MOD 3.1 solo/Alto Message_Decoder for SBD messages, format version 0.5-2.1
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11011     0  124 000000000000009  3426     0   4 2019/08/23, 14:02:30,  41.5603  -70.6398   3      0 14
SBD 11011     0  311 000000000000009  3428     0   2 2019/08/23, 14:03:03,  41.5603  -70.6398   4      0 8,14
SBD 11011     0  340 000000000000009  3427     0   3 2019/08/23, 14:02:49,  41.5603  -70.6398   3      0 13
SBD 11011     0  327 000000000000009  3429     0   1 2019/08/23, 14:06:14,  41.5413  -70.6851   5      0 9,10,11,15,13
SBD 11011     0  338 000000000000009  3430     0   0 2019/08/23, 14:06:26,  41.5334  -70.6398   5      0 4,14,14,5,14,0,1,2,3,6
SBD 11011     0  283 000000000000009  3431     0   5 2019/08/23, 16:45:21,  41.5413  -70.6851   2      0 6,14,14
SBD 11011     0  335 000000000000009  3432     0   4 2019/08/23, 16:45:32,  41.5334  -70.6398   2      0 6
SBD 11011     0  340 000000000000009  3433     0   3 2019/08/23, 16:45:46,  41.5413  -70.6851   2      0 13
SBD 11011     0  338 000000000000009  3434     0   2 2019/08/23, 16:45:58,  41.5413  -70.6851   2      0 8,15
SBD 11011     0  242 000000000000009  3435     0   1 2019/08/23, 16:46:11,  41.5334  -70.6398   2      0 10,11,13
SBD 11011     0  323 000000000000009  3436     0   0 2019/08/23, 16:46:23,  41.5683  -70.6851   2      0 7,5,14,0,1,2,3,9
SBD 11011     0  317 000000000000009  3437     0   4 2019/08/23, 16:46:41,  41.5683  -70.6851   2      0 14,0,4,14,14,9,10
SBD 11011     0  337 000000000000009  3438     0   2 2019/08/23, 16:47:28,  41.5603  -70.6398   3      0 8,14,11
# Raw packet information
#  Packets Bytes
P0      47  3799
# Sensor_ID bytes     % packets Description
          0    72  1.90       3         GPS
          1    65  1.71       2    Pressure
          2    58  1.53       2 Temperature
          3    58  1.53       2    Salinity
          4   121  3.19       2        Fall
          5   146  3.84       2        Rise
          6   518 13.64       3        Pump
          7    28  0.74       1      Unused
          8   834 21.95       3     Surface
          9   137  3.61       3  HR/Drift_P
         10   167  4.40       3  HR/Drift_T
         11   153  4.03       3  HR/Drift_S
         13   812 21.37       4  Parameters
         14   498 13.11      12 Engineering
         15   132  3.47       2     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G    0 0 2019/08/23 16:24 +41.53425  -70.64678   20  6 37 39 41  1.1     0
# Profile Data
#     C   Npts comp n_var
D0     0   10    1     3
#      Pres      Temp      Salinity
      1.120    26.1220    30.1920
      2.080    26.1250    30.1920
      3.040    26.1320    30.1910
      4.000    26.1320    30.1920
      5.000    26.1380    30.1920
      6.000    26.1420    30.1920
      7.000    26.1410    30.1920
      8.080    26.1420    30.1920
      9.440    26.1440    30.1920
      9.560    26.1450    30.1910
# High Resolution Data
#     C   Npts comp
H0     0   63    1
#      Pres      Temp      Salinity
      0.720    26.1230    30.1920
      0.800    26.1220    30.1920
      0.840    26.1240    30.1910
      0.920    26.1230    30.1930
      1.000    26.1240    30.1910
      1.080    26.1230    30.1920
      1.120    26.1230    30.1910
      1.200    26.1230    30.1930
      1.240    26.1210    30.1940
      1.360    26.1200    30.1940
      1.400    26.1190    30.1940
      1.480    26.1210    30.1920
      1.600    26.1210    30.1950
      1.720    26.1200    30.1930
      1.800    26.1200    30.1930
      1.880    26.1210    30.1920
      2.000    26.1220    30.1920
      2.080    26.1250    30.1910
      2.120    26.1310    30.1880
      2.200    26.1260    30.1950
      2.280    26.1260    30.1920
      2.360    26.1310    30.1890
      2.400    26.1280    30.1950
      2.440    26.1250    30.1950
      2.560    26.1190    30.1970
      2.640    26.1310    30.1850
      2.720    26.1310    30.1930
      2.840    26.1320    30.1920
      2.920    26.1310    30.1930
      3.000    26.1290    30.1930
      3.120    26.1350    30.1900
      3.200    26.1320    30.1930
      3.280    26.1320    30.1900
      3.360    26.1380    30.1880
      3.400    26.1380    30.1900
      3.480    26.1370    30.1920
      3.520    26.1340    30.1920
      3.600    26.1330    30.1910
      3.680    26.1340    30.1900
      3.760    26.1330    30.1920
      3.800    26.1300    30.1940
      3.880    26.1290    30.1930
      3.960    26.1270    30.1950
      4.040    26.1280    30.1940
      4.160    26.1310    30.1920
      4.240    26.1300    30.1910
      4.280    26.1360    30.1900
      4.400    26.1340    30.1930
      4.480    26.1360    30.1900
      4.560    26.1450    30.1850
      4.600    26.1430    30.1920
      4.680    26.1430    30.1910
      4.760    26.1350    30.1950
      4.800    26.1340    30.1900
      4.840    26.1350    30.1930
      4.880    26.1350    30.1910
      4.960    26.1350    30.1920
      5.040    26.1350    30.1930
      5.080    26.1310    30.1960
      5.160    26.1330    30.1930
      5.240    26.1360    30.1900
      5.320    26.1430    30.1920
      5.360    26.1420    30.1930
# Drift Data
#     C   Npts comp
I0     0    4    1
#      Pres      Temp      Salinity
      9.520    26.1480    30.1930
      9.520    26.1500    30.1920
      9.520    26.1490    30.1910
      9.520    26.1490    30.1910
# Surface Pressure Time Series
#    C  npts index Start (sec) relative to 1 Jan 2000
S    0  59    0   619881782 2019/08/23 13:23:02
# time      Pres      Temp       Psal   phase  Date
   124      0.280    25.4610  -999.0000   14 2019/08/23 13:23:02
  1218      0.070    21.9520  -999.0000   14 2019/08/23 13:41:16
  1281      0.080    21.9070  -999.0000   14 2019/08/23 13:42:19
  1292      0.070    21.8930  -999.0000   14 2019/08/23 13:42:30
  1302      0.070    21.8800  -999.0000   14 2019/08/23 13:42:40
  1312      0.060    21.8910  -999.0000   14 2019/08/23 13:42:50
  1323      0.070    21.8810  -999.0000   14 2019/08/23 13:43:01
  1333      0.070    21.8710  -999.0000   14 2019/08/23 13:43:11
  1343      0.070    21.8740  -999.0000   14 2019/08/23 13:43:21
  1353      0.070    21.8840  -999.0000   14 2019/08/23 13:43:31
  1364      0.070    21.8810  -999.0000   14 2019/08/23 13:43:42
  1374      0.060    21.8740  -999.0000   14 2019/08/23 13:43:52
  1384      0.070    21.8900  -999.0000   14 2019/08/23 13:44:02
  1395      0.070    21.8980  -999.0000   14 2019/08/23 13:44:13
  1405      0.060    21.9080  -999.0000   14 2019/08/23 13:44:23
  1424      0.080    21.9340  -999.0000   14 2019/08/23 13:44:42
  1434      0.070    21.9310  -999.0000   14 2019/08/23 13:44:52
  1444      0.080    21.9470  -999.0000   14 2019/08/23 13:45:02
  1455      0.080    21.9500  -999.0000   14 2019/08/23 13:45:13
  1465      0.060    21.9570  -999.0000   14 2019/08/23 13:45:23
  1475      0.060    21.9810  -999.0000   14 2019/08/23 13:45:33
  1485      0.070    21.9980  -999.0000   14 2019/08/23 13:45:43
  1496      0.070    22.0050  -999.0000   14 2019/08/23 13:45:54
  1506      0.070    22.0040  -999.0000   14 2019/08/23 13:46:04
  1516      0.070    22.0180  -999.0000   14 2019/08/23 13:46:14
  1527      0.080    22.0170  -999.0000   14 2019/08/23 13:46:25
  1537      0.070    22.0220  -999.0000   14 2019/08/23 13:46:35
  1547      0.070    22.0420  -999.0000   14 2019/08/23 13:46:45
  1557      0.070    22.0540  -999.0000   14 2019/08/23 13:46:55
  1568      0.070    22.0750  -999.0000   14 2019/08/23 13:47:06
  1573      0.060    22.0790  -999.0000   14 2019/08/23 13:47:11
   124      0.240    25.6310  -999.0000   14 2019/08/23 16:05:08
  1218      0.000    21.9630  -999.0000   14 2019/08/23 16:23:22
  1282      0.010    22.0560  -999.0000   14 2019/08/23 16:24:26
  1292      0.000    22.0120  -999.0000   14 2019/08/23 16:24:36
  1303      0.000    21.9880  -999.0000   14 2019/08/23 16:24:47
  1313      0.000    21.9910  -999.0000   14 2019/08/23 16:24:57
  1323      0.000    21.9670  -999.0000   14 2019/08/23 16:25:07
  1333      0.020    21.9910  -999.0000   14 2019/08/23 16:25:17
  1344      0.020    21.9790  -999.0000   14 2019/08/23 16:25:28
  1366      0.010    21.9600  -999.0000   14 2019/08/23 16:25:50
  1377      0.000    21.9130  -999.0000   14 2019/08/23 16:26:01
  1387      0.000    21.9200  -999.0000   14 2019/08/23 16:26:11
  1397      0.000    21.8950  -999.0000   14 2019/08/23 16:26:21
  1407      0.010    21.8670  -999.0000   14 2019/08/23 16:26:31
  1426      0.000    21.9110  -999.0000   14 2019/08/23 16:26:50
  1437      0.000    21.9110  -999.0000   14 2019/08/23 16:27:01
  1447      0.000    21.9110  -999.0000   14 2019/08/23 16:27:11
  1466      0.010    21.9450  -999.0000   14 2019/08/23 16:27:30
  1476      0.000    21.9420  -999.0000   14 2019/08/23 16:27:40
  1486      0.000    21.9560  -999.0000   14 2019/08/23 16:27:50
  1497      0.010    21.9700  -999.0000   14 2019/08/23 16:28:01
  1507      0.010    21.9490  -999.0000   14 2019/08/23 16:28:11
  1526      0.000    21.9650  -999.0000   14 2019/08/23 16:28:30
  1536     -0.010    21.9450  -999.0000   14 2019/08/23 16:28:40
  1547     -0.010    21.9380  -999.0000   14 2019/08/23 16:28:51
  1557      0.010    21.9490  -999.0000   14 2019/08/23 16:29:01
  1567      0.000    21.9610  -999.0000   14 2019/08/23 16:29:11
  1572      0.000    21.9770  -999.0000   14 2019/08/23 16:29:16
# Fall Time Series
#  Cycle  npts  Start
#               seconds        Date
F00    0  21    619881188 2019/08/23 13:13:08
# time    Pres   phase Date
     0    2.680    1    2019/08/23 13:13:08
    11    4.800   10    2019/08/23 13:13:19
    15    5.680   10    2019/08/23 13:13:23
    21    6.960    2    2019/08/23 13:13:29
    24    7.640   10    2019/08/23 13:13:32
    28    8.560   10    2019/08/23 13:13:36
    33    9.440   10    2019/08/23 13:13:41
   331    9.200   10    2019/08/23 13:18:39
  3220    0.040    1    2019/08/23 14:06:48
  3441    1.280   10    2019/08/23 14:10:29
  3449    2.600   10    2019/08/23 14:10:37
  3453    3.360   10    2019/08/23 14:10:41
  3457    4.160   10    2019/08/23 14:10:45
  3466    5.800   10    2019/08/23 14:10:54
  3472    6.960    2    2019/08/23 14:11:00
  3475    7.640   10    2019/08/23 14:11:03
  3479    8.480   10    2019/08/23 14:11:07
  3483    9.320   10    2019/08/23 14:11:11
  3785    9.560   10    2019/08/23 14:16:13
  4088    9.520   10    2019/08/23 14:21:16
  4101    9.560   10    2019/08/23 14:21:29
# Rise Time Series
#  Cycle  npts  Start
#               seconds        Date
R00    0  26    619881530 2019/08/23 13:18:50
# time    Pres   phase Date
     0    8.880    7    2019/08/23 13:18:50
    35    7.800   13    2019/08/23 13:19:25
    53    6.760   13    2019/08/23 13:19:43
    72    5.720   13    2019/08/23 13:20:02
    85    4.880   13    2019/08/23 13:20:15
   103    3.520   13    2019/08/23 13:20:33
   109    2.920   13    2019/08/23 13:20:39
   121    1.960   13    2019/08/23 13:20:51
   127    1.560    8    2019/08/23 13:20:57
  7371    9.520    7    2019/08/23 15:21:41
  7671    9.520   13    2019/08/23 15:26:41
  7972    9.520   13    2019/08/23 15:31:42
  8274    9.560   13    2019/08/23 15:36:44
  8575    9.520   13    2019/08/23 15:41:45
  8876    9.520   13    2019/08/23 15:46:46
  9178    9.520   13    2019/08/23 15:51:48
  9479    9.560   13    2019/08/23 15:56:49
  9712    8.840   13    2019/08/23 16:00:42
  9756    7.760   13    2019/08/23 16:01:26
  9774    6.920   13    2019/08/23 16:01:44
  9799    5.600   13    2019/08/23 16:02:09
  9811    4.800   13    2019/08/23 16:02:21
  9823    3.880   13    2019/08/23 16:02:33
  9836    2.920   13    2019/08/23 16:02:46
  9848    2.000   13    2019/08/23 16:02:58
  9853    1.480    8    2019/08/23 16:03:03
# Pump Time Series
#  Cycle  npts
B00    0  14
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    6.960   303   12.31     166  595     -9      2
    8.880    17   12.24     181  595     -9      7
    5.720     6   12.20     185  594     -9     13
    0.000   120   12.31     163  602     -9     14
    9.560     6   12.14     192  587     -9     13
    9.560     6   12.10     191  588     -9     13
    9.480     6   12.16     187  588     -9     13
    9.520     6   12.12     189  589     -9     13
    9.520     6   12.12     177  589     -9     13
    9.440     6   12.11     186  592     -9     13
    9.440     7   12.12     187  592     -9     13
    8.680     6   12.09     186  592     -9     13
    6.280     7   12.09     176  592     -9     13
    9.600     7   12.15     185  579     -9     13
# Argo Mission Data
#  Cycle Data  Miss Depth       Time                       Ascent Seeks Max Time           Cycle
#        Form  No   Prof  Park  Rise to Park to Prof Drift Speed        Surface  Seek      Time
A2    0  4.2   15     9     9    40    70       1     60    10     3    1550   10800      22.6100
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17    0 2019/08/23 14:06:45
# Pcase vacuums
   7.370   5.740
# Oil valve positions
   1.760   0.845   0.557
# Seek attempts and pump time
      0      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18    0  2019/08/23 14:21:30 9999/99/99 99:99:99
# Pcase vacuums
   5.740   5.740
# Average 
#   P      T      S
    9.520  26.149  30.192
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18    0  2019/08/23 15:21:41 9999/99/99 99:99:99
# Pcase vacuums
   5.740   5.910
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20    0  2019/08/23 13:20:58
# number of packets previous packets
      0               0
# CTD Status pres offset
 0x00000000   0.000
# Pcase vacuums
   0.000   0.000
# Air Pump max current time  max pressure
       0         0     0.000
# Air Valve Positions
   0.000   0.000
# Air Valve Calibrations
   0.000   0.000   0.000
# Oil Valve Positions
   0.000   0.000
# Oil Valve Calibrations
   0.000   0.000
# Encountered Error Codes
#       C number of Errors
EE20    0 0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00    0  2      0
# MC    Pres      Time  Pcase Vacuum
   1     3.320     15   5.740
   1     0.000    244   5.710
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M    0       197    2019/08/23 14:02:30
Parameter Table Param_Table-V17.txt |
     IFV = 6.8..0	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor) |
     MSN =    65535	(cycle number) |
     MPS =        0	(phase) |
     MPC =        0	(count) |
  MMM[0] =        3	(count) |
  MMM[1] =        2	(count) |
  MMM[2] =     1000	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =        9	(dBar) |
  DTD[1] =        9	(dBar) |
  DTD[2] =        9	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =     2000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =    30000	(ms) |
     MTS = 15552000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =    -1.65	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =        0	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =        5	(dBar) |
  DSD[2] =        5	(dBar) |
  DST[0] =     7200	(sec) |
  DST[1] =     3600	(sec) |
  DST[2] =     3600	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =        1	(dBar) |
     DDT =      300	(sec) |
  DFT[0] =       20	(sec) |
  DFT[1] =       20	(sec) |
  DFT[2] =       20	(sec) |
     DCM =        0	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      600	(sec) |
  DCT[1] =      600	(sec) |
  DCT[2] =      600	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =    10800	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        5	(dBar) |
     DDS =     3600	(sec) |
     PNS =        0	(cycle) |
  PTO[0] =     3600	(sec) |
  PTO[1] =   129600	(sec) |
  PTO[2] =    36000	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =    10800	(sec) |
  PFT[2] =      900	(sec) |
  PFD[0] =        9	(dBar) |
  PFD[1] =     1500	(dBar) |
  PFD[2] =      100	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =      420	(sec) |
  PFV[2] =       20	(sec) |
     PDT =      900	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =     1792	(cycle number) |
     ARR =        0	(target rise rate cm/sec) |
     APM =     2569	(sec) |
     APS =    24576	(sec) |
     APN =     1280	(sec) |
     APF =     1280	(sec) |
     ASI = 922746883	(ms) |
     ASD =    59392	(dBar) |
     ADD =      512	(dBar) |
     ADT =      257	(sec) |
     ARC =    11281	(?) |
     AHR =    37891	(cm/sec) |
     AHD =    59392	(dBar) |
     SFN =     2560	(cycle number) |
     SDC =        0	(dBar) |
     SGT =        0	(min) |
     SET =     3840	(min) |
     SBM =       20	(0=dis 1=en) |
     SBT =        1	(sec) |
     SDT = 738197504	(interval) |
     SSA =        3	(0=dis 1=en) |
     SSI = -402650623	(ms) |
     SDS =     7500	(sec) |
     SAM =       25	(0=dis 1=en) |
     SAX =    25604	(.001 psi) |
     SAN =    14340	(.001 psi) |
     SAO =    45057	(sec) |
     SAP =    11264	(sec) |
     SPT =    30720	(sec) |

