# Message Decoder for Solo2, Alto and Deep Solo Floats; v260708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11011    39  302 000000000000009  4083     0   6 2020/06/17, 21:30:29,  41.5197  -69.1690 128      2 6,14,14
SBD 11011    39  340 000000000000009  4084     0   5 2020/06/17, 21:30:48,  41.5197  -69.1690 137      2 13
SBD 11011    39  333 000000000000009  4085     0   4 2020/06/17, 21:31:01,  41.5197  -69.1690 135      2 8,15,13
SBD 11011    39  335 000000000000009  4086     0   3 2020/06/17, 21:31:12,  41.4927  -69.1690 140      2 8
SBD 11011    39  282 000000000000009  4087     0   2 2020/06/17, 21:31:25,  41.5187  -70.7009   7      0 10,11,0,1,2,3
SBD 11011    39  329 000000000000009  4088     0   1 2020/06/17, 21:32:22,  41.5266  -70.7462   7      0 10,11,7,5,14
SBD 11011    39  334 000000000000009  4089     0   0 2020/06/17, 21:33:31,  41.5108  -70.6557   8      0 0,4,14,14,9,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      25         2171           84        2255
# Sensor_ID bytes     % packets Description
          0    48  2.21       2         GPS
          1    32  1.47       1    Pressure
          2    36  1.66       1 Temperature
          3    32  1.47       1    Salinity
          4   133  6.13       1        Fall
          5    58  2.67       1        Rise
          6   202  9.30       1        Pump
          7    72  3.32       1 Drift_times
          8   502 23.12       2     Surface
          9   121  5.57       2  HR/Drift_P
         10   161  7.42       2  HR/Drift_T
         11   153  7.05       2  HR/Drift_S
         13   404 18.61       2  Parameters
         14   151  6.96       5 Engineering
         15    66  3.04       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   39 1 2020/06/15 18:27 +41.53422  -70.64682   40  8 27 32 39  1.1     0
G   39 2 2020/06/17 21:19 +41.53413  -70.64694   50  5 33 35 43  4.5     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    39    9    1     3         2
#      Pres      Temp    Salinity
      2.080    22.3830    30.4760
      2.960    21.8270    30.4540
      4.040    21.3840    30.4170
      5.120    21.0550    30.3960
      5.960    20.8140    30.3830
      7.040    20.4610    30.3630
      8.000    19.7910    30.3570
      9.040    18.9570    30.3460
      9.600    18.6240    30.3340
# High Resolution Data
#     C   Npts comp
H0    39   50     1
#      Pres      Temp    Salinity
      1.680    22.5580    30.4820
      1.800    22.5350    30.4750
      1.840    22.5170    30.4720
      1.840    22.4830    30.4790
      1.920    22.4560    30.4680
      2.080    22.4200    30.4740
      2.160    22.3730    30.4780
      2.240    22.3100    30.4820
      2.280    22.2380    30.4860
      2.360    22.1870    30.4700
      2.480    22.1370    30.4700
      2.600    22.0750    30.4760
      2.640    22.0110    30.4700
      2.680    21.9520    30.4670
      2.760    21.8990    30.4660
      2.920    21.8510    30.4600
      3.040    21.8150    30.4460
      3.080    21.7710    30.4510
      3.120    21.7280    30.4450
      3.200    21.6920    30.4440
      3.280    21.6650    30.4330
      3.440    21.6370    30.4340
      3.520    21.6010    30.4370
      3.560    21.5640    30.4320
      3.600    21.5290    30.4290
      3.680    21.5030    30.4260
      3.840    21.4790    30.4210
      3.880    21.4480    30.4250
      3.880    21.4180    30.4230
      4.000    21.3950    30.4130
      4.120    21.3590    30.4240
      4.200    21.3390    30.4110
      4.200    21.3180    30.4130
      4.240    21.2960    30.4110
      4.320    21.2810    30.4100
      4.440    21.2690    30.4050
      4.520    21.2560    30.4060
      4.480    21.2410    30.4050
      4.480    21.2270    30.4060
      4.600    21.2160    30.4020
      4.680    21.2000    30.4080
      4.720    21.1780    30.4090
      4.680    21.1640    30.4040
      4.720    21.1560    30.4010
      4.840    21.1450    30.4010
      4.880    21.1340    30.4000
      4.920    21.1190    30.4040
      4.920    21.1010    30.4030
      4.960    21.0890    30.4020
      5.000    21.0810    30.3960
# Drift Data
#      C Npts comp n_var
I0    39   96    1     4
#      Pres       Temp   Salinity     Time
      9.640    18.4600    30.3240        6
      9.600    18.4630    30.3230     1812
      9.600    18.4670    30.3230     3617
      9.640    18.4690    30.3230     5423
      9.640    18.4700    30.3230     7229
      9.600    18.4710    30.3230     9035
      9.640    18.4740    30.3220    10841
      9.640    18.4750    30.3230    12646
      9.600    18.4770    30.3220    14452
      9.640    18.4800    30.3230    16258
      9.640    18.4800    30.3230    18064
      9.640    18.4830    30.3230    19870
      9.640    18.4840    30.3230    21675
      9.640    18.4850    30.3230    23481
      9.600    18.4870    30.3230    25287
      9.640    18.4880    30.3250    27093
      9.640    18.4880    30.3240    28899
      9.640    18.4870    30.3250    30705
      9.640    18.4910    30.3240    32510
      9.600    18.4890    30.3260    34316
      9.640    18.4900    30.3260    36122
      9.640    18.4910    30.3250    37928
      9.640    18.4920    30.3260    39734
      9.600    18.4930    30.3240    41539
      9.640    18.4950    30.3240    43345
      9.640    18.4910    30.3280    45151
      9.640    18.4940    30.3260    46957
      9.640    18.4900    30.3280    48763
      9.600    18.4920    30.3270    50568
      9.640    18.4930    30.3260    52374
      9.640    18.4900    30.3270    54180
      9.640    18.4920    30.3260    55986
      9.640    18.4920    30.3260    57792
      9.640    18.4940    30.3250    59597
      9.640    18.4960    30.3240    61403
      9.640    18.4920    30.3270    63209
      9.640    18.4910    30.3280    65015
      9.640    18.4930    30.3280    66821
      9.640    18.4930    30.3290    68626
      9.640    18.4970    30.3270    70432
      9.600    18.4980    30.3270    72238
      9.640    18.5000    30.3280    74044
      9.600    18.5030    30.3270    75850
      9.600    18.5010    30.3300    77655
      9.600    18.5070    30.3280    79461
      9.600    18.5070    30.3290    81267
      9.640    18.5110    30.3290    83073
      9.600    18.5150    30.3290    84879
      9.600    18.5160    30.3290    86684
      9.600    18.5170    30.3300    88490
      9.600    18.5220    30.3300    90296
      9.600    18.5260    30.3290    92102
      9.640    18.5270    30.3300    93907
      9.600    18.5290    30.3310    95713
      9.600    18.5320    30.3310    97519
      9.600    18.5340    30.3320    99325
      9.600    18.5390    30.3310   101131
      9.600    18.5400    30.3320   102936
      9.600    18.5430    30.3320   104742
      9.600    18.5440    30.3320   106548
      9.600    18.5490    30.3310   108354
      9.600    18.5490    30.3310   110160
      9.600    18.5520    30.3320   111965
      9.600    18.5520    30.3330   113771
      9.600    18.5550    30.3320   115577
      9.600    18.5580    30.3310   117383
      9.600    18.5580    30.3320   119189
      9.600    18.5590    30.3320   120994
      9.600    18.5610    30.3330   122800
      9.600    18.5610    30.3330   124606
      9.600    18.5640    30.3340   126412
      9.600    18.5650    30.3330   128218
      9.600    18.5650    30.3330   130023
      9.600    18.5660    30.3340   131829
      9.600    18.5660    30.3340   133635
      9.600    18.5660    30.3340   135441
      9.600    18.5670    30.3330   137247
      9.600    18.5830    30.3330   139052
      9.600    18.5800    30.3420   140858
      9.600    18.6130    30.3330   142664
      9.600    18.5900    30.3360   144470
      9.600    18.5930    30.3370   146276
      9.600    18.5930    30.3370   148081
      9.600    18.5930    30.3350   149887
      9.600    18.5950    30.3350   151693
      9.600    18.5980    30.3360   153499
      9.600    18.6000    30.3340   155304
      9.600    18.6010    30.3350   157110
      9.600    18.6020    30.3350   158916
      9.600    18.6030    30.3350   160722
      9.600    18.6050    30.3350   162528
      9.600    18.6060    30.3360   164333
      9.560    18.6080    30.3360   166139
      9.560    18.6110    30.3360   167945
      9.600    18.6140    30.3360   169751
      9.600    18.6150    30.3360   171557
# Surface Pressure Time Series
#    C npts n_var               Start
S   39    54     6 2020/06/17 21:19:48
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    78      0.320     23.166      0.000      14 2020/06/17 21:19:48
    91      0.330     22.912      0.000      14 2020/06/17 21:20:01
   102      0.320     22.537      0.000      14 2020/06/17 21:20:12
   113      0.310     22.281      0.000      14 2020/06/17 21:20:23
   124      0.310     22.071      0.000      14 2020/06/17 21:20:34
   135      0.310     21.905      0.000      14 2020/06/17 21:20:45
   147      0.320     21.764      0.000      14 2020/06/17 21:20:57
   158      0.330     21.681      0.000      14 2020/06/17 21:21:08
   169      0.320     21.569      0.000      14 2020/06/17 21:21:19
   180      0.310     21.454      0.000      14 2020/06/17 21:21:30
   191      0.320     21.404      0.000      14 2020/06/17 21:21:41
   202      0.310     21.319      0.000      14 2020/06/17 21:21:52
   213      0.320     21.216      0.000      14 2020/06/17 21:22:03
   224      0.320     21.123      0.000      14 2020/06/17 21:22:14
   235      0.310     21.039      0.000      14 2020/06/17 21:22:25
   246      0.320     21.030      0.000      14 2020/06/17 21:22:36
   257      0.320     20.965      0.000      14 2020/06/17 21:22:47
   269      0.310     20.931      0.000      14 2020/06/17 21:22:59
   280      0.320     20.902      0.000      14 2020/06/17 21:23:10
   291      0.320     20.864      0.000      14 2020/06/17 21:23:21
   302      0.330     20.848      0.000      14 2020/06/17 21:23:32
   313      0.320     20.865      0.000      14 2020/06/17 21:23:43
   324      0.320     20.860      0.000      14 2020/06/17 21:23:54
   335      0.310     20.831      0.000      14 2020/06/17 21:24:05
   346      0.320     20.830      0.000      14 2020/06/17 21:24:16
   357      0.310     20.789      0.000      14 2020/06/17 21:24:27
   368      0.330     20.765      0.000      14 2020/06/17 21:24:38
   379      0.330     20.737      0.000      14 2020/06/17 21:24:49
   391      0.330     20.727      0.000      14 2020/06/17 21:25:01
   402      0.310     20.766      0.000      14 2020/06/17 21:25:12
   413      0.330     20.722      0.000      14 2020/06/17 21:25:23
   424      0.310     20.701      0.000      14 2020/06/17 21:25:34
   435      0.320     20.685      0.000      14 2020/06/17 21:25:45
   446      0.320     20.720      0.000      14 2020/06/17 21:25:56
   457      0.310     20.783      0.000      14 2020/06/17 21:26:07
   468      0.320     20.819      0.000      14 2020/06/17 21:26:18
   479      0.300     20.846      0.000      14 2020/06/17 21:26:29
   490      0.320     20.873      0.000      14 2020/06/17 21:26:40
   501      0.320     20.932      0.000      14 2020/06/17 21:26:51
   512      0.330     20.985      0.000      14 2020/06/17 21:27:02
   524      0.310     20.984      0.000      14 2020/06/17 21:27:14
   535      0.320     21.018      0.000      14 2020/06/17 21:27:25
   546      0.310     21.093      0.000      14 2020/06/17 21:27:36
   557      0.330     21.146      0.000      14 2020/06/17 21:27:47
   568      0.310     21.201      0.000      14 2020/06/17 21:27:58
   579      0.320     21.218      0.000      14 2020/06/17 21:28:09
   590      0.320     21.250      0.000      14 2020/06/17 21:28:20
   601      0.320     21.252      0.000      14 2020/06/17 21:28:31
   612      0.310     21.233      0.000      14 2020/06/17 21:28:42
   623      0.320     21.245      0.000      14 2020/06/17 21:28:53
   634      0.330     21.270      0.000      14 2020/06/17 21:29:04
   646      0.310     21.307      0.000      14 2020/06/17 21:29:16
   657      0.330     21.344      0.000      14 2020/06/17 21:29:27
   668      0.310     21.318      0.000      14 2020/06/17 21:29:38
# Fall Time Series
#  Cycle npts     Start                Date
F00   39   25 645560896 2020/06/15 18:28:16
# time     Pres phase                Date
     0    0.360     1 2020/06/15 18:28:16
    33    1.080    10 2020/06/15 18:28:49
    42    2.080    10 2020/06/15 18:28:58
    51    3.240    10 2020/06/15 18:29:07
    60    4.440    10 2020/06/15 18:29:16
    65    5.080    10 2020/06/15 18:29:21
    71    5.920     2 2020/06/15 18:29:27
    74    6.280    10 2020/06/15 18:29:30
    81    7.280    10 2020/06/15 18:29:37
    88    8.280    10 2020/06/15 18:29:44
    96    9.240    10 2020/06/15 18:29:52
   998    9.680    10 2020/06/15 18:44:54
  1879    9.680     3 2020/06/15 18:59:35
  1899    9.640    10 2020/06/15 18:59:55
  2798    9.640    10 2020/06/15 19:14:54
  3700    9.640    10 2020/06/15 19:29:56
  3718    9.640     3 2020/06/15 19:30:14
  4602    9.640    10 2020/06/15 19:44:58
  5503    9.640    10 2020/06/15 19:59:59
  5556    9.640     3 2020/06/15 20:00:52
  6403    9.640    10 2020/06/15 20:14:59
  7304    9.640    10 2020/06/15 20:30:00
  8206    9.640    10 2020/06/15 20:45:02
  9107    9.640    10 2020/06/15 21:00:03
  9196    9.640    10 2020/06/15 21:01:32
# Rise Time Series
#  Cycle npts     Start                Date
R00   39   10 645742907 2020/06/17 21:01:47
# time     Pres phase                Date
     0    9.600     7 2020/06/17 21:01:47
   762    8.920    13 2020/06/17 21:14:29
   804    7.840    13 2020/06/17 21:15:11
   864    6.960    13 2020/06/17 21:16:11
   909    5.920    13 2020/06/17 21:16:56
   957    4.920    13 2020/06/17 21:17:44
   975    3.880    13 2020/06/17 21:18:02
   990    2.680    13 2020/06/17 21:18:17
   999    1.920    13 2020/06/17 21:18:26
  1000    1.840     8 2020/06/17 21:18:27
# Pump Time Series
#   Cycle npts
B02    39   18
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     5.920    5   11.95     214    595    -9     2
     9.600   31   12.10     179    607    -9     7
     9.560   12   12.03     198    607    -9    13
     9.560   12   12.00     191    609    -9    13
     9.600   12   11.99     195    609    -9    13
     9.520   12   12.00     203    610    -9    13
     9.600   12   12.02     209    611    -9    13
     9.560   12   12.01     210    612    -9    13
     9.520   12   11.98     194    613    -9    13
     9.560   12   12.01     184    614    -9    13
     9.520   12   12.00     198    614    -9    13
     9.600   12   11.98     209    615    -9    13
     9.600   12   12.00     205    616    -9    13
     8.680   12   11.98     203    617    -9    13
     7.200   12   11.97     201    617    -9    13
     6.200   12   12.00     189    616    -9    13
     4.680   12   11.98     185    615    -9    13
     0.360   60   12.03     191    620    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    39  2.4    4  2000    10   600    32   600  2880      5     3    1725   30  2.9380
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   39 2020/06/15 18:28:13
# Pcase vacuums
   6.190   6.060
# Oil valve positions
   1.765   0.849   0.585
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   39 2020/06/15 21:01:33 9999/99/99 99:99:99
# Pcase vacuums
   6.060   6.060
# Average 
#   P      T      S
    9.600  18.529  30.329
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   39 2020/06/17 21:01:47 9999/99/99 99:99:99
# Pcase vacuums
   6.060   6.140
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   39 2020/06/17 21:18:28
# number of packets previous packets
      7              10
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.140   6.160
# Air Pump max current time  max pressure
       0         0     1.258
# Air Valve Positions
   1.154   0.000
# Air Valve Calibrations
   1.103   0.086   0.595
# Oil Valve Positions
   0.000   0.849
# Oil Valve Calibrations
   1.735   0.869
# Encountered Error Codes
#    Cycle Errors
EE20    39      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   39  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.360     65   5.950
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   39       197    2020/06/17 21:31:01
Parameter Table Param_Table-V17.txt |
     IFV =  7.2.9.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       39	(cycle number) |
     MPS =        2	(phase) |
     MPC =        2	(count) |
  MNM[0] =        3	(count) |
  MNM[1] =        3	(count) |
  MNM[2] =        3	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        3	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =     2000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =   150000	(ms) |
     MTS =    21600	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       38	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =        5	(dBar) |
  DSD[2] =        5	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =       60	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =        1	(sec) |
  DFT[1] =        1	(sec) |
  DFT[2] =        2	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =      120	(sec) |
     PNS =       38	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =     2048	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =        1	(sec) |
  PFT[2] =    36000	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =     1000	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        1	(sec) |
  PFV[2] =      300	(sec) |
     PDT =     1800	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       38	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       10	(sec) |
     ASI =  3276800	(ms) |
     ASD =     2000	(dBar) |
     ADD =        2	(dBar) |
     ADT =        1	(sec) |
     ARC =      900	(?) |
     AHR =     4500	(cm/sec) |
     AHD =     1000	(dBar) |
     SFN =       10	(cycle number) |
     SDC =       38	(dBar) |
     SGT =        0	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =     3840	(sec) |
     SSA =      300	(0=dis 1=en) |
     SSI =        0	(ms) |
     SDS =    10000	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     2561	(.001 psi) |
     SAN =     7500	(.001 psi) |
     SAO =     6500	(sec) |
     SAP =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |

