MOD 3.1 solo/Alto Message_Decoder for SBD messages, format version 0.5-2.1
#     SN Cycle Size   IMEI         MOMSN MTMSN PID   Date     Time      Latitude Longitude Err Status
SBD 11011    0  118 000000000000009  2690     0  5 2019/05/28, 15:28:00,  41.5108  -70.6557   2   0
SBD 11011    0  340 000000000000009  2691     0  4 2019/05/28, 15:28:12,  41.5108  -70.6557   2   0
SBD 11011    0  336 000000000000009  2692     0  3 2019/05/28, 15:28:25,  41.5108  -70.6557   2   0
SBD 11011    0  264 000000000000009  2693     0  2 2019/05/28, 15:28:40,  41.5108  -70.6557   3   0
SBD 11011    0  301 000000000000009  2694     0  1 2019/05/28, 15:32:05,  41.5334  -70.6398   3   0
SBD 11011    0  321 000000000000009  2695     0  0 2019/05/28, 15:32:16,  41.5334  -70.6398   3   0
SBD 11011    0  314 000000000000009  2696     0  5 2019/05/28, 19:42:44,  41.5683  -70.6851   3   0
SBD 11011    0  183 000000000000009  2697     0  4 2019/05/28, 19:44:41,  41.5413  -70.6851   4   0
SBD 11011    0  340 000000000000009  2698     0  3 2019/05/28, 19:44:50,  41.5413  -70.6851   4   0
SBD 11011    0  338 000000000000009  2699     0  2 2019/05/28, 19:45:01,  41.5413  -70.6851   4   0
SBD 11011    0  277 000000000000009  2700     0  1 2019/05/28, 19:48:36,  41.5334  -70.6398   3   0
SBD 11011    0  279 000000000000009  2701     0  0 2019/05/28, 19:48:58,  41.5334  -70.6398   4   0
SBD 11011    0  333 000000000000009  2702     0  5 2019/05/28, 19:49:08,  41.5334  -70.6398   4   0
SBD 11011    0  318 000000000000009  2703     0  2 2019/05/28, 19:49:21,  41.5334  -70.6398   4   0
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G    0 0 2019/05/28 19:22 +41.53438  -70.64677   10  5 31 35 41  0.2     0
# Profile Data
#     C   Npts comp n_var
D0     0    9    1     3
#      Pres      Temp      Salinity
      1.160    22.2080    30.5040
      2.000    20.8250    30.4880
      3.000    19.9470    30.4460
      4.040    19.4400    30.4370
      5.000    19.0810    30.4310
      6.040    18.7480    30.4260
      7.000    18.3120    30.4190
      8.040    16.6650    30.4310
      9.320    15.1260    30.4220
# High Resolution Data
#     C   Npts comp
H0     0   98    1
#      Pres      Temp      Salinity
      0.920    24.0600    30.3290
      0.920    22.3930    30.2930
      1.000    22.3770    30.3010
      1.000    22.3560    30.2920
      1.080    22.3390    30.2840
      1.080    22.3240    30.2770
      1.160    22.2950    30.2830
      1.160    22.2440    30.3060
      1.240    22.1920    30.3640
      1.280    22.1050    30.3990
      1.360    22.0150    30.3390
      1.440    21.9060    30.3490
      1.480    21.7550    30.3970
      1.520    21.5990    30.4200
      1.600    21.4500    30.4330
      1.680    21.3070    30.4260
      1.720    21.1650    30.4400
      1.760    21.0450    30.4150
      1.840    20.9440    30.4150
      1.920    20.8640    30.4250
      2.000    20.7870    30.4270
      2.040    20.7280    30.4320
      2.080    20.6630    30.4240
      2.160    20.6080    30.4250
      2.240    20.5470    30.4230
      2.280    20.4700    30.4160
      2.360    20.3980    30.4180
      2.400    20.3380    30.4160
      2.480    20.2870    30.4210
      2.520    20.2440    30.4250
      2.600    20.2080    30.4220
      2.640    20.1740    30.4160
      2.720    20.1400    30.4130
      2.760    20.1050    30.4120
      2.840    20.0660    30.4120
      2.920    20.0220    30.4100
      2.960    19.9790    30.4140
      3.040    19.9350    30.4170
      3.080    19.8910    30.4190
      3.120    19.8530    30.4240
      3.200    19.8160    30.4240
      3.240    19.7830    30.4250
      3.280    19.7550    30.4250
      3.360    19.7310    30.4200
      3.400    19.7100    30.4220
      3.480    19.6880    30.4240
      3.520    19.6660    30.4270
      3.560    19.6460    30.4260
      3.600    19.6210    30.4230
      3.640    19.5920    30.4210
      3.720    19.5640    30.4220
      3.760    19.5400    30.4220
      3.800    19.5200    30.4210
      3.840    19.5010    30.4300
      3.920    19.4830    30.4290
      3.960    19.4650    30.4270
      4.000    19.4490    30.4260
      4.040    19.4320    30.4250
      4.080    19.4160    30.4250
      4.120    19.3990    30.4240
      4.160    19.3840    30.4250
      4.200    19.3670    30.4250
      4.240    19.3500    30.4240
      4.280    19.3320    30.4240
      4.320    19.3150    30.4240
      4.400    19.2970    30.4230
      4.360    19.2800    30.4230
      4.440    19.2630    30.4230
      4.480    19.2470    30.4220
      4.520    19.2310    30.4220
      4.560    19.2160    30.4220
      4.600    19.2020    30.4220
      4.640    19.1890    30.4230
      4.680    19.1770    30.4230
      4.720    19.1650    30.4230
      4.760    19.1540    30.4220
      4.800    19.1450    30.4210
      4.840    19.1370    30.4210
      4.840    19.1280    30.4190
      4.920    19.1190    30.4190
      4.920    19.1090    30.4200
      4.960    19.0990    30.4200
      5.000    19.0870    30.4200
      5.040    19.0740    30.4200
      5.080    19.0630    30.4210
      5.120    19.0520    30.4220
      5.160    19.0420    30.4220
      5.160    19.0310    30.4210
      5.200    19.0200    30.4200
      5.240    19.0100    30.4200
      5.280    19.0010    30.4190
      5.320    18.9920    30.4180
      5.360    18.9830    30.4170
      5.400    18.9740    30.4180
      5.400    18.9660    30.4180
      5.480    18.9580    30.4180
      5.480    18.9500    30.4190
      5.520    18.9410    30.4190
# Drift Data
#     C   Npts comp
I0     0    6    1
#      Pres      Temp      Salinity
      9.400    15.0740    30.4260
      9.400    15.0740    30.4260
      9.400    15.0740    30.4260
      9.400    15.0740    30.4260
      9.400    15.0740    30.4260
      9.400    15.0740    30.4260
# Surface Pressure Time Series
#    C  npts index Start (sec) relative to 1 Jan 2000
S    0  59    0   612370849 2019/05/28 15:00:49
# time      Pres      Temp       Psal   phase  Date
    43      0.210    21.6020  -999.0000   14 2019/05/28 15:00:49
   477      0.040    21.0370  -999.0000   14 2019/05/28 15:08:03
   523      0.040    20.9540  -999.0000   14 2019/05/28 15:08:49
   534      0.040    20.8910  -999.0000   14 2019/05/28 15:09:00
   544      0.020    20.8510  -999.0000   14 2019/05/28 15:09:10
   554      0.030    20.8140  -999.0000   14 2019/05/28 15:09:20
   565      0.040    20.7910  -999.0000   14 2019/05/28 15:09:31
   575      0.040    20.7900  -999.0000   14 2019/05/28 15:09:41
   585      0.040    20.7930  -999.0000   14 2019/05/28 15:09:51
   595      0.010    20.8130  -999.0000   14 2019/05/28 15:10:01
   606      0.040    20.8200  -999.0000   14 2019/05/28 15:10:12
   616      0.040    20.8660  -999.0000   14 2019/05/28 15:10:22
   626      0.050    20.8990  -999.0000   14 2019/05/28 15:10:32
   637      0.060    20.9200  -999.0000   14 2019/05/28 15:10:43
   647      0.040    20.9350  -999.0000   14 2019/05/28 15:10:53
   657      0.040    20.9360  -999.0000   14 2019/05/28 15:11:03
   667      0.020    20.9460  -999.0000   14 2019/05/28 15:11:13
   678      0.030    20.9530  -999.0000   14 2019/05/28 15:11:24
   688      0.040    20.9550  -999.0000   14 2019/05/28 15:11:34
   698      0.040    20.9490  -999.0000   14 2019/05/28 15:11:44
   709      0.030    20.9460  -999.0000   14 2019/05/28 15:11:55
   719      0.030    20.9640  -999.0000   14 2019/05/28 15:12:05
   729      0.030    20.9680  -999.0000   14 2019/05/28 15:12:15
   739      0.040    20.9860  -999.0000   14 2019/05/28 15:12:25
   750      0.030    20.9760  -999.0000   14 2019/05/28 15:12:36
   760      0.040    20.9880  -999.0000   14 2019/05/28 15:12:46
   779      0.030    21.0070  -999.0000   14 2019/05/28 15:13:05
   789      0.030    20.9780  -999.0000   14 2019/05/28 15:13:15
   800      0.030    20.9910  -999.0000   14 2019/05/28 15:13:26
   810      0.050    21.0150  -999.0000   14 2019/05/28 15:13:36
   815      0.030    21.0250  -999.0000   14 2019/05/28 15:13:41
    35      0.120    22.3550  -999.0000   14 2019/05/28 19:15:17
   469     -0.090    19.5510  -999.0000   14 2019/05/28 19:22:31
   513     -0.090    19.5460  -999.0000   14 2019/05/28 19:23:15
   524     -0.090    19.5220  -999.0000   14 2019/05/28 19:23:26
   543     -0.110    19.5130  -999.0000   14 2019/05/28 19:23:45
   553     -0.080    19.4990  -999.0000   14 2019/05/28 19:23:55
   563     -0.090    19.4680  -999.0000   14 2019/05/28 19:24:05
   573     -0.090    19.4410  -999.0000   14 2019/05/28 19:24:15
   587     -0.090    19.4390  -999.0000   14 2019/05/28 19:24:29
   606     -0.100    19.4210  -999.0000   14 2019/05/28 19:24:48
   617     -0.100    19.4170  -999.0000   14 2019/05/28 19:24:59
   636     -0.090    19.4270  -999.0000   14 2019/05/28 19:25:18
   649     -0.080    19.4430  -999.0000   14 2019/05/28 19:25:31
   660     -0.090    19.4460  -999.0000   14 2019/05/28 19:25:42
   670     -0.090    19.4560  -999.0000   14 2019/05/28 19:25:52
   680     -0.090    19.4650  -999.0000   14 2019/05/28 19:26:02
   690     -0.090    19.4580  -999.0000   14 2019/05/28 19:26:12
   701     -0.090    19.4400  -999.0000   14 2019/05/28 19:26:23
   711     -0.090    19.4210  -999.0000   14 2019/05/28 19:26:33
   721     -0.100    19.4190  -999.0000   14 2019/05/28 19:26:43
   732     -0.080    19.4270  -999.0000   14 2019/05/28 19:26:54
   746     -0.080    19.4540  -999.0000   14 2019/05/28 19:27:08
   756     -0.090    19.4340  -999.0000   14 2019/05/28 19:27:18
   766     -0.080    19.4250  -999.0000   14 2019/05/28 19:27:28
   776     -0.080    19.4200  -999.0000   14 2019/05/28 19:27:38
   787     -0.090    19.4230  -999.0000   14 2019/05/28 19:27:49
   801     -0.090    19.4500  -999.0000   14 2019/05/28 19:28:03
   806     -0.090    19.4510  -999.0000   14 2019/05/28 19:28:08
# Fall Time Series
#  Cycle  npts  Start
#               seconds        Date
F00    0  29    612370413 2019/05/28 14:53:33
# time    Pres   phase Date
     0    9.440    1    2019/05/28 14:53:33
     8    9.440    2    2019/05/28 14:53:41
   155    8.840   10    2019/05/28 14:56:08
   171    7.720   10    2019/05/28 14:56:24
   184    6.760   10    2019/05/28 14:56:37
   196    5.880   10    2019/05/28 14:56:49
   212    4.720   10    2019/05/28 14:57:05
   225    3.640   10    2019/05/28 14:57:18
   233    2.840   10    2019/05/28 14:57:26
   241    2.000   10    2019/05/28 14:57:34
   254    0.680   10    2019/05/28 14:57:47
   315    0.280   10    2019/05/28 14:58:48
  2345    0.000    1    2019/05/28 15:32:38
  2545    1.200   10    2019/05/28 15:35:58
  2554    2.200   10    2019/05/28 15:36:07
  2562    3.240   10    2019/05/28 15:36:15
  2570    4.360   10    2019/05/28 15:36:23
  2579    5.480   10    2019/05/28 15:36:32
  2584    6.280    2    2019/05/28 15:36:37
  2588    6.720   10    2019/05/28 15:36:41
  2592    7.280   10    2019/05/28 15:36:45
  2600    8.240   10    2019/05/28 15:36:53
  2612    9.440   10    2019/05/28 15:37:05
  2915    9.440   10    2019/05/28 15:42:08
  3217    9.440   10    2019/05/28 15:47:10
  3520    9.440   10    2019/05/28 15:52:13
  3823    9.440   10    2019/05/28 15:57:16
  4126    9.400   10    2019/05/28 16:02:19
  4412    9.400   10    2019/05/28 16:07:05
# Rise Time Series
#  Cycle  npts  Start
#               seconds        Date
R00    0  21    612370748 2019/05/28 14:59:08
# time    Pres   phase Date
     0    0.240    7    2019/05/28 14:59:08
    56    0.240    8    2019/05/28 15:00:04
 14470    9.360    7    2019/05/28 19:00:18
 14533    9.400   13    2019/05/28 19:01:21
 14593    9.360   13    2019/05/28 19:02:21
 14659    9.320   13    2019/05/28 19:03:27
 14724    9.360   13    2019/05/28 19:04:32
 14784    9.400   13    2019/05/28 19:05:32
 14849    9.400   13    2019/05/28 19:06:37
 14909    9.360   13    2019/05/28 19:07:37
 14968    8.880   13    2019/05/28 19:08:36
 15001    7.880   13    2019/05/28 19:09:09
 15041    7.000   13    2019/05/28 19:09:49
 15067    7.000   13    2019/05/28 19:10:15
 15074    6.920   13    2019/05/28 19:10:22
 15133    5.880   13    2019/05/28 19:11:21
 15166    4.920   13    2019/05/28 19:11:54
 15193    3.920   13    2019/05/28 19:12:21
 15213    2.840   13    2019/05/28 19:12:41
 15233    1.600   13    2019/05/28 19:13:01
 15332    0.160    8    2019/05/28 19:14:40
# Pump Time Series
#  Cycle  npts
B00    0  18
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    9.440   300   12.07     155  622     -9      2
    0.240    53   12.14     146  623     -9      7
    0.000    30   12.15     150  625     -9     14
    6.280    21   12.02     191  602     -9      2
    9.360    18   12.08     216  625     -9      7
    9.400     7   12.12     221  626     -9     13
    9.280     7   12.11     204  626     -9     13
    9.320     7   12.09     220  627     -9     13
    9.120     7   12.08     214  633     -9     13
    8.040     7   12.08     201  633     -9     13
    7.080     7   12.10     201  633     -9     13
    6.920     7   12.11     205  632     -9     13
    6.720     7   12.06     213  632     -9     13
    6.000     7   12.07     201  630     -9     13
    5.160     7   12.07     205  629     -9     13
    3.920     7   12.05     212  628     -9     13
    2.040   100   12.21     179  627     -9     13
   -0.040    30   12.15     191  627     -9     14
# Argo Mission Data
#  Cycle Data  Miss Depth       Time                       Ascent Seeks Max Time           Cycle
#        Form  No   Prof  Park  Rise to Park to Prof Drift Speed        Surface  Seek      Time
A2    0  4.2   13    10 21184    40    65       1      0    10     3     930      60       0.1828
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17    0 2019/05/28 15:32:35
# Pcase vacuums
   7.220   6.170
# Oil valve positions
   1.756   0.848   0.563
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18    0  2019/05/28 16:07:06 9999/99/99 99:99:99
# Pcase vacuums
   6.170   6.250
# Average 
#   P      T      S
    9.400  15.074  30.426
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18    0  2019/05/28 19:00:18 9999/99/99 99:99:99
# Pcase vacuums
   6.250   6.260
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20    0  2019/05/28 15:00:06
# number of packets previous packets
      0               0
# CTD Status pres offset
 0x00000000   0.000
# Pcase vacuums
   0.000   0.000
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   0.000   0.000   0.000
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   0.000   0.000
# Encountered Error Codes
#       C number of Errors
EE20    0 0
# Alto Engineering: Surface
#       C  Mode  Status
EI17    0  0      5
# urgency attempts SBDs elapsed Time P min T test S test
  0      0     2      13260           1.600  -5.000   0.000
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00    0  2      0
# MC    Pres      Time  Pcase Vacuum
   1     9.440      2   6.060
   1    -0.040    232   5.970
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M    0       196    2019/05/28 15:28:00
Parameter Table Param_Table-V17.txt |
     IFV = 6.5..0	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor) |
     MSN =    65535	(cycle number) |
     MPS =        0	(phase) |
     MPC =        0	(count) |
  MMM[0] =     1000	(count) |
  MMM[1] =        0	(count) |
  MMM[2] =        0	(count) |
  MNS[0] =        0	(count) |
  MNS[1] =        0	(count) |
  MNS[2] =        0	(count) |
  DTD[0] =        5	(dBar) |
  DTD[1] =      400	(dBar) |
  DTD[2] =      400	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =      100	(HRD samples) |
     MTD =        1	(dBar) |
     MTT =    30000	(ms) |
     MTS =    10800	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        1	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        2	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =       30	(summer time threshold) |
  ILT[1] =       30	(winter time threshold) |
     ICN =        4	(shallow limit for test dBar) |
     ICX =        7	(deep limit for test dBar) |
     ICT =    16.40	(deg C) |
     ICS =        5	(sec) |
  ISS[0] =        0	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        0	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.00	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =        7	(summer descent depth dBar) |
  ICD[1] =        7	(winter descent depth dBar) |
  ICR[0] =       15	(summer minutes) |
  ICR[1] =        5	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =        0	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =      400	(dBar) |
  DSD[2] =      400	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     7200	(sec) |
  DST[2] =     7200	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =      130	(sec) |
  DFT[2] =      130	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =    10800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =       60	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        5	(dBar) |
     DDS =     3600	(sec) |
     PNS =        0	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =   532610	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =     1000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     3600	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =       10	(dBar) |
  PFD[1] =        0	(dBar) |
  PFD[2] =        0	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        0	(sec) |
  PFV[2] =        0	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =        0	(cycle number) |
     ARR =       10	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APS =       10	(sec) |
     APN =        5	(sec) |
     APF =       20	(sec) |
     ASI =     1000	(ms) |
     ASD =        2	(dBar) |
     ADD =        1	(dBar) |
     ADT =       60	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =        0	(cycle number) |
     SDC =        0	(dBar) |
     SGT =        5	(min) |
     SET =       20	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =     1000	(0=dis 1=en) |
     SSI =   655645	(ms) |
     SDS =    19481	(sec) |
     SAM =      100	(0=dis 1=en) |
     SAX =      420	(.001 psi) |
     SAN =      600	(.001 psi) |
     SAO =      300	(sec) |
     SAP =       30	(sec) |

