# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11011     9   88 000000000000009  3881     0   5 2020/04/15, 01:23:56,  41.5255  -70.5945  20      0 14
SBD 11011     9  340 000000000000009  3882     0   4 2020/04/15, 01:24:28,  41.5095  -70.5041  27      0 13
SBD 11011     9  313 000000000000009  3883     0   3 2020/04/15, 01:24:42,  41.4680  -70.5653  30      0 15,13,6,14
SBD 11011     9  340 000000000000009  3884     0   2 2020/04/15, 01:26:18,  41.5400  -70.5333 680      0 10,11,0,1,2,3,8
SBD 11011     9  320 000000000000009  3885     0   1 2020/04/15, 01:26:44,  41.5321  -70.4881 680      0 10,11,7,5,14
SBD 11011     9  332 000000000000009  3886     0   0 2020/04/15, 01:27:11,  41.5321  -70.4881  98      2 0,4,14,14,9,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      24         1661           72        1733
# Sensor_ID bytes     % packets Description
          0    48  2.89       2         GPS
          1    32  1.93       1    Pressure
          2    39  2.35       1 Temperature
          3    32  1.93       1    Salinity
          4   133  8.01       1        Fall
          5    58  3.49       1        Rise
          6   147  8.85       1        Pump
          7    72  4.33       1 Drift_times
          8    71  4.27       1     Surface
          9   119  7.16       2  HR/Drift_P
         10   156  9.39       2  HR/Drift_T
         11   133  8.01       2  HR/Drift_S
         13   404 24.32       2  Parameters
         14   151  9.09       5 Engineering
         15    66  3.97       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G    9 1 2020/04/12 22:26 +41.53423  -70.64676    0  4 38 39 41  4.8     0
G    9 2 2020/04/15 01:12 +41.53436  -70.64693   40  4  0 32 47  2.1     0
# Profile Data
#      C Npts comp n_var
D0     9    9    1     3
#      Pres      Temp    Salinity
      2.000    17.0910    30.1540
      3.040    16.5290    30.1060
      4.040    16.2330    30.0890
      5.040    15.9760    30.0790
      6.040    15.7660    30.0710
      7.040    15.5570    30.0630
      7.840    15.1870    30.0400
      9.400    12.6210    30.0370
      9.520    12.5070    30.0320
# High Resolution Data
#     C   Npts comp
H0     9   46     1
#      Pres      Temp    Salinity
      1.520    17.7650    30.2250
      1.560    17.5610    30.2130
      1.640    17.3930    30.1880
      1.760    17.2680    30.1680
      1.760    17.1600    30.1640
      1.960    17.0700    30.1550
      1.960    16.9970    30.1380
      2.120    16.9430    30.1330
      2.120    16.8900    30.1310
      2.280    16.8450    30.1270
      2.280    16.8020    30.1230
      2.440    16.7620    30.1230
      2.440    16.7260    30.1170
      2.600    16.6920    30.1120
      2.640    16.6550    30.1180
      2.760    16.6200    30.1170
      2.800    16.5870    30.1100
      2.880    16.5680    30.1050
      2.960    16.5450    30.1020
      3.040    16.5140    30.1090
      3.120    16.4810    30.1070
      3.280    16.4530    30.1050
      3.320    16.4250    30.0950
      3.400    16.4110    30.0910
      3.440    16.3910    30.0950
      3.520    16.3720    30.0930
      3.640    16.3500    30.0920
      3.720    16.3270    30.0940
      3.800    16.3020    30.0930
      3.840    16.2830    30.0910
      3.920    16.2590    30.0920
      4.000    16.2390    30.0890
      4.080    16.2180    30.0910
      4.200    16.2000    30.0870
      4.200    16.1780    30.0880
      4.320    16.1620    30.0850
      4.320    16.1430    30.0860
      4.480    16.1250    30.0870
      4.480    16.1060    30.0850
      4.640    16.0880    30.0850
      4.600    16.0650    30.0840
      4.720    16.0530    30.0790
      4.760    16.0400    30.0800
      4.880    16.0280    30.0810
      4.880    16.0130    30.0810
      5.000    16.0010    30.0780
# Drift Data
#      C Npts comp n_var
I0     9   96    1     4
#      Pres       Temp   Salinity     Time
      9.600    12.4760    30.0290        6
      9.600    12.4800    30.0270     1812
      9.600    12.4800    30.0280     3618
      9.600    12.4800    30.0270     5424
      9.600    12.4790    30.0270     7229
      9.600    12.4780    30.0270     9035
      9.600    12.4770    30.0280    10841
      9.600    12.4780    30.0270    12647
      9.560    12.4790    30.0260    14453
      9.600    12.4770    30.0280    16258
      9.560    12.4780    30.0270    18064
      9.600    12.4770    30.0270    19870
      9.560    12.4770    30.0270    21676
      9.560    12.4760    30.0280    23482
      9.560    12.4760    30.0270    25288
      9.560    12.4750    30.0280    27093
      9.560    12.4760    30.0270    28899
      9.560    12.4740    30.0280    30705
      9.560    12.4750    30.0270    32511
      9.560    12.4740    30.0270    34317
      9.560    12.4730    30.0280    36122
      9.520    12.4730    30.0280    37928
      9.520    12.4730    30.0270    39734
      9.520    12.4730    30.0270    41540
      9.560    12.4710    30.0280    43346
      9.520    12.4720    30.0270    45151
      9.520    12.4720    30.0270    46957
      9.520    12.4720    30.0270    48763
      9.480    12.4720    30.0270    50569
      9.480    12.4730    30.0260    52375
      9.480    12.4730    30.0270    54180
      9.440    12.4710    30.0280    55986
      9.440    12.4730    30.0280    57792
      9.440    12.4700    30.0290    59598
      9.440    12.4710    30.0280    61404
      9.440    12.4730    30.0270    63209
      9.400    12.4740    30.0270    65015
      9.400    12.4730    30.0280    66821
      9.400    12.4740    30.0280    68627
      9.360    12.4710    30.0280    70433
      9.400    12.4740    30.0270    72238
      9.360    12.4760    30.0270    74044
      9.400    12.4730    30.0290    75850
      9.400    12.4740    30.0280    77656
      9.360    12.4760    30.0280    79462
      9.400    12.4750    30.0290    81267
      9.400    12.4730    30.0300    83073
      9.400    12.4750    30.0290    84879
      9.400    12.4760    30.0290    86685
      9.400    12.4780    30.0280    88491
      9.400    12.4780    30.0290    90296
      9.440    12.4750    30.0300    92102
      9.400    12.4780    30.0290    93908
      9.440    12.4780    30.0290    95714
      9.400    12.4810    30.0290    97520
      9.440    12.4790    30.0300    99325
      9.440    12.4800    30.0300   101131
      9.440    12.4800    30.0300   102937
      9.440    12.4780    30.0320   104743
      9.440    12.4800    30.0310   106549
      9.440    12.4800    30.0300   108354
      9.480    12.4810    30.0300   110160
      9.440    12.4820    30.0300   111966
      9.480    12.4830    30.0300   113772
      9.440    12.4820    30.0290   115578
      9.480    12.4790    30.0310   117383
      9.480    12.4810    30.0310   119189
      9.480    12.4780    30.0320   120995
      9.480    12.4790    30.0310   122801
      9.520    12.4800    30.0300   124607
      9.520    12.4780    30.0310   126412
      9.520    12.4780    30.0310   128218
      9.520    12.4800    30.0300   130024
      9.480    12.4780    30.0310   131830
      9.520    12.4760    30.0320   133635
      9.520    12.4790    30.0310   135441
      9.520    12.4760    30.0320   137247
      9.520    12.4790    30.0320   139053
      9.520    12.4770    30.0320   140859
      9.520    12.4810    30.0310   142664
      9.520    12.4780    30.0320   144470
      9.520    12.4840    30.0300   146276
      9.520    12.4840    30.0320   148082
      9.520    12.4800    30.0330   149888
      9.520    12.4830    30.0330   151693
      9.480    12.4850    30.0320   153499
      9.520    12.4860    30.0320   155305
      9.480    12.4870    30.0330   157111
      9.520    12.4870    30.0320   158917
      9.520    12.4840    30.0340   160722
      9.480    12.4880    30.0310   162528
      9.520    12.4880    30.0320   164334
      9.520    12.4950    30.0310   166140
      9.520    12.4940    30.0340   167946
      9.520    12.4950    30.0320   169751
      9.520    12.4940    30.0330   171557
# Surface Pressure Time Series
#    C npts n_var               Start
S    9     7     6 2020/04/15 01:12:01
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    90      0.140     19.300   -999.000      14 2020/04/15 01:13:31
   103      0.140     19.138   -999.000      14 2020/04/15 01:13:44
   114      0.150     19.015   -999.000      14 2020/04/15 01:13:55
   125      0.140     18.916   -999.000      14 2020/04/15 01:14:06
   136      0.150     18.764   -999.000      14 2020/04/15 01:14:17
   147      0.140     18.626   -999.000      14 2020/04/15 01:14:28
   158      0.130     18.516   -999.000      14 2020/04/15 01:14:39
# Fall Time Series
#  Cycle npts     Start                Date
F00    9   25 640045588 2020/04/12 22:26:28
# time     Pres phase                Date
     0    0.240     1 2020/04/12 22:26:28
    38    1.240    10 2020/04/12 22:27:06
    47    2.040    10 2020/04/12 22:27:15
    60    3.400    10 2020/04/12 22:27:28
    69    4.320    10 2020/04/12 22:27:37
    78    5.320    10 2020/04/12 22:27:46
    84    6.000     2 2020/04/12 22:27:52
    87    6.320    10 2020/04/12 22:27:55
    94    7.120    10 2020/04/12 22:28:02
   104    8.280    10 2020/04/12 22:28:12
   112    9.000    10 2020/04/12 22:28:20
  1014    9.600    10 2020/04/12 22:43:22
  1893    9.600     3 2020/04/12 22:58:01
  1915    9.600    10 2020/04/12 22:58:23
  2817    9.600    10 2020/04/12 23:13:25
  3718    9.600    10 2020/04/12 23:28:26
  3731    9.600     3 2020/04/12 23:28:39
  4620    9.600    10 2020/04/12 23:43:28
  5521    9.600    10 2020/04/12 23:58:29
  5569    9.600     3 2020/04/12 23:59:17
  6423    9.600    10 2020/04/13 00:13:31
  7324    9.600    10 2020/04/13 00:28:32
  8225    9.600    10 2020/04/13 00:43:33
  9126    9.600    10 2020/04/13 00:58:34
  9207    9.600    10 2020/04/13 00:59:55
# Rise Time Series
#  Cycle npts     Start                Date
R00    9   10 640227609 2020/04/15 01:00:09
# time     Pres phase                Date
     0    9.520     7 2020/04/15 01:00:09
   496    8.960    13 2020/04/15 01:08:25
   538    7.920    13 2020/04/15 01:09:07
   622    6.960    13 2020/04/15 01:10:31
   649    5.920    13 2020/04/15 01:10:58
   670    4.880    13 2020/04/15 01:11:19
   685    3.800    13 2020/04/15 01:11:34
   697    2.800    13 2020/04/15 01:11:46
   709    1.760    13 2020/04/15 01:11:58
   710    1.760     8 2020/04/15 01:11:59
# Pump Time Series
#  Cycle  npts
B00    9  13
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    6.000     5   11.92     204  620     -9      2
    9.520    31   12.02     211  638     -9      7
    9.480    12   11.96     237  640     -9     13
    9.520    12   12.03     230  640     -9     13
    9.480    12   12.03     239  641     -9     13
    9.480    12   11.95     224  642     -9     13
    9.480    12   11.92     225  643     -9     13
    9.520    12   11.92     221  644     -9     13
    9.480    12   11.96     196  645     -9     13
    8.200    12   11.94     213  645     -9     13
    7.600    12   11.92     200  644     -9     13
    5.920    12   12.10     199  641     -9     13
    0.240    59   12.00     202  643     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2     9  2.4    4  2000    10   600    32   600  2880      5     3    1725   30  2.9380
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17    9 2020/04/12 22:26:25
# Pcase vacuums
   6.350   6.380
# Oil valve positions
   1.769   0.851   0.603
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18    9 2020/04/13 00:59:56 9999/99/99 99:99:99
# Pcase vacuums
   6.380   6.380
# Average 
#   P      T      S
    9.480  12.478  30.029
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18    9 2020/04/15 01:00:09 9999/99/99 99:99:99
# Pcase vacuums
   6.380   6.390
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20    9 2020/04/15 01:12:00
# number of packets previous packets
      5               8
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.390   6.360
# Air Pump max current time  max pressure
       0         0     1.292
# Air Valve Positions
   1.166   0.000
# Air Valve Calibrations
   1.103   0.086   0.595
# Oil Valve Positions
   0.000   0.852
# Oil Valve Calibrations
   1.735   0.869
# Encountered Error Codes
#    Cycle Errors
EE20     9      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00    9  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.240     79   6.200
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M    9       197    2020/04/15 01:24:42
Parameter Table Param_Table-V17.txt |
     IFV = 7.2.9.0	(IFV) |
     VER = 17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =        9	(cycle number) |
     MPS =        2	(phase) |
     MPC =        2	(count) |
  MMM[0] =        3	(count) |
  MMM[1] =        3	(count) |
  MMM[2] =        3	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        3	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =     2000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =   150000	(ms) |
     MTS =    21600	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =        8	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =        5	(dBar) |
  DSD[2] =        5	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =       60	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =        1	(sec) |
  DFT[1] =        1	(sec) |
  DFT[2] =        2	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =      120	(sec) |
     PNS =        8	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =     2048	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =        1	(sec) |
  PFT[2] =    36000	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =     1000	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        1	(sec) |
  PFV[2] =      300	(sec) |
     PDT =     1800	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =        8	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       10	(sec) |
     ASI =  3276800	(ms) |
     ASD =     2000	(dBar) |
     ADD =        2	(dBar) |
     ADT =        1	(sec) |
     ARC =      900	(?) |
     AHR =     4500	(cm/sec) |
     AHD =     1000	(dBar) |
     SFN =       10	(cycle number) |
     SDC =        8	(dBar) |
     SGT =        0	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =     3840	(sec) |
     SSA =      300	(0=dis 1=en) |
     SSI =        0	(ms) |
     SDS =    10000	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     2561	(.001 psi) |
     SAN =     7500	(.001 psi) |
     SAO =     6500	(sec) |
     SAP =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |

