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =       10	(dBar) |
  PFD[1] =        0	(dBar) |
  PFD[2] =        0	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        0	(sec) |
  PFV[2] =        0	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =       20	(cycle number) |
     ARR =       10	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       10	(sec) |
     APN =        5	(sec) |
     APF =       20	(sec) |
     ASI =     1000	(ms) |
     ASD =        2	(dBar) |
     ADD =        1	(dBar) |
     ADT =       60	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =       15	(cycle number) |
     SDC =        0	(dBar) |
     SGT =        5	(min) |
     SET =       20	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =     1000	(0=dis 1=en) |
     SSI =   655645	(ms) |
     SDS =    19481	(sec) |
     SAM =      100	(0=dis 1=en) |
     SAX =      420	(.001 psi) |
     SAN =      600	(.001 psi) |
     SAO =      300	(sec) |
     SAP =       30	(sec) |

