# Message Decoder for Solo2, Alto and Deep Solo Floats; v260708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11011    36  324 000000000000009  4062     0   6 2020/06/11, 12:09:31,  41.5413  -70.6851   3      0 6,14,14
SBD 11011    36  340 000000000000009  4063     0   5 2020/06/11, 12:10:24,  41.5413  -70.6851   3      0 13
SBD 11011    36  333 000000000000009  4064     0   4 2020/06/11, 12:10:34,  41.5413  -70.6851   3      0 8,15,13
SBD 11011    36  335 000000000000009  4065     0   3 2020/06/11, 12:10:47,  41.5334  -70.6398   3      0 8
SBD 11011    36  255 000000000000009  4066     0   2 2020/06/11, 12:10:58,  41.5334  -70.6398   3      0 10,11,0,1,2,3
SBD 11011    36  337 000000000000009  4067     0   1 2020/06/11, 12:11:15,  41.5683  -70.6851   3      0 10,11,7,5,14
SBD 11011    36  327 000000000000009  4068     0   0 2020/06/11, 12:11:28,  41.5603  -70.6398   3      0 0,4,14,14,9,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      25         2167           84        2251
# Sensor_ID bytes     % packets Description
          0    48  2.22       2         GPS
          1    33  1.52       1    Pressure
          2    37  1.71       1 Temperature
          3    33  1.52       1    Salinity
          4   133  6.14       1        Fall
          5    58  2.68       1        Rise
          6   224 10.34       1        Pump
          7    72  3.32       1 Drift_times
          8   502 23.17       2     Surface
          9   114  5.26       2  HR/Drift_P
         10   146  6.74       2  HR/Drift_T
         11   146  6.74       2  HR/Drift_S
         13   404 18.64       2  Parameters
         14   151  6.97       5 Engineering
         15    66  3.05       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   36 1 2020/06/09 09:04 +41.53425  -70.64678    0  6 32 35 41  1.5     0
G   36 2 2020/06/11 11:57 +41.53431  -70.64685   40  7 34 38 43  1.4     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    36   10    1     3         2
#      Pres      Temp    Salinity
      1.400    22.3130    30.4240
      1.960    22.0880    30.4350
      3.000    21.5290    30.3980
      4.080    21.1590    30.3760
      5.120    20.8710    30.3610
      5.800    20.7260    30.3440
      7.000    20.3590    30.3330
      7.800    19.6170    30.3000
      9.400    17.9510    30.3030
      9.560    17.8590    30.3000
# High Resolution Data
#     C   Npts comp
H0    36   35     1
#      Pres      Temp    Salinity
      1.400    22.3130    30.4240
      1.520    22.3080    30.4240
      1.560    22.3040    30.4220
      1.680    22.2880    30.4270
      1.840    22.2280    30.4530
      1.960    22.0880    30.4930
      2.040    21.9830    30.4330
      2.120    21.9280    30.4190
      2.280    21.8720    30.4170
      2.440    21.7930    30.4270
      2.560    21.7330    30.4040
      2.640    21.6810    30.4080
      2.720    21.6350    30.4010
      2.880    21.5820    30.4000
      3.040    21.5260    30.3980
      3.160    21.4700    30.4000
      3.200    21.4210    30.3930
      3.280    21.3740    30.3930
      3.440    21.3370    30.3850
      3.600    21.3070    30.3770
      3.720    21.2680    30.3870
      3.760    21.2250    30.3820
      3.880    21.1950    30.3730
      4.000    21.1680    30.3760
      4.160    21.1450    30.3710
      4.280    21.1180    30.3720
      4.360    21.0840    30.3760
      4.400    21.0540    30.3720
      4.480    21.0210    30.3740
      4.600    20.9990    30.3670
      4.720    20.9700    30.3720
      4.800    20.9430    30.3690
      4.840    20.9270    30.3620
      4.880    20.9110    30.3630
      5.000    20.9000    30.3590
# Drift Data
#      C Npts comp n_var
I0    36   96    1     4
#      Pres       Temp   Salinity     Time
      9.480    17.5140    30.2910        5
      9.480    17.5200    30.2890     1811
      9.480    17.5210    30.2890     3617
      9.480    17.5230    30.2900     5423
      9.480    17.5250    30.2900     7229
      9.480    17.5270    30.2900     9034
      9.480    17.5300    30.2900    10840
      9.480    17.5330    30.2900    12646
      9.480    17.5350    30.2900    14452
      9.480    17.5380    30.2900    16258
      9.480    17.5400    30.2900    18063
      9.480    17.5410    30.2910    19869
      9.480    17.5450    30.2910    21675
      9.440    17.5500    30.2900    23481
      9.440    17.5530    30.2900    25287
      9.480    17.5540    30.2910    27092
      9.440    17.5580    30.2910    28898
      9.480    17.5600    30.2920    30704
      9.480    17.5660    30.2900    32510
      9.440    17.5710    30.2910    34316
      9.440    17.5720    30.2920    36121
      9.440    17.5770    30.2920    37927
      9.480    17.5830    30.2910    39733
      9.480    17.5840    30.2940    41539
      9.480    17.5880    30.2930    43345
      9.480    17.5940    30.2930    45150
      9.480    17.5940    30.2950    46956
      9.480    17.6010    30.2930    48762
      9.480    17.6050    30.2940    50568
      9.520    17.6080    30.2940    52374
      9.480    17.6140    30.2940    54179
      9.480    17.6180    30.2940    55985
      9.480    17.6250    30.2920    57791
      9.480    17.6260    30.2940    59597
      9.480    17.6270    30.2960    61403
      9.480    17.6310    30.2960    63208
      9.520    17.6340    30.2950    65014
      9.480    17.6420    30.2940    66820
      9.520    17.6440    30.2950    68626
      9.480    17.6500    30.2940    70432
      9.520    17.6490    30.2960    72237
      9.480    17.6520    30.2960    74043
      9.520    17.6540    30.2980    75849
      9.520    17.6570    30.2970    77655
      9.520    17.6630    30.2960    79461
      9.520    17.6650    30.2960    81266
      9.480    17.6680    30.2970    83072
      9.520    17.6740    30.2960    84878
      9.520    17.6800    30.2930    86684
      9.520    17.6800    30.2960    88490
      9.520    17.6840    30.2960    90295
      9.520    17.6850    30.2960    92101
      9.520    17.6880    30.2980    93907
      9.520    17.6900    30.2990    95713
      9.520    17.6940    30.2990    97519
      9.520    17.6960    30.3000    99324
      9.520    17.7050    30.2960   101130
      9.520    17.7070    30.2970   102936
      9.520    17.7140    30.2970   104742
      9.520    17.7170    30.2980   106547
      9.520    17.7200    30.2980   108353
      9.560    17.7190    30.3020   110159
      9.520    17.7270    30.2980   111965
      9.520    17.7280    30.3010   113771
      9.520    17.7340    30.2990   115576
      9.520    17.7420    30.2950   117382
      9.560    17.7390    30.3010   119188
      9.560    17.7440    30.3010   120994
      9.520    17.7490    30.3000   122800
      9.560    17.7570    30.2980   124605
      9.520    17.7590    30.2990   126411
      9.520    17.7650    30.2990   128217
      9.560    17.7650    30.3000   130023
      9.520    17.7700    30.3010   131829
      9.520    17.7720    30.3010   133634
      9.560    17.7790    30.2980   135440
      9.520    17.7800    30.3000   137246
      9.520    17.8080    30.2990   139052
      9.560    17.7800    30.3040   140858
      9.560    17.8000    30.2990   142663
      9.560    17.8040    30.3020   144469
      9.520    17.8100    30.3000   146275
      9.520    17.8120    30.3000   148081
      9.520    17.8130    30.3020   149887
      9.560    17.8210    30.3000   151692
      9.520    17.8240    30.3000   153498
      9.520    17.8260    30.3000   155304
      9.560    17.8280    30.3010   157110
      9.520    17.8330    30.3000   158915
      9.560    17.8350    30.3010   160721
      9.520    17.8370    30.3010   162527
      9.560    17.8380    30.3030   164333
      9.520    17.8440    30.3000   166139
      9.560    17.8440    30.3020   167944
      9.520    17.8480    30.3010   169750
      9.560    17.8510    30.3010   171556
# Surface Pressure Time Series
#    C npts n_var               Start
S   36    54     6 2020/06/11 11:58:43
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    91      0.260     22.237      0.000      14 2020/06/11 11:58:43
   103      0.260     22.203      0.000      14 2020/06/11 11:58:55
   114      0.260     22.170      0.000      14 2020/06/11 11:59:06
   125      0.260     22.154      0.000      14 2020/06/11 11:59:17
   137      0.260     22.142      0.000      14 2020/06/11 11:59:29
   148      0.250     22.129      0.000      14 2020/06/11 11:59:40
   159      0.230     22.116      0.000      14 2020/06/11 11:59:51
   170      0.250     22.105      0.000      14 2020/06/11 12:00:02
   181      0.250     22.098      0.000      14 2020/06/11 12:00:13
   192      0.240     22.068      0.000      14 2020/06/11 12:00:24
   203      0.250     21.997      0.000      14 2020/06/11 12:00:35
   214      0.260     21.972      0.000      14 2020/06/11 12:00:46
   225      0.260     21.963      0.000      14 2020/06/11 12:00:57
   236      0.260     21.961      0.000      14 2020/06/11 12:01:08
   247      0.240     21.957      0.000      14 2020/06/11 12:01:19
   258      0.250     21.950      0.000      14 2020/06/11 12:01:30
   270      0.250     21.942      0.000      14 2020/06/11 12:01:42
   281      0.240     21.944      0.000      14 2020/06/11 12:01:53
   292      0.260     21.946      0.000      14 2020/06/11 12:02:04
   303      0.240     21.948      0.000      14 2020/06/11 12:02:15
   314      0.270     21.951      0.000      14 2020/06/11 12:02:26
   325      0.260     21.964      0.000      14 2020/06/11 12:02:37
   336      0.250     21.959      0.000      14 2020/06/11 12:02:48
   347      0.250     21.886      0.000      14 2020/06/11 12:02:59
   358      0.240     21.861      0.000      14 2020/06/11 12:03:10
   369      0.250     21.841      0.000      14 2020/06/11 12:03:21
   380      0.240     21.837      0.000      14 2020/06/11 12:03:32
   392      0.250     21.852      0.000      14 2020/06/11 12:03:44
   403      0.240     21.846      0.000      14 2020/06/11 12:03:55
   414      0.250     21.853      0.000      14 2020/06/11 12:04:06
   425      0.250     21.853      0.000      14 2020/06/11 12:04:17
   436      0.260     21.862      0.000      14 2020/06/11 12:04:28
   447      0.260     21.867      0.000      14 2020/06/11 12:04:39
   458      0.240     21.876      0.000      14 2020/06/11 12:04:50
   469      0.260     21.887      0.000      14 2020/06/11 12:05:01
   480      0.260     21.885      0.000      14 2020/06/11 12:05:12
   491      0.260     21.865      0.000      14 2020/06/11 12:05:23
   502      0.250     21.807      0.000      14 2020/06/11 12:05:34
   514      0.260     21.750      0.000      14 2020/06/11 12:05:46
   525      0.250     21.702      0.000      14 2020/06/11 12:05:57
   536      0.260     21.630      0.000      14 2020/06/11 12:06:08
   547      0.250     21.613      0.000      14 2020/06/11 12:06:19
   558      0.250     21.614      0.000      14 2020/06/11 12:06:30
   569      0.260     21.641      0.000      14 2020/06/11 12:06:41
   580      0.260     21.652      0.000      14 2020/06/11 12:06:52
   591      0.240     21.640      0.000      14 2020/06/11 12:07:03
   602      0.250     21.624      0.000      14 2020/06/11 12:07:14
   613      0.260     21.606      0.000      14 2020/06/11 12:07:25
   624      0.250     21.575      0.000      14 2020/06/11 12:07:36
   636      0.240     21.553      0.000      14 2020/06/11 12:07:48
   647      0.260     21.515      0.000      14 2020/06/11 12:07:59
   658      0.260     21.465      0.000      14 2020/06/11 12:08:10
   669      0.250     21.449      0.000      14 2020/06/11 12:08:21
   680      0.240     21.406      0.000      14 2020/06/11 12:08:32
# Fall Time Series
#  Cycle npts     Start                Date
F00   36   25 645008660 2020/06/09 09:04:20
# time     Pres phase                Date
     0    0.200     1 2020/06/09 09:04:20
    42    1.440    10 2020/06/09 09:05:02
    51    2.520    10 2020/06/09 09:05:11
    55    3.160    10 2020/06/09 09:05:15
    64    4.360    10 2020/06/09 09:05:24
    73    5.560    10 2020/06/09 09:05:33
    80    6.440     2 2020/06/09 09:05:40
    82    6.800    10 2020/06/09 09:05:42
    89    7.760    10 2020/06/09 09:05:49
    92    8.080    10 2020/06/09 09:05:52
   102    9.280    10 2020/06/09 09:06:02
  1004    9.480    10 2020/06/09 09:21:04
  1888    9.520     3 2020/06/09 09:35:48
  1905    9.520    10 2020/06/09 09:36:05
  2805    9.480    10 2020/06/09 09:51:05
  3706    9.480    10 2020/06/09 10:06:06
  3726    9.480     3 2020/06/09 10:06:26
  4608    9.480    10 2020/06/09 10:21:08
  5510    9.480    10 2020/06/09 10:36:10
  5565    9.480     3 2020/06/09 10:37:05
  6411    9.480    10 2020/06/09 10:51:11
  7313    9.480    10 2020/06/09 11:06:13
  8214    9.480    10 2020/06/09 11:21:14
  9116    9.480    10 2020/06/09 11:36:16
  9204    9.480    10 2020/06/09 11:37:44
# Rise Time Series
#  Cycle npts     Start                Date
R00   36   10 645190679 2020/06/11 11:37:59
# time     Pres phase                Date
     0    9.560     7 2020/06/11 11:37:59
   763    8.960    13 2020/06/11 11:50:42
   805    8.000    13 2020/06/11 11:51:24
   892    7.000    13 2020/06/11 11:52:51
   934    5.920    13 2020/06/11 11:53:33
  1123    4.800    13 2020/06/11 11:56:42
  1132    3.880    13 2020/06/11 11:56:51
  1141    2.880    13 2020/06/11 11:57:00
  1150    1.840    13 2020/06/11 11:57:09
  1151    1.680     8 2020/06/11 11:57:10
# Pump Time Series
#   Cycle npts
B02    36   20
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     6.440    4   11.97     177    601    -9     2
     9.560   31   12.25     189    611    -9     7
     9.560   12   12.20     209    612    -9    13
     9.560   12   12.04     202    613    -9    13
     9.560   12   12.05     220    614    -9    13
     9.560   12   12.09     186    615    -9    13
     9.520   12   12.13     176    616    -9    13
     9.520   12   12.04     200    616    -9    13
     9.560   12   12.14     175    617    -9    13
     9.560   12   12.02     210    618    -9    13
     9.520   12   12.00     200    619    -9    13
     9.560   12   12.03     190    620    -9    13
     9.560   12   12.07     209    621    -9    13
     8.760   12   12.01     189    621    -9    13
     7.640   12   12.12     188    621    -9    13
     6.800   12   12.04     191    621    -9    13
     5.760   12   12.04     190    619    -9    13
     5.720   12   12.00     196    619    -9    13
     5.720   12   12.04     196    618    -9    13
     0.200   60   12.01     188    623    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    36  2.4    4  2000    10   600    32   600  2880      5     3    1725   30  2.9380
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   36 2020/06/09 09:04:18
# Pcase vacuums
   6.200   6.100
# Oil valve positions
   1.763   0.853   0.585
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   36 2020/06/09 11:37:46 9999/99/99 99:99:99
# Pcase vacuums
   6.100   6.100
# Average 
#   P      T      S
    9.520  17.678  30.296
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   36 2020/06/11 11:37:59 9999/99/99 99:99:99
# Pcase vacuums
   6.100   6.160
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   36 2020/06/11 11:57:08
# number of packets previous packets
      7              25
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.160   6.220
# Air Pump max current time  max pressure
       0         0     1.264
# Air Valve Positions
   1.155   0.000
# Air Valve Calibrations
   1.103   0.086   0.595
# Oil Valve Positions
   0.000   0.851
# Oil Valve Calibrations
   1.735   0.869
# Encountered Error Codes
#    Cycle Errors
EE20    36      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   36  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.200     74   6.000
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   36       197    2020/06/11 12:10:34
Parameter Table Param_Table-V17.txt |
     IFV =  7.2.9.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       36	(cycle number) |
     MPS =        2	(phase) |
     MPC =        2	(count) |
  MNM[0] =        3	(count) |
  MNM[1] =        3	(count) |
  MNM[2] =        3	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        3	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =     2000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =   150000	(ms) |
     MTS =    21600	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       35	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =        5	(dBar) |
  DSD[2] =        5	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =       60	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =        1	(sec) |
  DFT[1] =        1	(sec) |
  DFT[2] =        2	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =      120	(sec) |
     PNS =       35	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =     2048	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =        1	(sec) |
  PFT[2] =    36000	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =     1000	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        1	(sec) |
  PFV[2] =      300	(sec) |
     PDT =     1800	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       35	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       10	(sec) |
     ASI =  3276800	(ms) |
     ASD =     2000	(dBar) |
     ADD =        2	(dBar) |
     ADT =        1	(sec) |
     ARC =      900	(?) |
     AHR =     4500	(cm/sec) |
     AHD =     1000	(dBar) |
     SFN =       10	(cycle number) |
     SDC =       35	(dBar) |
     SGT =        0	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =     3840	(sec) |
     SSA =      300	(0=dis 1=en) |
     SSI =        0	(ms) |
     SDS =    10000	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     2561	(.001 psi) |
     SAN =     7500	(.001 psi) |
     SAO =     6500	(sec) |
     SAP =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |

