# Message Decoder for Solo2, Alto and Deep Solo Floats; v260708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11011    33  335 000000000000009  4041     0   6 2020/06/05, 02:43:00,  41.5108  -70.6557   6      0 6,14,14
SBD 11011    33  340 000000000000009  4042     0   5 2020/06/05, 02:43:12,  41.5108  -70.6557   7      0 13
SBD 11011    33  333 000000000000009  4043     0   4 2020/06/05, 02:46:20,  41.5334  -70.6398  12      0 8,15,13
SBD 11011    33  335 000000000000009  4044     0   3 2020/06/05, 02:46:32,  41.5334  -70.6398  10      0 8
SBD 11011    33  292 000000000000009  4045     0   2 2020/06/05, 02:46:51,  41.5334  -70.6398   9      0 9,10,11,0,1,2,3
SBD 11011    33  333 000000000000009  4046     0   1 2020/06/05, 02:47:16,  41.5334  -70.6398   7      0 10,11,7,5
SBD 11011    33  304 000000000000009  4047     0   0 2020/06/05, 02:47:49,  41.5334  -70.6398   5      0 0,4,14,14,9,14
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      25         2188           84        2272
# Sensor_ID bytes     % packets Description
          0    48  2.19       2         GPS
          1    33  1.51       1    Pressure
          2    37  1.69       1 Temperature
          3    33  1.51       1    Salinity
          4   133  6.08       1        Fall
          5    58  2.65       1        Rise
          6   235 10.74       1        Pump
          7    72  3.29       1 Drift_times
          8   502 22.94       2     Surface
          9   113  5.16       2  HR/Drift_P
         10   167  7.63       2  HR/Drift_T
         11   136  6.22       2  HR/Drift_S
         13   404 18.46       2  Parameters
         14   151  6.90       5 Engineering
         15    66  3.02       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G   33 1 2020/06/02 23:37 +41.53430  -70.64688    0  5 23 36 45  1.6     0
G   33 2 2020/06/05 02:31 +41.53403  -70.64677   30  5 28 34 43  2.8     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0    33   10    1     3         2
#      Pres      Temp    Salinity
      1.440    22.0770    30.4190
      1.960    21.5960    30.4140
      2.960    20.9190    30.3930
      4.000    20.4430    30.3540
      5.080    20.0690    30.3400
      5.680    19.9280    30.3180
      7.040    19.4140    30.2990
      7.920    18.1920    30.2800
      9.440    16.7160    30.2690
      9.520    16.6880    30.2680
# High Resolution Data
#     C   Npts comp
H0    33   33     1
#      Pres      Temp    Salinity
      1.440    22.0770    30.4190
      1.560    22.0150    30.3890
      1.720    21.9280    30.4120
      1.840    21.7890    30.4390
      1.880    21.6370    30.4440
      2.040    21.4890    30.4450
      2.200    21.3630    30.4220
      2.240    21.2910    30.3900
      2.360    21.2590    30.3730
      2.560    21.1970    30.4000
      2.680    21.1020    30.4090
      2.720    21.0210    30.3940
      2.920    20.9670    30.3800
      3.040    20.8930    30.3970
      3.120    20.8080    30.3880
      3.240    20.7220    30.3940
      3.400    20.6430    30.3850
      3.520    20.5650    30.3750
      3.560    20.5120    30.3620
      3.720    20.4980    30.3390
      3.880    20.4960    30.3350
      3.920    20.4720    30.3540
      4.040    20.4430    30.3560
      4.200    20.4180    30.3510
      4.240    20.3830    30.3540
      4.360    20.3460    30.3580
      4.480    20.3000    30.3600
      4.520    20.2480    30.3600
      4.640    20.2080    30.3550
      4.800    20.1630    30.3480
      4.800    20.1220    30.3500
      4.880    20.0920    30.3450
      5.000    20.0680    30.3410
# Drift Data
#      C Npts comp n_var
I0    33   96    1     4
#      Pres       Temp   Salinity     Time
      9.440    16.2820    30.2620        6
      9.440    16.2860    30.2620     1812
      9.440    16.2910    30.2610     3618
      9.440    16.2940    30.2630     5423
      9.440    16.2980    30.2630     7229
      9.440    16.3000    30.2630     9035
      9.440    16.3040    30.2620    10841
      9.440    16.3080    30.2620    12647
      9.400    16.3120    30.2620    14452
      9.400    16.3160    30.2620    16258
      9.400    16.3160    30.2630    18064
      9.400    16.3210    30.2620    19870
      9.400    16.3240    30.2620    21676
      9.400    16.3250    30.2640    23481
      9.400    16.3290    30.2640    25287
      9.400    16.3320    30.2630    27093
      9.400    16.3340    30.2640    28899
      9.400    16.3380    30.2630    30705
      9.400    16.3390    30.2630    32510
      9.400    16.3430    30.2630    34316
      9.400    16.3450    30.2640    36122
      9.400    16.3470    30.2640    37928
      9.400    16.3530    30.2620    39734
      9.360    16.3540    30.2650    41539
      9.400    16.3580    30.2660    43345
      9.360    16.3590    30.2670    45151
      9.360    16.3670    30.2640    46957
      9.360    16.3680    30.2650    48763
      9.360    16.3700    30.2660    50568
      9.360    16.3750    30.2650    52374
      9.360    16.3810    30.2630    54180
      9.360    16.3790    30.2670    55986
      9.360    16.3850    30.2660    57792
      9.360    16.3910    30.2650    59597
      9.360    16.3920    30.2670    61403
      9.360    16.3950    30.2680    63209
      9.360    16.4020    30.2660    65015
      9.360    16.4040    30.2680    66821
      9.360    16.4110    30.2660    68626
      9.360    16.4120    30.2680    70432
      9.360    16.4180    30.2660    72238
      9.360    16.4200    30.2680    74044
      9.360    16.4250    30.2680    75850
      9.360    16.4340    30.2650    77655
      9.360    16.4340    30.2670    79461
      9.360    16.4370    30.2680    81267
      9.400    16.4420    30.2690    83073
      9.360    16.4490    30.2670    84879
      9.360    16.4510    30.2680    86684
      9.360    16.4580    30.2660    88490
      9.400    16.4590    30.2690    90296
      9.360    16.4610    30.2700    92102
      9.360    16.4690    30.2690    93908
      9.360    16.4710    30.2700    95713
      9.400    16.4770    30.2680    97519
      9.400    16.4790    30.2690    99325
      9.400    16.4830    30.2700   101131
      9.400    16.4850    30.2710   102937
      9.400    16.4960    30.2670   104742
      9.400    16.4960    30.2680   106548
      9.400    16.5000    30.2690   108354
      9.400    16.5040    30.2680   110160
      9.440    16.5060    30.2690   111966
      9.400    16.5110    30.2680   113771
      9.440    16.5150    30.2680   115577
      9.440    16.5150    30.2700   117383
      9.400    16.5180    30.2700   119189
      9.400    16.5240    30.2690   120995
      9.440    16.5260    30.2710   122800
      9.440    16.5290    30.2710   124606
      9.440    16.5300    30.2720   126412
      9.440    16.5390    30.2710   128218
      9.440    16.5410    30.2710   130023
      9.440    16.5470    30.2700   131829
      9.440    16.5480    30.2710   133635
      9.440    16.5530    30.2710   135441
      9.440    16.5580    30.2710   137247
      9.440    16.5870    30.2780   139052
      9.440    16.5590    30.2720   140858
      9.440    16.5710    30.2760   142664
      9.440    16.5980    30.2670   144470
      9.440    16.5910    30.2710   146276
      9.440    16.5990    30.2710   148081
      9.440    16.6090    30.2700   149887
      9.440    16.6130    30.2710   151693
      9.440    16.6200    30.2710   153499
      9.440    16.6250    30.2710   155305
      9.480    16.6300    30.2710   157110
      9.440    16.6360    30.2700   158916
      9.440    16.6420    30.2690   160722
      9.440    16.6480    30.2690   162528
      9.480    16.6530    30.2700   164334
      9.480    16.6570    30.2710   166139
      9.480    16.6640    30.2700   167945
      9.440    16.6710    30.2700   169751
      9.440    16.6770    30.2700   171557
# Surface Pressure Time Series
#    C npts n_var               Start
S   33    54     6 2020/06/05 02:31:57
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    88      0.180     22.446      0.000      14 2020/06/05 02:31:57
   101      0.190     22.267      0.000      14 2020/06/05 02:32:10
   112      0.170     22.129      0.000      14 2020/06/05 02:32:21
   123      0.180     22.010      0.000      14 2020/06/05 02:32:32
   134      0.170     21.871      0.000      14 2020/06/05 02:32:43
   145      0.180     21.751      0.000      14 2020/06/05 02:32:54
   156      0.180     21.666      0.000      14 2020/06/05 02:33:05
   167      0.170     21.559      0.000      14 2020/06/05 02:33:16
   178      0.180     21.468      0.000      14 2020/06/05 02:33:27
   190      0.180     21.401      0.000      14 2020/06/05 02:33:39
   201      0.170     21.314      0.000      14 2020/06/05 02:33:50
   212      0.160     21.201      0.000      14 2020/06/05 02:34:01
   223      0.160     21.040      0.000      14 2020/06/05 02:34:12
   234      0.170     20.929      0.000      14 2020/06/05 02:34:23
   245      0.170     20.806      0.000      14 2020/06/05 02:34:34
   256      0.170     20.716      0.000      14 2020/06/05 02:34:45
   267      0.170     20.644      0.000      14 2020/06/05 02:34:56
   278      0.180     20.613      0.000      14 2020/06/05 02:35:07
   289      0.180     20.579      0.000      14 2020/06/05 02:35:18
   300      0.180     20.572      0.000      14 2020/06/05 02:35:29
   312      0.180     20.581      0.000      14 2020/06/05 02:35:41
   323      0.170     20.606      0.000      14 2020/06/05 02:35:52
   334      0.180     20.629      0.000      14 2020/06/05 02:36:03
   345      0.190     20.647      0.000      14 2020/06/05 02:36:14
   356      0.180     20.637      0.000      14 2020/06/05 02:36:25
   367      0.170     20.614      0.000      14 2020/06/05 02:36:36
   378      0.180     20.613      0.000      14 2020/06/05 02:36:47
   389      0.170     20.605      0.000      14 2020/06/05 02:36:58
   400      0.170     20.619      0.000      14 2020/06/05 02:37:09
   411      0.180     20.595      0.000      14 2020/06/05 02:37:20
   422      0.180     20.579      0.000      14 2020/06/05 02:37:31
   433      0.180     20.563      0.000      14 2020/06/05 02:37:42
   445      0.170     20.562      0.000      14 2020/06/05 02:37:54
   456      0.170     20.575      0.000      14 2020/06/05 02:38:05
   467      0.180     20.580      0.000      14 2020/06/05 02:38:16
   478      0.160     20.548      0.000      14 2020/06/05 02:38:27
   489      0.180     20.529      0.000      14 2020/06/05 02:38:38
   500      0.180     20.486      0.000      14 2020/06/05 02:38:49
   511      0.180     20.425      0.000      14 2020/06/05 02:39:00
   522      0.180     20.397      0.000      14 2020/06/05 02:39:11
   533      0.180     20.417      0.000      14 2020/06/05 02:39:22
   544      0.170     20.425      0.000      14 2020/06/05 02:39:33
   555      0.190     20.437      0.000      14 2020/06/05 02:39:44
   567      0.210     20.462      0.000      14 2020/06/05 02:39:56
   578      0.180     20.454      0.000      14 2020/06/05 02:40:07
   589      0.160     20.471      0.000      14 2020/06/05 02:40:18
   600      0.180     20.499      0.000      14 2020/06/05 02:40:29
   611      0.170     20.543      0.000      14 2020/06/05 02:40:40
   622      0.180     20.554      0.000      14 2020/06/05 02:40:51
   633      0.170     20.529      0.000      14 2020/06/05 02:41:02
   644      0.180     20.501      0.000      14 2020/06/05 02:41:13
   655      0.170     20.442      0.000      14 2020/06/05 02:41:24
   666      0.160     20.435      0.000      14 2020/06/05 02:41:35
   677      0.170     20.439      0.000      14 2020/06/05 02:41:46
# Fall Time Series
#  Cycle npts     Start                Date
F00   33   25 644456255 2020/06/02 23:37:35
# time     Pres phase                Date
     0    0.160     1 2020/06/02 23:37:35
    38    1.240    10 2020/06/02 23:38:13
    47    2.280    10 2020/06/02 23:38:22
    56    3.400    10 2020/06/02 23:38:31
    65    4.520    10 2020/06/02 23:38:40
    69    5.120    10 2020/06/02 23:38:44
    75    5.920     2 2020/06/02 23:38:50
    78    6.240    10 2020/06/02 23:38:53
    85    7.160    10 2020/06/02 23:39:00
    93    8.040    10 2020/06/02 23:39:08
   103    9.160    10 2020/06/02 23:39:18
  1005    9.480    10 2020/06/02 23:54:20
  1884    9.480     3 2020/06/03 00:08:59
  1906    9.480    10 2020/06/03 00:09:21
  2808    9.440    10 2020/06/03 00:24:23
  3710    9.480    10 2020/06/03 00:39:25
  3722    9.480     3 2020/06/03 00:39:37
  4611    9.440    10 2020/06/03 00:54:26
  5513    9.480    10 2020/06/03 01:09:28
  5561    9.480     3 2020/06/03 01:10:16
  6414    9.480    10 2020/06/03 01:24:29
  7316    9.480    10 2020/06/03 01:39:31
  8218    9.440    10 2020/06/03 01:54:33
  9119    9.440    10 2020/06/03 02:09:34
  9200    9.440    10 2020/06/03 02:10:55
# Rise Time Series
#  Cycle npts     Start                Date
R00   33   10 644638269 2020/06/05 02:11:09
# time     Pres phase                Date
     0    9.480     7 2020/06/05 02:11:09
   745    8.920    13 2020/06/05 02:23:34
   778    8.000    13 2020/06/05 02:24:07
   862    7.000    13 2020/06/05 02:25:31
   916    6.000    13 2020/06/05 02:26:25
  1129    4.880    13 2020/06/05 02:29:58
  1141    3.720    13 2020/06/05 02:30:10
  1150    2.680    13 2020/06/05 02:30:19
  1156    1.880    13 2020/06/05 02:30:25
  1157    1.840     8 2020/06/05 02:30:26
# Pump Time Series
#   Cycle npts
B02    33   21
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     5.920    5   11.91     197    604    -9     2
     9.480   31   11.97     207    616    -9     7
     9.440   12   12.04     196    617    -9    13
     9.440   12   12.17     205    618    -9    13
     9.440   12   11.98     210    619    -9    13
     9.480   12   11.99     212    619    -9    13
     9.520   12   12.00     223    621    -9    13
     9.440   12   12.07     228    622    -9    13
     9.520   12   12.14     213    622    -9    13
     9.440   12   11.99     204    623    -9    13
     9.480   12   12.09     189    624    -9    13
     9.440   12   12.01     190    625    -9    13
     9.320   12   12.02     178    625    -9    13
     8.080   12   11.94     195    626    -9    13
     7.120   12   12.10     183    626    -9    13
     6.320   12   11.95     193    625    -9    13
     5.560   12   12.04     177    623    -9    13
     5.600   12   12.11     204    622    -9    13
     5.560   12   12.02     222    622    -9    13
     2.240    4   11.99     181    621    -9    13
     0.160   60   12.03     164    627    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2    33  2.4    4  2000    10   600    32   600  2880      5     3    1725   30  2.9380
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17   33 2020/06/02 23:37:32
# Pcase vacuums
   6.240   6.160
# Oil valve positions
   1.762   0.848   0.585
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18   33 2020/06/03 02:10:56 9999/99/99 99:99:99
# Pcase vacuums
   6.160   6.160
# Average 
#   P      T      S
    9.400  16.456  30.267
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18   33 2020/06/05 02:11:09 9999/99/99 99:99:99
# Pcase vacuums
   6.150   6.210
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20   33 2020/06/05 02:30:24
# number of packets previous packets
      7              20
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.210   6.230
# Air Pump max current time  max pressure
       0         0     1.275
# Air Valve Positions
   1.158   0.000
# Air Valve Calibrations
   1.103   0.086   0.595
# Oil Valve Positions
   0.000   0.850
# Oil Valve Calibrations
   1.735   0.869
# Encountered Error Codes
#    Cycle Errors
EE20    33      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00   33  1      0
# MC    Pres      Time  Pcase Vacuum
   1     0.160     70   6.030
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M   33       197    2020/06/05 02:46:20
Parameter Table Param_Table-V17.txt |
     IFV =  7.2.9.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =       33	(cycle number) |
     MPS =        2	(phase) |
     MPC =        2	(count) |
  MNM[0] =        3	(count) |
  MNM[1] =        3	(count) |
  MNM[2] =        3	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        3	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =     2000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =   150000	(ms) |
     MTS =    21600	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =       32	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =        5	(dBar) |
  DSD[2] =        5	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =       60	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =        1	(sec) |
  DFT[1] =        1	(sec) |
  DFT[2] =        2	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =      120	(sec) |
     PNS =       32	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =     2048	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =        1	(sec) |
  PFT[2] =    36000	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =     1000	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        1	(sec) |
  PFV[2] =      300	(sec) |
     PDT =     1800	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =       32	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       10	(sec) |
     ASI =  3276800	(ms) |
     ASD =     2000	(dBar) |
     ADD =        2	(dBar) |
     ADT =        1	(sec) |
     ARC =      900	(?) |
     AHR =     4500	(cm/sec) |
     AHD =     1000	(dBar) |
     SFN =       10	(cycle number) |
     SDC =       32	(dBar) |
     SGT =        0	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =     3840	(sec) |
     SSA =      300	(0=dis 1=en) |
     SSI =        0	(ms) |
     SDS =    10000	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     2561	(.001 psi) |
     SAN =     7500	(.001 psi) |
     SAO =     6500	(sec) |
     SAP =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |

